The impedances of the individual components from the 3IFO EOM (before modification) were tested.
Each component was modeled by LISO. The LISO model (in PDF and txt) are attached at the end of the entry.
There are three inductors taken from the EOM unit. They showed the Q ranging from 150~300.
Their impedances are compared with the coil taken from the 9MHz port of the spare EOM (=current LHO EOM).
The inductance of the 8.7MHz inductor indicated higher L but still higher Q.
Todd made a replica of the 45.3MHz coil. He used a silver plated wire and it actually showed highest Q of ~400.
The crystal capacitances were measured by attaching a test rig on the DB15 connector of the crystal housing. The rig was calibrated such that the impedances of the attched components on the rig were measured. They showed somewhat similar feature with parasitic resonances at ~50MHz. Above this frequnecy the capacitance went down (i.e. Abs(Z) went up). This indicates there are stray series LCR in pararrel to the crystal. Not sure what is the cause of this.
The central (24.1MHz) port showed smaller capacitance. This probably means the plates for the central port is smaller. Not sure the actual dimensions of the plates for this unit.