To set the demod phase for RF CARM, sensed at REFL2 (REFL 166I), it suffices to set the demod phase for REFL2 to be the optimal phase for controlling SRCL in a no-arm state.
For POX33, the ideal phase for single arm locking does not yield a zero-offset CARM signal. So the offset needs to be manipulated digitally.
With no DARM offset, sweeping CARM shows an asymmetry between the state where we lock to a DARM spring and the state with a DARM anti-spring. This is why we have a link between the DARM and CARM optical springs.
For each DARM detune direction (positive or negative, spring or anti-spring), there is only one CARM direction which can yield a DC-based error signal lock with a CARM offset but no DARM offset, which is what we want.
I've plotted TRX, TRY, PD12I and PD11Q. Arm powers after locking increase for a few tens of minutes, peak out, and then decrease before lock is lost.
I should have mentioned that the AS port camera image seems to get progressively uglier over the course of these locks. Maybe we can use the JoeCam to make a movie of it.
locks last for about an hour. this was true last night as well (see "arm power curve" entries). the second lock shown here evolves differently for unknown reasons. the jumps in the arm powers of the first lock are due to turning on DC readout. length-to-angle needs tuning.
attached plot shows MC_IN1/MC_IN2. needs work.
This is supposed to be a measurement of the relative gain of the MCL and AO paths in the CM servo. We expect there to
be a more steep slope (ideally 1/f). Somehow the magnitude is very shallow and so the crossover is not stable. Possible
causes? Saturations in the measurement, broken whitening filters, extremely bad delay in the digital system? needs work.
the align script was run after the third lock here. it would have been interesting to see the arm powers in a 4th lock
It seems that the MC3 problem is intermittent (one-day trend attached). I tried to take advantage of a "clean MC3" night, but the watch script would usually fail at the transition to DC CARM and DARM. It got past this twice and then failed later, during powering up. I need to check the handoff.
Recently the watch script was having difficulty grabbing a lock for more than a few seconds. Rob discovered that the violin notch filters which were activated in the script were causing the instability. We're not sure why yet. The script seems significantly more stable with that step commented out.
We worked on tuning the DD handoff tonight. We checked the DD PD alignments and they looked fine. First I tuned the 3 demod phases to minimize offsets. Then I noticed that the post-handoff MICH xfer function needed an increase in gain to look like the pre-handoff xfer function (which has a UGF of about 25 Hz). I increased the MICH PD9_Q gain from 2 to 7 in the input matrix. But, the handoff to PRC still failed, so tomorrow we will try to find out why.
In the plot, ref0 is before MICH handoff, and ref1 is after MICH handoff. There is also a PRC trace (before PRC handooff).
Rana, Alberto, Pete
We have the DD handoff nominally working. Sometimes, increasing the SRC gain at the end makes MICH get unstable. This could be due to a non-diagonal term in the matrix, or possibly because the DRM locks in a funky mode sometimes.
To get the DD handoff working, first we tuned demod phases in order to zero the offsets in the PD signals handed-off-to. Based on transer function measurements, I set the PRC PD6_I element to 0.1, and set the PD8_I signal to 0, since it didn't seem to be contributing much. We also commented out the MICH gain increase at the end of the DD_handoff script.
It could still be more stable, but it seems to work most of the time.
I played with the DD handoff during the day. The DRM dark port was flickering like a candle flame in Dracula's castle. The demod offsets for the handoff signals looked fine. After MICH handoff, the MICH_CTRL started to get unstable at some low frequency, maybe 3 Hz (I didn't measure). So I increased the MICH gain from 0.1 to 0.17 and it settled down. PRC and SRC went fine. Then the DD_handoff script raised the MICH gain to 0.7, and an instability started to grow in MICH_CTRL (at some higher frequency). I decreased the MICH gain from 0.7 to 0.5, and it settled down and stayed stable.
We were stymied early in the evening by a surreptitiously placed, verbo-visually obfuscated command in the drstep script.
tdsavg 5 C1:LSC-PD4_DC_IN1
was causing grievous woe in the cm_step script. It turned out to fail intermittently at the command line, as did other LSC channels. (But non-LSC channels seem to be OK.) So we power cycled c1lsc (we couldn't ssh).
Then we noticed that computers were out of sync again (several timing fields said 16383 in the C0DAQ_RFMNETWORK screen). We restarted c1iscey, c1iscex, c1lsc, c1susvme1, and c1susvme2. The timing fields went back to 0. But the tdsavg command still intermittently said "ERROR: LDAQ - SendRequest - bad NDS status: 13".
The channel C1:LSC-SRM_OUT16 seems to work with tdsavg every time.
Let us know if you know how to fix this.
Did you try restarting the framebuilder?
What you type is in bold:
op440m> telnet fb40m 8087
After fixing the tp problem, I tried locking again. Grabbing and DD handoff, no problem. Died earlier than last night, handing off CARM to REFL_DC, around arm power of 4 or so. Seems to happen after turning off the moving zero, Rob says it might be touchy in daytime.
Lock acquisition is proceeding smoothly for the most part, but there is a very consistent failure point near the end of the cm_step script.
Near the end of the procedure, while in RF common mode, the sensing for the MCL path of the common mode servo is transitioned from a REFL 166I signal which comes into the LSC whitening board from the demodulator, to another copy of the signal which has passed through the common mode board, and is coming out of the Length output of the common mode board. We do this because the signal which comes through the CM board sees the switchable low-frequency boost filter, and so both paths of the CM servo (AO and MCL) can get that filter switched on at the same time.
The problem is occurring after this transition, which works reliably. However, when the script tries to remove the final CARM offset, and bring the offset to zero, lock is abruptly lost. DARM, CM, and the crossover all look stable, and no excess noise appears while looking at the DARM, CARM, MCF spectra. But lock is always lost right about the same offset.
I've seen this before. At that time, the problem was gone spontaneously the next day.
You could stop just before the offset reaches zero and then try to slowly reduce the offset manually to see where is the threshold.
Well, it hasn't gone away yet. It happened Sat, Mon, and Tues afternoon, as well as Friday. The threshold varies slightly, but is always around ~200-300 cnts. I've tried reducing the offset with the signal coming from the CM board and the signal not going through the CM board, I've also tried jumping the signal to zero (rather than a gradual reduction).
Tonight we'll measure the MC length and set the modulation frequencies, and maybe try some MZ tweaking to do RFAMMon minimization.
I had trouble getting the SRC handoff from SD to DD to work. I tried different gains, flipping the PD7 & 8 demod phases by 180 degrees, and messing with the output matrix to reduce cross-couplings in the state with MICH & PRC on DD and SRC on SD. Eventually I decided to try to make the DRM matrix diagonalization work.
It does, mostly. The handoff is now stable, and the loops all have UGFs around 100Hz. So, tonight anyways, it's possible to run senseDRM and then loadDRMImatrixData.m and run the resulting tdswrite command, and have a working handoff. I had to eliminate a few PDs (PD5 & PD10) to get it to work properly.
It would be nice if this script would measure all the PDs and allow individual setting of loop UGFs and measurement frequencies.
Lock is still being lost, right at the end of the process when trying to reduce the CARM offset to zero.
The IFO is locked, at the operating point (zero CARM offset). The problem with reducing the residual CARM offset in the last stage appears to have been because the common mode gain was getting too high, and so the loop was going unstable at high frequencies.
The cm_step script is currently a confusing mess, with all the debugging statements. I'll clean it up this afternoon and check that it still works.
Last night Rob ran senseDRM and loadDRMImatrixData and came up with the following for the input matrix:
tdswrite C1:LSC-ITMTRX_b2 0.065778 \
C1:LSC-ITMTRX_d2 2.2709 \
C1:LSC-ITMTRX_f2 2.9361 \
C1:LSC-ITMTRX_122 0.42826 \
C1:LSC-ITMTRX_b3 -0.064839 \
C1:LSC-ITMTRX_d3 -0.016913 \
C1:LSC-ITMTRX_f3 -0.021576 \
C1:LSC-ITMTRX_123 -0.0025243 \
C1:LSC-ITMTRX_b5 0.3719 \
C1:LSC-ITMTRX_d5 1.3109 \
C1:LSC-ITMTRX_f5 -0.16412 \
C1:LSC-ITMTRX_125 0.39574 \
C1:LSC-ITMTRX_33 0 \
C1:LSC-ITMTRX_42 0 \
Today, I reran these and got the following, and DD_handoff remained happy:
tdswrite C1:LSC-ITMTRX_b2 -0.10329 \
C1:LSC-ITMTRX_d2 2.0344 \
C1:LSC-ITMTRX_f2 3.2804 \
C1:LSC-ITMTRX_122 0.22516 \
C1:LSC-ITMTRX_b3 -0.076292 \
C1:LSC-ITMTRX_d3 -0.014603 \
C1:LSC-ITMTRX_f3 -0.12101 \
C1:LSC-ITMTRX_123 0.0054128 \
C1:LSC-ITMTRX_b5 0.33521 \
C1:LSC-ITMTRX_d5 1.1425 \
C1:LSC-ITMTRX_f5 -0.32759 \
C1:LSC-ITMTRX_125 0.25877 \
C1:LSC-ITMTRX_33 0 \
C1:LSC-ITMTRX_42 0 \
I wanted to remeasure with the canonical output matrix (-0.7 from MICH to PRM and 0.7 from MICH to SRM), but the DRM freaked out when MICH to PRM went below -0.3.
With the common mode servo bandwidth above 30kHz and the BOOST on (1), I was able to switch on the test mass dewhitening. Finally.
Here's a noise spectrum of the RSE interferometer, in anti-spring mode, with RF readout. I'd say the calibration is "loose."
I used the Buonanno & Chen modification of the KLMTV IFO transfer functions to model the DARM opto-mechanical response. I just guessed at the quadrature, and normalized the optical gain at the frequency of the calibration line used (927Hz, not visible on the plot).
Tonight I tried to lock the interferometer. At the first attempts the arm power didn't go above about 4. The mode cleaner seemed to be not well aligned and it lost lock or got stuck on a wrong mode. I had to run the MC_UP and MC_DOWN scripts to lock it again.
After that the locking proceed more smoothly; at least till a power level in the arms of about 60. Then again the mode cleaner lost lock and I had to run the scripts again. Without the MCWFS servo off the MC reflected power is still rather high (about 1.7); also even when the WFS servo is engaged the reflected power is about 0.5, versus 0.3 that it should be.
Those are both signs of a not very good alignment. Tomorrow I'll have to work on the injection periscope on the PSL table to try to fix that.
Since lately the alignment of the input beam to the interferometer has changed, I went checking the alignment of the beam on the photodiodea. They were all fine except for pd9, that is AS DD 199. Here the DC is totally null. The beam seems to go right on the diode but the scope on the PD's DC output shows no power. This is really strange and bad.
It's railed. This is what halted locking progess on Monday night, as this channel is used for the offloadMCF script, which slowly feeds back a CARM signal to the ETMs to prevent the VCO from saturating.
Attached is a 5 day trend, which shows that the channel went dead a few days ago. All the channels shown are being collected from the same ICS110B (I think), but only some are dead. It looks like they went dead around the time of the "All computers down" from Sunday.
After inspecting PD9 with the viewer and the cards, the beam looks like it is aligned to the photodiode althought there is no signal at the DC output of the photodetector. So I checked the spectrum for PD9_i and Q (see attachments) and it seems that those channels are actually seeing the beam. I'm going to check the alignemtn again and see the efefct on the spectra to make sure that the beam is really hitting the PD.
I aligned PD9. Here are the spectra confirming that.
Attached are the channels being recorded from the ICS110B in 1Y2 (the IOO rack). Channels 12, 13, 16, 17, 22, 24, 25 appear to have gone dead after the computer problems on Sunday.
This has been fixed by one of the two most powerful & useful IFO debugging techniques: rebooting. I keyed the crate in 1Y2.
I found the alignment biases for the PRM and the SRM in a funny state. It seemed like they had been "saved" in badly misaligned position, so the restore scripts on the IFO configure screen were not working. I've manually put them into a better alignment.
After the mini boot fest that Jenne did today, I checked whether that fixed the overflow issues we yesterday prevented the alignemnt of the arms.
I ran the alignment script for the arms getting 0.85 for TRX and 0.75 for TRY: low values.
After I ran the script ,C1SUSVME1 and C1SUSVME2 started having problems with the FE SYNC (counter at 16378). I rebooted those two and fix the sync problem but the transmitted powers didn't improve.
Are we still having problem due to MC misalignment?
I've now also trended the MOPA output power for the last 200 days to check a possible correlation with the FSS reflected power. See attachment.
The trend shows that the laser power has decayed but it seems that the FSS reflected power has done it even faster: 30% drop in the FSS vs 7% for the MOPA in the last 60 days (attachment n.2).
Here we trended also the PMC and the MZ. The drop in the PMC happens at the same rate as the MOPA's.
That let us think that the FSS transmitteed power has gone down because of the reference cavity progressive misalignment to the laser beam.
We need to adjust that alignment sometime.
The drop in the NPRO output power (upper row, 3rd plot: Ch10 C1:PSL_126MOPA_126MON) accompained an increase of "fuzziness" in PMCTRANSPD and both coincided in time with the day we tempoarirly removed the flap from the laser chiller's chiller (July 14 2009).
Spent a lot of time aligning tonight. The BS is not staying put--sometimes after a lock loss it gets badly mis-aligned.
DD handoff is working, after putting beam on REFL diodes and running senseDRM script.