[ Yuki, Gautam ]
I calibrated PZT mirrors. The ROUGH result was attached. (Note that some errors and trivial couplings coming from inclination of QPD were not considered here. This should be revised and posted again.)
The PZT mirrors I calibrated were:
I did the calibration as follows:
The calibration factor was
CVI-pitch: 0.089 mrad/V
CVI-yaw: 0.096 mrad/V
Laseroptic-pitch: 0.062 mrad/V
Laseroptic-yaw: 0.070 mrad/V
Previous calibration of the same mirrors, elog:40/8967
Interim Procedure Report:
The current setup of AUX Y-arm Green locking has to be improved because:
What to do
I improved Anti-Imaging board (D000186-Rev.D), which will be put between DAC port and PZT driver board.
It had notches at f = 16.6 kHz and 32.7 kHz, you can see them in the plot attached. So I replaced some resistors as follows:
Then the notch moved to 65.9 kHz (> sampling frequency of DAC = 64 kHz, good!).
(The plot enlarged around the notch frequency and the plot of all channels will be posted later.)
All electronics and optics seem to be ready.
I made a cable which connects DAC port (40 pins) and AI board (25 pins). I will check if it works.
Tomorrow I will change setup for improvement of AUX Y-end green locking. Any optics for IR will not be moved in my design, so this work doesn't affect Y-arm locking with main beam.
While doing this work, I will do:
Before changing setup at Y-end table, I measured the status value of the former setup as follows. These values will be compared to those of upgraded setup.
(These numbers are shown in the attachment #2.)
The setup I designed is here. It can bring 100% mode-matching and good separation of degrees of TEM01, however I found a probrem. The picture of setup is attached #3. You can see the reflection angle at Y7 and Y8 is not appropriate. I will consider the schematic again.
The SHG crystal has the conversion efficiency of ~2%W (i.e. if you have 1W input @1064, you get 2% conversion efficiency ->20mW@532nm)
It is not possible to produce 0.58mW@532nm from 20.9mW@1064nm because this is already 2.8% efficiency.
I measured it with the wrong setting of a powermeter. The correct ones are here:
The calculated conversion efficiency of SHG crystal is 1.2%W.
What about just copying the Xend layout? I think it has good MM (per calculations), reasonable (in)sensitivity to component positions, good Gouy phase separation, and I think it is good to have the same layout at both ends. Since the green waist has the same size and location in the doubling crystal, it should be possible to adapt the X end solution to the Yend table pretty easily I think.
I designed a new layout. It has good mode-matching efficiency, reasonable sensitivity to component positions, good Gouy phase separation. I'm setting optics in the Y-end table. The layout will be optimized again after finishing (rough) installation. (The picture will be posted later)
To facilitate Yuki's alignment of the EY green beam into the Yarm cavity, I have changed the LSC triggering and PowNorm settings to use only the reflected light from the cavity to do the locking of Arm Cavity length to PSL. Running the configure script should restore the usual TRY triggering settings. Also, the X arm optics were macroscopically misaligned in order to be able to lock in this configuration.
I finished installation of optics in the Y-end and recovered green locking. Current ALS-TRY_OUTPUT is about 0.25, which is lower than before. So I still continue the alignment of the beam. The simulation code was attached. (Sorry. The optic shown as QWP2 is NOT QWP. It's HWP.)
After installation I measured these power again.
There is a little power loss. That may be because of adding one lens in the beam path. I think it is allowable margin.
While pointing Yuki to the c1asx servo system, I noticed that the filter file for c1asx is missing in the usual chans directory. Why? Backups for it exist in the filter_archive subdirectory. But there is no current file. Clearly this doesn't seems to affect the realtime code execution as the ASX model seems to run just fine. I copied the latest backup version from the archive area into the chans directory for now.
Setting up c1asy:
Now Yuki can work on copying the simulink model (copy c1asx structure) and implementing the autoalignment servo.
[ Yuki, Gautam, Steve ]
To align the green beam in Y-end these hardware were installed:
I made sure that DAC CH9~16 and cable to AI-board worked correctly.
When we applied +100V to PZT driver and connected DAC, AI-board and PZT drive, the output voltage of the driver was not correct. I'll check it tomorrow.
I connected DAC - AIboard - PZTdriver - PZT mirrors and made sure the PZT mirrors were moving when changing the signal from DAC. Tomorrow I will prepare alignment servo with green beam for Y-arm.
Final Procedure Report for Green Locking in YARM:
To do for Green Locking in YARM:
The auto-alignment servo should be completed. This servo requires many parameters to be optimized: demodulation frequency, demodulation phase, servo gain (for each M1/2 PIT/YAW), and matrix elements which can remove PIT-YAW coupling.
Our 4 ion pumps were closed off for a lomg time. I estmated their pressure to be around ~1 Torr. After talking with Koji we decided not to vent them.
It'd be still useful to wire their position sensors. But make sure we do not actuate the valves.
The cryo pump was regenerated to 1e-4 Torr about 2 years ago. It's pressure can be ~ 2 Torr with charcoal powder. It is a dirty system at room temperature.
Do not actuate VC1 and VC2, and keep its manual valve closed.
IF someone feels we should vent them for some reason, let us know here in the elog before Monday morning.
Wiring of the power, Ethernet, and indicator lights for the vacuum Acromag chassis is complete. Even though this crate will only use +24V DC, I wired the +/-15V connector and indicator lights as well to conform to the LIGO standard. There was no wiring diagram available, so I had to reverse-engineer the wiring from the partially complete c1susaux crate. Attached is a diagram for future use. The crate is ready to begin software developing on Monday.
Chub & Steve,
We swapped in our replacement of Varian V70D "bear-can" turbo as factory clean.
The new Agilent TwisTorr 84 FS turbo pump [ model x3502-64002, sn IT17346059 ] with intake screen, fan, vent valve. The controller [ model 3508-64001, sn IT1737C383 ] and a larger drypump IDP-7, [ model x3807-64010, sn MY17170019 ] was installed.
Next things to do:
donatella was one of our last workstations running ubuntu12. we installed SL7 on there today
Then there are ~20 errors related to gds-crtools that look like this:Error: Package: gds-crtools-2.18.12-1.el7.x86_64 (lscsoft-production) Requires: libMatrix.so.6.14()(64bit)
I'm making a controlled shutdown of the vac controls to add new ADC channels. Will advise when it's back up.
ADC work finished for the day. The vac controls are back up, with all valves CLOSED and all pumps OFF.
The schematic of the homodyne configuration is shown below.
Following are the list of components
One set of fiber is now kept along the arm of the interferometer
Fiber coupled (3 No's)
Free space ( 2 No's)
I drew out some idea of how we might use a single OMC to clean both paths of the BHD after mixing, without being susceptible to polarization-dependent effects within the OMC. Basically, can we send the two legs of the BHD into the OMC counterpropagating. I've attached a diagram.
I think one issue would be scattered light, since any backscatter directly couples into the counterpropagating mode, and thus directly to the PD. However, unless the polarization of the scattered light rotates it would not scatter back to the IFO. And, since the LO and signal mix before the OMC, this scattered light would not directly add phase noise.
Maybe more problematic would be that if the rejection at the PBS (or the polarization rotation) isn't perfect, light from the LO directly couples into the dark port. Can we get away with a Faraday isolator before the OMC?
SURFs had trouble connecting paola to martian via wireless.
Of course, it requires a fixed IP but it had not it yet. So I went to chiara and gave 192.168.113.110 as "paolawl". Note that the wired connection has .111 and it is "paola".
Followed the instruction on http://nodus.ligo.caltech.edu:8080/40m/14121
I think a Faraday rotator rotates the polarizations in a same way for both forward and backward beam, and it's not like in this figure.
And the transmission through multiple faradays will also be a big issue.
This afternoon, Gautam helped me setup Giada to access the GigE installed for MC2. Unlike Paola, which was being used earlier, Giada has a better battery life and doesn't shutdown when the charger is unplugged. Gautam configured Giada to enable its wireless connection to Martian, just like Koji had configured Paola (https://nodus.ligo.caltech.edu:8081/40m/14672). We also rerouted the ethernet cable we were using with the PoE adaptor from Netgear Switch in 1x2 to 1x6.
Kruthi noticed that she could not login to rossa from giada.
I checked /etc/resolv.conf and it was
so obviously it is useless to refer localhost (i.e. giada) as a nameserver.
I copied our usual resolv.conf to giada as following:
Giada's ssh known_host had unupdated entry for rossa, so I had to clean it up, but after that we can connect to rossa from giada just by "ssh rossa".
I think rana did some more changes to this workstation to make it useful for commissioning activities - but the MEDM screens were still white blanks. The problem was that the firewalld wasn't disabled (last two steps of the KThorne setup wiki). I disabled it. Now donatella can run MEDM, ndscope and StripTool. DTT doesn't work to get online data because of a "Synchronization Error", I'm not bothering with this for now. I think Kruthi successfully demonstrated the fetching of offline data with DTT.
I noticed recently that Megatron was running Ubuntu 12, so I've started its OS upgrade.
Megatron and IMC autolocking will be down for awhile, so we should use a different 'script' computer this week.
Mon Dec 9 14:52:58 2019
upgrade to Ubuntu 14 complete; now upgrading to 16
Megatron is now running Ubuntu 18.04 LTS.
We should probably be able to load all the LSC software on there by adding the appropriate Debian repos.
I have re-enabled the cron jobs in the crontab.
The MC Autolocker and the PSL NPRO Slow/Temperature control are run using 'initctl', so I'll leave that up to Shruti to run/test.
[Yehonathan, Jon, Shruti]
Since the PMC would not lock, we initially burt-restored the c1psl machine to the last available shapshot (Dec 10th 2019), but it still would not lock.
Then, it was burt-restored to midnight of Dec 1st, 2019, after which it could be locked.
I have completed the new EPICS channel database for the c1psl and c1ioo channels (now combined into the new c1psl Acromag machine). I've tested a small subset of channels on the electronics bench to confirm that the addressing and analog channel calibrations are correct in a general sense. At this point, we are handing the chassis off to Chub to complete the wiring of the Acromag terminals to Dsub feedthroughs. At the 40m meeting today, we identified Feb. 17-22 as a potential window for installation in the interferometer (Gautam is out of town then). Below are some implementaton details for future reference.
For analog input (ai) channels, the Acromag outputs raw values ranging from +/-30,000 counts, but the EPICS IOC interprets the data type as ranging from +/-2^15 = 32,768. Similarly, for analog output (ao) channels, the Acromag expects a drive signal in the range +/-30,000 counts. To achieve proper scaling, Johannes had previously changed the EGUF and EGUL fields from +/-10 V to +/-10.923 V. However, changing the engineering fields makes it much harder for a human to read off the real physical I/O range of the channel.
A better way to achieve the correct scaling is to simply set the field ASLO=1.09225 (65,536 / 60,001) in addition to the normal EGUF and EGUL field values (+/-10 V). Setting this field forces a rescaling of the number of raw counts that works as so (assuming a 16-bit bipolar ADC or DAC, as are the Acromags):
OVAL = (RVAL * ASLO + AOFF + 2^15) * (EGUF - EGUL) / 2^16 + EGUL
In the above mapping, OVAL is the value of the channel in engineering units (e.g., V) and RVAL is its raw value in counts. It is not the case that either the ASLO/AOFF or EGUF/EGUL fields are used, but not both. The ASLO/AOFF parameters are always applied (but their default values are ASLO=1 and AOFF=0, so have no effect unless changed). The EGUF and EGUL parameters are then additionally applied if the field LINR="LINEAR" is set.
This conversion allows the engineering fields to remain unchanged from the real physical range. The ASLO value is the same for both analog input and output channels. I have implemented this on all the new c1psl and c1ioo channels and confirmed it to work using a calibrated input voltage source.
upgrade was done
cronjob testing wasn't one by one 😢
burt snapshots were gone
i brought them back home 🏠
The burt snapshotting is still not so reliable - for whatever reason, the number of snapshot files that actually get written looks random. For example, the 14:19 backup today got all the snaps, but 15:19 did not. There are no obvious red flags in either the cron job logs or the autoburt log files. I also don't see any clues when I run the script in a shell. It'll be good if someone can take a look at this.
To facilitate wiring the c1psl chassis and scripting loopback tests, I've compiled a distilled spreadsheet with the Acromag-to-breakout board wiring, broken down by connector. This information is extractable from the master spreadsheet, but not easily. There were also a few apparent typos which are fixed here.
The wiring assignments at the time of writing are attached below. Here is the link to the latest spreadsheet.
The 24 V Sorenson (2nd from bottom) in the small rack west of 1x2 was repurposed to 12V 600 mA, and was run to a terminal block on the north side of 1X1. Cables were routed underneath 1X1 and 1X2 to the terminal blocks. 12V was then routed to the PSL table and banana clip terminals were added.
I've also been noticing that the IMC Autolocker scripts are running rather sluggishly on Megatron recently. Some evidence - on Feb 11 2019, the time between the mcup script starting and finishing is ~10 seconds (I don't post the raw log output here to keep the elog short). However, post upgrade, the mean time is more like ~45-50 seconds. Rana mentioned he didn't install any of the modern LIGO software tools post upgrade, so maybe we are using some ancient EPICS binaries. I suspect the cron job for the burt snapshot is also just timing out due to the high latency in channel access. Rana is doing the software install on the new rossa, and once he verifies things are working, we will try implementing the same solution on megatron. The machine is an old Sun Microsystems one, but the system diagnostics don't signal any CPU timeouts or memory overflows, so I'm thinking the problem is software related...
There were a bunch of medm processes stalled on megatron (connected with screenshot taking). To see if they were interfering with the other scripts, I killed all of the medm processes, and commented out the line in the crontab that runs the screenshots every 10 mins. Let's see if this improves stability.
We found that the caput commands were taking much longer to execute on megatron than on pianosa (for example). Suspecting that this had something to do with the fact that megatron was using EPICS binaries from the shared NFS drive which were compiled for a much older OS, I installed the latest stable release of EPICS on megatron. The new caput commands execute much faster. I also added the local EPICS directory to the head of the $PATH variable used by the MC autolocker and FSS Slow scripts, so that they use the new caput command. But mcup is still slow - maybe my new path definition isn't picked up and it is still using the NFS binaries? To be looked into...
After some discussion with Gautam, I decided to build more spare channels into the new c1psl machine. This is anticipation of adding new laser and ISS channels in the near future, to avoid having to disconnect the installed chassis and pull it out of the rack. The spare channels will be wired to DB37M feedthroughs on the front side of the chassis, with enough wire length to be able to pull the breakout boards out of the front to reconfigure their wiring as needed (e.g., split off channels onto a separate connector).
To have enough overhead, this will require installing 1 additional ADC unit (XT1221) and 1 additional DAC (XT1541). We have enough spare BIO channels among the existing units (both sinking and sourcing). This will give us:
The updated c1psl chassis wiring assignments are attached. It adds 4 new DB37M connectors for the spare channels (highlighted in yellow) and fixes one typo Jordan found while wiring today. The most current spreadsheet is available here.
The vac system is going down at 11 am today for planned maintenance:
We will advise when the work is completed.
This work is finally complete. The dry pump replacement was finished quickly but the controls updates required some substantial debugging.
For one, the mailer code I had been given to install would not run against Python 3.4 on c1vac, the version run by the vac controls since about a year ago. There were some missing dependencies that proved difficult to install (related to Debian Jessie becoming unsupported). I ultimately solved the problem by migrating the whole system to Python 3.5. Getting the Python keyring working within systemd (for email account authentication) also took some time.
Edit: The new interlock flag channel is named C1:Vac-interlock_flag.
Along the way, I discovered why the interlocks had been failing to auto-close the PSL shutter: The interlock was pointed to the channel C1:AUX-PSL_ShutterRqst. During the recent c1psl upgrade, we renamed this channel C1:PSL-PSL_ShutterRqst. This has been fixed.
The main volume is being pumped down, for now still in a TP3-backed configuration. As of 8:30 pm the pressure had fallen back to the upper 1E-6 range. The interlock protection is fully restored. Any time an interlock is triggered in the future, the system will send an immediate notification to 40m mailing list. 👍
I propose we go for all CAPS for all channel names. The lower case names is just a holdover from Steve/Alan from the 90's. All other systems are all CAPS.
It avoids us having to force them all to UPPER in the scripts and channel lists.
I looked into how the new UPS devices suggested by Chub would communicate with the vac interlocks. There are several possible ways, listed in order of preference:
I recommend we proceed with ordering the Tripp Lite 36HW20 for TP1 and Tripp Lite 1AYA6 for TP2 and TP3 (and other 120V electronics). As far as I can tell, the only difference between the two 120V options is that the 6FXN4 model is TAA-compliant.
Chub has placed the order for two new UPS units (115V for TP2/3 and a 220V version for TP1).
They will arrive within the next two weeks.
I have been working on analyzing the seismic data obtained from the 3 seismometers present in the lab. I noticed while looking at the combined time series and the gain plots of the 3 seismometers that there is some error in the calibration of the BS seismometer. The EX and the EY seismometers seem to be well-calibrated as opposed to the BS seismometer.
The calibration factors have been determined to be :
The seismometers each have 3 channels i.e X, Y, and Z for measuring the displacements in all the 3 directions. The X channels of the three seismometers should more or less be coherent in the absence of any seismic excitation with the gain amongst all the similar channels being 1. So is the case with the Y and Z channels. After analyzing multiple datasets, it was observed that the values of all the three channels of the BS seismometer differed very significantly from their corresponding channels in the EX and the EY seismometers and they were not calibrated in the region that they were found to be coherent as well.
Note: All the frequency domain plots that have been calculated are for a sampling rate of 32 Hz. The plots were found to be extremely coherent in a certain frequency range i.e ~0.1 Hz to 2 Hz so this frequency range is used to understand the relative calibration errors. The spread around the function is because of the error caused by coherence values differing from unity and the averages performed for the Welch function. 9 averages have been performed for the following analysis keeping in mind the needed frequency resolution(~0.01Hz) and the accuracy of the power calculated at every frequency.
The gain in the given frequency range is ~3. The phase plotting also shows a 180-degree phase as opposed to 0 so a negative sign would also be required in the calibration factor. Thus the calibration factor for the Y channel of the BS seismometer should be around ~3.
The mean value of the gain in the given frequency range is the desired calibration factor and the error would be the mean of the error for the gain dataset chosen which is caused due to factors mentioned above.
Note: The standard error envelope plotted in the attached graphs is calculated as follows :
1. Divide the data into n segments according to the resolution wanted for the Welch averaging to be performed later.
2. Calculate PSD for every segment (no averaging).
3. Calculate the standard error for every value in the data segment by looking at distribution formed by the n number values we obtain by taking that respective value from every segment.
The BS seismometer is a different model than the EX and the EY seismometers which might be a major cause as to why we need special calibration for the BS seismometer while EX and EY are fine. The sign flip in the BS-Y seismometer may cause a lot of errors in future data acquisitions. The time series plots in Attachment #4 shows an evident DC offset present in the data. All of the information mentioned above indicates that there is some electrical or mechanical defect present in the seismometer and may require a reset. Kindly let me know if and when the seismometer is reset so that I can calibrate it again.
The vac system is going down now for planned repairs [ELOG 15499]. It will likely take most of the day. Will advise when it's back up.
Vacuum work is completed. The TP2 and TP3 interlocks have been overhauled as proposed in ELOG 15499 and seem to be performing reliably. We're now back in the nominal system state, with TP2 again backing for TP1 and TP3 pumping the annuli. I'll post the full implementation details in the morning.
I did not get to setting up the new UPS units. That will have to be scheduled for another day.