I began moving the AA and AI chassis over to 1X1/1X2 as outlined in the elog.
The chassis were mostly filled with empty cables. There was one cable attached to the output of a QPD interface board, but there was nothing attached to the input so it was clearly not in use and I disconnected it.
I also attach a picture of some of the SMA connectors I had to rotate to accommodate the chassis in their new locations.
The chassis are installed, and the anti-imaging chassis can be seen second from the top; the anti-aliasing chassis can be seen 7th from the top.
I need to breakout the SCSI on the back of the AA chassis, because ADC breakout board only has a DB36 adapter available; the other cables are occupied by the signals from the WFS dewhitening outputs.
I ran a BNC from the PD on the AS table along the cable rack to a free ADC channel on the LSC whitening board. I lay the BNC on top of the other cables in the rack, so as not to disturb anything. I also was careful not to touch the other cables on the LSC whitening board when I plugged in my BNC. The PD now reads out to... a mystery channel. The mystery channel goes then to c1lsc ADC0 channels 9-16 (since the BNC goes to input 8, it should be #16). To find the channel, I opened the c1lsc model and found that adc0 channel 15 (0-indexed in the model) goes to a terminator.
Rather than mess with the LSC model, Gautam freed up C1:ALS-BEATY_FINE_I, and I'm reading out the AS signal there.
I misaligned the x-arm then re-installed the AS PO PD, using the scope to center the beam then connecting it to the BNC to (first the mystery channel, then BEATY). I turned off all the lights.
I went to misalign the x-arms, but the some of the control channels are white boxed. The only working screen is on pianosa.
The noise on the AS signal is much larger than that on the MC trans signal, and the DC difference for misaligned vs locked states is much less than the RMS (spectrum attached); the coherence between MC trans and AS is low. However, after estimating that for ~30ppm the locked vs misaligned states should only be ~0.3-0.4% different, and double checking that we are well above ADC and dark noise (blocked the beam, took another spectrum) and not saturating the PD, these observations started to make more sense.
To make the measurement in cds, I also made the following changes to a copy opf Johannes' assess_armloss_refl.py that I placed in /opt/rtcds/caltech/c1/scripts/lossmap_scripts/armloss_cds/ :
I started taking a measurement, but quickly realized that the mode cleaner has been locked to a higher order mode for about an hour, so I spend some time moving the MC. It would repeatedly lock on the 00 mode, but the alignment must be bad because the transmission fluctuates between 300 and 1400, and the lock only lasts about 5 minutes.
Prep for this work:
I was trying to get some pics of the optics as a zeroth-level reference for the pre-vent loss with the single arms locked, but since our SL7 upgrade, the sensoray won't work anymore . I'll try fixing this during the daytime.
The 40m vacuum envelope has one large single O-ring on the OOC west side. All other doors have double O-ring with annuloses.
There are 3 spacers to protect o-ring. They should not be removed!
The Cryo-pump static seal to VC1 also viton. All gate valves and right angle valve plates have single viton o-ring seal.
Small single viton o-rings on all optical quality viewports.
Helium will permiate through these fast. Leak checking time is limited to 5-10 minutes.
All other seals are copper gaskits. We have 2 manual right angle with METAL-dynamic seal [ VATRING ] as VV1 & RV1
Back to loss measurements.
I replaced the PD I've been using for the AS beam.
I misaligned the x arm.
I tried to lock the y arm, but PRC was locked so I could was unable. Gautam reminded me where the config scripts are.
The armloss measurement script needed two additional modifications:
I ran successfully the loss measurement script for the x and y arms. I'm getting losses of ~100ppm from the first estimates.
I made the following changes to the lossmap script:
When the optic aligns itself not at the ideal position, I'm noticing that it often locks on a 01. When the cavity is then misaligned and restored, it can no longer obtain lock. To fix this, I've moved my 'save' commands to just before the loop begins. This means the script may take longer to run, but as long as the cavity is initially locked and well aligned, this should make it more robust against wandering off and never reacquiring lock.
I left the lossmap script running for the x-arm. Next would be to run it for the y arm, but I see that after stepping to a few positions the lock is again lost. It's still trying to run, but if you want to stop it no data already taken will be lost. To stop it, go to the remaining terminal open on rossa and ctrl+c
the analysis needs:
I made additional measurements on the x and y arms, at 5 offset positions for each arm (along with 6 measurements at the "zeroed" position).
I've begun prepping the IFO for the vent, and completed most of the IFO related items on the checklist. The power into the MC has been cut, but the low-power autolocker has not been checked. I will finish up tomorrow and post the go ahead. PSL shutter is closed for tonight.
Following the checklist, I did these:
@Steve & Chub, we are ready to vent tomorrow (Monday Nov 19).
Vent 80 is nearly complete; the instrument is almost to atmosphere. All four ion pump gate valves have been disconnected, though the position sensors are still connected,and all annulus valves are open. The controllers of TP1 and TP3 have been disconnected from AC power. VC1 and VC2 have been disconnected and must remained closed. Currently, the RGA is being vented through the needle valve and the RGA had been shut off at the beginning of the vent preparations. VM1 and VM3 could not be actuated. The condition status is still listed as Unidentified because of the disconnected valves.
Gautam, Aaron, Chub and Steve,
The vent 81 is completed.
4 ion pumps and cryo pump are at ~ 1-4 Torr (estimated as we have no gauges there), all other parts of the vacuum envelope are at atm. P2 & P3 gauges are out of order.
V1 and VM1 are in a locked state. We suspect this is because of some interlock logic.
TP1 and TP3 controllers are turned off.
Valve conditions as shown: ready to be opened or closed or moved or rewired. To re-iterate: VC1, VC2, and the Ion Pump valves shouldn't be re-connected during the vac upgrade.
Thanks for all of your help.
As I was turning off the lights in the VEA, I heard a rattling sound from near the PSL enclosure. I followed it to a valve - I couldn't see a label on this valve in my brief effort to find one, but it is on the south-west corner of the IMC table, so maybe VABSSCI or VABSSCO? The power cable is somehow spliced with an attachment that looks to be bringing gas in/out of the valve (See Attachment #1), and the nut on the bottom was loose, the whole power cable + mettal attachment was responsible for the rattling. I finger-tightened the nut and the sound went away.
I checked the IMC alignment following the vent, for which the manual beam block placed on the PSL table was removed. The alignment is okay, after minor touchup, the MC Trans was ~1200 cts which is roughly what it was pre-vent. I've closed the PSL shutter again.
Rana, Aaron, Gautam
The old Zojirushi has died. We have received and comissioned our new Technivoorm Mocha Master today. It is good.
I finished running the cabling for the OMC, which involved running 7x 50ft DB9 cables from the OMC_NORTH rack to the 1X2 rack, laying cables over others on the tray. I tried not to move other cables to the extent I could, and I didn't run the new cables under any old cables. I attach a sketch diagram of where these cables are going, not inclusive of the entire DAC/ADC signal path.
I also had to open up the AA board (D050387, D050374), because it had an IPC connector rather than the DB37 that I needed to connect. The DAC sends signals to a breakout board that is in use (D080302) and had a DB37 output free (though note this carries only 4 DAC channels). I opened up the AA board and it had two IPC 40s connected to an adapter to the final IPC 70 output. I replaced the IPC40 connectors with DB37 breakouts, and made a new slot (I couldn't find a DB37 punch, so this is not great...) on the front panel for one of them, so I can attach it to the breakout board.
I noticed there were many unused wires, so I had to confirm that I had the wiring correct (still haven't confirmed by driving the channels, but will do). There was no DCC for D080302, but I grabbed the diagrams for the whitening boards it was connected to (D020432) and for the AA board I was opening up as well as checked out elog 8814, and I think I got it. I'll confirm this manually and make a diagram if it's not fake news.
Attachment #1 is a block diagram depicting the pathway by which the vertex DOF control signals can couple into DARM (adapted from a similar diagram in Gabriele's Virgo note on the subject). I've also indicated some points where noise can couple into either loop. In general, there are sensing noises that couple in at the error point of the loop, and actuation noises that couple in at the control point. In this linear picture, each block represents a (possibly time varying) transfer function. So we can write out the node-to-node transfer functions and evaluate the various couplings.
The motivation is to see if we can first simulate with some realistic noise and time-varying couplings (and then possibly test on the realtime system) the effectiveness of the filter denoted by "FF" in canceling out the shot noise from the auxiliary loop being re-injected into the DARM loop via the DARM sensor. Does this look correct?
With Chub's help, I've setup a mini cleanroom at EY - Attachment #1. The HEPA unit is running on high now. All surfaces were wiped with isopropanol, we can wipe everything down again on Monday and replace the foil.
I replaced the projector bulb. Previous bulb was shattered.
I've started testing the OMC channels I'll use.
I needed to update the model, because I was getting "Unable to setup testpoint" errors for the DAC channels that I had created earlier, and didn't have any ADC channels yet defined. I attach a screenshot of the new model. I ran
Gautam, Aaron, Chub & Steve,
ETMY heavy door replaced by light one.
We did the following: measured 950 particles/cf min of 0.5 micron at SP table, wiped crane and it's cable, wiped chamber,
placed heavy door on clean merostate covered stand, dry wiped o-rings and isopropanol wiped Aluminum light cover
[steve, rana, gautam]
Rana pointed out that the OSEM cabling, because of lack of a plastic shielding, is grounded directly to the table on which it is resting. A glass baking dish at the base of the seismic stack prevents electrical shorting to the chamber. However, there are some LEMO/BNC cables as well on the east side of the stack, whose BNC ends are just lying on the base of the stack. We should use this opportunity to think about whether anything needs to be done / what the influence of this kind of grounding is (if any) on actuator noise.
Steve also pointed out that we should replace the rubber pads which the vacuum chamber is resting on (Attachment #1, not from this vent, but just to indicate what's what). These serve the purpose of relieving small amounts of strain the chamber may experience relative to the beam tube, thus helping preserve the vacuum joints b/w chamber and tube. But after (~20?) years of being under compression, Steve thinks that the rubber no longer has any elasticity, and so should be replaced.
Exceptions: cryo pump and 4 ion pumps
Vac Status: The vac rack power was recycled yesterday and power to controller TP1,2 and 3 restored. atm3
VME is OFF. Power to all other instrument are ON. 23.9Vdc 0.2A
ETMY sus tower with locked optic in HEPA tent at east end is standing by for action.
[koji, gautam, jon, steve]
I wanted to set up an RTCDS model to understand this problem better. Attachment #1 is the simulink diagram of the signal flow. The idea will be to put in the appropriate filter shapes into the various filter blocks denoting the DARM and auxiliary DoF plants, controllers and actuators, and then use awggui / diaggui to inject some noises and see if in this idealized model I can achieve good subtraction. Then we can build up to applying a time varying cross coupling between DARM and the vertex DoF, and see how good the adaptive FF works. Still need to setup some MEDM screens to make working with the test system easier.
I figured c1omc would be the least invasive model to set this upon without risking losing any of our IR/green alignment references. Compile and install went smooth, see Attachment #2. The c1omc model was clocking 4us before, now it's using 7us.
Attachment #3 shows the top level of the OMC model, while Attachment #4 shows the MEDM screen.
* Note to self: when closing a loop inside the realtime model, there has to be a delay block somewhere in the loop, else a compilation error is thrown.
Recently we wondered at the meeting what the IMC round trip loss was. I had done several ringdowns in the winter of 2017, but because the incident light on the cavity wasn't being extinguished completely (the AOM 0th order beam is used), the full Isogaio et. al. analysis could not be applied (there were FSS induced features in the reflection ringdown signal). Nevertheless, I fitted the transmission ringdowns. They looked like clean exponentials, and judging by the reflection signals (see previous elogs in this thread), the first ~20us of data is a clean exponential, so I figured we may get some rough value of the loss by just fitting the transmission data.
The fitted storage time is .However, this number isn't commensurate with the 40m IMC spec of a critically coupled cavity with 2000ppm transmissivity for the input and output couplers.
Attachment #1: Expected storage time for a lossless cavity, with round-trip length ~27m. MC2 is assumed to be perfectly reflecting. The IMC length is known to better than 100 Hz uncertainty because the marconi RF modulation signal is set accordingly. For the 40m spec, I would expect storage times of ~40 usec, but I measure almost 30% longer, at ~60 usec.
Attachment #2: Fits and residuals from the 10 datasets I had collected. This isn't a super informative plot because there are 10 datasets and fits, but to eye, the fits are good, and the diagonal elements of the covariance matrix output by scipy's curve_fit back this up. The function used to fit the t > 0 portions of these signals (because the light was extinguished at t=0 by actuating on the AOM) is , where A and tau are the fitted parameters. In the residuals, the same artefacts visible in the reflection signal are seen.
Attachment #3: Scatter plot of the data. Width of circles are proportional to fit error on individual measurements (i just scaled the marker size arbitrarily to be able to visually see the difference in uncertainty, the width doesn't exactly indicate the error), while the dahsed lines are the global mean and +/- 1 sigma levels.
Attachment #4: Cavity pole measurement. Using this, I get an estimate of the loss that is a much more believable .
need to vary start/stop times in fit to test for systematics
I need to hookup +/- 24 V supplies to the OMC whitening/dewhitening boxes that have been added to 1X2.
There are trailing +24V fuse slots, so I will extend that row to leave the same number of slots open.
While removing one +24V wire to add to the daisy chain, I let the wire brush an exposed conductor on the ground side, causing a spark. FSS_PCDRIVE and FSS_FAST are at different levels than before this spark. The 24V sorensens have the same currents as before according to the labels. Gautam advised me to remove the final fuse in the daisy chain before adding additional links.
gautam: we peeled off some outdated labels from the Sorensens in 1X1 such that each unit now has only 1 label visible reflecting the voltage and current. Aaron will post a photo after his work.
I started putting together some code to implement some ideas we discussed at the Tuesday meeting here. Pipeline isn't setup yet, but i think it's commented okay so if people want to play around with it, the code lives on the 40m gitlab.
Initial results and conclusions:
There still seems to be some data quality issues with the ringdown data I have, so I don't think we really gain anything from running this analysis on the data I have already collected - but in the future, we can do the ringdown with complete extinguishing of the input light, and repeat the analysis.
As for whether we should clean the IMC mirrors - I'm going to see how much power comes out at the REFL port (with PRM aligned) this afternoon, and compare to the input power. This technique suffers from uncertainty in the Faraday insertion loss, isolation and IMC parameters, but I am hoping we can at least set a bound on what the IMC loss is.
Both were measured using the FieldMate power meter. I was hesitant to use the Ophir power meter as there is a label on it that warns against exceeding 100 mW. I can't find anything in the elog/wiki about the measured inesrtion loss / isolation of the input faraday, but this seems like a pretty low amount of light to get back from PRM. The IMC visibility using the MC_REFL DC values is ~87%. Assuming perfect transmission of the 87% of the 97mW that's coupled into the IMC, and assuming a further 5% loss between the Faraday rejected port and the AP table, the Faraday insertion loss would be ~30%. Realistically, the IMC transmission is lower. There is also some part of the light picked off for IPPOS. Judging by the shape of the REFL spot on the camera, it doesn't look clipped to me.
Either way, seems like we are only getting ~half of the 1W we send in on the back of PRM. So maybe it's worth it to investigate the situation in the IOO chamber during this vent.
c1psl, c1susaux,c1iool0,caux crates were keyed. Also, the physical shutter on the PSL NPRO, which was closed last Monday for the Sundance crew filming, was opened and the PMC was locked. PMC remains locked, but there is no light going into the IMC.
I did some ray tracing and determined that the aux beam will enter the OMC after losing some power in reflection on OMPO (couldn't find this spec on the wiki, I remember something like 90-10 or 50-50) and the SRM (R~0.9), and then transmission through OMPO. This gives us something like 8%-23% of the aux light going to the OMC, depending on the OMPO transmission. This elog tells me the aux power before the recombination BS is ~37mW, ~3.7mW onto SRM, which is consistent with the OMPO being 90-10, and would mean the aux power onto the OMC is ~3mW, plenty for aligning into the OMC.
Since the dewhitening board I'd intended to use isn't working (see elog) , I'm gong to scan the OMC length with a function generator while adjusting the alignment by hand, as was briefly attempted during the last vent.
I couldn't identify a PD on the AP table that was the one I had used during the last vent, I suspect I coopted the very same PD for the arm loss measurements. It is a PDA520, which has a large (100mm^2) area so I've repurposed it again to catch the OMC prompt reflection during the mode scans. I've mounted it approximately where I expect the refl beam to exit the AS chamber.
I brought over the cart that usually lives at 1X1 to help me organize materials near the OMC chamber for opening.
I replaced the banana connectors we'd been using to send HV to the HV driver with soldered wires going to the final locking connector only, so now the 150V is on a safe cable.
I powered up the DCPD sat box and again confirmed that it's working. I sent a 500Hz sine wave through the sat box and confirmed that I can see the signal in the DCPD channels I've defined in cds. I gave the TT and OMC-L PZT channels bad assignments on the ADC (right now, what reads as 'OMC_PZT_MON' is actually the unfiltered output from the sat box, while the DCPD channels are for the filtered outputs of the box), because the way the signals are grouped on the cables I can't attach all of them at once. For this vent, I'll only really need the DCPD outputs, and since I have confirmed that I can read out both of those I'll fix up the HV driver mon channels later.
I kept having trouble keeping the power LEDs on the dewhitening board 'on'. I did the following:
1. I noticed that the dewhitening board was drawing a lot of current (>500mA), so I initially thought that the indicators were just turning on until I blew the fuse. I couldn't find the electronics diagrams for this board, so I was using analagous boards' diagrams and wasn't sure how much current to expect to draw. I swapped out for 1A fuses (only for the electronics I was adding to the system).
2. Now the +24V indicator on the dewhitening board wasn't turning on, and the -24V supply was alternatively drawing ~500mA and 0mA in a ~1Hz square wave. Thinking I could be dropping voltage along the path to the board, I swapped out the cables leading to the whitening/dewhitening boards with 16AWG (was 18AWG). This didn't seem to help.
3. Since the whitening board seemed to be consistently powered on, I removed the dewhitening board to see if there was a problem with it. Indeed, I'd burned out the +24V supply electronics--two resisters were broken entirely, and the breadboard near the voltage regulator had been visibly heated.
I noticed that the +/-15V currents are slightly higher than the labels, but didn't notice whether they were already different before I began this work.
I also noticed one pair of wires in the area of 1X1 I was working that wasn't attached to power (or anything). I didn't know what it was for, so I've attached a picture.
Disclaimer: This is almost certainly some user error on my part.
I've been trying to get this running for a couple of days, but am struggling to understand some behavior I've been seeing with DTT.
I wanted to measure some transfer functions in the simulated model I set up.
To see if this is just a feature in the simulated model, I tried measuring the "plant" filter in the C1:LSC-PRCL filter bank (which is also just a pendulum TF), and run into the same error. I also tried running the DTT template on donatella (Ubuntu12) and pianosa (SL7), and get the same error, so this must be something I'm doing wrong with the way the measurement is being run / setup. I couldn't find any mention of similar problems in the SimPlant elogs I looked through, does anyone have an idea as to what's going on here?
* I can't get the "import" feature of DTT to work - I go through the GUI prompts to import an ASCII txt file exported from FOTON but nothing selectable shows up in DTT once the import dialog closes (which I presume means that the import was successful). Are we using an outdated version of DTT (GDS-2.15.1)? But Attachment #1 shows the measured part of the pendulum TF, and is consistent with what is expected until the measurement terminates with a synchronization error.
the import problem is fixed - when importing, you have to give names to the two channels that define the TF you're importing (these can be arbitrary since the ASCII file doesn't have any channel name information). once i did that, the import works. you can see that while the measurement ran, the foton TF matches the DTT measured counterpart.
11 Dec 2pm: After discussing with Jamie and Gabriele, I also tried changing the # of points, start frequency etc, but run into the same error (though admittedly I only tried 4 combinations of these, so not exhaustive).
Taking another look at the datasheet, I don't think LM7812 is an appropriate replacement and I think the LM2940CT-12 is supposed to supply 1A, so it's possible the problem actually is on the power board, not on the dewhitening board. The board takes +/- 15V, not +/- 24...
D050368 Anti-Imaging Chassis
D050368 Adl SUS/SEI Anti-Image filter board
S/N 100-102 Assembled by screaming circuits. Begin testing 4/3/06
S/N xxx Mohana returned it to the shop. No S/N or traveler. Put in shop inventory 4/24/06
S/N 103 Rev 01. Returned from Screaming circuits 7/10/06. complete except for C28, C29
S/N 104-106 Rev 01. Returned from Screaming circuits 7/10/06. complete except for C28, C29 Needs DRV-135’s installed
S/N 107-111 Rev 02 (32768 Hz) Back from assembly 7/14/06
S/N 112-113 Rev 03 (65536 Hz) assembled into chassis and waiting for test 1/29/07
S/N 114 Rev 03 (65536 Hz) assembled and ready for test 020507
D050512 RBS Interface Chassis Power Supply Board (Just an entry. There is no file)
RBS Interface Chassis Power Board D050512-00
Koji gave me some tips on testing this board that I wanted to write down, notes probably a bit intermingled with my thoughts. Thanks Koji, also for the DCC and equipment logging!
I set up a function generator to drive OMC-L, and have the two DCPD mons and the OMC REFL PD sent to an oscilloscope. I need to select a cds channel over which to read the REFL signal.
The two DCPD mon channels have very different behaviors on the PD mons at the sat box (see attachment). PD1 has an obvious periodicity, PD2 has less noise overall and looks more white. I don't yet understand this, and whether it is caused by real light, something at the PDs, or something at the sat box.
I've again gone through the operations that will happen with the OMC chamber vented. Here's how it'll go, with some of the open questions that I'm discussing with Gautam or whoever is around the 40m:
Talked with Gautam for a good while about the above plan. In trying to figure out why the DCPD sat box appears to have a different TF for the two PDs (seems to be some loose cabling problem at the mons, because wiggling the cables changed this), we determined that the AA chassis also wasn't behaving as expected--driving the expected channels (28-31) with a sine wave yields some signal at the 100Hz driving frequency, but all save ch31 were noisy. We also still saw the 100Hz when the chassis was unplugged. I will continue pursuing this, but in the meantime I'm making an IDE40 to DB37 connector so I can drive the ADC channels directly with the DAC channels I've defined (need to match pinouts for D080303 to D080302). I also will make a new SCSI to DB37 adapter that is more robust than mentioned here. I also need to replace the cable carrying HV to the OMC-L driver, so that it doesn't have a wire-to-wire solder joint.
We moved a razor blade on the AP table so it is no longer blocking the aux beam. We checked the alignment of aux into the AS port. AUX and AS are not colinear anywhere on the AP table, and despite confirming that the main AS beam is still being reflected off of the OMC input mirror, the returning AUX beam does not reach the AP table (and probably is not reaching the OMC). AUX needs to be realigned such that it is colinear with the AS beam. It would be good if in this configuration, the SRM is held close to its position when the interferometer is locked, but the TTs should provide us some (~2.5mrad) actuation. Gautam will do this alignment and I will calculate whether the TTs will be able to compensate for any misalignment of the SRM.
Here is the new plan and minimal things to do for the door opening tomorrow:
That is the first, minimal sequence of steps, which I plan to complete tomorrow. After aligned into the OMC, the alignment into the DCPDs shouldn't need modification. Barring work needed to align from OMC to DCPDs, I think most other work with the OMC can be done in-air.
NDscope is now running on pianosa. To be really useful, we need the templates, so I've made /users/Templates/NDScope_templates where these will be stored. Perhaps someone can write a parser to convert dataviewer .xml to something ndscope can understand. To get it installed, I had to run:
sudo yum install ndscope
sudo yum install python34-gpstime
sudo yum install python34-dateutil
sudo yum install python34-requests
I also changed the pythonpath variable to include the python3.4 site-packages library in .bashrc
Let's install Jamie's new Data Viewer
I found that the BS/PRM OL SUM channels were reading close to 0. So I went to the optical table, and found that there was no beam from the HeNe. I tried power-cycling the controller, there was no effect. From the trend data, it looks like there was a slow decay over ~400000 seconds (~ 5 days) and then an abrupt shutoff. This is not ideal, because we would have liked to use the Oplevs as a DC alignment reference during the ventI plan to use the AS camera to recover some sort of good Michelson alignment, and then if we want to, we can switch out the HeNe.
*How can I export PDF from NDscope?
I did the following:
At 11:13 am there was a ~2-3 second interruption of all power at the 40m.
I checked that nobody was in any of the lab areas at the time of the outage.
I walked along both arms of the 40m and looked for any indicator lights or unusual activity. I took photos of the power supplies that I encountered, attached. I tried to be somewhat complete, but didn't have a list of things in mind to check, so I may have missed something.
I noticed an electrical buzzing that seemed to emanate from one of the AC adapters on the vacuum rack. I've attached a photo of which one, the buzzing changes when I touch the case of the adapter. I did not modify anything on the vacuum rack. There is also
Most of the cds channels are still down. I am going through the wiki for procedures on what to log when the power goes off, and will follow the procedures here to get some useful channels.
After several combinations of soft/hard reboots for FB, FEs and expansion chassis, we managed to recover the nominal RTCDS status post power outage. The final reboots were undertaken by the rebootC1LSC.sh script while we went to Hotel Constance. Upon returning, Koji found all the lights to be green. Some remarks:
sudo systemctl start open-mx.service
sudo systemctl start mx.service
sudo systemctl start daqd_*
The PSL (Edwin) remains in an interlock-triggered state. We are not sure what is causing this, but the laser cannot be powered on until this is resolved.
Bob, Aaron, and I removed the door from the OMC chamber this morning. Everything went well.
I did a walkaround and checked the status of all the interlock switches I could find based on the SOP and interlock wiring diagram, but the PSL remains interlocked. I don't want to futz around with AC power lines so I will wait for Koji before debugging further. All the "Danger" signs at the VEA entry points aren't on, suggesting to me that the problem lies pretty far upstream in the wiring, possibly at the AC line input? The Red lights around the PSL enclosure, which are supposed to signal if the enclosure doors are not properly closed, also do not turn on, supporting this hypothesis...
I confirmed that there is nothing wrong with the laser itself - i manually shorted the interlock pins on the rear of the controller and the laser turned on fine, but I am not comfortable operating in this hacky way so I have restored the interlock connections until we decide the next course of action...
In order to see the AS beam a bit more clearly in our low-power config, I swapped out the ND=1.0 filter on the AS camera for ND=0.5.
[Gautam, Aaron, Koji]
The PSL interlock system was fixed and now the 40m lab is laser hazard as usual.
- The schematic diagram of the interlock system D1200192
- We have opened the interlock box. Immediately we found that the DC switching supply (OMRON S82K-00712) is not functioning anymore. (Attachment #1)
- We could not remove the module as the power supply was attached on the DIN rail. We decided to leave the broken supply there (it is still AC powered with no DC output).
- Instead, we brought a DC supply adapter from somewhere and chopped the head so that we can hook it up on the crimping-type quick connects. In Attachment #1, the gray is +12V, and the orange and black lines are GND.
- Upon the inspection, the wires of the "door interlock reset button" fell off and the momentary switch (GRAYHILL 30-05-01-502-03) got broken. So it was replaced with another momentary swicth, which is way smaller than the original unfortunately. (Attachments 2 and 3)
- Once the DC supply adapter was pluged to an AC tap, we heard the sounds of the relays working, and we recovered the laser hazard lamps, PSL door alerm lamps. Also it was confirmed that the PSL innolight is operatable now.
- BTW, there is the big switch box on the wall close to the PSL enclosure. Some of the green lamps were gone. We found that we have plenty of spare lamps and relays inside of the box. So we replaced the bulbs and know the A.C. lights are functioning. (Attachments 4 & 5)
I completed testing of the AI board mentioned above. In addition to the blown fuse, there were two problems:
After this, I tested the TF of all channels. For the most part, I found the expected 3rd order ~7500Hz cheby with notches at ~16kHz and 32kHz. However, some of the channels had shallower or deeper notches. By ~32kHz, I was below the resolution on the spectrum analyzer. Perhaps I just have nonideal settings? I'll attach a few representative examples.
I reinstalled the chassis at 1X2, but haven't connected power.
I turned on AUX, and aligned the aux beam to be centered on the first optic the AS beam sees on the AP table. I then turned off the AUX laser.
We looked into the /frames situation a bit tonight. Here is a summary:
Plan of action:
BTW - the last chiara (shared drive) backup was October 16 6 am. dmesg showed a bunch of errors, Koji is now running fsck in a tmux session on chiara, let's see if that repairs the errors. We missed the opportunity to swap in the 4TB backup disk, so we will do this at the next opportunity.
I replaced the 2'' AUX-AS combining BS with a 2'' HR mirror for 1064. I aligned the AUX beam from the new HR mirror into the next iris, so AUX passes through irises both before and after the new optic. Now, AS does not go out to the AS PDs.
I mounted the old BS on the SP table in a random orientation.
I also dumped the beam transmitted through one of the AUX steering mirrors before the new HR mirror.
I replaced the 2'' AUX-AS combining BS with a freshly mounted 2'' HR mirror for 1064. The mirror is labelled 'Y1-2037-45-P', and had a comment on its case: 'V'. I aligned the AUX beam from the new HR mirror into the next iris, so AUX passes through irises both before and after the new optic. Now, AS does not go out to the AS PDs.