I took a relative intensity noise (RIN) measurement of the ECDL, by feeding the 0th order output of the AOM to the SR785. The RF power driving the AOM was set to 0 dBm. The RIN at 1 Hz is about 3x10-5, which is consistent with informal measurements we took on 08/13. From my understanding this noise looks pretty low, which is good. I will consult with Paco and add more discussion or conclusions, if any.