Here is the analyzed data from Test 3 of the A1 JPL PD.
Note about the QE measurement:
Dark current is the output from the Keithley scan - the vertical scale is correct in Amps [ignore question mark]
Dark noise spectra are included for different bias levels and at different temperatures. Stll need to add ADC noise floor for these plots.
Notes from Test 3
6-Oct-2021: done with cryocooler off and temperature increasing
PREAMP GAIN = 1E3
SR560 gain = 500
LD temp set point = 20.2kOhm