I re-centered the ITMX & ITMY Optical lever beams today since they were off. First I aligned the beam into the vacuum so that it went through the center of the on table optics and then tweaked the receiver optics alignment.
There are several bad practices on these which probably makes them drift:
According to the datasheets, the laser has a beam diameter of 0.6 mm and a divergence angle of 1.3/2 mrad. So we can just calculate the right lens positions next time and not have to experiment with the whole visible laser lens kit.
For next Wednesday's cleanup, someone should volunteer to make the mounts more stable for the ITMs.