40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 OMC elog Not logged in Message ID: 179     Entry time: Fri Feb 28 19:50:11 2014
 Author: Koji Type: General Category: General Subject: Misalignment ABCD matrix for the aLIGO OMC

Relationship between mirror misalignment and cavity mode shift was calculated.
The technique described in T0900647 by Sam Waldman was used.

The angles and displacement of the mirrors and beams are defined in the attached figure.

x1 = 0.893134 α + 1.10676 β + 1.32252 γ + 1.24619 δ
𝛳1 = 0.75864 α - 0.75864 β - 0.271075 γ + 0.271075 δ

x2 = 1.10676 α + 0.893134 β + 1.24619 γ + 1.32252 δ
𝛳2 = 0.75864 α + 1.24136 β - 0.271075 γ + 0.271075 δ

x3 = 1.32252 α + 1.24619 β + 1.1691 γ + 1.39962 δ
𝛳3 = -0.271075 α + 0.271075 β + 0.818668 γ - 0.818668 δ

x4 = 1.24619 α + 1.32252 β + 1.39962 γ + 1.1691 δ
𝛳4 = -1.24136 α - 0.75864 β - 0.271075 γ + 0.271075 δ

Assuming the flat mirrors are fixed:
If I want to move the x3 mirror up by 1mm without moving x4, the solution is 