Not logged in
Wed Oct 8 19:17:55 2008
Laser power is slowly recovering
This afternoon we (rich, steve, yoichi) shutdown the laser for the DC-DC converter installation.
(we decided not to do so. Detail will be posted soon.)
After we turned on the laser again,the laser power has been recovering but very slowly.
At the time of writing, the laser power is 2.6W (MOPA_AMPMON).
I think it is because the chiller temparature has not yet settled down (it went up to 25C and slowly coming down, now at 22C).
It will take some hours until the power fully comes back.
Right now I confirmed that at least the MC locks.