The 10' x 6' optical table is seating on 6 tripods. Each tripod leg has 3 feet. They are supported by jack screws against copper plates on the concrete floor. These set screws allowed us to set the height, level and distribute the load of the table.
To lock this position of the table grouting is used around the jack screws and between the concrete floor and the tripod feet.
Once the grout set the load will be carried by the hole feet of 4" x 3.25" x3 = 39 sq inches per tripod leg
These ~4" thick grouted islands of 8-9" OD will be covered by 4" concrete. The area between the legs will be filled with 8" thick concrete.
Cement based grout, early height change 0 - 4%, hardened height change 0 - 0.3%
Atm2-3, pouring grout
Atm4, south west leg is done
Tomorrow: setting up ribbed bars, building frame to hold concrete and pour concrete
The preparation for pouring concrete is continuing. The grout forms were removed. Half inch ribbed steel bars were epoxied into our 4" thick floor.
The concrete holding sides- forms are cut to size. They will be installed on Monday morning and pouring concrete should follow.
We can be planning for cleaning up on Tuesday
Atm1, epoxy used bars
Atm2, south west tripod feet grouted
Atm3, ribbed bars are getting ready
C1: PEM-count channels are dead since August 12, 2010
Bertine Robby, Bertin and Jerry completed the installation by pouring concrete yesterday and cleaning up today.
Atm1-2, steel bars, grouted feet and side forms are ready
Atm3, bounding agent is applied for better bounding
Atm4, pumping concrete
Atm5, shaker is used to fill-all
Atm6, late after noon the forms- sides were removed, NOTE: actual table to concrete distance is 12.75"
Atm7, construction people left and the tile man is on the way
The tile patching should complete today so tomorrow we can remove plastic covers.
The tile work was done yesterday after noon.
This morning Mike Gerfen and me lowered the enclosure frame to normal height.
Keven- janitor and I removed plastic covers from chambers, racks, SP, MC2 and clean tool boxes.
The afternoon Jenne, Kiwamu, Joe and Aiden cleaned the enclosure inside out. The particle count measured zero inside the enclosure with HEPAs on when the covers were
removed. The MOPA and all other components were happy to see us in excellent condition.
This table height is very user friendly!
Safety grounds were reconnected.
Atm1, new tiles around the concrete slab
Atm2, frame lowered with low cross bars reinstalled
Atm3, the enclosure frame's north west foot is connected to ground
Atm4, PSL optical table is connected to ground at the north east corner through 1 Mohm
Atm5, PSL optical table level at a stimulating, back-friendly height
I posted the crane safety document on the 40m wiki, vacuum page as 26 August 2010
Please add your comments and corrections.
The South End Crane will be balanced on Tuesday, 31 August 2010
This will mean that the back door of the south arm will be open on and off. Air quality will bad.
Please plan accordingly.
The PSL -leg concrete was sealed with a single coat of SCOFIELD Cureseal-S to minimize shedding of particles.
The optical table was covered and optics were removed from the shelf. Accelerometers were turned off.
I lowered room 101 thermostat setting from 74F to 72F
Our toilets were replaced by a green Gerber 1.6 gpf ( 6 lpf ) last week.
Crane balancing at the south end is continuing today.
The crane I -beam now leveled at all degrees of rotation. The lower hinge was moved southward about 1/4 of an inch. Performance was tested at 2000 lbs
Atm1, work in progress
Atm2, load test at 1 Ton
Atm3, service report
I hooked up interlock to the Innolight 2W 1064 nm in the enclosure. The manual shutter is closed on this unit.
SAFETY GLASSES REQUIRED !
I played with the particle counter this morning. Now it's running and reading the correct numbers on the old COCHECKLIST.adl medm screen.
However, it can not be seen with dataviewer.
Koji just rebooted the frame builder and the problem was solved.
Innolight 2W 1064nm main laser was turned off for more enclosure related work.
The electrician re rooting the power- conduit to the interlock under the floor and bring it through the new concrete tunnel.
Emergency laser power shut off switch was tested at entry door 104M. It worked fine.
No laser glasses required. Laser will be turned back on around 11 am today.
Atm1, AC power line to enclosure and interlock was extended.
Atm2, 24V line to interlock rerouted
Atm3, job is completed
SAFETY GLASSES REQUIRED ! SAFETY GLASSES REQUIRED!
The laser is turned on. Injection current set to 0.871A.
Actually some one turned the laser off last night and did not enter it into the elog ! Burned toast award is granted !
Please come forward voluntarily.
I turned off/on the power to the accelerometers in order to re rout their connections. I found cable connector body-nut #3 loose to Accelerometer 2X This connector should be checked for solid performance.
Cleaned up cables on the top and bottom. Vacuumed both areas. We still have some remaining shading from the MOPA umbilical and more unknown BNC cables hanging around.
The PSL out 2" OD beam guide tube was cut 1.5" shorter to 13.5"
The 10" OD 0.25" wall Al tube was replaced by a lighter, not anodized and thinner wall 0.094" tube of 15.5" lenght, that is 0.75" shorter.
The new position of the PSL table made these cuts necessary.
John Miller has arrived from Australia with 3 bags of Wagonga Coffee. Trade bargaining has started on
250 mgs of Sumatran Mandehling, Timur and Papua New Guine.
The squeezing open frame rack was moved from the south side of the PSL enclosure to the north side of the SP table.
AC power breaker is PC-2 #1
Blocked PSL output beam into IFO
Checked: HV at IOO & OMC are off, jam nuts in position,
Closed V1 and VM3, opened VV1 to N2 regulator
We are venting at 1 Torr/min rate
The vertex crane drive is overheating, it stopped functioning. Service man will be here tomorrow morning.
I crane was just turned on for for may be about 5 minutes. The vertical drive was fine for a while, but the horizontal did not worked at all.
The crane is tagged out again and the controller box is cooling down.
Atm1, Vertex crane controller unit on horizontal I-beam.
Module CCES-407 1HP 2.3A 460V/3 phase on the right was replaced. Speed was increased to 30 Hz
Atm2, See burned insulation on black wire that was shorting to the large suppressor resistor on Atm3 One can see the pit marks on the right end of it.
This large power dissipater resistor got hot as the malfunctioning controller 407 broke down. Touching black power wire insulation melted and made a short.
So the heat was created and finally the fuses were blown. The unnecessary large fuses were replaced by small ones.
Fortunately we had one spare controller on hand that made this repair to be done fast.
We used the new Crane Safety Check list from LIGO-E1000379-v1 the first time yesterday before doors were removed.
The control room's north west corner smelled like propane gas yesterday around 16:30
We all agreed that the smell was real and I called the safety office. I was told that they received 6 other calls from different parts of the campus.
The smell disappeared in about a half an hour.
Our visiting graduate student Yuta Michimura received 40m specific basic safety training today.
Accelerometers xyz were moved from IOO-south/west corner to under PSL table. They were turned off for ~20 minutes.
Guralp was also moved eastward about 2 ft. It is not leveled.
This is part of the preparation to remove access connector.
Kiwamu, Koji, Yutah, Rana, Jan and Steve
We removed the west access connector this afternoon.
Our new undergrad Joonho Lee received 40m specific basic safety training yesterday.
Yuta and Joonho are scheduled to participate in the LIGO-laser safety tutelage with Peter K on Oct 12
We got two small RF filter for the PMC from Valera They are made by http://www.larkengineering.com/ "MC35.5-3-AB" sma, 29300-01
The 2 W Innolight shutter is closed and enclosure door removed. Beam path blocked. Do not change this condition.
The PSL output beam guide was upgraded from 2" to 8" OD . It is green ready. Shutter is open.
Prof Alan Weistein guided the 24 student through the 40m. His performance was rated as an enthusiastic 9.5
Innolight PSL laser is set to @2.1A , ~1.6W output ! Please scan out when finished!
The output monitoring pick up window W2-LW-2-2050-UV-1064-45S is in place. IOO_ANG_OPD and IOO_POS_QPD roughly aligned.
The PMC alignment and/ or mode matching is bad. PMC reflected is 50%, throughput 600mW
Remember to block PSL output ! into IFO
This BNC cable is crying for help. Please do not do this to me. It should be reported to the abused center.Throw this cable into the garbage now.
I'm cleaning out to make room for our new optical cabinet. Are we keeping these? There are ~20 pieces of 10" od 1" wide tapes and large number of cassettes.
AJW, Zucker, Stuart A and Koji were notified in this matter.
Alan suggested to save data of Bruce Allen paper of observation of binary neutron stars in the 40m on 1994 November 14-20 and save back up tapes of his period in the 40m.
Mike: reels are not readable any more, it is time to let go
www.avalontest.com has fixed the 25MHz oscillation. Contact: Jim Burnham 760-536-0191
Actually, NOT FIXED.
Fire-smoke sensors in the vertex area #2-31, 2-30 east, 2-32 south/MC2 and 2-37 old control room area are turned off to accommodate the welding
activity of folding crane. These sensors will be reactivated at 3:30pm today.
Stay out of the 40m lab: IFO room till 6 pm today.
The new cord wheel was to big. They be back tomorrow?
The lab is open.
Pedro of the electrical shop helped to retrace wires from the racks to the breakers. Many wrong labels were replaced. These labels are still representing the south arm as Y and the east arm as X.
Racks: 1X1,2,3,4 and 1Y3,4,5,6,7 and 9 are connected through manual disconnect swithes to AC power breaker PC-1 on the west wall of IFO room 104.
This panel PC-1 is connected to T1 isolation transformer in room 101 Panel L - # 38, 40, 42
Racks: 1Y1 & 2 psl and mc are connected through manual disconnect switches to AC power breaker PC-2 just west of the psl enclosure.
This panel is connected to T2 isolation transformer in room 101 Panel L - # 32, 34, 36
Our new postdoc Suresh Doravari received 40m specific safety training last week.
The south end of IFO room 104 was reshuffled. The flow bench was moved all the way to the south end to create more room for the crane reach. The old large drawing cabinet was moved under the vacuum tube.
The new clear view through Acrylic-plexyglass cabinet is anchored. It has powder finished coating so it will not out-gas too much.The back side shelf mounting holes are sealed off with kapton tape. The doors still needs some gaskit seals.
8 pieces of 10" CF blanks were shipped to MIT
Koji, Suresh and Steve,
The beam scanner is back from repair. Koji and Suresh helped to move the I/O address jumpers on the interface card from default position 300 to 320
It is working again.
Physical Electronics: Vac-Seal #288-6000 arriving Monday. Our last bag used last week had a p/n 1002003 with expiration date .....2009 ? and it was stored in the
refrigerator all times. We are getting 68 small bags.
We should have precision gluing notes of epoxies/procedures used on our sus upgrade.
Please remember to cover the optical tables !
This condition is created when we run out of nitrogen that holds the valves in set positions. The Maglev and other turbos are still running in this configuration so one has to pay attention that the foreline valves are open.
Reset error message on vacuum screen and opened V4 & V5 at 3pm today.