Today I tested the remaining Acromag channels and retested the non-functioning channels found yesterday, which Chub repaired this morning. We're still not quite ready for an in situ test. Here are the issues that remain.
I further diagnosed these channels by connecting a calibrated DC voltage source directly to the ADC terminals. The EPICS channels do sense this voltage, so the problem is isolated to the wiring between the ADC and DB37 feedthrough.
No output signal
To further diagnose these channels, I connected a voltmeter directly to the DAC terminals and toggled each channel output. The DACs are outputting the correct voltage, so these problems are also isolated to the wiring between DAC and feedthrough.
In testing the DC bias channels, I did not check the sign of the output signal, but only that the output had the correct magnitude. As a result my bench test is insensitive to situations where either two degrees of freedom are crossed or there is a polarity reversal. However, my susPython scripting tests for exactly this, fetching and applying all the relevant signal gains between pitch/yaw input and coil bias output. It would be very time consuming to propagate all these gains by hand, so I've elected to wait for the automated in situ test.
For the new c1susaux, Gautam and I moved the watchdog channels from autoBurt.req to a new file named autoBurt_watchdogs.req. When the new modbus service starts, it loads the state contained in autoBurt.snap. We thought it best for the watchdogs to not be automatically enabled at this stage, but for an operator to manually have to do this. By moving the watchdog channels to a separate snap file, the entire SUS state can be loaded while leaving just the watchdogs disabled.
This same modification should be made to the ETMX and ETMY machines.
For the in-situ test, I decided that we will use the physical SRM to test the c1susaux Acromag replacement crate functionality for all 8 optics (PRM, BS, ITMX, ITMY, SRM, MC1, MC2, MC3). To facilitate this, I moved the backplane connector of the SRM SUS PD whitening board from the P1 connector to P2, per Koji's mods at ~5:10PM local time. Watchdog was shutdown, and the backplane connectors for the SRM coil driver board was also disconnected (this is interfaced now to the Acromag chassis).
I had to remove the backplane connector for the BS coil driver board in order to have access to the SRM backplane connector. Room in the back of these eurocrate boxes is tight in the existing config...
At ~6pm, I manually powered down c1susaux (as I did not know of any way to turn off the EPICS server run by the old VME crate in a software way). The point was to be able to easily interface with the MEDM screens. So the slow channels prefixed C1:SUS-* are now being served by the Supermicro called c1susaux2.
A critical wiring error was found. The channel mapping prepared by Johannes lists the watchdog enable BIO channels as "C1:SUS-<OPTIC>_<COIL>_ENABLE", which go to pins 23A-27A on the P1 connector, with returns on the corresponding C pins. However, we use the "TEST" inputs of the coil driver boards for sending in the FAST actuation signals. The correct BIO channels for switching this input is actually "C1:SUS-<OPTIC>_<COIL>_TEST", which go to pins 28A-32A on the P1 connector. For todays tests, I voted to fix this inside the Acromag crate for the SRM channels, and do our tests. Chub will unfortunately have to fix the remaining 7 optics, see Attachment #1 for the corrections required. I apportion 70% of the blame to Johannes for the wrong channel assignment, and accept 30% for not checking it myself.
The good news: the tests for the SRM channels all passed!
Additionally, I confirmed that the watchdog tripped when the RMS OSEM PD voltage exceeded 200 counts. Ideally we'd have liked to test the stability of the EPICS server, but we have shut it down and brought the crate back out to the electronics bench for Chub to work on tomorrow.
I restarted the old VME c1susaux at 915pm local time as I didn't want to leave the watchdogs in an undefined state. Unsurprisingly, ITMY is stuck. Also, the BS (cable #22) and SRM (cable #40) coil drivers are physically disconnected at the front DB15 output because of the undefined backplane inputs. I also re-opened the PSL shutter.
We briefly talked about the bounce and roll modes of the SOS optic at the meeting today.
Attachment #1: BR modes for ETMY from my free-swinging run on 17 April. The LL coil has a very different behavior from the others.
Attachment #2: BR modes for ETMY from my free-swinging run on 18 April, which had a macroscopically different bias voltage for the PIT/YAW sliders. Here too, the LL coil has a very different behavior from the others.
Attachment #3: BR modes for ETMX from my free-swinging run on 27 Feb. There are many peaks in addition to the prominent ones visible here, compared to ITMY. The OSEM PD noise floor for UR and SIDE is mysteriously x2 lower than for the other 3 OSEMs???
In all three cases, a bounce mode around 16.4 Hz and a roll mode around 24.0 Hz are visible. The ratio between these is not sqrt(2), but is ~1.46, which is ~3% larger. But when I look at the database, I see that in the past, the bounce and roll modes were in fact at close to these frequencies.
Because of my negligence and rushing the closeout procedure, I don't have a great close-out picture of the magnet positions in the face OSEMs, the best I can find is Attachment #4. We tried to replicate the OSEM arrangement (orientation of leads from the OSEM body) from July 2018 as closely as possible.
I will investigate the side coil actuation strength tomorrow, but if anyone can think of more in-air tests we should do, please post your thoughts/poetry here.
Today we installed the c1susaux Acromag chassis and controller computer in the 1X4 rack. As noted in 14580 the prototype Acromag chassis had to first be removed to make room in the rack. The signal feedthroughs were connected to the eurocrates by 10' DB-37 cables via adapters to 96-pin DIN.
Once installed, we ran a scripted set of suspension actuation tests using PyIFOTest. BS, PRM, SRM, MC1, MC2, and MC3 all passed these tests. We were unable to test ITMX and ITMY because both appear to be stuck. Gautam will shake them loose on Monday.
Although the new c1susaux is now mounted in the rack, there is more that needs to be done to make the installation permanent:
On Monday we plan to continue with additional scripted tests of the suspensions.
gautam - some more notes:
A concern was raised about the two ETMs and ITMX having the opposite response (relative to the other 7 SOS optics) in the OSEM PDmon channel in response to a given polarity of PIT/YAW offset being applied to the coils. Jon has factored into account all the digital gains in the actuation part of the CDS system in making this conclusion. I raised the possibility of the OSEM coil winding direction being opposite on the 15 OSEMs of the ETMs and ITMX, but I think it is more likely that the magnets are just glued on opposite to what they are "supposed" to be. See Attachment #6 of this elog (you'll have to rotate the photo either in your head or in your viewer) and note that it is opposite to what is specified in the assembly procedure, page 8. The net magnetic quadrupole moment is still 0, but the direction of actuation in response to current in the coil in a given direction would be opposite. I can't find magnet polarities for all the 10 SOS optics, but this hypothesis fits all the evidence so far..
Yesterday Gautam and I ran final tests of the eight suspensions controlled by c1susaux, using PyIFOTest. All of the optics pass a set of basic signal-routing tests, which are described in more detail below. The only issue found was with ITMX having an apparent DC bias polarity reversal (all four front coils) relative to the other seven susaux optics. However, further investigation found that ETMX and ETMY have the same reversal, and there is documentation pointing to the magnets being oppositely-oriented on these two optics. It seems likely that this is the case for ITMX as well.
I conclude that all the new c1susaux wiring/EPICS interfacing works correctly. There are of course other tests that can still be scripted, but at this point I'm satisfied that the new Acromag machine itself is correctly installed. PyIFOTest has been morphed into a powerful general framework for automating IFO tests. Anything involving fast/slow IO can now be easily scripted. I highly encourage others to think of more applications this may have at the 40m.
The code is currently located in /users/jon/pyifotest although we should find a permanent location for it. From the root level it is executed as
$ ./IFOTest <PARAMETER_FILE>
where PARAMETER_FILE is the filepath to a YAML config file containing the test parameters. I've created a config file for each of the suspended optics. They are located in the root-level directory and follow the naming convention SUS-<OPTIC>.yaml.
The code climbs a hierarchical "ladder" of actuation/readback-paired tests, with the test at each level depending on signals validated in the preceding level. At the base is the fast data system, which provides an independent reference against which the slow channels are tested. There are currently three scripted tests for the slow SUS channels, listed in order of execution:
I found the 8 vertex watchdogs tripped today morning. The ETMs were fine, suggesting this was not an actual earthquake. I suspect it was connected to this remote work? Was there a reason why they were left tripped?
On a side note - I don't think we log the watchdog state explicitly. We can infer whether the optic is damped by looking at the OSEM sensor time series, but do we want to record the watchdog state to frames?
Chub and I crossed off some of these items today morning. The last bullet was addressed by Jon yesterday. I added a couple of new bullets.
The new power connectors will arrive next week, at which point we will install them. Note that there is no 24V Sorensen available, only 20V.
I am running a test on the 2W Mephisto for which I wanted the diagnostics connector plugged in again and Acromag channels to record them. So we set up the highly non-ideal but temporary set up shown in Attachment #1. This will be cleaned up by Monday evening latest.
update 1630 Monday 5/6: the sketchy PSL acromag setup has been disassembled.
Yes, this was a consequence of the systemd scripting I was setting up. Unlike the old susaux system, we decided for safety NOT to allow the modbus IOC to automatically enable the coil outputs. Thus when the modbus service starts/restarts, it automatically restores all state except the watchdog channels, which are left in their default disabled state. They then have to be manully enabled by an operator, as I should have done after finishing testing.
I collected some free-swinging data from earlier today evening. There are still only 3 peaks visible in the ASDs, see Attachment #1.
Plan for tomorrow:
TBH, I don't have any clear ideas as to what we are supposed to do to to fix the problem (or even what the problem is). So here is my plan for now:
I anticipate that these will throw up some more clues
I setup the usual mini-cleanroom setup around the ETMY chamber. Then I carried out the investigative plan outlined here.
Main finding: I saw a fiber of what looks like first contact on the bottom left (as viewed from HR side) of ETMY, connecting the optic to the cage. See Attachment #1. I don't know that this can explain the problem with the missing eigenmode, it's not a hard constraint. Seems like something that should be addressed in any case. How do we want to remove this? Just use a tweezer and pull it off, or apply a larger FC patch and then pull it off? I'm pretty sure it's first contact and not a piece of PEEK mesh because I can see it is adhered to the HR side of the optic, but couldn't capture that detail in a photo.
There weren't any obvious problem with the magnet positioning inside the OSEM, or the suspension wire. All the EQ stop tips were >3mm away from the optic.
I also backed out the bottom EQ stops on the far (south side) of the optic by ~2 full turns of the screw. Taking another free-swinging dataset now to see if anything has changed. I will upload all the photos I took, with annotations, to the gPhotos later today eve. Light doors back on at ~1730.
Update 10pm: the photos have been uploaded. I've added a "description" to each photo which should convey the message of that particualr shot, it shows up in my browser on the bottom left of the photo but can also be accessed by clicking the "info" icon. Please have a look and comment if something sticks out as odd / requires correction.
Update 1045pm: I looked at the freeswinging data from earlier today. Still only 3 peaks around 1 Hz.
A pair of tweezer is OK as long as there is no magnets around. You need to (somewhat) constrain the mirror with the EQ stops so that you can pull the fiber without dragging the mirror.
I used a pair of tweezers to remove the stray fiber of first contact. As Koji predicted, this was rather dry and so it didn't have the usual elasticity, so while I was able to pull most of it off, there is a small spot remaining on the HR surface of the ETM. We will remove this with a fresh application of a small patch of FC.
I the meantime, I'm curious if this has actually fixed the suspension woes, so yet another round of freeswinging data collection is ongoing. From the first 5 mins, looks positive, I see 4 peaks around 1Hz !
Update 730pm: There are now four well-defined peaks around 1 Hz. Together with the Bounce and Roll modes, that makes six. The peak at 0.92 Hz, which I believe corresponds to the Yaw eigenmode, is significantly lower than the other three. I want to get some info about the input matrix but there was some NDS dropout and large segments of data aren't available using the python nds fetch method, so I am trying again, kicked ETMY at 1828 PDT. It may be that we could benefit from some adjustment of the OSEM positions, the coupling of bounce mode to LL is high. Also the SIDE/POS resonances aren't obviously deconvolved. The stray first contact has to be removed too. But overall I think it was a successful removal, and the suspension characteristics are more in line with what is "expected".
Here is my analysis. I think there are still some problems with this suspension.
Attachment #1: Time domain plots of the ringdown. The LL coil has peak response ~half of the other face OSEMs. I checked that the signal isn't being railed, the lowest level is > 100 cts.
Attachment #2: Complex TF from UL to the other coils. While there are four peaks now, looking at the phase information, it isn't possible to clearly disentangle PIT or YAW motion - in fact, for all peaks, there are at least three face shadow sensors which report the same phase. The gains are also pretty poorly balanced - e.g. for the 0.77 Hz peak, the magnitude of UR->UL is ~0.3, while LR->UL is ~3. Is it reasonable that there is a factor of 10 imbalance?
Attachment #3: Nevertheless, I assumed the following mapping of the peaks (quoted f0 is from a lorentzian fit) and attempted to find the input matrix that best convers the Sensor basis into the Euler basis.
Unsurprisingly, the elements of this matrix are very different from unity (I have to fix the normalization of the rows).
Attachment #4: Pre and post diagonalization spectra. The null stream certainly looks cleaner, but then again, this is by design so I'm not sure if this matrix is useful to implement.
In case anyone wants to repeat the analysis, the suspension was kicked at 1828 PDT today and this analysis uses 15000 seconds of data from then onwards.
Update 18 May 3pm: Attachment #5 better presentation of the data shown in Attachment #2, the remark about the odd phasing of the coils is more clearly seen in this zoomed in view. Attachment #6 shows Lorentzian fits to the peaks - the Qs are comparable to that seen for the other optics, although the Q for the 0.77 Hz peak is rather low.
At ~930am, I vented the IY annulus by opening VAEV. I checked the particle count, seemed within the guidelines to allow door opening so I went ahead and loosened the bolts on the ITMY chamber.
Chub and I took the heavy door off with the vertex crane at ~1015am, and put the light door on.
Diagnosis plan is mainly inspection for now: take pictures of all OSEM/magnet positionings. Once we analyze those, we can decide which OSEMs we want to adjust in the holders (if any). I shut down the ITMY and SRM watchdogs in anticipation of in-chamber work.
Not related to this work: Since the annuli aren't being pumped on, the pressure has been slowly rising over the week. The unopened annuli are still at <1 torr, and the PAN region is at ~2 mtorr.
To investigate my mapping of the eigenfrequencies to eigenmodes, I checked the Oplev spectra for the last few hours, when the Oplev spot has been on the QPD (but the optic is undamped).
So, while I conclude that my first-contact residue removal removed a constraint from the system (hence the pendulum dynamics are accurate and there are 6 eigenmodes), more thought is needed in judging what is the appropriate course of action.
With Chub providing illumination via the camera viewport, I was able to take photos of ITMY this morning. All the magnets look well clear of the OSEMs, with the possible exception of UR. I will adjust the position of this OSEM slightly. To test if this fix is effective, I will then cycle the bias voltage to the ITM between 0 and the maximum allowed, and check if the optic gets stuck.
Following the observation that the response in the LL shadow sensor was lower than that of the others, I decided to pull it out a little to move the signal level with nominal DC bias voltage applied was closer to half the open-voltage. I also chose to rotate the SIDE OSEM by ~20 degrees CCW in its holder (viewed from the south side of the EY chamber), to match more closely its position from a photo prior to the haphazhard vent of the summer of 2018. For the SIDE OSEM, the theoretical "best" alignment in order to be insensitive to POS motion is the shadow sensor beam being horizontal - but without some shimming of the OSEM in the holder, I can't get the magnet clear of the teflon inside the OSEM.
While I was inside the chamber, I attempted to minimize the Bounce/Roll mode coupling to the LL and SIDE OSEM channels, by rotating the Coil inside the holder while keeping the shadow sensor voltage at half-light. To monitor the coupling "live", I set up DTT with 0.3 Hz bandwidth and 3 exponentially weighted averages. For the LL coil, I went through pi radians of rotation either side of the equilibrium, but saw no significant change in the coupling - I don't understand why.
In any case, this wasn't the most important objective so I pushed ahead with recovering half-light levels for all the shadow sensors and closed up with the light doors. I kicked the optic again at 1712:14 PDT, let's see what the matrix looks like now.
before starting this work, i had to key the unresponsive c1auxey VME crate.
For good measure:
So the primary vent objectives have been achieved, I think.
Tomorrow and later this week:
While we have the chance:
Unrelated to this work: megatron is responding to ping but isn't ssh-able. I also noticed earlier to day that the IMC autolocker blinky wasn't blinking. So it probably requries a hard reboot. I left the lab for tonight so I'll reboot it tomorrow, but no nds data access in the meantime...
We executed this plan. Photos are here. Summary:
So if nothing, we got to practise this new wiping technique with OSEMs in situ successfully.
Yesterday we noticed that the POS and SIDE eigenmodes were degenerate (with 1mHz spectral resolution). Moreover, the YAW peak had shifted down by ~500 mHz compared to earlier this week, although there was still good separation between PIT and YAW in the Oplev error signals. Ideas were (i) check if EQ stops were not backed out sufficiently, and (ii) look for any fibers/other constraints in the system. Today morning, I inspected the optic again. I felt the EQ stop viton tips were a bit close to the optic, so I backed them out further. Apart from this, I adjusted the LR and SIDE OSEM position in their respective holders to make the sensor voltages closer to half-light. Kicked the optic again just now, let's see if there is any change.
If everything goes smoothly, I think we should plan for the heavy doors going back on and commencing the pumpdown tomorrow. After discussion with Koji, we came to the conclusion that it isn't necessary to investigate IPANG (high likelihood of it falling off the steering optics during the pumpdown) / AS beam clipping (no strong evidence that this is a problem) for this vent.
Update 1235: Indeed, the eigenmodes are back to their positions from earlier this week. Indeed, the POS and SIDE modes are actually better separated! So, the OSEM/magnet and EQstop/optic interactions are non-negligible in the analysis of the dynamics of the pendulum.
So Cal Earthquake. All suspension watchdogs tripped.
Tried to recover the OSEM damping.
=> The watchdogs for all suspensions except for ITMX were restored. ITMX seems to be stuck. No further action by me for now.
Koji came to the lab to align the IMC/IFO, but found the mirrors are dancing around. Kruthi told me that there was M7.1 EQ at Ridgecrest. Looks like there are aftershocks of this EQ going on. So we need to wait for an hour to start the alignment work.
ITMX and ETMX are stuck.
- ITM unstuck now
- IMC briefly locked at TEM00
A series of aftershocks came. I could unstick ITMX by turning on the damping during one of the aftershocks.
Between the aftershocks, MC1~3 were aligned to the previous dof values. This allowed the IMC flashing. Once I got the lock of a low order TEM mode, it was easy to recover the alignment to have a weak TEM00.
Now at least temporarily the full alignment of the IMC was recovered.
In fact, ETMX was not stuck until the M7.1 EQ today. After that it got stuck, but during the after shocks, all the OSEMs occasionally showed full swing of the light levels. So I believe the magnets are OK.
We unstuck ETMX by shaking the stack. Most effective was to apply large periodic human sized force to the north STACIS mounts.
At first, we noticed that the face OSEMs showed nearly zero variation.
We tried unsticking it through the usual ways of putting large excitations through AWG into the pit/yaw/side DOFs. This produced only ~0.2 microns of motion as seen by the OSEMs.
After the stack shake, we used the IFO ALIGN sliders to get the oplev beam back on the QPD.
The ETMX sensor trends observed before and after the earthquake are attached.
** plots deleted; SOMEONE, tried to take raster images and turn them into PDF as if this would somehow satisfy our vetor graphics requirement. Boo. lpots must be actual vector graphics PDF
After this activity, the DC bias voltage required on ETMX to restore good X arm cavity alignment has changed by ~1.3 V. Assuming a full actuation range of 30 mrad for +/- 10 V, this implies that the pitch alignment of the stack has changed by ~2 mrad? Or maybe the suspension wires shifted in the standoff grooves by a small amount? This is ~x10 larger than the typical change imparted while working on the table, e.g. during a vent.
Main point is that this kind of range requirement should probably be factored in when thinking about the high-voltage coil driver actuation.
Arnaud and I moved one of the two spare TT suspensions from the south clean cabinet to the bake lab clean room. The main purpose was to inspect the contents of the packaging. According to the label, this suspension was cleaned to Class A standards, so we tried to be clean while handling it (frocks, gloves, masks etc). We found that the foil wrapping contained one suspension cage, with what looked like all the parts in a semi-assembled state. There were no OSEMs or electronics together with the suspension cage. Pictures were taken and uploaded to gPhoto. Arnaud is going to plan his tests, so in the meantime, this unit has been stored in Cabinet #6 in the bake lab cleanroom.
We noticed that the PRM watchdog was tripping frequently. This is a period of enhanced seismic activity. The reason PRM in particular trips often is because the SIDE OSEM has 5x increased transimpedance. We implemented a workaround by modifying the watchdog tripping condition to scale the SD channel RMS by a factor of 0.2 (relative to the UL and LL channels). We restarted the modbus process on c1susaux and tested that the new logic works. Here is the relevant snippet of code:
# PRM Side is special, see elog 14745
field(DESC,"Tests whether RMS too high")
field(INPA,"C1:SUS-PRM_ULPD_VAR NPP NMS")
field(INPB,"C1:SUS-PRM_PD_MAX_VAR NPP NMS")
field(INPC,"C1:SUS-PRM_LLPD_VAR NPP NMS")
field(INPD,"C1:SUS-PRM_SDPD_VAR NPP NMS")
The db file has a note about this as well so that future debuggers aren't mystified by a factor of 0.2.
All suspension watchdogs were tripped ~90mins ago. I restored the damping. IMC is locked.
ITMX was stuck. I set it free. But notice that the UL Sensor RMS is higher than the other 4? I thought ITMY UL was problematic, but maybe ITMX has also failed, or maybe it's coincidence? Something for IFOtest to figure out I guess. I don't think there is a cable switch between ITMX/ITMY as when I move the ITMX actuators, the ITMX sensors respond and I can also see the optic moving on the camera.
Took me a while to figure out what's going on because we don't have the seis BLRMS - i moved the usual projector striptool traces to the TV screen for better diagnostic ability.
Update 16 July 1515: Even though the RMS is computed from the slow readback channels, for diagnosis, I looked at the spectra of the fast PD monitoring channels (i.e. *_SENSOR_*) for ITMX - looks like the increased UL RMS is coming from enhanced BR-mode coupling and not of any issues with the whitening switching (which seems to work as advertised, see Attachment #3, where the LL traces are meant to be representative of LL, LR, SD and UR channels).
There were several small/medium earthquakes in Ridgecrest and one medium one in Blackhawk CA at about 2000 UTC (i.e. ~ 2 hours ago), one of which caused BS, ITMY, and ETM watchdogs to trip. I restored the damping just now.
Rana and I talked about some (genius) options for the large range DC bias actuation on the SOS, which do not require us to supply high-voltage to the OSEMs from outside the vacuum.
What we came up with (these are pretty vague ideas at the moment):
For the thermal option, I remembered that (exactly a year ago to the day!) when we were doing cavity mode scans, once the heaters were turned on, I needed to apply significant correction to the DC bias voltage to bring the cavity alignment back to normal. The mechanism of this wasn't exactly clear to me - furthermore, we don't have a FLIRcam picture of where the heater radiation patter was centered prior to my re-centering of it on the optic earlier this year, so we don't know what exactly we were heating. Nevertheless, I decided to look at the trend data from that night's work - see Attachment #1. This is a minute trend of some ETMY channels from 0000 UTC on 18 July 2018, for 24 hours. Some remarks:
Also see this elog by Terra.
Attachment #2 shows the results from today's heating. I did 4 steps, which are obvious in the data - I=0.6A, I=0.76A, I=0.9A, and I=1.05A.
In science, one usually tries to implement some kind of interpretation. so as to translate the natural world into meaning.
On Friday, Milind and I performed the pitch adjustment test Rana had asked us to do. Only 1 blue beam in case of ITMX and two in case of ETMY, ETMX and ITMY were accessible. Milind (of mass 72 kg as of 10 May 2019) stood on each of the accessible blue beams of the test mass chambers for one minute and I recorded the corresponding gps time. Before moving to the next beam, we spared more than a minute for relaxation after the standing end time. Following are the recorded gps times.
Standing start time (gps)
Standing end time (gps)
PS: For each blue beam relaxation time ~ 1 min after the standing end time
Koji suggested systematic investigation of the ETMX suspension electronics. The tests to be done are:
So the ETMX satellite box is unplugged now, starting 530 pm PDT.
The satellite box was reconnected and the suspension was left with watchdog off but OSEM roughly centered. We will watch for glitches over the weekend.
Looking at the sensor and oplev trends over the weekend, there was only one event where the optic seems to have been macroscopically misaligned, at ~11:05:00 UTC on Oct 19 (early Saturday morning PDT). I attach a plot of the 2kHz time series data that has the mean value subtracted and a 0.6-1.2 Hz notch filter applied to remove the pendulum motion for better visualization. The y-axis calibration for the top plot assumes 1 ct ~= 1 um. This "glitch" seems to have a timescale of a few seconds, which is consistent with what we see on the CCD monitors when the cavity is locked - the alignment drifts away over a few seconds.
As usual, this tells us nothing conclusive. Anyways, I am re-enabling the watchdog and pushing on with locking activity and hope the suspension cooperates.
While I was trying to lock the PRMI this evening, I noticed that I couldn't move the REFL beamspot on the CCD field of view by adjusting the slow bias voltages to the PRM. Other suspensions controlled by c1susaux seem to respond okay so at first glance it isn't a problem with the Acromag. Looking at the OSEM sensor input levels, I noticed that UL is much lower than the others - see Attachment #1, seems to have happened ~100 days ago. I plugged the tester box in to check if the problem is with the electronics or if this is an actual shorting of some pins on the physical OSEM as we had in the past. So PRM watchdog is shutdown for now and there is no control of the optic available as the cables are detached. I will replace the connections later in the evening.
Since I couldn't find anything wrong, I plugged the suspension back in - and voila, the suspect UL PD voltage level came back to a level consistent with the others! See Attachment #2.
Anyway, I had some hours of data with the tester box plugged in - see Attachment #3 for a comparison of the shadow sensor readout with the tester box (all black traces) vs with the suspension plugged in, local damping loops active (coloured traces). The sensing noise re-injection will depend on the specifics of the local damping loop shapes but I suspect it will limit feedforward subtraction possibilities at low frequencies.
However, I continue to have problems aligning the optic using the slow bias sliders (but the fast ones work just fine) - problem seems to be EPICS related. In Attachment #4, I show that even though I change the soft PITCH bias voltage adjust channel for the PRM, the linked channels which control the actual voltages to the coils take several seconds to show any response, and do so asynchronously. I tried restarting the modbus process on c1susaux, but the problem persists. Perhaps it needs a reboot of the computer and/or the acromag chassis? I note that the same problem exists for the BS and PRM suspensions, but not for ITMX or ITMY (didn't check the IMC optics). Perhaps a particular Acromag DAC unit is faulty / has issues with the internal subnet?
Sigh... hard loch
Looks like a M=4.6 earthquate in Barstow,CA tripped all the suspensions. ITMX got stuck. I restored the local damping on all the suspensions just now, and freed ITMX. Looks like all the suspensions damp okay, so I think we didn't suffer any lasting damage. IMC was re-aligned and is now locked.
In the last couple days, as the IMC ringdowns have been going on, we have noticed that the MC is behaving bad. Misaligning, drifting, etc.
Gautam told me a horror story about him, Koji, and melted wires inside the sat boxes.
I said, "Its getting too hot in there. So let's take the lids off!"
So then we:
After some minutes, we saw no drifting. So maybe my theory of "hot heatsink partially shorting a coil current to GND through partially melted ribbon cable" makes sense? IF this seems better after a month, lets de-lid all the optics.
Let's look at some longer trends and be very careful next to MC2 for the next 3 days! I have put a dangerous mousetrap there to catch anyone who walks near the vacuum chamber.
gautam: the grounding situation per my assessment is that the shield of all the IDC cables are connected to a common metal strip at 1X5 - but in my survey, I didn't see any grounding of this strip to a common ground.
An earthquake around 330 UTC (=730pm yesterday eve) tripped ITMX, ITMY and ETMX watchdogs. ITMX got stuck. I released the stuck optic and re-enabled the local damping loops just now.
ETMX was grossly misaligned.
I re-aligned it and the X arm now locks.
7:00PM with Koji
Both the alignment of the X and Y arms was recovered.
~>z avg 10 C1:LSC-TRX_OUT C1:LSC-TRY_OUT
We are running ass for the X arm to recover the X arm alignment.
Meanwhile, i want to block the Y arm trans PD (Thorlabs). To do it, the PD<->QPD thresholds were changed from 5.0/3.0 to 0.5/0.3.
I returned the triggering threshold to normal values (5/3).
This EQ in Nevada seems to have tripped all watchdogs. ITMX was stuck. It was released, and all the watchdogs were restored. Now the IMC is locked.
This EQ seems to have knocked all suspensions out. ITMX was stuck. It is now released, and the IMC is locked again. It looks like there are some serious aftershocks going on so let's keep an eye on things.
I found that there is an issue with the MC1 slow bias voltages.
I usually offload the DC part of the output voltage from the WFS servos to the slow bias voltage sliders, so as to preserve maximum actuation range from the fast system. However, today, I found that this servo wasn't working well at all. So I dug a little deeper. Looking at the EPICS database records:
We can limit the EPICS values giving some parameters to the channels. cf https://epics.anl.gov/tech-talk/2012/msg00147.php
But this does not solve the MC1 issue. Only we can do right now is to make the output resister half, for example.
This earthquake tripped all suspensions and ITMX got stuck. The watchdogs were restored and the stuck optic was released. The IFO was re-aligned, POX/POY and PRMI on carrier locking all work okay.