Looking back through the elog, 1mtorr is the pressure at which it is deemed safe to send the full power beam into the IMC. After replacing the HR mirror in the MCREFL path with a 10% reflective BS, I just cranked the power back up. IMC is locked. With the increased exposure on the MC2T camera, lots of new scattered light has become visible.
While proceeding with the interferometer recovery, I noticed that there appeared to be no light on WFS2. I confirmed on the AP table that the beam was indeed hitting the QPD, but the DC quadrants are all returning 0. Looking back, it appears that the failure happened on Monday 26 October at ~6pm local time. For now, I hand-aligned the IMC and centered the beams on the WFS1 and MC2T QPDs - MCT is ~15000 cts and MC REFL DC is ~0.1, all consistent with the best numbers I've been able to obtain in the past. I don't think the servo will work without 1 sensor without some retuning of the output matrix.
It would appear that both the DC and RF outputs of WFS2 are affected - I dithered the MC2 optic in pitch (with the WFS loop disabled) at 3.33 Hz, the transmission and WFS1 sensors see the dither but not WFS2. It could be that I'm just not well centerd on the PD, but by eye, I am, so it would appear that the problem is present in both the DC and RF signal paths. I am not going into the PD head debugging today.
I want to test out an AS port WFS now that I have all the parts in hand - I guess the Michelson / PRMI will suffice until I make the ALS noise good again, and anyways, there is much assembly work to be done. Overnight, I'm repeating the suspension eigenmode measurement.
The results from the ringdown are attached - in summary:
I had to go through five SR560s in the lab yesterday evening to find one that had the expected 4 nV/rtHz input noise and worked on battery power. To confirm that the batteries were charged, I left 4 of them plugged in overnight. Today, I confirmed that the little indicator light on the back is in "Maintain" and not "Charge". However, when I unplug the power cord, they immediately turn off.
One of the units has a large DC output offset voltage even when the input is terminated (though it is not present with the input itself set to "GND" rather than DC/AC). Do we want to send this in for repair? Can we replace the batteries ourselves?
I now think the excess noise in this circuit could be coming from the KEPCO switching power supply (in fact, the supplies are linear, and specd for a voltage ripple at the level of <0.002% of the output - this is pretty good I think, hard to find much better).
All component references are w.r.t. the schematic. For this test, I decided to stuff a fresh channel on the board, with new components, just to rule out some funky behavior of the channel I had already stuffed. I decoupled the HV amplifier stage and the Acromag DAC noise filtering stages by leaving R3 open. Then, I shorted the non-inverting input of the PA95 (i.e. TP3) to GND, with a jumper cable. Then I measured the noise at TP5, using the AC coupling pomona box (although in principle, there is no need for this as the DC voltage should be zero, but I opted to use it just in case). The characteristic bump in the spectra at ~100Hz-1kHz was still evident, see the bottom row of Attachment #1. The expected voltage noise in this configuration, according to my SPICE model, is ~10 nV/rtHz, see the analysis note.
As a second test, I decided to measure the voltage noise of the power supply - there isn't a convenient monitor point on the circuit to directly probe the +/- HV supply rails (I didn't want any exposed HV conductors on the PCB) - so I measured the voltage noise at the 3-pin connector supplying power to the 2U chassis (i.e. the circuit itself was disconnected for this measurement, I'm measuring the noise of the supply itself). The output is supposedly differential - so I used the SR785 input "Float" mode, and used the Pomona AC coupling box once again to block the large DC voltage and avoid damage to the SR785. The results are summarized in the top row of Attachment #1.
The shape of the spectra suggests to me that the power supply noise is polluting the output noise - Koji suggested measuring the coherence between the channels, I'll try and do this in a safe way but I'm hesitant to use hacky clips for the High Voltage. The PA95 datasheet says nothing about its PSRR, and seems like the Spice model doesn't include it either. It would seem that a PSRR of <60dB at 100 Hz would explain the excess noise seen in the output. Typically, for other Op-Amps, the PSRR falls off as 1/f. The CMRR (which is distinct from the PSRR) is spec'd at 98 dB at DC, and for other OpAmps, I've seen that the CMRR is typically higher than the PSRR. I'm trying to make a case here that it's not unreasonable if the PA95 has a PSRR <= 60dB @100 Hz.
So what are the possible coupling mechanisms and how can we mitigate it?
What do the analog electronics experts think? I may be completely off the rails and imagining things here.
Update 2130: I measured the coherence between the positive supply rail and the output, under the same conditions (i.e. HV stage isolated, input shorted to ground). See Attachment #2 - the coherence does mirror the "bump" seen in the output voltage noise - but the coherence is. only 0.1, even with 100 averages, suggesting the coupling is not directly linear - anyways, I think it's worth it to try adding some extra decoupling, I'm sourcing the HV 10uF capacitors now.
Shruti picked it up @4pm.
It is stored along with the cables that arrived a few weeks ago, awaiting the gauges which are now expected next week sometime.
Where do we want to install the interface and readout electronics for the AS port WFS? Options are:
There isn't much difference in terms of cable length that will be required - I believe the AS WFS is going to go on the AP table even in the new optical layout and not on the ITMY in-air oplev table?
The project requires a large number of new electronics modules. Here is a short update and some questions I had:
Approximately half of the assembly of the various electronics is now complete. The basic electrical testing of the interface chassis and demod chassis are also done (i.e. they get power, the LEDs light up, and are stable for a few minutes). Detailed noise and TF characterization will have to be done.
Attachment #1 - Proposed mods for 40m RF freqs.
Attachment #2 - Modelled TFs for the case where all the notches are stuffed, and where only the 2f notch is stuffed.
Attachment #3 - Modelled TFs for the case where all the notches are stuffed, and where only the 2f notch is stuffed.
Any other red flags anyone sees before I finish stuffing the board?
WFS head and housing. Need to finalize the RF transimpedance gain (i.e. the LC resonant part), and also decide which notches we want to stuff.
Optics --> Cabinet at south end (Attachment #1)
Scanned datasheets--> wiki. It would be good if someone can check the specs against what was ordered.
Five Agilent pressure gauges were delivered to the 40m. It is stored with the controller and cables in the office area. This completes the inventory for the gauge replacement - we have all the ordered parts in hand (though. not necessarily all the adaptor flanges etc). I'll see if I can find some cabinet space in the VEA to store these, the clutter is getting out of hand again...
in addition, the spare gate valve from LHO was also delivered today to the 40m. It is stored at EX with the other spare valves.
On the call last week, I claimed that there isn't much hope of directly measuring Ponderomotive Squeezing in aLIGO without some significant configurational changes. Here, I attempt to quantify this statement a bit, and explicitly state what I mean by "significant configurational changes".
The I/O relations will generally look something like:
The. magnitudes of the matrix elements C_12 and C_21 (i.e. phase to amplitude and amplitude to phase coupling coefficients) will encode the strength of the Ponderomotive squeezing.
For the inital study, let's assume DC readout (since there isn't a homodyne readout yet even in Advanced LIGO). This amounts to setting in the I/O relations, where the former angle is the "homodyne phase" and the latter is the "SRC detuning". For DC readout, the LO quadrature is fixed relative to the signal - for example, in the usual RSE operation, . So the quadrature we will read out will be purely (or nearly so, for small detunings around RSE operation). The displacement noises will couple in via the matrix element. Attachment #1 and Attachment #2 show the off-diagonal elements of the "C" matrix for detunings of the SRC near RSE and SR operation respectively. You can see that the optomechanical coupling decays pretty rapidly above ~40 Hz.
In this particular case, there is no benefit to detuning the SRC, because we are assuming the homodyne angle is fixed, which is not an unreasonable assumption as the quadrature of the LO light is fixed relative to the signal in DC readout (not sure what the residual fluctuation in this quantity is). But presumably it is at the mrad level, so the pollution due to the orthogonal anti-squeezed quadrture can be ignored for a first pass I think. I also assume ~10 degrees of detuning is possible with the Finesse ~15 SRC, as the linewidth is ~12 degrees.
To see how this would look in an actual measurement, I took the data from Lee's ponderomotive squeezing paper, as an estimate for the classical noises, and plotted the quantum noise models for a few representative SRC detunings near RSE operation - see Attachment #3. The curves labelled for various phis are the quantum noise models for those SRC detunings, assuming DC readout. I fudged the power into the IFO to make my modelled quantum noise curve at RSE line up with the high frequency part of the "Measured DARM" curve. To measure Ponderomotive Squeezing unambiguously, we need the quantum noise curve to "dip" as is seen around 40 Hz for an SRC tuning of 80 degrees, and that to be the dominant noise source. Evidently, this is not the case.
The case for balanced homodyne readout:
I haven't analyzed it in detail yet - but it may be possible that if we can access other quadratures, we might benefit from rotating away from the DARM quadrature - the strength of the optomechanical coupling would decrease, as demonstrated in Attachments #1 and #2, but the coupling of classical noise would be reduced as well, so we may be able to win overall. I'll briefly investigate whether a robust measurement can be made at the site once the BHD is implemented.
I am confused by the discussion during the call today. I revisited Hartmut's paper - the circuit in Fig 6 is essentially what I am calling "only 2f_2 notch stuffed" in my previous elog. Qualitatively, the plot I presented in Attachment #2 of the preceeding elog in this thread shows the expected behavior as in Fig 8 of the paper - the impedance seen by the photodiode is indeed lower. In Attachment #1, I show the comparison - the "V(anode)/I(I1)" curve is analogous to the "PD anode" curve in Hartmut's paper, and the "V(vout)/I(I1)" curve is analogous to the "1f-out" curve. I also plot the sensitivity analysis (Attachment #2), by varying the photodiode junction capacitance between 100pF and 200 pF (both values inclusive) in 20 pF steps. There is some variation at 55 MHz, but it is unlikely that the capacitance will change so much during normal operation?
I understand the motivation behind stuffing the other notches, to reduce intermodulation effects. But the impression I got from the call was that somehow, the model I presented was wrong. Can someone help me identify the mistake?
I didn't bother to export the LTspice data and make a matplotlib plot for this quick analysis, so pardon the poor presentation. The colors run from green=100pF to grey=200pF.
An 8 channel whitening chassis was prepared and tested. I measured:
Whitening chassis. Waiting for front panels to arrive, PCBs and interface board are in hand, stuffed and ready to go. A question here is how we want to control the whitening - it's going to be rather difficult to have fast switchable whitening. I think we can just fix the whitening state. Another option would be to control the whitening using Acromag BIO channels.
FYI, there is this. Seems pretty well maintained, and so might be more useful in the long run. The available catalog of instruments is quite impressive - TC200 temp controller and SRS345 func gen are included and are things we use in the lab. maybe you can make a pull request to add MDT694B (there is some nice API already built I think). We should also put our netgpibdata stuff and the vacuum gauge control (basically everything that isn't rtcds) on there (unless there is some intellectual property rights issues that the Caltech lawyers have to sort out).
Given the similarities between the MDT694B (single channel piezo controller) and TC200 (temperature controller) serial interfaces, I added the pyserial driver here.
*Warning* this first version of the driver remains untested
As discussed at the meeting, I commenced the recovery of the CDS status at 1750 local time.
Single arm POX/POY locking was checked, but not much more. Our IMC WFS are still out of service so I hand aligned the IMC a bit, IMC REFL DC went from ~0.3 to ~0.12, which is the usual nominal level.
The summary pages were in a sad state of disrepair - the daily jobs haven't been running for > 1 month. I only noticed today because Jordan wanted to look at some vacuum trends and I thought summary pages is nice for long term lookback. I rebooted it just now, seems to be running. @Tega, maybe you want to set up some kind of scripted health check that also sends an alert.
I'm thinking of making some modifications to the RF distribution box in 1X2, so as to have an extra 55 MHz pickoff. Koji already proposed some improvements to the layout in 2015. I've marked up his "Possible Improvement" page of the document in Attachment #1, with my proposed modifications. I believe it will be possible to get 15-16 dBm of signal into a 4 way RF splitter in the quad demod chassis. With the insertion loss of the splitter, we can have 9-10 dBm of LO reaching each demod board, which will then be boosted to +20 dBm by the Teledyne on board. The PE4140 mixer claims to require only -7 dBm of LO signal. So we have quite a bit of headroom here - as long as we limit the RF signal to 0dBm (=0.5 Vpp from the LMH6431 opamp at 55 MHz, we shouldn't be having a much larger signal anyways), we should be just fine with 15 dBm of LO power (which is what we will have after the division into the I and Q paths, and nominal insertion losses in the transmission path). These numbers may be slight overestimates given the possible degradation of the RF amps over the last 10 years, but shouldn't be a show-stopper.
Do the RF electronics experts agree with my assessment? If so, I will start working on these mods tomorrow. Technically, the splitter can be added outside the box, but it may be neater if we package it inside the box.
The latest greatest UPS has been delivered. I will move it to near the vacuum rack in its packaging for storage. It weighs >100lbs so care will have to be taken when installing - can the rack even support this?
Since we will have several new 1U / 2U aLIGO style electronics chassis installed in the racks, it is desirable to have a more compact power distribution solution than the fusable terminal blocks we use currently.
I did a quick walkaround of the lab and the electronics rack today. I estimate that we will need 5 units of the 24 V and 5 units of the 18 V power strips. Each end will need 1 each of 18 V and 24 V strips. The 1Y1/1Y2/1Y3 (LSC/OMC/BHD sus) area will be served by 1 each 18 V and 24 V. The 1X1/1X2 (IOO) area will be served by 1 each 18 V and 24 V. The 1X5/1X6 (SUS Shadow sensor / Coil driver) area will be served by 1 each of 18 V and 24 V. So I think we should get 7 pcs of each to have 2 spares.
Most of the chassis which will be installed in large numbers (AA, AI, whitening) supports 24V DC input. A few units, like the WFS interface head, OMC driver, OMC QPD interface, require 18V. It is not so clear what the input voltage for the Satellite box and Coil Drivers should be. For the former, an unregulated tap-off of the supply voltage is used to power the LT1021 reference and a transistor that is used to generate the LED drive current for the OSEMs. For the latter, the OPA544 high current opamp used to drive the coil current has its supply rails powered by again, an unregulated tap-off of the supply voltage. Doesn't seem like a great idea to drive any ICs with the unregulated switching supply voltage from a noise point of view, particularly given the recent experience with the HV coil driver testing and the PSRR, but I think it's a bit late in the game to do anything about this. The datasheet specs ~50 dB of PSRR on the negative rail, but we have a couple of decoupling caps close to the IC and this IC is itself in a feedback loop with the low noise AD8671 IC so maybe this won't be much of an issue.
For the purposes of this discussion, I think both Satellite Amp and Coil Driver chassis can be driven with +/- 24 V DC.
On a side note - after the upgrade will the "Satellite Amplifiers" be in the racks, and not close to the flange as they currently are? Or are we gonna have some mini racks next to the chambers? Not sure what the config is at the sites, and if the circuits are designed to drive long cables.
I removed the Frequency Generation box from the 1X2 rack. For the time being, the PSL shutter is closed, since none of the cavities can be locked without the RF modulation source anyways.
Prior to removal, I did the following:
One thing I noticed was that we're using very stiff coax cabling (RG405) inside this box? Do we need to stick with this option? Or can we use the more flexible RG316? I guess RG405 is lower loss, so it's better. I can't actually find any measurement of the shielding performance in my quick google searching but I think the claim on the call yesterday was that RG405 with its solder soaked braids offer superior shielding.
Before doing any modification you should check how much the distributed powers are at the ports.
Also your modification will change the relative phase between 11MHz and 55MHz.
Can you characterize how much phase difference you have between them, maybe using the modulation of the main marconi? And you might want to adjust it to keep the previous value (or any new value) after the modification by adding a cable inside?
I'm open to either approach. If the full replacement requires a lot of machining, maybe I will stick to just the 55 MHz line. But if only a couple of new holes are required, it might be advantageous to do the revamp while we have the box out? What do you think?
BTW, now that I look more closely at the RF chain, I have several questions:
I guess it is feasible to have +17 dBm of 55 MHz signal to plug into the Quad Demod chassis - e.g. drive the 55 MHz input with 20 dBm, pick off 3dBm to the front panel for ASC. Then we can even have several "spare" 55 MHz outputs and still satisfy the 9 dBm input that the ZHL-2 in the 55 MHz chain wants (though again, isn't this dangerously close to the 1dB compression point?). The design doc claims to have done some Optickle modeling, so I guess there isn't really any issue?
Are you going to full replacement of the 55MHz system? Or just remove the 7dBm and then implement the proposed modification for the 55MHz line?
This turned out to be a much more involved project than I expected. The layout is complete now, but I found several potentially damaged sections of cabling (the stiff cables don't have proper strain relief near the connectors). I will make fresh cables tomorrow before re-installing the unit in the rack. Several changes have been made to the layout so I will post more complete details after characterization and testing.
I was poring over minicircuits datasheets today, and I learned that the minicircuits bandpass filters (SBP10.7 and SBP60) are not bi-directional! The datasheet clearly indicates that the Male SMA connector is the input and the Female SMA connector is the output. Almost all the filters were installed the other way around 😱 . I'll install them the right way around now.
This work is now complete. The box was characterized and re-installed in 1X2. I am able to (briefly) lock the IMC and see PDH fringes in POX and POY so the lowest order checks pass.
Even though I did not deliberately change anything in the 29.5 MHz path, and I confirmed that the level at the output is the expected 13 dBm, I had to lower then IN1 gain of the IMC servo by 2dB to have a stable lock - should confirm if this is indeed due to higher optical gain at the IMC error point, or some electrical funkiness. I'm not delving into a detailed loop characterization today - but since my work involved all elements in the RF modulation chain, some detailed characterization of all the locking loops should be done - I will do this in the coming week.
After tweaking the servo gains for the POX/POY loops, I am able to realize the single arm locks as well (though I haven't dont the characterization of the loops yet).
I'm leaving the PSL shutter open, and allowing the IMC autolocker to run. The WFS loops remain disabled for now until I have a chance to check the RF path as well.
Unrelated to this work: Koji's swapping back of the backplane cards seems to have fixed the WFS2 issue - I now see the expected DC readbacks. I didn't check the RF readbacks tonight.
Update 7 Dec 2020 1 pm: A ZHL-2 with heat sink attached and a 11.06 MHz Wenzel source were removed from the box as part of this work (the former was no longer required and the latter wasn't being used at all). They have been stored in the RF electronics cabinet along the east arm.
The MC1 suspension has begun to show evidence of glitches again, from Friday/Saturday. You can look at the suspension Vmon tab a few days ago and see that the excess fuzz in the Vmon was not there before. The extra motion is also clearly evident on the MCREFL spot. I noticed this on Saturday evening as I was trying to recover the IMC locking, but I thought it might be Millikan so I didn't look into it further. Usually this is symptomatic of some Satellite box issues. I am not going to attempt to debug this anymore.
There seems to be significant phase loss in the TTFSS path, which is limiting the IMC OLTF to <100 kHz.
See Attachment #1 and #2. The former shows the phase loss, while the latter is just to confirm that the optical gain of the error point is roughly the same, since I noticed this after working on and replacing the RF frequency distribution unit. Unfortunately there have been many other changes also (e.g. the work that Rana and Koji did at the IMC rack, swapping of backplane controls etc etc - maybe they have an OLTF measurement from the time they were working?) so I don't know which is to blame. Off the top of my head, I don't see how the RF source can change the phase lag of the IMC servo at 100 kHz. The only part of the IMC RF chain that I touched was the short cable inside the unit that routes the output of the Wenzel source to the front panel SMA feedthrough. I confirmed with a power meter that the power level of the 29.5 MHz signal at that point is the same before and after my work.
The time domain demod monitor point signals appear somewhat noisier in todays measurement compared to some old data I had from 2018, but I think this isn't significant. Once the SR785 becomes available, I will measure the error point spectrum as well to confirm. One thing I noticed was that like many of our 1U/2U chassis units, the feedthrough returns are shorted to the chassis on the RF source box (and hence presumably also to the rack). The design doc for this box makes many statements about the precautions taken to avoid this, but stops short of saying if the desired behavior was realized, and I can't find anything about it in the elog. Can someone confirm that the shields of all the connectors on the box were ever properly isolated? My suspicion is that the shorting is happening where the all-metal N-feedthroughs touch the drilled surfaces on the front panel - while the front and back surfaces of the panel are insulating, the machined surfaces are not.
This is an unacceptable state but no clear ideas of how to troubleshoot quickly (without going piece by piece into the IMC servo chain) occur to me. I still don't understand how the freq source work could have resulted in this problem but I'm probably overlooking something basic. I'm also wondering why the differential receiving at the TTFSS error point did not require a gain adjustment of the IMC servo? Shouldn't the differential-receiving-single-ended-sending have resulted in an overall x0.5 gain?
Update 8 Dec 1200: To test the hypothesis, I bypassed the SR560 based differential receiving and restored the original config. I am then able to run with the original gain settings, and you see in Attachment #4 that the IMC OLTF UGF is back above 100 kHz. It is still a little lower than it was in June 2019, not sure why. There must be some saturation issues somewhere in the signal chain because I cannot preserve the differential receiving and retain 100 kHz UGF, either by raising the "VCO gain" on the MC servo board, setting the SR560 to G=2, or raising the "Common Gain Adjust" on the FSS box by 6 dB. I don't have a good explanation for why this worked for some weeks and failed now - maybe some issue with the SR560? We don't have many working units so I didn't try switching it.
So either there is a whole mess of lines or the frequency noise suppression is limited. Sigh.
In favor of keeping the same servo gains, I tuned the digital demod phases for the POX and POY photodiode signals to put as much of the PDH error signal in the _I quadrature as possible. The changes are summarized below:
The old locking settings seem to work fine again. This setting isn't set by the ifoconfigure scripts when they do the burt restore - do we want it to be?
Attachments #1 and #2 show some spectra and TFs for the POX/POY loops. In Attachment #2, the reference traces are from the past, while the live traces are from today. In fact, to have the same UGF as the reference traces (from ~1 year ago), I had to also raise the digital servo loop gain by ~20%. Not sure if this can be put down to a lower modulation depth - at least, at the output on the freq ref box, I measured the same output power (at the 0dB variable attenuator gain setting we nominally run in) before and after the changes. But I haven't done an optical measurement of the modulation depth yet. There is also a hint of lesser phase available at ~100 Hz now compared to a year ago.
I measured the modulation depth at 11 MHz andf 55 MHz using an optical beat + PLL setup. Both numbers are ~0.2 rad, which is consistent with previous numbers. More careful analysis forthcoming, but I think this supports my claim that the optical gain for the PDH locking loops should not have decreased.
I updated the ndscope on rossa to a bleeding edge version (0.7.9+dev0) which has many of the fixes I've requested in recent times (e.g. direct PDF export, see Attachment #1). As usual if you find issue, report it on the issue tracker. The basic functionality for looking at signals seems to be okay so this shouldn't adversely impact locking efforts.
In hindsight - I decided to roll-back to 0.7.9, and have the bleeding edge as a separate binary. So if you call ndscope from the command line, you should still get 0.7.9 and not the bleeding edge.
The ITMX Oplev (installed in March 2019) was near end of life judging by the SUM channel (see Attachment #1). I replaced it yesterday evening with a new HeNe head. Output power was ~3.25 mW. The head was labelled appropriately and the Oplev spot was recentered on its QPD. The lifetime of ~20 months is short but recall that this HeNe had already been employed as a fiber illuminator at EX and so maybe this is okay.
Loop UGFs and stability margins seem acceptable to me, see Attachment #2-#3.
Continuting the IFO recovery - I am unable to recover similar levels of TRX RIN as I had before. Attachment #1 shows that the TRX RIN is ~4x higher in RMS than TRY RIN (the latter is commensurate with what we had previously). The excess is dominated by some low frequency (~1 Hz) fluctuations. The coherence structure is confusing - why is TRY RIN coherent with IMC transmission at ~2 Hz but not TRX? But anyways, doesn't look like its intensity fluctuations on the incident light (unsurprisingly, since the TRY RIN was okay). I thought it may be because of insufficient low-frequency loop gain - but the loop shape is the same for TRX and TRY. I confirmed that the loop UGF is similar now (red trace in Attachment #2) as it was ~1 month ago (black trace in Attachment #2). Seismometers don't suggest excess motion at 1 Hz. I don't think the modulation depth at 11 MHz is to blame either. As I showed earlier, the spectrum of the error point is comparable now as it was previously.
What am I missing?
I suspect what happened here is that the IP didn't get updated when we went from the 131.215.113.xxx system to 192.168.113.xxx system. I fixed it now and can access the web interface. This system is now ready for remote debugging (from inside the martian network obviously). The IP is 192.168.113.90.
Managed to pull this operation off without crashing the RFM network, phew.
BTW, a windows laptop that used to be in the VEA (I last remember it being on the table near MC2 which was cleared sometime to hold the spare suspensions) is missing. Anyone know where this is ?
As discussed at the meeting, I decided to effect a satellite box swap for the misbehaving MC1 unit. I looked back at the summary pages Vmon for the SRM channels, and found that in the last month or so, there wasn't any significant evidence of glitchiness. So I decided to effect that swap at ~4pm today. The sequence of steps was:
One thing I was reminded of is that the motion of the MC1 optic by controlling the bias sliders is highly cross-coupled in pitch and yaw, it is almost diagonal. If this is true for the fast actuation path too, that's not great. I didn't check it just now.
While I was working on this, I took the opportunity to also check the functionality of the RF path of the IMC WFS. Both WFS heads seem to now respond to angular motion of the IMC mirror - I once again dithered MC2 and looked at the demodulated signals, and see variation at the dither frequency, see Attachment #1. However, the signals seem highly polluted with strong 60 Hz and harmonics, see the zoomed-in time domain trace in Attachment #2. This should be fixed. Also, the WFS loop needs some re-tuning. In the current config, it actually makes the MC2T RIN worse, see Attachment #3 (reference traces are with WFS loop enabled, live traces are with the loop disabled - sorry for the confusing notation, I overwrote the patched version of DTT that I got from Erik that allows the user legend feature, working on getting that back).
Around 7pm, the UPS at the vacuum rack seems to have failed. Don't ask me why I decided to check the vacuum screen 10 mins after the failure happened, but the point is, this was a silent failure so the protocols need to be looked into.
Going to the rack, I saw (unsurprisingly) that the 120V UPS was off.
For now, I think this is a safe state to leave the system in. Unless I hear otherwise, I will leave it so - I will be in the lab another hour tonight (~10pm).
Some photos and a screen-cap of the Vac medm screen attached.
I don't buy this story - P2 only briefly burped around GPStime 1291608000 which is around 8pm local time, which is when I was recovering the system.
Today. Jordan talked to Jon Feicht - apparently there is some kind of valve in the TP2 forepump, which only opens ~15-20 seconds after turning the pump on. So the loud sound I was hearing yesterday was just some transient phenomenon. So today morning at ~9am, we turned on TP2. Once again, PTP2 pressure hovered around 500 torr for about 15-20 seconds. Then it started to drop, although both Jordan and I felt that the time it took for the pressure to drop in the range 5 mtorr - 1 mtorr was unusually long. Jordan suspects some "soft-start" feature of the Turbo Pumps, which maybe spins up the pump in a more controlled way than usual after an event like a power failure. Maybe that explains why the pressure dropped so slowly? One thing is for sure - the TP2 controller displayed "TOO HIGH LOAD" yesterday when I tried the first restart (before migrating everything to the older UPS unit). This is what led me to interpret the loud sound on startup of TP2 to indicate some issue with the forepump - as it turns out, this is just the internal valve not being opened.
Anyway, we left TP2 on for a few hours, pumping only on the little volume between it and V4, and PTP2 remained stable at 20 mtorr. So we judged it's okay to open V4. For today, we will leave the system with both TP2 and TP3 backing TP1. Given the lack of any real evidence of a failure from TP2, I have no reason to believe there is elevated risk.
As for prioritising UPS swap - my opinion is that it's better to just replace the batteries in the UPS that has worked for years. We can run a parallel reliability test of the new UPS and once it has demonstrated stability for some reasonable time (>4 months), we can do the swap.
I was able to clear the FAULT indicator on the new UPS by running a "self-test". pressing and holding the "mute" button on the front panel initiates this test according to the manual, and if all is well, it will clear the FAULT indicator, which it did. I'm still not trusting this unit and have left all units powered by the old UPS.
Update 1100 Dec 11: The config remained stable overnight so today I reverted to the nominal config of TP3 pumping the annuli and TP2 backing TP1 which pumps the main volume (through the partially open RV2).
According to the Tripp Lite manual, the FAULT icon indicates "the battery-supported outlets are overloaded." The failure of the TP2 dry pump appears to have caused this. After the dry pump failure, the rising pressure in the TP2 foreline caused TP2's current draw to increase way above its normal operating range. Attachment 1 shows anomalously high TP2 current and foreline pressure in the minutes just before the failure. The critical system-wide failure is that this overloaded the UPS before overloading TP2's internal protection circuitry, which would have shut down the pump, triggering interlocks and auto-notifications.
I gave one Noliac PZT from the two spare in the metal PMC kit to Paco. There is one spare left in the kit.
I acquired several spare OSEMs (in unknown condition) from Paco. They are stored alongside the shipment from UF.
The assembly of the head is nearly complete, I thought I'd do some characterization before packaging everything up too nicely. I noticed that the tapped holes in the base are odd-sized. According to the official aLIGO drawing, these are supposed to be 4-40 tapped, but I find that something in between 2-56 and 4-40 is required - so it's a metric hole? Maybe we used some other DCC document to manufacture these parts - does anyone know the exact drawings used? In the meantime, the circuit is placed inside the enclosure with the back panel left open to allow some tuning of the trim caps. The front panel piece for mounting the SMA feedthroughs hasn't been delivered yet so hardware-wise, that's the last missing piece (apart from the aforementioned screws).
Attachment #1 - the circuit as stuffed for the RF frequencies of relevance to the 40m.
Attachment #2 - measured TF from the "Test Input" to Quadrant #1 "RF Hi" output.
Update 11 Dec: For whatever reason, whoever made this box decided to tap 4-40 holes from the bottom (i.e. on the side of the base plate), and didn't thread the holes all the way through, which is why I was unable to get a 4-40 screw in there. To be fair the drawing doesn't specify the depth of the 4-40 holes to be tapped. All the taps we have in the lab have a maximum thread length of 9/16" whereas we need something with at least 0.8" thread length. I'll ask Joe Benson at the physics workshop if he has something I can use, and if not, I'll just drill a counterbore on the bottom side and use the taps we have to go all the way through and hopefully that does the job.
The front panel I designed for the SMA feedthroughs arrived today. Unfortunately, it is impossible for the D-sub shaped holes in this box to accommodate 8 insulated SMA feedthroughs (2 per quadrant for RF low and RF high) - while the actual SMA connector doesn't occupy so much space, the plastic mold around the connector and the nut to hold it are much too bulky. For the AS WFS application, we will only need 4 so that will work, but if someone wants all 8 outputs (plus an optional 9th for the "Test Input"), a custom molded feedthrough will have to be designed.
As for the 170 MHz feature - my open loop modeling in Spice doesn't suggest a lack of phase margin at that frequency so I'm not sure what the cause is there. If this is true, just increasing the gain won't solve the issue (since there is no instability at least by the phase margin metric). Could be a problem with the "Test Input" path I guess. I confirmed it is present in all 4 quadrants.
I installed a DC power strip (24 V variant, 12 outlets available) on the NW upright of the 1X1 rack. This is for the AS WFS. Seems to work, all outlets get +/- 24 V DC.
The FSS_RMTEMP channel is very noisy after this work. I'll look into it, but probably some Acromag grounding issue.
In the afternoon, Jordan and I also laid out 4x SMA LMR240 cables and 1x DB15 M/F cable from 1X2 to the NE corner of the AP table via the overhead cable trays.
I think the WFS head performs satisfactorily.
Details and remarks:
If the RF experts see some red flags / think there are more tests that need to be performed, please let me know. Big thanks to Chub for patiently supporting this build effort, I'm pleasantly surprised it worked.
As I was working on the IFO re-alignment just now, the rfm errors popped up again. I don't see any useful diagnostics on the web interface.
Do we want to take this opportunity to configure jumpers and set up the rogue master as Rolf suggested? Of course there's no guarantee that will fix anything, and may possibly make it impossible to recover the current state...
I installed 4 chassis in the rack 1X2 (characterization on the E-bench was deemed satisfactory, I will upload the analysis later). I ran out of hardware to make power cables so only 2 of them are powered right now (1 32ch AA chassis and 1 WFS head interface). The current limit on the +24V Sorensens was raised to allow for similar margin to the limit with the increased current draw.
While I definitely bumped various cables, I don't seem to have done any lasting damage to the CDS system (the RFM errors remain of course).
The CDS model change required to get the AS WFS signals into the RTCDS system are rather invasive.
In terms of computational load, the c1ioo model seems to be able to handle the extra load no issues - ~35us/60us per cycle. The RFM model shows no extra computational time.
After this work, the IMC locking and POX/POY locking, and dither alignment servos are working okay. So I have some confidence that my invasive work hasn't completely destroyed everything. There is some hardware around the rear of 1X2 that I will clear tomorrow.
Koji fixed the problematic channel - the issue was a bad solder joint on the input resistors to the THS4131. The board was re-installed. I also made a custom 2x4-pin LEMO-->DB9 cable, so we are now recording the PMC and FSS ERR/CTRL channel diagnostics again (spectra tomorrow). Note that Ch32 is recording some sort of DuoTone signal and so is not usable. This is due to a misconfiguration - ADC0 CH31 is the one which is supposed to be reserved for this timing signal, and not ADC1 as we currently have. When we swap the c1ioo hosts, we should fix this issue.
I also did most of the work to make the MEDM screens for the revised ASC topology, tried to mirror the site screens where possible. The overview screen remains to be done. I also loaded the anti-whitening filters (z:p 150:15) at the demodulated WFS input signal entry points. We don't have remote whitening switching capability at this time, so I'll test the switching manually at some point.
The main issue is that in the AA chassis I built, Ch14 (with the first channel as Ch1) has the output saturated to 28V (differential). I'm not sure what kind of overvoltage protection the ADC has - we frequently have the inputs exceed the spec'd +/-20 V (e.g. when the whitening filters are engaged and the cavity is fringing), but pending further investigation, I am removing the SCSI connection from the rear of the AA chassis.
Last night, I briefly spoke with Koji about the CDS upgrade plan. This is a follow up.
Each server needs a minimum of two peripheral devices added to the PCIe bus:
As for the second issue, the main question is if we had an open PCIe slot on the c1iscex machine to install a Dolphin card. Looks like the 2 standard slots are taken (see Attachment #1), but a "low profile" slot is available. I can't find what the exact models of the Supermicro servers installed back in 2010 are, but maybe it's this one? It's a good match visually anyways. The manual says a "riser card" is required. I don't know if such a riser is already installed.
Questions I have, Rolf is probably the best person to answer:
I don't have omnigraffle - what about uploading the source doc in a format that the excellent (and free) draw.io can handle? I think we can do a much better job of making this diagram reflect reality. There should also be a corresponding diagram for the Acromag system (but that doesn't have to be tied to this task). Megatron (scripts machine) and nodus should be added to that diagram as well.
Please send me any omissions or corrections to the layout.
the replacement was done this afternoon. The red "Replace Battery" indicator is no longer on.
As part of the hunt why the X arm IR transmission RIN is anomalously high, I noticed that the BS Oplev Servo periodically kicks the optic around - the summary pages are the best illustration of this happening. Looking back in time, these seem to have started ~Nov 23 2020. The HeNe power output has been degrading, see Attachment #1, but this is not yet at the point where the head usually needs replacing. The RIN spectrum doesn't look anomalous to me, see Attachment #2 (the whitening situation for the quadrants is different for the BS and the TMs, which explains the HF noise). I also measured the loop UGFs (using swept sine) - seems funky, I can't get the same coherence now (live traces) between 10-30 Hz that I could before (reference traces) with the same drive amplitude, and the TF that I do measure has a weird flattening out at higher frequencies that I can't explain, see Attachment #3.
The excess RIN is almost exactly in the band that we expect our Oplevs to stabilize the angular motion of the optics in, so maybe needs more investigation - I will upload the loop suppression of the error point later. So far, I don't see any clean evidence of the BS Oplev HeNe being the culprit, so I'm a bit hesitant to just swap out the head...
I've noticed that there is some phase loss in the POX/POY locking loops - see Attachment #1, live traces are from a recent measurement while the references are from Nov 4 2018. Hard to imagine a true delay being responsible to cause so much phase loss at 100 Hz. Attachment #2 shows my best effort loop modeling, I think I've got all the pieces, but maybe I missed something (I assume the analog whitening / digital anti-whitening are perfectly balanced, anyway this wasn't messed with anytime recently)? The fitter wants to add 560 us (!) of delay, which is almost 10 clock cycles on the RTS, and even so, the fit is poor (I constrain the fitter to a maximum of 600 us delay so maybe this isn't the best diagnostic). Anyway, how can this change be explained? The recent works I can think of that could have affected the LSC sensing were (i) RF source box re-working, and (ii) vent. But I can't imagine how either of these would introduce phase loss in the LSC sensing. Note that the digital demod phase has been tuned to put all the PDH signal in the "I" quadrature, which is the condition in which the measurement was taken.
Probably this isn't gonna affect locking efforts (unless it's symptomatic of some other larger problem).