As I mentioned in my previous post, the signal from the plate's displacement is strongly coupled with the coil's signal, so that our system is unstable. In fact, I calculated the transfer function of this "feedback loop" of the coils and found it to be about 2mV per V, roughly the magnitude of the feedback signal of the plate. We now use DCs coils to provide the feedback loop and want to find the conversion between volts applied from the DC coils and the force and only care about certain readings.
In the above figure, I represent the plate with the circle. Sensors and coils are in black and lie above the plate, while motors are in purple and lie below. As you can see in our arrangement, the DC coils are above the DC motors, so it is safe to ignore readings from the strain gauges that are not at or neighboring with the coils. Then, I calculated the conversion factors between applied V in the coils and applied force on the plate.
If prior post I showed the measurements between Volts in coils and measured mV for the strain gauges [mV=Volts*slope (mV/V)]
I also posted the measurements between weight/force and measured mV for the strain gauges [mV=Force*slope(mV/N)]
I found how volts in the coils correlate to applied force by combinging the two equations:
F=Volts*slope(mV/V) / slope (mV/N) = Volts * slope (N/V)
To give an example, I look at the AC1 coil. I have measured the response of the B1, B2, and B3 strain gauges. I also know how B1 and B2 strain gauges responded to the weights I put on AC1 (here, I ignore the 3rd reading from B3 strain gauge, because it is further away as seen in the above figure). Thus, I will get two readings (one through each, B1 and B2, motor) for how AC1 coil signal correlates to force applied by the AC1 coil. These numbers should in principle agree, or at least be close. Here are my findings:
ΔN (by DC3 coil) = 0.0239 (N/V) * V (by DC3 coil) ; measured through B2
ΔN (by AC3 coil) = 0.00055 (N/V) * V (by AC3 coil) ; measured through B2
ΔN (by AC3 coil) = 0.0023 (N/V) * V (by AC3 coil) ; measured through B3
ΔN (by DC2 coil) = 0.0023 (N/V) * V (by DC2 coil) ; measured through B3
ΔN (by AC2 coil) = 0.0022 (N/V) * V (by AC2 coil) ; measured through B1
ΔN (by AC2 coil) = 0.00198 (N/V) * V (by AC2 coil) ; measured through B3
ΔN (by DC1 coil) = 0.0016 (N/V) * V (by DC1 coil) ; measured through B1
ΔN (by AC1 coil) = 0.00078 (N/V) * V (by AC1 coil) ; measured through B1
ΔN (by AC1 coil) = 0.00363 (N/V) * V (by AC1 coil) ; measured through B2
The coefficient for DC3 seems not to fit the norm shown by the rest data.
Correction of coupling signal
I thought that, knowing the signal from the coils, we could feed its opposite to the sensors to cancel its effect. In practice, we would our feedback loop to look as the picture on the left part in the figure below. I can rearrange it to show it more clearly that the G_{c} and G_{c} would simply add and cancel. We can do this cancellation within our digital feedback loop. Specifically, we can add the term G_{c}G_{S} to cancel the coupling signal of the coils. Haixing agreed and we will try this tomorrow.
