On Monday, after I did some inventory of all the parts we have received from various companies, Dmass helped me mount the RIO lasers into their mounts so that I could get started with the optical setup. We cleaned the surfaces with methanol, applied a small layer of silver thermal compound, and then screwed them in.
I then borrowed the following to run the lasers:
After finding the right cables, I powered up the lasers and verified the P-I curve for each as listed on the spec sheets.
I then built a quick (temporary) optical beat setup, combining the two beams on an 1811. I had the temperatures (actually, thermistor resistances) set to what was listed as the testing set point on the datasheet, and as soon as I overlapped the beams and focused them onto the PD, there was already a strong ~50 MHz optical beat.
I have spent some time since then trying to lock various kinds of PLLs, both to interrogate the free-running frequency noise and to get used to controlling the lasers. Some things I've tried:
The first two were not helped by the fairly basic loop shaping afforded by attenuators and an SR560.
I think my next step will be to simply use the I-Q demodulation method (like I did to measure the no-FM Marconi noise in ATF:1877) to measure the frequency noise. I'll compare that to what I get with the Zurich PLL.