Following the method in Peter's Elog,
I edited c1psl.db to include the following:
I restarted c1psl (had to go hit the physical reset button since it didn't come back after telnet-ing and "reboot"ing) to make this take effect.
Next step is to tell the PMC screen to look at this _LOCALC rather than _LODET, and the screen will be calibrated into dBm.
Right now, the screen is as it always has been, because after relooking at the calibration, I no longer believe it. This calibration claimes -19dBm for an LOmon value of 0.1200, when I actually measured +16dBm for this LOmon value. So I've screwed something up in doing my MatLAB calibration. I'll fix it tomorrow, and put in the correct calibration before I change the PMC screen.
RefCav, PMC, MC are all back and locked after my shenanigans.
Vacuum normal valve condition was changed to accommodate SRS-RGA calibration.
VM1 was closed to isolate the RGA from the IFO.
Vacuum valve configuration is back to VACUUM NORMAL condition. RGA calibration completed.
RGA scan attached is the backgroud of the rga with std cal leak open, sn 08581
Krypton at amu 84 and Argon at amu 40 are the cal signals.
compiled using the 'gcc' compiler instead of the 'ANSI C' compiler that is recommended in the README (which, I notice,
is now missing from Ben Johnsons web page!). Let's see how long this runs.
We found that DMF/ was not an SVN working copy, so I wiped out the SVN version, imported the on-disk copy, moved it to DMFold/ and then checked out the SVN version.
We can delete DMFold/ whenever we are happy with the SVN copy.
nodus was hanging because it was trying to mount the cit40m account from rigel and rigel was not responding.
Neither I nor Yoichi can recall what the cit40m account does for us on nodus anymore and so I commented it out of the nodus /etc/vfstab.
nodus may still need a boot to make it pay attention. I was unable to do a 'umount' to get rid of the rigel parasite. But mainly I don't want anything in
the 40m to depend on the LIGO GC system if at all possible.
I really don't understand why my programs that I used to use to get data from the HP Spectrum Analyzer and the Marconi frequency generator don't work anymore.
I spent hours trying to debug the code but I can't sort the problem out.
The main problem seem to be with the function recv from the socket library. Somehow it can't anymore get any data from the instruments. The thing I can't understand, though, is that if called directly from the python terminal it works fine!
In particular the problem is with the following lines in my code:
tmp = netSock.recv(1024)
Tried a lot of tickering but it didn't work.
I attach the two scripts I've been using. One (sweepfrequencyPRC.py) calls the other (HP4395PRC.py).
They worked egregiously for weeks in the past. Don't know what happened since then.
This morning Joe looked at my code and made me notice that for some reason the query to the Spectrum Analyzer made by netSock.recv(1024) contained two answers. It was like the buffer contained the answer two different queries.
After some experiment I found that basically the GPIB interface wasn't switching from the "auto 1" to the "auto 0" mode as it should. I rewrote part of the code and that seemed have solved the problem.
Still don't understand why it used to work in the past and then it stopped.
tmp = netSock.recv(1024)
I have calibrated the PMC LO Mon (C1:PSL-PMC_LODET) on the PMC's EPICS screen, by inputting different RF LO levels into the LO input of the PMC servo board.
Since the RF output adjust slider on the PMC's Phase Shifter screen doesn't do a whole lot (see elog 1471), I used a combination of attenuators and the slider to achieve different LO levels. I measured the level of the attenuated RF out of the LO board using the 4395A in spectrum analyzer mode, with the units in dBm, with 50dB attenuation to make it stop complaining about being overloaded. For each row in the table I measured the RF level using the 4395, then plugged the cable back into the PMC servo board to get the EPICS screen's reading.
The last 2 columns of the table below are the 'settings' I used to get the given RF LO level.
When the new mixers that Steve ordered come in (tomorrow hopefully), I'll put in a Level 13 mixer in place of the current Level 23 mixer that we have. Also, Rana suggested increasing the gain on the op-amp which is read out as the LO Mon so that 13dBm looks like 1V. To do this, it looks like I'll need to increase the gain by ~80.
I don't know who left the X arm locked, but I just ran the Align Full IFO script, so everything is good in case Yoichi/someone comes in to lock the IFO this weekend.
I noticed that the ISS Mean Value and CS Saturation were both RED and unhappy. (The alarms were going off, and they were both red on the MEDM screen). None of the MEDM settings seemed off kilter, so we went out to take a look at the PSL table.
Rob checked that light is indeed going to both of the ISS photodiodes (Morag and Siobhan). Next we checked that all the cables were good, and that the power to the ISS box was plugged in. In this process, Rob wiggled all the cables to check that they were plugged in. Just after doing this, the Mean Value and CS Sat were happy again. Rob thinks the current shunt connection might be bad, but we don't really know which one it was since all of the cables were jiggled between our checking the screens.
Right now, everything is happy again, but as with all bad-cabling-problems, we'll probably see this one again.
I don't know why in particular the connection decided to spaz out this afternoon...I don't think anyone opened the PSL table before Rob and I went to investigate. I was working on the PMC servo (checking the LO levels...to be posted in a couple minutes), but didn't have anything to do with the ISS. After I was done, I put everything back, and locked the PMC and the MC, and everything was good, until some time later when the ISS started flipping out.
I hereby award the previous rainbow transfer functions the plot innovation of the month award for its use of optical frequency to denote CARM offset.
The attached movie here shows the sensing matrix (minus MICH) as a function of CARM offset. There are 3 CARM signals plotted:
GREEN - tonights starting CARM signal - REFL_DC
RED - my favorite CARM signal - REFL 166 I
CYAN - runner up CARM signal - POX 33 I
I tried to play an .avi file on allegra. In a normal universe this would be easy, but because its linux I was foiled.
The default video player (Totem) doesn't play .avi or .wmv format. The patches for this work in Suse but not Fedora. Kubuntu but not CentOS, etc.I also tried installing Kplayer, Kaffeine, mplayer, xine, Aktion, Realplay, Helix, etc. They all had compatibility issues with various things but usuallylibdvdread or some gstreamer plugin.So I pressed the BIG update button. This has now started and allegra may never recover. The auto update wouldn't work in default mode becauseof the libdvdread and gstreamer-ugly plugins, so I unchecked those boxes. I think we're going to have this problem as long as we used any kind ofadvanced gstreamer stuff for the GigE cameras (which is unavoidable).
I've plotted some transfer functions showing the response at POB DC to laser frequency (phase) noise. There are transfer functions for multiple CARM offsets. Basically, the transfer function looks like the DARM transfer function when the CARM is at zero offset, and is super-wonky elsewhere. POB-DC is not a good CARM signal for intermediate stages of lock acquisition in a dual-recycled interferometer. We should look into switching back to REFL-DC.
Here are the corresponding transfer functions for REFL-DC.
Last night, we put the IFO in FP Michelson configuration. We took transfer functions of CARM and DARM, first using CM excitations directly on the ETMs, and then using modulations of the laser frequency via MC excitation. We found that there was basically no coupling into DARM using the MC excitation, but that there was coherence in DARM using the ETM excitation. Therefore, I tuned the ETM common mode in the output matrix. I did this by taking transfer functions of PD1_Q with PD2_I (see attached plot). I changed the drdown_bang script to set C1:LSC-BTMTRX_14 0.98 and C1:LSC-BTMTRX_24 1.02.
Cold cathode gauge CC4 is reading normal.
CC1 is glitching, it is probably dirty.
CC2 is fluctuating too much and it is cutting out for 6-7 minutes. It must be insulated by deposits and there is no emission current.
I think the same goes for P1
They will have to be replaced at the next vent
While waiting for the installation of the 32-bit Matlab 2009a to finish, I tried updating our seisBLRMS.m code.
Although DMF is in SVN, we forgot to check it out and so the directory where we have been doing our mods is not a working copy and our changes have not been captured: Shame.
We will probably have to wipe out the existing SVN trunk of DMF and re-import the directory after checking with Yoichi for SVN compliance.
Also wrote a script: LSC/x2mc, which will transition from regular ETM based X Arm locking to the MC2 based locking. It ran once OK, but I get a segfault on the 'trianglewave' which was trying to run the 'ezcastep' perl script which was calling 'ezcastep.bin'.
I also restarted the seisBLRMS.m on a terminal on Mafalda in the new Matlab 2009a to see if it loses its NDS connection like it did with 2007a. I also reduced the 'delay' parameter to 4 minutes and the 'interval' to 1 minute. This should be so that the total delay is now 5 minutes between seismic noise and seismic trend.
Old Guralp is hooked back up, the new one is sitting next to it, disconnected for now.
After many, many "it'll be there in 2 weeks" from the Guralp people, our seismometer is finally back!
I have it plugged into the Guralp breakout box's Channel 1xyz (so I have unplugged the other Guralp). Both of the Guralp's are currently sitting under the MC1/MC3 chamber.
Before we can have both Guralps up and running, I need to stuff the next 3 channels of the breakout box (back in the fall, I only had Caryn do 1x, 1y, 1z, and now I need 2x, 2y and 2z done with the fancy low-noise resistors), so all the gains match between the 2 sets of channels.
I'm leaving the new Guralp plugged in so we can see how it behaves for the next couple days, until I take out the breakout box for stuffing.
Rga scan of day 231 since pumpdown pd66-m-d231
m stands for maglev pumping speed, vacuum normal condition of valves,
cc4 cold cathode gauge at the rga location,
cc1 is real ifo pressure from the 24" tube at the pumpspool,
PEM-count temp: vac envelope temp at the top of IOO chamber
Thanks to Joe B who made the SRS RGA working with linux
Last data file logged at 2008 Oct 24 with old Dycor unit
First data file logged at 2009 Feb 10 with SRS
The Caltech gasoline storage tank is being upgraded.
They are jack hammering and digging with bulldozer 50 yards south of ETMY
Eric Gustafson is handling the old HP4291A rehabilitation. Tarac picked both units up today.
March of 2008 Tucker Electronics failed to fix it's intermittent ~25MHz 0.5V oscillation at the swept sine output
See 40m-elog id:398 on 3-24-2008 by Rob Ward
ETMY sus damping was found to be tripped.
It was retored.
All fluorecent light were turned off. Please try to conserve some energy.