Here's a plot of the 15-day output of the SLED.
Currently there is an 980nm FC/APC fiber-optic patch-cord attached to the SLED. It occurred to me this morning that even though the patch cord is angle-cleaved, there may be some back-reflection than desired because the SLED output is 830nm (or thereabouts) while the patch cord is rated for 980nm.
I'm going to turn off the SLED until I get an 830nm patch-cord and try it then.
Correction: I removed the fiber-optic connector and put the plastic cap back on the SLED output. The mode over-lap (in terms of area) from the reflection off the cap with the output from the fiber is about 1 part in 1000. So even with 100% reflection, there is less than the 0.3% danger level coupled back into the fiber. The SLED is on again.