I added some new channels to the Athena DAQ that record the diagnostic channels from the Superlum SLED.
The ioc that handles the EPICS channels is on tcs_daq(10.0.1.34) in /target/TCS_westbridge.db
The channels are added to the frame builder in /cvs/cds/caltech/chans/daq/C4TCS.ini
Currently, the driver for the SLED is ON but the current to the SLED is off. This is to check that the zero value of the PD_VOLTAGE signal doesn't wander.
Also, the input noise of the Athena is around +/- 10 counts (where 2^15 counts = 10V) which is a pretty poor 3mV.