40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log Not logged in Message ID: 10098     Entry time: Wed Jun 25 09:16:52 2014
 Author: Harry Type: Update Category: General Subject: Razorblade Measurements

Purpose

To use a razorblade to measure beam waist at multiple points along the optical axis, so as to later extrapolate the modal profile of the entire beam. This information will then be used to effectively couple AUX laser light to fibers for use in the frequency offset locking apparatus.

Data Acquisition

1) Step the micrometer-controlled razorblade across the beam at a given value of Z, along optical axis, in the plane orthogonal to it (arbitrarily called X).

2) At each value of X, record the corresponding output of a photodiode, (Thorlabs PD A55) here given in mV.

3) Repeat process at multiple points along Z

Analysis

Data from each iteration in the X were fitted to the error function shown below.

V(x) = A*(erf((x-m)/s)+c)

In the Y, they were fitted to:

V(x) = -A*(erf((x-m)/s)+c)

'A' corresponds to an amplitude, 'm' to a mean, 's' to a σ, and 'c' to an offset.

(Only because in Y measurements, the blade progressed toward eclipsing the beam, as opposed to in the X where it progressively revealed the beam.

These fits can be solved for x = (erf-1((V/A)-c)*s)+m1  which can be calculated at the points (Vmax/e2) and (Vmax*(1-1/e2)). The difference between these points will yield beam waist, w(z).

Conclusion

Calculations yielded waists of: X1=66.43um, X2=67.73um, X3=49.45um, Y1=61.20um, Y2=58.70, Y3=58.89

These data seem suspect, and shall be subjected to further analysis.

 Attachment 1: 40m.zip  2.844 MB
ELOG V3.1.3-