We had a look this morning at the status of the seismometer array, so that we can get it all put together. While we were looking at the Guralp at the Yend, we noticed that it was pointing the wrong way. The North-South nubbins were pointed East-West, so X and Y coming out of the seismometer were backward.
To fix the Yend's Guralp, we powered off the Guralp readout box, rotated the seismometer, re-leveled it, and then turned the power back on. Now X from the seismometer lines up with the X data channel, and similarly for Y.
The Yend Guralp has all of the cabling needed, and is installed on the granite slab. This seismometer doesn't need any more work for now. When we get around to it, we'll need to do some kind of thermal insulation, but other than that, it's good to go.
The Xend will also have a Guralp (Zach still has it in the Gyro lab for now). We have the long cable that should go from the readout box to the slab that we'll need to put into the cable tray. The short cable from the slab's plate to the seismometer is already in place. For this seismometer, we should just need to plop the instrument in place and lay the cable in the overhead cable trays. We should also remove the now obsolete STS-2 cable while we're doing that. So, the Xend seismometer station doesn't need too much work.
The corner station will need more work. Zach made for us the long cable, although he still has it in the Gyro lab, so when we get the seismometer and cable back, we'll need to lay that cable in the overhead trays. The short cable from the slab's plate to the seismometer does not exist yet. We want to make sure that we can feed the finished cable and connector through the hole in the slab, and then we'll solder it up out here on the EE bench. I think this is how Den was doing things. If not, we'll have to do the soldering in-situ, which we don't want. So, for the corner station, we need to make the short cable, lay the long cable, get the T-240 back from Zach and put it on the slab, re-install the readout box that Zach has, etc, etc. We should also make sure that the spaghetti pot fits on the slab, underneath the piece of metal that's sticking out over the slab. We think that it's the same amount of clearance that the Yend pot has, so it should be okay, but we'll check. The O-ring seems to be sitting on the MC2 chamber, so we should remember that.
Neither the Xend nor the corner station had the yellow dog clamps, so we'll have to figure out where Den / Steve have hidden them.
EDIT: We have checked, and the Guralp connector, which is larger than the Trillium connector, fits through the hole in the slab (with some disassembly), so we can solder together the short cable out here on the EE bench, and install it separately. Eeeeexxxxxcelllent.