In order to resume testing the curvatures of the mirrors, the same mirror as the previous one was tested.
The result looks consistent with the previous measurement.
It seems that there has been some locking offset. Actually, the split peaks in the TF@83MHz indicates
the existence of the offset. Next time, it should be adjusted at the beginning.
Curved mirror SN: C1
RoC: 2.5785 +/- 0.000042 [m]
=> 2.5830, 2.5638 => sqrt(RoC1*RoC2) = 2.5734 m
=> 2.5844, 2.5666 => sqrt(RoC1*RoC2) = 2.5755 m