[Aidan, Jordan, Radhika]
Radhika and Jordan identified some particulates (hair and flecks of foil) on the O-ring on the IR labs dewer. Additionally, we saw a scratch in the O-ring groove and a nick on the metal of the base of the dewer where it meets the O-ring. All were in the leaky vicinity previously identified by the He testing.
We set up a cradle to hold the dewer while we are working on it. Still needs vertical supports.
R & J replaced the O-ring with a new one with Crytox applied.
Purchased from Lesker, for venting of IR Labs Cryostat:
- P/N C35103000 (KF10 Valve for up-to-air)
- P/N QF25XQF10 (KF25-KF10 Reducer)
- P/N QF25-100-T (KF25 Tee)
(to be connected in between the Chamber Tee and the Gate Valve)
Should be in next week!