Yesterday Jeff and Chub worked on the cabling of the OMC. It turned out that the gender of the cable connectors
going from the cavity side to the connector bracket on top of the OMC were opposite from what is needed.
This way, the connectors can't fixed on the cable harness, thus they are free during the shipping.
We considered several ideas to mitigate this issue and decided to swap the gender of the Mighty Mouse connectors.
In order to check this operation may cause the shortage of the cable length, we made the fitting of the cables.
They seem all long enough for Chub to replace the Mighty Mouse connectors with the proper gender.
We also checked the polarity of the PZT wires. We marked the positive side of the PZT by a knot at the wire end.