We did an ingenious checkup of the whitening board tonight.
I've restored all connections at that we messed with at the LSC rack to their original positions.
The TT alignment seems to be drifting around more than usual after we disconnected one of the channels - when I came in today afternoon, the spot on the AS camera had drifted by ~1 spot diameter so I had to manually re-align TT1.
Based on my tests, everything on the Demod board seems to work as expected. I need to think more about what else could be happening here - specifically do a more direct test on the whitening board.
sudo dd if=SL-7.3-x86_64-2017-01-20-LiveCD.iso of=/dev/sdf
Debian doesn't like EPICS. Or our XY plots of beam spots...Sad!
No, not confused on that point. We just will not be testing OS versions at the 40m or running multiple OS's on our workstations. As I've said before, we will only move to so-called 'reference' systems once they've been in use for a long time.
Ubuntu16 is not to my knowledge used for any CDS system anywhere. I'm not sure how you expect to have better support for that. There are no pre-compiled packages of any kind available for Ubuntu16. Good luck, you big smelly doofuses. Nyah, nyah, nyah.
K Thorne recommends that we use SL7.3 with the 'xfce' window manager instead of the Debian family of products, so we'll try it out on allegra and rossa to see how it works for us. Hopefully the LLO CDS team will be the tip of the spear on solving the usual software problems we have when we "~up" grade.
nominal changed from 22 to 23 dB to minimize PC drive RMS
previous loop gain measurement is sort of bogus (made on SR785); need some 4395 loop measurements and checking of crossover and error point spectrum
I have moved the USB flash drives from the electronics bench back into the middle drawer of the cabinet next to the AC which is west of the fridge. Drawer re-enlabeled.
Today I tried debugging the mysterious increase in REFL55 signal levels in the DRMI configuration. I focused on the demod board, because last week, I had tried routing these signals through different channels on the whitening board, and saw the same effect.
I did a quick check by switching the output of the REFL55 demod board to the inputs normally used by AS55 signals on the whitening board. Setting the whitening gain to +18dB for these channels had the same effect - ADC overflow galore. So looks like the whitening board isn't to blame. I will have to check the demod board out.
All connections have been restored untill further debugging later in the evening.
A couple of minutes ago, Larry W swapped the fibers to our 40m Edgeswitch (BROCADE FWS 648G) to a faster connection. This is the switch to which our gateway machine, NODUS, is connected. The actual swap itself happened at the core router in Bridge, and took only a few seconds. After the switch, I double checked that I was able to ssh into nodus from my laptop, and Larry informed me that everything is working as expected on his end.
Larry also tells us that the other edgeswitch at the 40m (Foundry Networks), to which most of our GC network machines are connected, is a 100MBPS switch, and so we should re-route the connections from this switch to the BROCADE switch at our convenience to take advantage of the faster connection.
I trained a deep neural network (DNN) to reconstruct MICH and PRCL degrees of freedom in the PRMI configuration. For details on the DNN architecture please refer to G1701455 or G1701589. Or if you really want all the details you can look at the code. I used the following signals as input to the DNN: POPDC, POP22_Q, ASDC, REFL11_I/Q, REFL55_I/Q, AS55_I/Q.
Gautam took some PRMI data in free swinging and driven configuration:
In contrast to the Fabry-Perot cavity case, we don't have a direct measurement of the real PRCL/MICH degrees of freedom, so it's more difficult to assess if the DNN is working well.
All MICH and PRCL values are wrapped into the unique region [-lambda/4, lambda/4]^2. It's even a bit more complicated than simpling wrapping. Indeed, MICH is periodic over [-lambda/2, lambda/2]. However, the Michelson interferometer reflectivity (as seen from PRC) in the first half of the segment is the same as in the second half, except for a change in sign. This change of sign in Michelson reflectivity can be compensated by moving PRCL by lambda/4, thus generating a pi phase shift in the PRC round trip propagation that compensate for the MICH sign change. Therefore, the unit cell of unique values for all signals can be taken as [-lambda/4, lambda/4] x [-lambda/4, lambda/4] for MICH x PRCL. But when we hit the border of the MICH region, PRCL is also affected by addtion of lambda/4. Graphically, the square regions A B C below are all equivalent, as well as more that are not highlighted:
This makes it a bit hard to un-wrap the resonstructed signal, especially when you add in the factor that in the reconstruction the wrapping is "soft".
The plot below shows an example of the time domain reconstruction of MICH/PRCL during the free swinging period.
It's hard to tell if the positions look reasonable, with all the wrapping going on.
Here's an attempt at validating the DNN reconstruction. Using the reconstructed MICH/PRCL signal, I can create a 2d map of the values of the optical signals. I binned the reconstructed MICH/PRCL in a 51x51 grid, and computed the mean value of all optical signals for each bin. The result is shown in the plot below, directly compared with the expectation from a simulation.
The power signals (POP_DC, AS_DC, PO22_Q) looks reasonably good. REFL11_I/Q also looks good (please note that due to an early mistake in my code, I reversed the convention for I/Q, so PRCL signal is maximized in Q instead than in I). The 55MHz signals look a bit less clear...
MC autolocker and FSS loops were stuck because c1psl was unresponsive. I rebooted it and did a burtrestore to enable PSL locking. Then the IMC locked fine.
c1susaux and c1iscaux were also unresponsive so I keyed those crates as well, after taking the usual steps to avoid ITMX getting stuck - but it still got stuck when the Sat. Box. connectors were reconnected after the reboot, so I had to shake it loose with bias slider jiggling. This is annoying and also not very robust. I am afraid we are going to knock the ITMX magnets off at some point. Is this problem indicative of the fact that the ITMX magnets were somehow glued on in a skewed way? Or can we make the situation better by just tweaking the OSEM-holding fixtures on the cage?
In any case, I've started listing stuff down here for things we may want to do when we vent next.
I changed the heater circuit described in this elog to a current sink. The new and old circuits are shown in the attachment. The heater is and is currently 24Ω; the sense resistor is currently 6Ω. The op-amp is still an OP27 and the MOSFET is still an IRF630.
The current through the old circuit was saturating because the gate voltage on the MOSFET was saturating at the op-amp supply rails. This is because the source voltage is relatively high: .
In the new circuit the source voltage is lower and the op-amp can thus drive a large enough to draw more current (until the power supply saturates at 25V/30Ω = 0.8A in this case). The source and DAC voltages are equal in this case and so the current is . Since this is the same current through the heater, the drain voltage is . I observed this behavior in this circuit until the power supply saturated at 0.8A. Note that when this happens and the gate voltage saturates at the supply rails in an attempt to supply the necessary current.
there is a large (~1 cm) aperture Pockels cell that Frank Siefert was using for making pulses to damage photo diodes. There is a DEI Pulser unit near the entrance to the QIL in Bridge which can drive it.
I'll look for it tomorrow, but I haven't given up on the AOMs yet. I swapped in the ISOMET modulator today and saw the same behavior, both in 0th and 1st order. The fall time is pretty much identical. Gautam saw no such thing in the PSL AOM using the same photodetector.
In the meantime I prepared the fiber mode-matching but realized in the process that I had mixed up some lenses. As a result the beam did not have a waist at the AOM location and thus didn't have the intended size, although I doubt that this would cause the slower decay. I'll fix it tomorrow, along with setting up the fiber injection, beat note with the PSL, and routing the fiber if possible.
I don't understand why the 1st order diffracted beam doesn't go to zero when you shut off the drive. My guess is that the standing acoustic wave in the AO crystal needs some time to decay: f = 40 MHz, tau = 1 usec... Q ~ 100. Perhaps, the crystal is damped by the PZT and ther output impedance of the mini-circuits switch is different from the AO driver.
In any case, if you need a faster shut off, or want something that more cleanly goes to zero, there is a large (~1 cm) aperture Pockels cell that Frank Siefert was using for making pulses to damage photo diodes. There is a DEI Pulser unit near the entrance to the QIL in Bridge which can drive it.
I worked with Kevin and Gautam to create a heater circuit. The first attachment is Kevin's schematic of the circuit. The OP amp connects to the gate of the power MOSFET, and the power supply connects to the drain, while the source goes into the heater. We set the power supply voltage to 22V and varied the voltage of the input to the OP amp. At 6V to the OP amp, we got a current of 0.35A flowing through the heater and resistor. This was the peak current we got due to the OP amp being saturated (an increase in either of the power supplies did not change the current), but when we increased the voltage of the supply rails of the OP amp from 15V to 20V, we got a current of 0.5A. We would want a higher current than this, so we will need to get a different OP amp with a higher max voltage rating, and a resistor that can take more power than this one (it currently takes 5W of power, and is the best one we could find).
Kevin and I created a simulation of this circuit using CircuitLab to understand why the current was so low (second attachment). The horizontal axis is the voltage we supply to the OP amp. The blue line shows the voltage at the point between the output of the OP amp and the gate of the MOSFET. The orange line is the voltage at the point between the source of the MOSFET and the heater. The brown line is the voltage at the point between the heater and resistor. Thus, we can see that saturation occurs at about 2.1V. At that point, the gate-source voltage is the difference between the blue curve and the orange curve, which is about 4V, which is what we measured. Likewise, the voltage across the heater is the difference between the orange curve and the brown curve, which comes out to around 8V, which is also what we measured. Lastly, the voltage across the resistor is the brown curve, which is about 2V, which matches our observations. The circuit works as it should, but saturates too soon to get a high enough current out of it.
Gautam noted that it is important to measure the current correctly. We can't just use an ammeter and place it across the resistor or heater, because the internal resistance of the ammeter (~0.5 ohm) is comparable to the resistance we want to measure, so the current gets split between the circuit and the ammeter and we get an equivalent resistance of 1/R = 1/R0 + 1/Ra, where R0 is the resistance of the part we want to measure the current across, and Ra is the ammeter resistance. Thus, the new resistance will be lower and the ammeter will show a higher current value than what is actually there. So to accurately measure the current, we must place the ammeter in series with the part we want to measure. We initially got a 1A reading on the heater, which was not correct, and our setup did not heat up at all basically. When we placed the ammeter in series with the heater, we got only 0.35A.
The last two images are the setup for testing of the heater. We wrapped it around an aluminum piece and covered it with a few layers of insulating material. We can stick a thermometer in between the insulation and heater to see the temperature change. In later tests, we may insulate the whole piece so that less heat gets dissipated. In addition, we used a heat sink and thermal paste to secure the MOSFET to it, as it got very hot.
Our next steps will be to get a resistor and an OP amp that are better suited for our purposes. We will also run simulations with components that we choose to make sure that it can provide the desired current of 1A (the maximum output of the power supply is 24V, and the heater is 24 ohm, so max current is 1A). Kevin is working on that now.
RP1 and RP3 roughing pump manual of Leybold D30A oily rotory pump
Fore pump of TP2 & TP3 Varian SH-100 Dry Scroll
TP2 and TP3 small turbo drag pump Varian 969-9361
TP2 and TP3 turbo controller Varian 969-9505
TP1 magnetically suspended turbo pump Osaka TG390MCAB, sn360 and controller TC010M and note : this pump running on 208VAC single phaseIt is not on the UPS !
Osaka Maglev Manual and Osaka Controller Communication Wiring
VC1 cryo pump CTI-Cryogenics Cryo Torr 8 sn 8g23925 SAFETY note: compressor single phase 208VAC and the head driver 3 phase 208VAC Compressor and driver have each separate power cord!
Installed at 40m wiki also
The V1 gate valve specs installed at 40m wiki page. VAT model number 10846-UE44-0007 Our main volume pumping goes through this 8" id gate valve V1 to Maglev turbo or Cryo pump to VC1
The ion pumps have 6" id gate valves:VAT 10844-UE44-AAY1, Pneumatic actuator with position indicator and double acting solenoid valve 115V 60Hz Purchased 1999 Dec 22
UHV gate valves 2.5" id. VAT 10836-UE44 Pneumatic actuator with position indicator and double acting solenoid valve 115V 60 hz, IFO to RGA VM1 & RGA to Maglev VM2
mini UHV gate valve 1.5" id. VAT 01032-UE01 2016 cataloge page 14, manual - no position indicator, VM4 next to manual adjustable fine leak valve to RGA
UHV angle valve 1.5" id, model VAT 28432-GE41, Viton plate seal, pneumatic actuator with position indicator & solenoid valve 115V & single acting closing spring MEDM screen: VM3,VC2, V3,V4,V5,V6,VA6,V7 & annuloses Each chamber annulos has 2 valves.
UHV angle valve 1.5" id, model VAT 57132-GE05 go page 208, Metal tip seal, manual actuating only with position indicator, MEDM screen: roughing RV1 and venting VV1 hand wheel needed to close to torque spec
UHV angle valve 1.5" id. model VAT 28432-GE01 Viton plate seal, manual operation only at IT gauges Hornet & Super Bee and ion pumps roughing ports. These are not labeled.
The Cryo pump interlock wiring was added too
Note: all moving valve plate seals are single.
Last night, while we were working on the ALS, I noticed the GTRY spot moving around (in PITCH) on the CCD monitor in the control room at ~1-2Hz. The operating condition was that the arm was locked to the IR, and the PSL green shutter was closed, so that only the arm transmissions were visible on the CCD screens. There was no such noticable movement of the GTRX spot. When looking at the out-of-loop ALS nosie in this configuration (but now with the PSL green shutter open of course), the Y arm ALS noise at low frequencies was much worse than the X arm.
Today, I looked into this a little more. I first checked that the Y-endtable enclosure was closed off as usual (as I had done some tweaking to the green input pointing some days ago). There are various green ghost beams on the Y-endtable. When time permits, we should make an effort to cleanly dump these. But the enclosure was closed as usual.
Then I looked at the in-loop Oplev error signal spectra for the ITMY and ETMY Oplev loops. There was high coherence between ETMYP Oplev error signal and GTRY. So I took a loop transfer function measurement - the upper UGF was around 3.5Hz. I increased the loop gain such that the upper UGF was around 4.5Hz, with phase margin ~30degrees. Doing so visibly reduced the angular movement of the GTRY spot on the CCD. Attachment #1 shows the Oplev loop TF after the gain increase, while Attachment #2 compares the GTRX and GTRY spectra (DC value is approximately the same for both, around 0.4). GTRY still seems a bit noisier at low frequencies, but the out-of-loop ALS noise for the Y arm now lines up much more closely with its reference trace from a known good time.
Y-arm ALS wasn't so stellar tonight, especially at low frequencies. I can see the GTRY spot moving on the CCD monitor, so something is wonky. To be investigated.
I was having a chat with EricQ about this today, just noting some points from our discussion down here so that I remember to look into this tomorrow.
Can we make use of the Jetstor raid array for some kind of consolidated 40m CDS backup system? Once we've gotten everything of interest out of it...
It is unclear when this was last done, and since I modified the coil driver electronics for the ITMs and BS recently, I figured it would be useful to get this calibration done. The primary motivation was to see if we could resolve the discrepancy between the current ALS noise (using POX as a sensor) compared to the Izumi et. al. plot.
Because we are planning to change the coil driver electronics further soon anyways, we decided to do the calibration at a single frequency for tonight. For future reference, the extension of this method to calibrate the actuator over a wider range of frequencies is here. The procedure followed, and the relevant numbers from tonight, are as follows.
Once these calibrations were updated, we decided to control the arms with ALS, and look at the POX spectrum. Y-arm ALS wasn't so stellar tonight, especially at low frequencies. I can see the GTRY spot moving on the CCD monitor, so something is wonky. To be investigated. But the X arm ALS noise looked pretty good.
Seems like updating the calibration did the job; see the attached comparison plot.
I moved the axuiliary NPRO to the PSL table today and started setting up the optics.
The Faraday Isolator was showing a pretty unclean mode at the output so I took the polarizers off to take a look through them, and found that the front polarizer is either out of place or damaged (there is a straight edge visible right in the middle of the aperture, but the way the polarizer is packaged prevents me from inspecting it closer). I proceeded without it but left space so an FI can be added in the future. The same goes for the broadband EOM.
There are two spare AOMs (ISOMET and Intraaction, both resonant at 40MHz) available before we have to resort to the one currently installed in the PSL.
I installed the Intraaction AOM first and looked at the switching speed of its first order diffracted beam using both its commercial driver and a combination of minicircuits components. Both show similar behavior. The fall time of the initial step is ~110ns in both cases, but it doesn't decay rapidly no light but a slower exponential. Need to check the 0 order beam and also the other AOM.
In addition to bootable full disk backups, it would be wise to make sure the important service configuration files from each machine are version controlled in the 40m SVN. Things like apache files on nodus, martian hosts and DHCP files on chiara, nds2 configuration and init scripts on megatron, etc. This can make future OS/hardware upgrades easier too.
What are the critical filesystems? I've also indicated the size of these disks and the volume currently used, and the current backup situation.
Not backed up
LDAS pulls files from nodus daily via rsync, so there's no cron job for us to manage. We just allow incoming rsync.
Local backup on /media/40mBackup on chiara via daily cronjob
Remote backup to ldas-cit.ligo.caltech.edu::40m/cvs via daily cronjob on nodus
Currently mounted on Megatron, not backed up.
Then there is Optimus, but I don't think there is anything critical on it.
So, based on my understanding, we need to back up a whole bunch of stuff, particularly the boot disks and root filesystems for Chiara, Megatron and Nodus. We should also test that the backups we make are useful (i.e. we can recover current operating state in the event of a disk failure).
Please edit this elog if I have made a mistake. I also don't have any idea about whether there is any sort of backup for the slow computing system code.
The RGA was turned on 7 days ago. It's 46 C now. The X-arm room tem ~20 C
IFO pressure 6.5e-6 Torr at IT-Hornet gauge. Valve configuration vacuum normal.
It seems like the main contribution to the RMS comes from the high frequency bump. When using the ALS loop to lock the arm to the beat, only the stuff below ~100 Hz will matter. Interesting to see what that noise budget will show. Perhaps the discrepancy between inloop and out of loop will go down.
I included the 55 MHz sideband and higher order modes in my training examples. To keep things simple, I just assumed there are higher order modes up to n+m=4 in the input beam. The power in each HOM is randomly chosen from a random gaussian distribution with width determined from experimental cavity scans. I used a value of 0.913+-0.01 rad for the Gouy phase (again estimated from cavity scans, but in reasonable agreement with the nominal radius of curvature of ETMX)
Results are improved. The plot belows show the performance of the neural network on 100s of experimental data
For reference, the plots below show the performance of the same network on simulated data (that includes sensing noise but no higher order modes)
Is it better to mount the box in the PSL under the existing shelf, or in a nearby PSL rack?
Further characterization needs to be done, but the results of this test are encouraging. If we are able to get this kind of out of loop ALS noise with the IR beat, perhaps we can avoid having to frequently fine-tune the green beat alignment on the PSL table. It would also be ideal to mount this whole 1U setup in an electronics rack instead of leaving it on the PSL table
Attachment #1 - Photo of the revamped beat setup. The top panel has to be installed. New features include:
Attachment #2 - Power budget inside the box. Some of these FC/APC connectors seem to not offer good coupling between the two fibers. Specifically, the one on the front panel meant to accept the PSL light input fiber seems particularly bad. Right now, the PSL light is entering the box through one of the front panel connectors marked "PSL + X out". I've also indicated the beat amplitude measured with an RF analyzer. Need to do the math now to confirm if these match the expected amplitudes based on the power levels measured.
Attachment #3 - We repeated the measurement detailed here. The X arm (locked to IR) was used for this test. The "X" delay line electronics were connected to the X green beat PD, while the "Y" delay line electronics were connected to the X IR beat PD. I divided the phase tracker Hz calibration factor by 2 to get IR Hz for the Y arm channels. IR beat was at ~38MHz, green beat was at ~76MHz. The broadband excess noise seen in the previous test is no longer present. Indeed, below ~20Hz, the IR beat seems less noisy. So seems like the cleaning / electronics revamp did some good.
Further characterization needs to be done, but the results of this test are encouraging. If we are able to get this kind of out of loop ALS noise with the IR beat, perhaps we can avoid having to frequently fine-tune the green beat alignment on the PSL table. It would also be ideal to mount this whole 1U setup in an electronics rack instead of leaving it on the PSL table.
Photos + power budget + plan of action for using this box to characterize the green PDH locking to follow.
GV Edit: I've added better photos to the 40m Google Photos page. I've also started a wiki page for this box / the proposed IR ALS system. For the moment, all that is there is the datasheet to the Fiber Couplers used, I will populate this more as I further characterize the setup.
Looks like MC1 got another big kick just under 4 hours ago. None of the other optics show any evidence of a glitch so it seems unlikely that this was some sort of global event. It's been well behaved for ~2weeks now. IMC was unlocked. I manually re-aligned MC1, at which point the autolocker was able to lock the IMC.
Looking at this plot, it seems that LR and UL coils seem to have the largest kicks. UR barely saw it. Not sure what (if anything) to make of this - apparently the optic moved by ~20urad with the UR magnet approximately the pivot.
I tried some DRMI locking again tonight, but had no success. Here is the story.
Looks like I will have to embark on the REFL55 LSC electronics investigation. I was able to successfully lock the PRC on carrier and sideband, and the Michelson lock also seems to work fine, all of which seem to point to a hardware problem with the REFL55 signal chain.
In brief, I trained a deep neural network (DNN) to recosntuct the cavity length, using as input only the transmitted power and the reflection PDH signals. The training was performed with simulated data, computed along 0.25s long trajectories sampled at 8kHz, with random ending point in the [-lambda/4, lambda/4] unique region and with random velocity.
The goal of thsi work is to validate the whole approach of length reconstruction witn DNN in the Fabry-Perot case, by comparing the DNN reconstruction with the ALS caivity lenght measurement. The final target is to deploy a system to lock PRMI and DRMI. Actually, the Fabry-Perot cavity problem is harder for a DNN: the cavity linewidth is quite narrow, forcing me to use very high sampling frequency (8kHz) to be able to capture a few samples at each resonance crossing. I'm using a recurrent neural network (RNN), in the input layers of the DNN, and this is traine using truncated backpropagation in time (TBPT): during training each layer of RNN is unrolled into as many copies as there are input time samples (8192 * 0.25 = 2048). So in practice I'm training a DNN with >2000 layers! The limit here is computational, mostly the GPU memory. That's why I'm not able to use longer data stretches.
But in brief, the DNN reconstruction is performing well for the first attempt.
In the results shown below, I'm using a pre-trained network with parameters that do not match very well the actual data, in particular for the distribution of mirror velocity and the sensing noises. I'm working on improving the training.
I used the following parameters for the Fabry-Perot cavity:
The uncertaint is assumed to be the 90% confidence level of a gaussian distribution. The DNN is trained on 100000 examples, each one a 0.25/8kHz long trajectory with random velocity between 0.1 and 5 um/s, and ending point distributed as follow: 33% uniform on the [-lambda/4, lambda/4] region, plus 33% gaussian distribution peaked at the center with 5 nm width. In addition there are 33% more static examples, distributed near the center.
For each point along the trajectory, the signals TRA, POX11_I and POX11_Q are computed and used as input to the DNN.
Gautam collected about 10 minutes of data with the free swinging cavity, with ALS locked on the arm. Some more data were collected with the cavity driven, to increase the motion. I used the driven dataset in the analysis below.
The ALS signal is calibrated in green Hz. After converting it to meters, I checked the calibration by measuring the distance between carrier peaks. It turned out that the ALS signal is undercalibrated by about 26%. After correcting for this, I found that there is a small non-linearity in the ALS response over multiple FSR. So I binned the ALS signal over the entire range and averaged the TRA power in each bin, to get the transmission signals as a function of ALS (in nm) below:
I used a peak detection algorithm to extract the carrier and 11 MHz sideband peaks, and compared them with the nominal positions. The difference between the expected and measured peak positions as a function of the ALS signal is shown below, with a quadratic fit that I used to improve the ALS calibration
The result is
The ALS calibrated z error from the peak position is of the order of 3 nm (one sigma)
Using the calibrated ALS signal, I computed the cavity length velocity. The histogram below shows that this is well described by a gaussian with width of about 3 um/s. In my DNN training I used a different velocity distribution, but this shouldn't have a big impact. I'm retraining with a different distirbution.
The plot below shows a stretch of time domain DNN reconstruction, compared with the ALS calibrated signal. The DNN output is limited in the [-lambda/4, lambda/4] region, so the ALS signal is also wrapped in the same region. In general the DNN reconstruction follows reasonably well the real motion, mostly failing when the velocity is small and the cavity is simultanously out of resonance. This is a limitation that i see also in simulation, and it is due to the short training time of 0.25s.
I did not hand-pick a good period, this is representative of the average performance. To get a better understanding of the performance, here's a histogram of the error for 100 seconds of data:
The central peak was fitted with a gaussian, just to give a rough idea of its width, although the tails are much wider. A more interesting plot is the hisrogram below of the reconstructed position as a function of the ALS position, Ideally one would expect a perfect diagonal. The result isn't too far from the expectation:
The largest off diagonal peak is at (-27, 125) and marked with the red cross. Its origin is more clear in the plot below, which shows the mean, RMS and maximum error as a function of the cavity length. The second peak corresponds to where the 55 MHz sideband resonate. In my training model, there were no 55 MHz sidebands nor higher order modes.
The DNN reconstruction performance is already quite good, considering that the DNN couldn't be trained optimally because of computation power limitations. This is a validation of the whole idea of training the DNN offline on a simulation and then deploy the system online.
I'm working to improve the results by
However I won't spend too much time on this, since I think the idea has been already validated.
A couple of weeks ago, I was trying to modernize the python version of the FSS Slow temperature control loops, when I accidentally ended up deleting it . There was no svn backup. So the old Perl PID script has been running for the last few days.
Today, I checked out the latest version that Andrew and co. have running in the PSL lab. I had to make some important modifications for the script to work for the 40m setup.
python FSSSlow.py -i FSSSlowPy.ini
Then I stopped the Perl process on megatron by running
sudo initctl stop FSSslow
and started the Python process by running
sudo initctl start FSSslowPy
I have now committed the files FSSSlow.py and FSSSlowPy.ini to the 40m svn. Things seem to be stable for the last 20 mins or so, let's keep an eye on this though - although we had been running the Python PID loop for some months, this version is a slightly modified one.
The initctl stuff still isn't very robust - I think both the Autolocker and the FSS slow servos have to be manually restarted if megatron is shutdown/restarted for whatever reason. It doesn't seem to be a problem with the initctl routine itself - looking at the logs, I can see that init is trying to start both processes, but is failing to do so each time. To be investigated. The wiki procedure to restart this process is up to date.
GV Edit 0000 25 Aug 2017: I had to add a line to the script that checks MC transmission before enabling the PID loop. Change has been committed to svn. Now, when the MC loses lock or if the PSL shutter is kept closed for an extended period of time, the temperature loop doesn't rail.
Since the single arm locking and dither alignment seemed to work alright after the CDS overhaul, I decided to try some recycling cavity locking tonight.
Why should this have changed? I was just on the AS table and did re-center the beam onto the REFL 55 RFPD, but I had also done this in April/May when I was last doing DRMI locking. But I can't explain the apparent factor of ~4 increase in light level. I think I have some measurements of the light levels at various PDs from April 2017, I will see how the present levels line up.
Of course dataviever won't cooperate when I am trying to monitor testpoints.
I may be missing something obvious, but I am quitting for tonight, will look into this more tomorrow.
Unrelated to this work: looking at the GTRY spot on the CCD monitor, there seems to be some excess angular motion. Not sure where this is coming from. In the past, this sort of problem has been symptomatic of something going wonky with the Oplev loops. But I took loop measurements for ITMY and ETMY PIT and YAW, they look normal. I will investigate further when I am doing some more ALS work.
I completed the revamp of the box, and re-installed the box on the PSL table today. I think it would be ideal to install this on one of the electronic racks, perhaps 1X2 would be best. We would have to re-route the fibers from the PSL table to 1X2, but I think they have sufficient length, and this way, the whole arrangement is much cleaner.
Did a quick check to make sure I could see beat notes for both arms. I will now attempt to measure the ALS noise with this revamped box, to see if the improved power supply and grounding arrangement, as well as fiber cleaning, has had any effect.
For quick reference: here is the AM/PM measurement done when we re-installed the repaired Innolight NPRO on the new X endtable.
Didn't someone look at what the OLG req. should be for these servos at some point? I wonder if we can make a parallel digital path that we switch on after green lock. Then we could make this a simple 1/f box and just add in the digital path (take analog control signal into ADC, filter, and then sum into the control point further down the path to the laser) for the low frequency boost.
After getting the go ahead from Jamie, I recompiled all the FE models against the same version of RCG that we tested on the c1iscex models.
To do so:
IFO alignment needs to be redone, but at least we now have a (admittedly rounabout way) of getting testpoints. Did a quick check for "nan-s" on the ASC screen, saw none. So I am re-enabling watchdogs for all optics.
GV 23 August 9am: Last night, I re-aligned the TMs for single arm locks. Before the model restarts, I had saved the good alignment on the EPICs sliders, but the gain of x3 on the coil driver filter banks have to be manually turned on at the moment (i.e. the safe.snap file has them off). ALS noise looked good for both arms, so just for fun, I tried transitioning control of both arms to ALS (in the CARM/DARM basis as we do when we lock DRFPMI, using the Transition_IR_ALS.py script), and was successful.
Here is what was done (Jamie will correct me if I am mistaken).
So while we are in a better state now, the problem isn't fully solved.
Comment: seems like there is an in-built timeout for testpoints opened with DTT - if the measurement is inactive for some time (unsure how much exactly but something like 5mins), the testpoint is automatically closed.
I surveyed the lab today to see what we may need to buy for the AS laser setup.
NPRO 200 mW + Driver
Faraday Isolator from cabinet
ISOMET Model 1201E: This is a free space AOM I found in the modulator cabinet. It needs to be driven at 40MHz (to be confirmed) with ~6W of electrical power. For a 500 micron beam it can allegedly achieve rise times of '93' [units not specified, could this be nanoseconds?]. I did not find a dedicated driver for it, however there was a 5W minicircuits amplifier ZHL-5W-1 in the RF cabinet and a switch ZSDR-230, which has a typical switch time of 2 microseconds, however I'm not sure how this translates to rise/fall times of the deflected power. It seems we have everything to set this up, so we'll by the end of the week if we can use a combination of these things or if we need to buy additional driver electronics.
New Focus model 4004 broadband phase modulator which is labeled as dusty, and in fact quite dirty when looking through. We should attempt to clean this thing and maybe we can use it here or at the ends.
Probably all the optics we need for the PSL table setup.
Beat PD: How about one of these: EOT ET-3000A? I didn't find a broadband PD for the beat with the PSL
Fiber Stuff: coupler & polarization maintaining fiber 20m & collimator. There are a couple options here, which we can discuss in the meeting.
Faraday Isolator: If we want to inject P-polarization. If S is okay we can use a polarizing plate beamsplitter instead.
Possibly some large lenses for mode-matching to IFO (TBD)
I had some trouble getting the daqd processes up and running again using Jamie's instructions.
With Jamie's help however, they are back up and running now. The problem was that the mx infrastructure didn't come back up on its own. So prior to running sudo systemctl restart daqd_*, Jamie ran sudo systemctl start mx. This seems to have done the trick.
c1iscey was still showing red fields on the CDS overview screen so Jamie did a soft reboot. The machine came back up cleanly, so I restarted all the models. But the indicator lights were still red. Apparently the mx processes weren't running on c1iscey. The way to fix this is to run sudo systemctl start mx_stream. Now everything is green.
Now we are going to work on trying the fix Rolf suggested on c1iscex.
It turns out the problem was just a bent pin on the SCSI cable, likely from having to stretch things a bit to reach optimus from the RAID unit.
I hooked it up to megatron, and it was automatically recognized and mounted.
I had to turn off the new FB machine and remove it from the rack to be able to access megatron though, since it was just sitting on top. FB needs a rail to sit on!
At a cursory glance, the filesystem appears intact. I have copied over the achived DRFPMI frame files to my user directory for now, and Gautam is going to look into getting those permanently stored on the LDAS copy of 40m frames, so that we can have some redundancy.
Also, during this time, one of the HDDs in the RAID unit failed its SMART tests, so the RAID unit wanted it replaced. There were some spare drives in a little box directly under the unit, so I've installed one and am currently incorporating it back into the RAID.
There are two more backup drives in the box. We're running a RAID 5 configuration, so we can only lose one drive at a time before data is lost.
Attachment #1 shows the results of my measurements tonight (SR785 data in Attachment #2). Both loops have a UGF of ~10kHz, with ~55 degrees of phase margin.
Excitation was injected via SR560 at the PDH error point, amplitude was 35mV. According to the LED indicators on these boxes, the low frequency boost stages were ON. Gain knob of the X end PDH box was at 6.5, that of the Y end PDH box was at 4.9. I need to check the schematics to interpret these numbers. GV Edit: According to this elog, these numbers mean that the overall gain of the X end PDH box is approx. 25dB, while that of the Y end PDH box is approx. 15dB. I believe the Y end Lightwave NPRO has an actuator discriminant ~5MHz/V, while the X end Innolight is more like 1MHz/V.
Not sure what to make of the X PDH loop measurement being so much noisier than the Y end, I need to think about this.
More detailed analysis to follow.
I am now going to measure the OLTFs of both green PDH loops to check that the overall loop gain is okay, and also check the measurement against EricQ's LISO model of the (modified) AUX green PDH servos. Results to follow.
I worked a little bit on the Y arm ALS today.
Some weeks ago, I had moved some of the Green steering optics on the PSL table around, in order to flip some mirror mounts and try and get angles of incidence closer to ~45deg on some of the steering mirrors. As a result of this work, I can see some light on the GTRY CCD when the X green shutter is open. It is unclear if there is also some scattered light on the RFPDs. I will post pictures + a more detailed investigation of the situation on the PSL table later, there are multiple stray green beams on the PSL table which should probably be dumped.
As I was writing this elog, I saw the X green lock drop abruptly. During this time, the X arm stayed locked to the IR, and the Y arm beat on the control room network analyzer did not jump (at least not by an amount visible to the eye). Toggling the X end shutter a few times, the green TEM00 lock was re-acquired, but the beatnote has moved on the control room analyzer by ~40MHz. On Friday evening however, the X green lock held for >1 hour. Need to keep an eye on this.
In case you want to use it, I had profiled the Lightwave NPRO sometime back, and we were even using it as the AUX X laser for a short period of time.
As for using the AS laser for mode spectroscopy: don't we want to match the beam into the cavity as best as possible, and then use some technique to disturb the input mode (like the dental tooth scraper technique from Chris Mueller's thesis)?
Johannes and I did an arm scan of the X arm today (arm controlled with ALS, monitoring IR transmission) - only 2 IR FSRs were scanned, but there should be sufficient information in there to extract the modulation depth and mode matching - can we use Kaustubh's/Naomi's code?. The Y arm ALS needs to be touched up so I don't have a Y arm scan yet. Note that to get a good arm scan measurement, the High Gain Thorlabs PD should be used as the transmission PD.
There are three methods we (will soon) have available to evaluate the round-trip dissipative losses in the arms that do not suffer from the ITM loss dominance:
The DC method comparing reflectivities has been used in the past and is relatively easy to do. After the recent vacuum troubles the first step should be to re-perform these as CDS permits (needs some ASS functionality and of course the MC to behave). It wouldn't hurt to know the parameters this depends on, aka mode overlap and modulation depths with better certainty. Maybe the SURF scripts for mode-spectroscopy can be applied?
With the new CCD cameras calibrated, pre-vent we can determine the magnitude of the large-angle scatter loss (assuming isotropic scatter) of ETMX and possibly ETMY. Can we look past ETMX/ETMY from the viewports? Then we can probably also look at the small angle scatter of ITMX and ITMY. If not, once we open one of the chambers there's the option of installing mirrors as close as possible to the main beam path. The easiest is probably to look at ITMX, since there is plenty of space in the XEND chamber, and the camera is already installed.
This requires a lot of up-front work. We decided to use the spare 200mW NPRO. It will be placed on the PSL table and injected into an optical fiber, which terminates on the AS table. The again free space beam there needs to be sort-of mode-matched into the SRC ("sort-of" because mode-spectroscopy). We want to be able to phaselock this secondary beam to the PSL with at least a couple kHz bandwidth and also completely extinguish the beam on time-scales of a few microseconds. We will likely need to purchase a few components that we can salvage from other labs, I'm still going through the inventory and will know more soon (more detailed post to follow). We need to settle for the polarization we want to send in from the back.
At Rolf/Rich Abbott's request, we performed a check of the UPS today.
Steve believed that the UPS was functioning as it should, and the recent accidental vent was because the UPS batteries were insufficiently charged when the test was performed. Today, we decided to try testing the UPS.
We first closed V1, VM1 and VA6 using the MEDM screen. We prepared to pull power on all these valves by loosening the power connections (but not detaching them). [During this process, I lost the screw holding the power cord fixed to the gate valve V1 - we are looking for a replacement right now but it seems to be an odd size. It is cable tied for now.]
The battery charge indicator LEDs on the UPS indicated that the batteries were fully charged.
Next, we hit the "Test" button on the UPS - it has to be held down for ~3 seconds for the test to be actually initiated, seems to be a safety feature of the UPS. Once the test is underway, the LED indicators on the UPS will indicate that the loading is on the UPS batteries. The test itself lasts for ~5seconds, after which the UPS automatically reverts to the nominal configuration of supplying power from the main line (no additional user input is required).
In this test, one of the five battery charge indicator LEDs went off (5 ON LEDs indicate full charge).
So on the basis of this test, it would seem that the UPS is functioning as expected. It remains to be investigated if the various hardware/software interlocks in place will initiate the right sequence of valve closures when required.
Never hit O on the Vacuum UPS !
Note: the " all off " configuration should be all valves closed ! This should be fixed now.
In case of emergency you can close V1 with disconnecting it's actuating power as shown on Atm3 if you have peumatic pressure 60 PSI
In the aftermath of the accidental vent, it looks like the RGA was shutdown.
We followed the instructions in this elog to restart the RGA.
Seems to be working now, Steve says we just need to wait for it to warm up before we can collect a reliable scan.
We have good RGA scan now. There was no scan for 3 months.
On Friday, I cleaned up the circuit so that there are only three connections needed (+15V, -15V, GND) and a BNC connector for reading the output. Today, I added in bypass capacitors. The small yellow ones are 0.1 microF ceramic, and the large ones are 100 microF electrolytic. They are used to stabilize the +15V and -15V inputs to the OP amp and minimize fluctuations, since it doesn't have a regulator for stability. I have also attached the circuit diagram for the OP amp only, where 1 are the electrolytic and 2 are the ceramic. The temperature is still about 2 degrees off, but if that difference is constant for all temperatures in our range we can just calibrate it later.
Here is a helpful link on bypass capacitors (thanks to Kevin for sending it to me).
As a note, the electrolytic capacitors do have a polarity, so it is important to place them correctly (the negative side is towards the lower voltage potential, and not always towards ground).
Got it to work. One of the connections was faulty. I decided to check the temperature measured against a thermometer. The sensor showed 26.1 C, but the thermometer showed 25.8 C after I let them both cool down after heating them up. The temperature of the thermometer was dropping at the time of measurement, but the temperature of the sensor was not. This is still a rough version of the final sensor, so I'm not sure what exactly causes this discrepancy.
Tried taking the circuit from the breadboard to the PCB. I attached all the components to adapters that would allow them to be connected to the PCB. From the first picture, the first component is AD586, the second is AD590, and the third is LT1012, along with a resistor across it. I then soldered the connections between the components, as can be seen in the second picture. When I tested out this version of the circuit by hooking it up to the DC source, I got a reading of ~-15V. I will have to check all the connections to make sure there is contact where there should be one, and no contact where there shouldn't be. I had issues attaching the tiny AD590 and LT1012 to its adaptor, so the issue may lie there as well. I'll also check that each component is in working order as well.
Once I figure out where my error is, my plan is to build two more of these and place a metal object such that it contacts only the surface of the AD590s. This would allow me to compare the three values to the actual temperature of the metal, which would then tell me how accurate this setup is.
Note on the resistor: I measured all the resistors and chose three that had exactly 10.00k Ohm. The voltage detected is dependent on the resistor, so if we are to take three identical copies, I ensured that there would be no error due to the resistors being a little different.
They are synchronised tiny glitches. They are not mechanical.
My motivation tonight was to get an up-to-date spectrum of a calibrated measurement of the out-of-loop displacement of an arm locked on ALS (using the PDH signal as the out-of-loop sensor) to compare the performance of ALS control noise with the Izumi et al green locking paper.
I was able to fish out the PSD from the paper from the 40m svn, but the comparison as plotted looks kind of fishy. I don't see why the noise from 10-60Hz should be so different/worse. We updated the POX counts to meters conversion by looking at the Hz-calibrated ALSX signal and a ~800Hz line injected on ETMX.
Leaving LSC mode OFF for now while CDS is still under investigation
Not really related to this work: We saw that the safe.snap file for c1oaf seems to have gotten overwritten at some point. I restored the EPICS values from a known good time, and over-wrote the safe.snap file.