ID 
Date 
Author 
Type 
Category 
Subject 
5871

Fri Nov 11 10:30:27 2011 
rana  Update  Adaptive Filtering  MC_F 
There should be a whitening filter in the Pentek Generic DAQ board (Eurocard with 8 differential LEMO inputs). It used to be that the MC_L channel came in through here and I believe it has 2 stages of 150:15 pole:zero filters.
I don't remember if it is one or two stages, but this should be easy to measure with a function generator or by driving this input using the MC2 UL Coil monitor and doing the transfer function in DTT (as Koji and Jenne did for the demod boards). 
5879

Sat Nov 12 02:00:36 2011 
Mirko  Update  Adaptive Filtering  MCF and other signals  Regarding http://nodus.ligo.caltech.edu:8080/40m/5867 and http://nodus.ligo.caltech.edu:8080/40m/5869 :
MC_F signal:
The measurements on p. 5867 were done using the ADC attached to the PEM computer. There was a big difference between the MC_F signal recorded directly after the server board and the signal just before the FSS board as recorded by a PEM channel.
To understand how this happens we measured the signal at different places with a spec. analyzer:
1. WIth a locked MC measuring the signal just before the PEM ADCs (meaning after a 60ft BNC cable)
2. Same position, but unlocked and seemingly dark MC
3. Locked MC, signal just before the FSS box
4. MC_F signal that is usually going into the Pentek Generic board and is recorded in C1:IOO
=> The 60ft BNC cable adds a considerable amount of noise, but doesn't fundamentally change the signal. It is weird that the signal is white from approx. 4Hz on.
Due to Jenne's measurement ( http://nodus.ligo.caltech.edu:8080/40m/5848 ) we know the TF from MCL through PD, mixer Pentek and into C1:IOO looks like this:
This is with the double HP from 15Hz on that should be in the Pentek. So one might expect a less white signal going into the FSS board...
PEM ADCs
The dark noise in the PEM ADCs is actually a factor 10 higher than in the IOO ADCs. Still ok wrt the the seismometers.
We also tried to measure essentially the dark noise of the whole seismometer readout (seismometer box, then ADC). That seemed ok, but is of limited value since the seismometer electronics behave a bit strange when there is no seismometer connected.

5882

Sat Nov 12 02:46:13 2011 
Den  Update  Adaptive Filtering  stacks and ground  We measured the coherence between the seismometer near the MC2 stack and accelerometers on the vacuum tank where MC2 is. Because accelerometers produce small signals at low frequencies, which are comparable with adc noise, we amplified the accelerometer signal by a factor of 20. We could not do it more because though adc has 40 V range, the black box that follows the channel sockets can transmit only 2.5 V max amplitude signal. Probably, this was done because old adc accepted 2 V max amplitude.
We were able to found some coherence at 0.11 Hz though the accelerometer signal is rather noisy. So to consider stack as a noise source is still possible. This measurement should be better done with two seismometers, one on the floor, the other on the stack. From the figure we can also see that tilt affects the x and y seismometer signals from 0.1 Hz. Green line (zcomponent) is much lower that red and blue lines (x and y). Tilt affects on horisontal axes of the seismometer much more than on vertical.
What we also think about is that at low frequencies mirrors start to move approximately the same and seismometers can help us to figure out small reletive displacement of the mirrors which form the MC length. We can estimate the critical frequency by presenting the ground motion as interference of surface waves with different velocities and amplitudes. For only 1 wave we have for the relation of MC length to the seismometer read out ~sin (2*pi*f/v*L). f  frequency of the wave, v speed, L  length between the mirrors. We can see that below 1 Hz we have ~sin (f/2). At this point seismometer signal could lose coherence with MC length signal. We could try to subtract seismometer signals from corresponding axes, but gur1 and sts1 has different calibrations. Moreover, the noise floor of the seismometers might not allow us to measure the differential signal. We'll try to simulate this scenario and find seismometer calibration or measure it. We are basicly interested only in the ratio of calibraion fucntions of 2 seismometers. 
5900

Tue Nov 15 22:31:39 2011 
Mirko  Update  Adaptive Filtering  Towards wiener filtering and improved OAFing 
[Jenne, Mirko]
1. We should help the OAF by compensating for the actuator TF:
The actuator TF, from adaptive filter output to MC2, through PD, mixer, Pentek and into C1:IOO looks like this:
If we assume a whiteish error signal that the adaptive code tries to compensate for its job gets extra complicated because it has to invert this TF. So we really should compensate for that. Easiest place for that is the CORR filter directly behind the adaptive code block.
Using the TF measurement from above I used the vectfit (" /cvs/cds/caltech/apps/mDV/extra/firfit_forFotonMirkoComplex.m" ) to get fit a corresponding digital filter:
If we invert swap the zeros and poles in the digital filter we get the inverted TF.
(Todo: Figure out how to invert the TF. Just switching the poles and zeros doesn't work).
2. Wiener filtering
The idea was to use the adaptive filtering only for small corrections to the wiener filtering. So we really should try to get the wiener filtering going.
Howto:
1. Get data for STS1X and GUR1X and MC_F in matlab. E.g. via ligodv
2. Check the MC was in lock the entire time.
3, Filter MC_F with the actuator TF, so the wiener filter knows about that and compensates for it
4. Calculate the wiener filter " h1winolevLigoDV.m "
5. Export the data to the workspace. It is also saved to the disc as "h1filtcoeffTS.mat". Make sure there are first the witnesses, then MC_F
6. Execute " /cvs/cds/caltech/apps/mDV/extras/LHO/firfit_for_FotonMirko.m" while one directory higher.
7. Copy the digital filter in SOS form that is printed into the matlab command line and put it into the corresponding filter in the OAF model via foton.
With data from 111115 04:00 to 05:45. Sampling freq. 256Hz. 8000 Taps => length = 30.2s. Prefiltered to notch the 60Hz line in MC_F, but not compensation the actuator TF. This results in the following wiener filter and corresponding SOS filter to be copied into foton.
STS1X:
GUR1X:

5919

Wed Nov 16 23:50:40 2011 
Den  Update  Adaptive Filtering  seismic noise injection  [Micro, Den]
Analyzing coherence of seismic noise and mode cleaner length we've figured out that at some days the coherence below 1 Hz is still present. For example, at Nov 13 we can see some coherence compared to most other dates when we are not able to see coherence as shown on the figure. On the top plot  psd of MC_L and GUR1_X at Nov 13 (red and blue) and Nov 15 (black and cyan). On the bottom plot is presented coherence between MC_L and GUR1_X on Nov 13 (red) and Nov 15 (black)
We can divide the psd plot for 2 parts  below 1 Hz and above 1 Hz. Above 1 Hz seismic noise on Nov 15 (cyan) was higher then on Nov 13 (blue) and correspondently MC_L at that region was higher on Nov 15. Below 1 Hz seismic noise was higher on Nov 13 but MC_L is still lower that on Nov 15. That is surprising. From the coherence plot we can say that once we have some more seismic noise than usually, we immediately see coherence.
Because of this we wanted to find out the level of the X noise that makes seismic noise invisible. We injected seismic noise by doing smooth physical exercises near MC_2 (1.5 m and 3 m apart). The MC_2 was in lock during the experiment.
In the coherence plot we can see that coherence between GUR1_X and MC_L increased with noise injection. The highest coherenced we obtained sittind down and standing up smoothly near MC_2 at distance 1.5 m. We did not want to come clother and break the lock. This measurement tells us that the X noise is approximately 34 times higher than seismic noise in the range 0.1  1 Hz. That means that it is approximately 1e6  1e8 m/sqrt(Hz) in this region. This noise goes down at frequencies from 2 Hz and not seen because of seismic noise. Actually, seismic noise can be filtered out with the Wiener filter and then we'll see the spectrum of X noise.
We now try to figure out the method to estimate the contribution of OSEM noise to the X noise. 
5932

Thu Nov 17 22:24:19 2011 
Den  Update  Adaptive Filtering  MC1_COIL  Analyzing coherence between MC length and signals on MC1, MC2 and MC3 coils we have noticed that MC1 COIL signal is not coherent to MC length at all at interesting frequencies 0.1  1 Hz.
We try to explain this phenomena.

6065

Sat Dec 3 18:29:20 2011 
Den  Update  Adaptive Filtering  coherence  I've looked through the coherence between the MC length and seismometers after the ifstatement problem was fixed. Coherence improved for all seismometers but is still not 1. It is possible that contribution from X, Y, Z directions split the coherence between them but at ~0.203 Hz we do not see much coherence for all these directions.
I looked at the coherence between MC2 OSEM signal and MC_F when the AUTO LOCKER is ON and OFF. I thought that we'll ses the same coherence for both regimes as laser is locker to the MC length. However, I figured out the coherence is worse when AUTO LOCKER is ON at frequencies 0.20.3 Hz.
The first idea that comes to mind is that when feedback to the laser is provided, the pressure to the mirrors from the laser beam is changed. 
6068

Mon Dec 5 02:55:30 2011 
Den  Update  Adaptive Filtering  C1OAF  I've added filter banks for correction path in the C1OAF model to use AA filters. I compiled and installed the new version. I runs but does not sync. Probably, I've made a mistake in the some names of epics channels. Leave it for now, figure out tomorrow. If someone needs an old version, it is in the /opt/rtcds/caltech/c1/userapps/trunk/isc/c1/models/c1oaf_BACKUP20111204.mdl file. Corresponding medm screen is in the /opt/rtcds/caltech/c1/userapps/trunk/isc/common/medm/OAF_OVERVIEW.adl file. 
6070

Mon Dec 5 10:13:13 2011 
Jenne  Update  Adaptive Filtering  C1OAF 
Quote: 
I've added filter banks for correction path in the C1OAF model to use AA filters. I compiled and installed the new version. I runs but does not sync. Probably, I've made a mistake in the some names of epics channels. Leave it for now, figure out tomorrow. If someone needs an old version, it is in the /opt/rtcds/caltech/c1/userapps/trunk/isc/c1/models/c1oaf_BACKUP20111204.mdl file. Corresponding medm screen is in the /opt/rtcds/caltech/c1/userapps/trunk/isc/common/medm/OAF_OVERVIEW.adl file.

The general rule in the 40m is that if it's not an 'emergency', i.e. something is wrong with the computers and preventing the main locking work, no model recompilingtype activities at nighttime.
Also, if you do things and recompile, you need to do an svn checkin. That's where backups are kept. We don't want to clutter folders with backups anymore. 
6078

Wed Dec 7 00:11:58 2011 
Den  Update  Adaptive Filtering  OfflineAF  I did offline adaptive filtering with yesterday's 3 hours of MCF and GUR1X data. It turns out that normalizedlms can strongly outperform static Wiener filtering!
This is interesting. It might be something inside MC_F that Wiener static does not see. I think the problem is either with seismometer noise or tilt. 
6093

Fri Dec 9 13:28:09 2011 
Den  Update  Adaptive Filtering  C1OAF  I tried to figure out why red NO SYNC label became present in the C1OAF_GDS_TP screen after I added AA filters to the C1OAF model.
C1OAF model contains 8 libraries C1OAF_ADAPT for 8 DOF. I changed C1OAF_ADAPT library to C1OAF_ADAPT_AA library where I added 28 AA filters for 28 witness channels. It turns out that if I use this library for all 8 DOF then I see NO SYNC label, if only for one DOF (MCL) then I see green IOP label. This means that using AA filters for each DOF too much channels of filters are created for online system to operate. I think there is some number inside the code that one can not exceed. Analyzing compilation output after "make c1oaf" I figured out that without using AA filters we have 632 filters and using AA we have 856 filters.
For now I'll use AA filters for MCL only. 
6094

Fri Dec 9 14:33:16 2011 
Alex Ivanov  Update  Adaptive Filtering  C1OAF 
Quote: 
I tried to figure out why red NO SYNC label became present in the C1OAF_GDS_TP screen after I added AA filters to the C1OAF model.
C1OAF model contains 8 libraries C1OAF_ADAPT for 8 DOF. I changed C1OAF_ADAPT library to C1OAF_ADAPT_AA library where I added 28 AA filters for 28 witness channels. It turns out that if I use this library for all 8 DOF then I see NO SYNC label, if only for one DOF (MCL) then I see green IOP label. This means that using AA filters for each DOF too much channels of filters are created for online system to operate. I think there is some number inside the code that one can not exceed. Analyzing compilation output after "make c1oaf" I figured out that without using AA filters we have 632 filters and using AA we have 856 filters.
For now I'll use AA filters for MCL only.

I have a feeling we are not fitting into preallocated memory space in the shared memory between the frontend process and the epics process. Filter module data is overwriting some other data and that's why we are not getting a sync light. I suggest we upgrade to 2.4 code first and then we will figure out a way to expand memory areas to fit 856 filters. 
6100

Fri Dec 9 17:53:31 2011 
Den  Update  Adaptive Filtering  C1OAF  [Jenne, Den]
AA filters for witness channels are added to the oaf model. It is working now and the number of memory used is not critical. NO SYNC is not present any more. 
6110

Tue Dec 13 01:20:38 2011 
Den  Update  Adaptive Filtering  Modifications to LSC, RFM models, added OAF model 
Quote: 
[Jenne, Mirko, with supervision from Jamie]
We are starting to create the new OAF model, so that it works with the new CDS system.

Why did you place Matt's code inside the simulink library and use the same library for all DOFs? I think this won't work out. Inside the .c code there are static variables. If all DOF use the same ADAPT_XFCODE() function, it means that they all mess there signals and coefficients with each other! Or the RCD during the compilation creates a copy of the function with the name of a library name in front? For example, ADAPT_MCL_ADAPT_XFCODE(). But then in the RCG manual it is claimed to name the .c file the same.
This problem can be fixed by creating .c files with proper names for each DOF. But here a memory question may arise. For 1 DOF we now have 28 witness channel. If we have a several minute filter, we use 28 * 10^{4}(filter length) * 3 (FIR coefficients, adapt input, corr input) * 8 (number of bytes in 1 double) = 6.7 Mb / DoF. For 8 DOF we'll allocate ~55 Mb of memory in the kernel. The c1lsc cache size is 6 Mb per cpu. So we are definitely out of cache and it will take some time for a processor to communicate with ram. I wonder if it is OKEY for us to allocate this amount of memory as static arrays inside the kernel.
Now we use 6.7 Mb of memory because it seems to be a mistake with placing the same function for all DOF and we actually allocate for 1.

6137

Mon Dec 19 17:17:02 2011 
Den  Update  Adaptive Filtering  filter tap dependence  Online filter diverges. I did offline simulations with current ccode. Offline filter also diverges, even in the simplest case
witness = randn(1e6, 1); target = witness + 0.01*randn(1e6, 1);
I tried to create a new implementation of FXLMS algorithm as a c code. Then with this c code I did offline filtering with MCL and GUR signals and compared the error signals depending on the length of the filter.
One can see the code at the svn
adaptOnline  start here and choose algorithm
adaptive_filtering  Matlab implementation of AF
current_version.c  current version of the Filter (Matt's)
fxlms_filter.c  new version of the FXLMS filter
oaf.c  agent between Matlab and C (edited Matt's file)
Data samples can be found at nodus /users/den/wiener_filtering/data 
6199

Sun Jan 15 10:28:02 2012 
Den  Update  Adaptive Filtering  delays  We can account for delays in the oaf system by compensating it in the adaptive path of the filter. But using only this procedure is not enough. Parameters mu and tau should be chosen accurately:
w = (1  tau) * w;
w += mu * dw / norm;
NLMS algorithm without considering delays works well for mode cleaner length and gur1 seismometer signals, significantly reducing MC_F with parameters mu=1, tau=0. These parameters are considered because nlms algorithm should converge with the highest speed when mu=1. However, if the system has a delay so at time moment n:
error_signal [n] = desired_signal [n]  filter_output [ndelay];
then the adaptive filter diverges for the same parameters mu=1 and tau=0 even for delay=1. For that reason we make the same calculations with tau = 1e4 and tau = 1e2 without reducing mu conserving the adaptation rate and get the same result as nlms algorithm without delays. Next figure shows MC_F signal, error after applying enlms filter with tau=1e4 and tau=1e2. "e" is added to show that a small number (epsilon) is added to the norm of the signal in order to prevent the filter from diverging in the beginning of the process when the norm is not welldetermined yet.
The test was done offline with the sampling frequency 2048 Hz, without downsampling and any filters. We can see that tau=1e4 is still not enough, tau=1e3 or tau=1e2 is as good as nlms without delays, tau=1e1 and high are also bad.
Correctly choosing tau we have some freedom for delay compensation in the adaptation path. This is important as we do not know exactly what is the delay in the real system. We can measure it approximately. In order to figure out the range of reasonable delay errors we make a test with delay = 1, but to the adaptation path we give delays from 0 to 10. It turns out that adaptation path delays greater then 5 make the filter diverge, delays in the range 03 produce a reasonable error. In the figure below errors with adaptation path delays = 1 (correct) and 3 are presented.

6200

Sun Jan 15 11:40:30 2012 
Den  Update  Adaptive Filtering  downsampling  Here for the downsampling process we use a lowpass Bessel digital filter of order 6, normalized cutoff frequency = 0.1. In the plot presented below we compare the results with downsampling ratio = 1, 2, 4.
We can see that increasing the downsampling ratio, we increase the error of the filter. Moreover, the error at some particular frequency f seems to depend on the ratio f/Fs, where Fs  sampling frequency (2048 Hz) devided by downsampling ratio. Error is the same for all curves below 1 Hz but then begins to increase as we increase the sownsampling ratio. In order to figure out what the problem is  mistake in the filter code, inaccurate upsample algorithm or this is NLMS particularity, I've changed sampling frequency in the chans/daq/C1PEM.ini and C1IOO.ini files from 2048 Hz to 512 Hz for corresponding channels. Now, we compare the error from the filter working with 2048 Hz frequency, downsampling ratio = 4, lowpass filter = Bessel of order 6, normalized cutoff = 0.1 and filter working with 512 Hz sampling frequency, without downsampling and with corresponding Bessel lowpass filter with normalized frequency 0.4.
MC_F measurement at 2048 Hz was done during the day, for that reason red curved is slightly higher then green in the resonance frequencies. But still we can see that these two cases are very much alike. For this reason, it seems that NLMS filter works better with higher sampling frequencies. 
6201

Sun Jan 15 12:18:00 2012 
Den  Update  Adaptive Filtering  running time  In order to figure out what downsampling ratio we can take, we need to determine the running time of the fxlms_filter() function. If the filter length is equal to 5000, downsampling ratio is equal to 1, number of witness channels is 1 then with ordinary compilation without speed optimization one call runs for 0.054 ms (milli seconds). The test was done on the 3 GHz Intel processor. With speed optimization flags the situation is better
O1 0.014 ms, O3 0.012 ms
However, Alex said that speed optimization is not supported at RCG because it produce unstable execution time for some reason. However, by default the kernel should optimize for size Os. With this flag the running time is also 0.012 ms. We should check if the frontend machine compilers indeed use Os flag during the compilation and also play with speed optimization flags. Flags O3 and Os together might also give some speed improvement.
But for now we have time value = 0.012 ms as running time for 5000 coefficient filter, 1 witness channel and downsample ratio = 1. Now, we need to check how this time is scaled if we change the parameters.
5000 cofficients  0.012 ms
10000 coefficients  0.024 ms
15000 coefficients  0.036 ms
20000 coefficients  0.048 ms
We can see that filter length scaling is linear. Now we check downsampling ratio
ratio=1  0.048 ms
ratio=2  0.024 ms
ratio=4  0.012 ms
Running time on the dependance of downsample ratio is also linear as well as on the dependence of the number of witness channels and degrees of freedom.
If we want to filter 8 DOF with approximately 10 witness channels for each DOF, then 5000 length filter will make 1 cycle for ~1 ms, that is good enough to make the downsample ratio equal to 4.
Things get a little bit complicated when the code is called for the first time. Some time is taken to initialize variables and check input data. As a result the running time of the first cycle is ~0.1 ms for 1 DOF that is ~10 times more then running time of an ordinary cycle. This effect takes place at this moment when one presses reset button in the c1oaf model  the filter becomes suspended for a while. To solve this problem the initialization should be divided by several (~10) parts. 
6296

Sat Feb 18 17:01:26 2012 
Den  Update  Adaptive Filtering  static variables  In order to prevent different DOF from redetermining static variables in the adaptive code, I've created a separate code for each DOF with the name ADAPT_XFCODE_{$DOF}.c
I've provided the links for these files in the c1oaf.mdl, compiled and run it. Now there are no conflicts between DOFs. 
6297

Sat Feb 18 18:29:38 2012 
Den  Update  Adaptive Filtering  online filtering  I tried to filter MC_F from seismic noise measured by GUR1 seismometer. I've used 8000 tap filter, downsample ratio=8, delay=1. In the Figure the output of the filter is presented with MC_F signal.
We can see that output is close to the MC_F, but the phase for some reason is not zero. It should not be at 1 Hz  10 Hz due to the actuator. But below these frequencies I do not see any reasons for the output phase to differ from MC_F phase. But it is possible, the phase of the actuator is evaluated very rough and the adaptive filter can't match it. 
6332

Tue Feb 28 16:12:59 2012 
Den  Update  Adaptive Filtering  lunch talk  Just to be clear what I said at the meeting, I write all this down here. Adaptive filtering of real signals (MC_F and GUR1_X) with all noises inside is
This is offline filtering but with real signals and with the Ccode that is compiled at the 40m now. We can reduce the MC_F signal by ~100 below 10 Hz, but the problem is that reducing the adaptation gain, the error increases. As a result when we move towards FxLMS algorithm with AA, AI and downsampling, we have to take the gain equal to ~1e2 and we do not reduce any noise.
The second demonstration of this problem is static Wiener filtering. This is the result
We can see that adaptive filtering outperforms the "optimal" filtering. This is because an adaptive filter can follow the changes of coefficients immediately while the Wiener filter averages them. This is the mathematical formulation:
mcl_real = coeff _real* seismic_noise_real + other_noise
mcl_real  the real length of the MC,
coeff_real  real coefficients, that represent the transfer function between seismic noise and MC length,
other_noise  noise uncorrelated to the seismic noise seismic_noise_real
But in the world of our measured signals we have the equation
mcl_measured = coeff * seismic_noise_measured + other_noise
mcl_measured = TF_mcl * mcl_real
seismic_noise_measured = TF_seis * seismic_noise_real
where TF_mcl and TF_seis  transfer functions from the real world to measurements.
It seems to me that TF_mcl or TF_seis are not constants and for that reason the TF between measured seismic noise and mcl is not constant. But it is exactly what an adaptive or Wiener filter tries to define:
coeff(time) = average(coeff(time)) + delta(coeff(time))
The result of applying average(coeff) is the green line in the Figure 2  error after applying the Wiener filtering.
delta(coeff)  the changing part of the transfer function is caught by the adaptive filtering. The lower the gain, the lower is the capability of adaptive filter to catch these changes. Theoretically. the error after applying adaptive filter can be presented like this:
E(error*error) = E(other_noises*other_noises) + 1/(2mu)*mu*E(other_noises*other_noises) + 1/ {mu*(2mu)} * Tr(Q) * A
where mu = adaptation gain
Q  covariance matrix of delta(coeff)
A  norm of the seismic signal
The first term in this equation is the dispersion of other noises, the second term is the error of the adaptive filter due to nonzero gain, the third term is due to the changes in the transfer function  we can see that it is proportional to 1/mu. This term explains why the error increases while mu decreases.
Now I'm looking for the part in the path of the signals where the transfer function can change. As I mentioned above, this is not a change in the real world, it is the change in the measured signlals. My first guess is the quantization error  we do not have not enough counts. If this is not the case, I'll move to other things of the signal path. 
6490

Thu Apr 5 18:24:55 2012 
Den  Configuration  Adaptive Filtering  oaf starts to work  Today I tried to make the lms filter to work online. I played around with the signals (GUR1_X and MC_F) to prewhiten them and in the end the following configuration worked out:
1. mu = 0.03, tau = 1e5, downsample=8, nCoeff = 4000, delay = 5 (sampleandhold delay is not included in the new code, it should be added here!)
2. witness pass: AA32 = cheby1("LowPass", 4, 1, 32) AND 0.1:0
3. witness adaptation path: AA32 AND AI32 = cheby1("LowPass", 4, 1, 32) AND 0.1:0
4. error path: AA32 AND 0.1:0 AND anti_1Hz. Before I added anti_1Hz filter oaf did nothing. This filter tries to approximate the actuator transfer function. Note, it is not in the witness adaptation path. This is some sort of whitening.
5. correction path: AI32, gain = 1
Convergence time ~ 5 mins. The performance of the filter is far not perfect compared to the offline implementation. But it deals with a stack though.

6491

Fri Apr 6 09:57:24 2012 
Den  Update  Adaptive Filtering  static starts to work  I made static filter to work to evaluate the actuator TF.. Here is the result of static filtering:
What I did:
I did offline simulation of the MC_F Wiener filtering using 2 witness signals  GUR1X and GUR1Y. I've downsampled the data from 2048 to 128 Hz and applied the Wiener filter with 10000 for each witness channel:
Result of the filtering Filter coefficients for gur1x and then gur1y
Gur1x > MC_F transfer function Gur1y > MC_F transfer function
Then using vectfit I approximated obtained transfer functions in the region 0.5  20 Hz. I used a window function and then weights to get a more precise result in this range using only 8 poles and zeros.
I obtained the zpkmodel for each witness channel and entered it into the FOTON splitting it into 2 parts before that because FOTON does not like too long filters. These zpkmodels are at the C1:OAFSTATIC_STATMTX_8_8 and C1:OAFSTATIC_STATMTX_8_9 filter banks.
GUR1X:
z =
7.527339430029315 +31.603999069997801i
7.527339430029230 31.603999069997823i
27.897703898191267 + 0.000000000000071i
6.437806394766186 + 9.893955654289517i
6.437806394766159  9.893955654289510i
1.114401249545640 + 5.479278396987240i
0.176877296954015 + 0.000000000000006i
1.114401249545616  5.479278396987245i
p =
0.407251778925379 + 6.263247012022007i
0.407251778925379  6.263247012022007i
0.230672968859081 + 6.846868757063707i
0.230672968859081  6.846868757063707i
2.871419857491615 +13.707864827517826i
2.871419857491615 13.707864827517826i
2.134260618362721 +18.319129999777648i
2.134260618362721 18.319129999777648i
k =
4.113285626223658e04
GUR1Y
z =
17.961416874092624 +13.631821252434328i
17.961416874092642 13.631821252434353i
8.788634771726304 + 7.653357335975781i
8.788634771726285  7.653357335975777i
0.037906973323273 + 5.133348020858679i
0.164348392996182 + 3.588803405511463i
0.164348392996187  3.588803405511474i
0.037906973323277  5.133348020858679i
p =
0.027577318242359 + 5.174655410828068i
0.027577318242359  5.174655410828068i
0.500384298611703 + 6.310552036591990i
0.500384298611703  6.310552036591990i
0.237055716999485 + 6.881204941979009i
0.237055716999485  6.881204941979009i
1.408223271160550 +14.874570175309771i
1.408223271160550 14.874570175309771i
k =
2.723835471763049e04
Then I approximated the reversed actuator TF and placed it to the C1:OAFSUS_MC2_OUT filter bank. The gain to the static filter output is 1.
P.S. Also the static matrix was filled with 1 for some reason. Here is the script to fix it if if will be bad again
for i in {1..8}
do
for j in {1..28}
do
element="C1:OAFSTATIC_STATMTX_"$i"_"$j"_GAIN"
ezcawrite $element 0
done
done

6492

Fri Apr 6 10:31:07 2012 
Den  Update  Adaptive Filtering  static and adaptive  I've run static and adaptive filters simultaneously. AA32 filters rotate the phase of the witness signals GUR1X and GUR1Y and now the performance of the static filter is worse. Next time I'll recalculate Wiener filter coefficients taking this into account. But still 2 filters together can deal with a stack better.

6493

Fri Apr 6 11:14:34 2012 
Jenne  Update  Adaptive Filtering  static and adaptive 
Quote: 
I've run static and adaptive filters simultaneously. AA32 filters rotate the phase of the witness signals GUR1X and GUR1Y and now the performance of the static filter is worse. Next time I'll recalculate Wiener filter coefficients taking this into account. But still 2 filters together can deal with a stack better.

This is super awesome! I'm totally excited!! 
6551

Thu Apr 19 22:18:24 2012 
Den  Update  Adaptive Filtering  oaf algorithm: old vs new  Here are the issues that I found not quite accurate in the old oaf code:
1. There is no need to calculate the norm of the witness signal every time from zero:
norm += (*pBufAdapt) * (*pBufAdapt); // add to the norm
Every step the witness signal vector is the same except the first and last values
wit[i].norm += histAdpt[nCoeff]*histAdpt[nCoeff]  histAdpt[0]*histAdpt[0];
This step will reduce the number of multiplications and summations from 3*M/k to 2*M/k, M  filter length, k  downsample ratio.
2. Old code filter corrects filter coefficients with a delay equal to k=downsample ratio (pretty big):
witness o o o o o o o o o o o o o o o o o o o o o
error o o o o o o o o o o o o o o o o o o o o o
We want the filter to work at green points and skip red points computing output and correcting coefficients at this time (downsample ratio in this example is 4). Old code
 grabs error signal
 calculates output during next k1 red points and 1 green point
 corrects coefficients using this error during next k1 red points and 1 green point
But LMS algorithm should correct coefficients according to the latest error. As we calculate output and correct coefficients before the latest error signal will be available, we should change the order:
 grabs error signal
 corrects coefficients using this error during next k1 red points and 1 green point
 calculates output during next k1 red points and 1 green point
This scheme is completely equivalent to the ordinary LMS algorithm, as now we correct coefficients according to the latest error signal and so do not add any delay.
3. So long soft start is probably not needed for the LMS filter, it makes the filter to converge longer
// modify adaptation gain for soft start
if( state.iWait < state.nFIR )
{
adaptGain = 0.0;
decayRate = 1.0; // clear FIR coeffs after reset
}
As far as I understand this is done to prevent the filter from huge coefficients variations in the beginning when norm is not calculated yet. Instead we can just introduce some small
epsilon = 10.0;
to prevent the filter from divergence in the beginning
delta = mu * error / (wit[i].norm + epsilon);
Though some soft start might be needed by not so long  it will take several minutes before the adaptation gain will take it's specified value. 
6553

Fri Apr 20 23:02:25 2012 
Den  Update  Adaptive Filtering  frequency domain filter  DFTLMS is a frequency domain adaptive filter that demonstrates faster convergence compared to the timedomain LMS filter. I've tested Discrete Fourier Transform (DFTLMS) filter. It converts witness signal to the frequency domain using DFT and corrects the eigenvalues of the covariance matrix to make them as equal to each other as possible (does prewhitenning of the witness signal).
Left plot compares learning curves for time domain LMS and DFTLMS algorithms on the simulated data from seismometers and mcl (number of averages = 30) Right plot shows the evolution of the filter coefficients norm (Euclidean norms of the coefficient vector). Though LMS algorithm works in the time domain and DFTLMS in the frequency domain, the coefficient vectors must have the same length, because we Fourier Transform is achieved by applying a unitary operator => vector norm must not change.
Plots show that both algorithms converge to the same coefficients vector norm, but DFTLMS does it much faster then LMS.
Online realization:
Good news: algorithm complexity is linear in filter length. Though the algorithm does Fourier transform, its complexity is still O(M), M  number of coefficients. Simulations show that DFTLMS is ~89 times slower then LMS. This is not so bad, may be we can do even slightly better.
Bad news: downsample process is not simple. Due to Fourier transform, the filter needs the whole witness signal vector before calculating the output. This is sad and in contrast with LMS algorithm where we could start to calculate the new output immediately after computing the previous output. We either need to calculate the whole output immediately or introduce delay in the output or approximate Fourier transform with some previous witness signal values.
Realization in the kernel: I asked Alex about complex numbers, exponents, sin and cos functions in the kernel c and he answers that we do not have complex numbers, about exp, cos, sin he is not sure. But for DFTLMS algorithm we are able to get round of these difficulties. Complex numbers will be presented as 2 real numbers. Then exp (a) = cos(a) + i*sin(a). All what we need for DFTLMS are sin(2 * pi * k / M) and cos(2 * pi * k / M), k=0,1,2,...,M1. Fortunately, M  (filter length) is big enough, typical value pi/M ~ 0.001 and we can calculate sin(2*pi/M) and cos(2*pi/M) using Taylor series. As the argument is small, 56 terms will be enough to get precision ~1e20. Then we build the whole table of cos and sin according to induction cos(2*pi/M*k) = cos(2*pi/M*(k1))cos(2*pi/M)  sin(2*pi/M*(k1))sin(2*pi/M), sin(2*pi/M*k) = cos(2*pi/M*(k1))sin(2*pi/M) + sin(2*pi/M*(k1))cos(2*pi/M). We should do it only once, so the algorithm will build these values in the beginning during first several iterations, then will use them.
The main problem is downsampling. I need to think more about it. 
6642

Fri May 11 23:33:41 2012 
Den  Update  Adaptive Filtering  offline vs online  I've compared offline Wiener filtering with online static + adaptive filtering for MC_F with GUR1_XYZ and GUR2_XYZ as witness signals
Note: online filter works up to 32 Hz (AI filter at 32 Hz is used). There is no subtraction up from this frequency, just MC_F was measured in different times for online and offline filtering. This difference in MC_F in frequency range 20100 Hz showed up again as before with microphone testing. One can see it in 1 minute. Smth is noisy.
Reasons why online filter is worse then offline:
1. FIR > IIR conversion produces error. Now I'm using VECTFIT approximation with 16 poles (splitting into 2 filter banks), this not enough. I tried to use 50 and split them into 5 filter banks, but this scheme is not working: zpk > sos conversion produces error and the result filter works completely wrong.
2. Actuator TF. VECTFIT works very good here  we have only 1 resonance. However, it should be measured precisely.
3. Account for AA and AI filters that rotate the phase at 110 Hz by ~ 10 degrees. 
7000

Sat Jul 21 18:04:02 2012 
Den  Update  Adaptive Filtering  frequency domain filter  I've implemented online frequency domain filter and applied it to MC_F.
+
Magnitude of the filter output at 1 Hz is the same as MC_F. This means that it is not hard for FIR to match the resonance. The problem is with the phase. We can not match the resonance exactly. If the resonance is at f_{0 }and we match at f_{0} +/ df then in the frequency range (f_{0}, f_{0 }+/ df) the phase is not matched for 180. I guess the filter does not diverge because df is small but also the filter can not account for this huge phase lag. We need to slightly change the simulated actuator TF and see how the filter will react._{
} 
7234

Mon Aug 20 13:02:57 2012 
Den  Update  Adaptive Filtering  1 Hz resonance  Static filter was adjusted to filter 1 Hz resonance in MCL and it could do it. Stack is not great in this experiment due to the phase mismatch. I'll fix it.

7252

Wed Aug 22 20:33:51 2012 
Den  Update  Adaptive Filtering  MC_L in ARMS  Jenne and I did adaptive filtering of MC_L and measured how X and Y ARM control signals change compared to nonfiltered MC_L. We did the test during 1.5 Hz seismic noise activity and adaptive filter was able to subtract it. However, it adds noise at high frequencies, It is not seen in MC_L but it is present in the ARMs control signals.
I'll investigate this problem. May be we need to reduce adaptation gain. In this experiment it was 0.1 and adaptive filter convergence time was equal to 12 mins.

7589

Mon Oct 22 20:44:49 2012 
Ayaka  Update  Adaptive Filtering  microphone noise  I will do some experiments on acoustic noise canceling during my stay.
Now I am planning to cancel acoustic noise from PMC and see how the acoustic noise work and how we should place microphones.
First, I measured the noise in microphones and its circuit.
blue, green, red, solid lines; microphone signals
blue, green, red, dashed lines; uncoherent noise in signals
yellow, black, solid lines; circuit noise (signal input is open, not connected to the microphones)
We can see the acoustic signal above 1 Hz, and the circuit does not seem to limit its sensitivity. But I do not know why yellow and black is so different. I will check it tomorrow. 
7592

Tue Oct 23 00:51:41 2012 
Jamie  Update  Adaptive Filtering  microphone noise 
Quote: 
I will do some experiments on acoustic noise canceling during my stay.
Now I am planning to cancel acoustic noise from PMC and see how the acoustic noise work and how we should place microphones.a
First, I measured the noise in microphones and its circuit.
blue, green, red, solid lines; microphone signals
blue, green, red, dashed lines; uncoherent noise in signals
yellow, black, solid lines; circuit noise (signal input is open, not connected to the microphones)
We can see the acoustic signal above 1 Hz, and the circuit does not seem to limit its sensitivity. But I do not know why yellow and black is so different. I will check it tomorrow.

Hi, Ayaka. It would be good if you could give a little bit more detail about this plot:
 What exactly are the "signals"? Are you making a sound somehow? If so, what is producing the sound? What is it's spectrum?
 Are the blue/green/red traces from three different microphones?
 Coherence usually implies a comparison between two signals. Is something being compared in the dashed traces?
 Are the yellow and black traces from different amplifiers?
 What are the units of the Y axis?

7596

Tue Oct 23 10:24:42 2012 
Ayaka  Update  Adaptive Filtering  microphone noise 
Quote: 
Quote: 
I will do some experiments on acoustic noise canceling during my stay.
Now I am planning to cancel acoustic noise from PMC and see how the acoustic noise work and how we should place microphones.a
First, I measured the noise in microphones and its circuit.
blue, green, red, solid lines; microphone signals
blue, green, red, dashed lines; uncoherent noise in signals
yellow, black, solid lines; circuit noise (signal input is open, not connected to the microphones)
We can see the acoustic signal above 1 Hz, and the circuit does not seem to limit its sensitivity. But I do not know why yellow and black is so different. I will check it tomorrow.

Hi, Ayaka. It would be good if you could give a little bit more detail about this plot:
 What exactly are the "signals"? Are you making a sound somehow? If so, what is producing the sound? What is it's spectrum?
 Are the blue/green/red traces from three different microphones?
 Coherence usually implies a comparison between two signals. Is something being compared in the dashed traces?
 Are the yellow and black traces from different amplifiers?
 What are the units of the Y axis?

Sorry for my poor explanation.
I measured this by the same way as you measured the instrumental noise of seismometers.
I put the three microphones at the same place so that the three can hear the same sound. I did not make any sounds, just put them in the lab.
The signals from microphones are all amplified by the circuit.
And I took the correlations of each signals and two others and got the noise (dashed lines) by subtracting the correlated signal from the original signal.
So,
The signal is the acoustic sound in the lab, amplified by the circuit.
Three lines are from three different microphones.
Dashed lines are subtraction of coherent signal from the original.
Yellow and black lines are from different amplifiers in the same circuit box. The circuit has 6 channels.
I did not calibrate the signals I got by DTT since I do not know the calibration factor now. It is just the number I got from the real time system.

7607

Wed Oct 24 14:15:34 2012 
Ayaka  Update  Adaptive Filtering  microphone noise  Previous results
I am measuring the noise level of the microphones. The circuit does not seems to limit their sensitivities but the circuit's noise seems to be different from other channels.
Measurement
I measured the circuit noise of all 6 channels. (input open)
(mic_open.png)
The noise level is about 10 times different from the others.
Comparing the acoustic signal, microphone+circuit noise, and ADC noise;
(mic_noise.png)
 blue; acoustic signal
 green; microphone+circuit noise
 red; circuit (the data was not took simultaneously.)
 sky blue; ADC noise
To do
I will remake the circuit though the circuit does not limit the sensitivity. I would like to make sure that the circuit does not affect badly and to make the circuit noise level the same.
At the same time, I will get the PMC control signal and see coherence between it and acoustic sound. 
7609

Wed Oct 24 15:29:52 2012 
rana  Update  Adaptive Filtering  microphone noise  We have to change the sample rate and AA filter for the mic channels before going too far with the circuit design. 
7610

Wed Oct 24 17:02:01 2012 
Jenne  Update  Adaptive Filtering  microphone noise 
Quote: 
We have to change the sample rate and AA filter for the mic channels before going too far with the circuit design.

To save the mic channels at higher than 2k (which we should do), we either have to move them to a different model, change the rate of the PEM model, or see if you can save data faster than the model runs (which I can't imagine is possible). 
7614

Wed Oct 24 22:20:24 2012 
Den  Update  Adaptive Filtering  microphone noise 
Quote: 
We have to change the sample rate and AA filter for the mic channels before going too far with the circuit design.

PEM model is running at 64K now. It turned out to be tricky to increase the rate:
 BLRMS are computationally expensive and original pem model did not start at any frequency higher then 16k ( at 16k cpu meter readings were 59/60 ). Also when we go higher then 16k, frontend gives the model less resources. I guess it is assumed that this model is iop and won't need too much time. So in the end I had to delete BLRMS blocks for all channels except for GUR2Z and MIC1.
 Foton files are modified during model compilation: lines with sampling rate and declaration of filters in the beginning of the file are changed only. Sosrepresentation and commands are the same. I hoped that filter commands will let me change sosrepresentation quickly. I've opened Foton and saved the file. However, Foton modified commands in such a way that the ratio of poles and zeros to sampling rate is preserved. I guess all filters have to be replaced or this process should be done in another way.
 BLRMS block uses lowpass filters below 0.01 Hz, increasing the sampling rate by a factor of 32 might make calculations incorrect. I'll check it.
We should also increase cut off frequency of the lowpass filter in the microphone preamplifier from 2 kHz up to ~2030 kHz. 
7621

Thu Oct 25 09:53:23 2012 
Ayaka  Update  Adaptive Filtering  microphone noise 
Quote: 
Quote: 
We have to change the sample rate and AA filter for the mic channels before going too far with the circuit design.

PEM model is running at 64K now. It turned out to be tricky to increase the rate:
 BLRMS are computationally expensive and original pem model did not start at any frequency higher then 16k ( at 16k cpu meter readings were 59/60 ). Also when we go higher then 16k, frontend gives the model less resources. I guess it is assumed that this model is iop and won't need too much time. So in the end I had to delete BLRMS blocks for all channels except for GUR2Z and MIC1.
 Foton files are modified during model compilation: lines with sampling rate and declaration of filters in the beginning of the file are changed only. Sosrepresentation and commands are the same. I hoped that filter commands will let me change sosrepresentation quickly. I've opened Foton and saved the file. However, Foton modified commands in such a way that the ratio of poles and zeros to sampling rate is preserved. I guess all filters have to be replaced or this process should be done in another way.
 BLRMS block uses lowpass filters below 0.01 Hz, increasing the sampling rate by a factor of 32 might make calculations incorrect. I'll check it.
We should also increase cut off frequency of the lowpass filter in the microphone preamplifier from 2 kHz up to ~2030 kHz.

Thank you for changing the sample rate!
Also we have to change the AntiAliasing filter, as Jamie said.
Now my question is, whether S/N ratio is enough at high frequencies or not. The quality of EM172 microphone is good according to the data sheet. But as you can see in previous picture, the S/N ratio around 1kHz is not so good, though we can see some peaks, e.g. the sound that a fan will make. I have to check it later.
And, is it possible to do online adaptive noise cancellation with a high sampling rate such that computationally expensive algorithms cannot be run? 
7622

Thu Oct 25 10:03:38 2012 
rana  Update  Adaptive Filtering  microphone noise  That's no good  we need BLRMS channels for many PEM channels, not just two. And the channel names should have the same name as they had in the past so that we can look at long term BLRMS trends.
I suggest:
 Have a separate model for Mics and Magnetometers. This model should run at 32 kHz and not have low frequency poles and zeros. Still would have acoustic frequency BLRMS.
 Have a low frequency (f_sample = 2 kHz) model for seis an acc. Seismometers run out of poop by 100 Hz, but we want to have the ACC signal up to 800 Hz since we do have optical mount resonances up to there.
 Never remove or rename the BLRMS channels  this makes it too hard to keep long term trends.
 Do a simple noise analysis to make sure we are matching the noise of the preamps to the noise / range of the ADCs.
 Immediately stop using bench supplies for the power. Use ONLY fused, power lines from the 1U rack supplies.

7623

Thu Oct 25 14:39:14 2012 
Den  Update  Adaptive Filtering  microphone noise 
Quote: 
That's no good  we need BLRMS channels for many PEM channels, not just two. And the channel names should have the same name as they had in the past so that we can look at long term BLRMS trends.
I suggest:
 Have a separate model for Mics and Magnetometers. This model should run at 32 kHz and not have low frequency poles and zeros. Still would have acoustic frequency BLRMS.
 Have a low frequency (f_sample = 2 kHz) model for seis an acc. Seismometers run out of poop by 100 Hz, but we want to have the ACC signal up to 800 Hz since we do have optical mount resonances up to there.
 Never remove or rename the BLRMS channels  this makes it too hard to keep long term trends.
 Do a simple noise analysis to make sure we are matching the noise of the preamps to the noise / range of the ADCs.
 Immediately stop using bench supplies for the power. Use ONLY fused, power lines from the 1U rack supplies.

Ayaka, Den
C1PEM model is back to 2K.
We created a new C1MIC model for microphones that will run at 32K. C1SUS machine is full, we have to think about rearrangement.
For now, we created DQ channels for microphones inside iop model, so we can subtract noise offline.
We provided 025 kHz bandwidth noise to AA board and saw the same signal in the output of ADC in the corresponding channel. So cutoff frequency is higher then 25 kHz. There is a label on the AA board that all filters are removed. What does this mean?
We've turned off AA bench power supply, prepare to use fused from 1U. 
7633

Fri Oct 26 18:25:02 2012 
Ayaka  Update  Adaptive Filtering  Microphone noise again  [Raji, Ayaka]
Thanks to Den, power supplies for microphone circuit are changed.
So I measured the microphone noise again by the same way as I did last time.
solid lines: acoustic noise
dashed lines: uncoherent noise
black line: circuit noise (microphone unconnected)
The circuit noise improves so much, but many line noises appeared.
Where do these lines (40, 80, 200 Hz...) come from?
These does not change if we changed the microphones...
Anyway, I have to change the circuit (because of the lowpass filter). I can check if the circuit I will remake will give some effects on these lines. 
7634

Fri Oct 26 19:06:14 2012 
Den  Update  Adaptive Filtering  Microphone noise again 
Quote: 
The circuit noise improves so much, but many line noises appeared.
Where do these lines (40, 80, 200 Hz...) come from?
These does not change if we changed the microphones...
Anyway, I have to change the circuit (because of the lowpass filter). I can check if the circuit I will remake will give some effects on these lines.

I do not think that 1U rack power supply influenced on the preamp noise level as there is a 12 V regulator inside. Lines that you see might be just acoustic noise produced by cpu fans. Usually, they rotate at ~25003000 rpm => frequency is ~4050 Hz + harmonics. Microphones should be in an isolation box to minimize noise coming from the rack. This test was already done before and described here.
I think we need to build a new box for many channels (32, for example, to match adc). The question is how many microphones do we need to locate around one stack to subtract acoustic noise. Once we know this number, we group microphones, use 1 cable with many twisted pairs for a group and suspend them in an organized way. 
7636

Mon Oct 29 08:41:22 2012 
Ayaka  Update  Adaptive Filtering  Microphone noise again 
Quote: 
Quote: 
The circuit noise improves so much, but many line noises appeared.
Where do these lines (40, 80, 200 Hz...) come from?
These does not change if we changed the microphones...
Anyway, I have to change the circuit (because of the lowpass filter). I can check if the circuit I will remake will give some effects on these lines.

I do not think that 1U rack power supply influenced on the preamp noise level as there is a 12 V regulator inside. Lines that you see might be just acoustic noise produced by cpu fans. Usually, they rotate at ~25003000 rpm => frequency is ~4050 Hz + harmonics. Microphones should be in an isolation box to minimize noise coming from the rack. This test was already done before and described here.
I think we need to build a new box for many channels (32, for example, to match adc). The question is how many microphones do we need to locate around one stack to subtract acoustic noise. Once we know this number, we group microphones, use 1 cable with many twisted pairs for a group and suspend them in an organized way.

I do not think they are acoustic sounds. If so, there should be coherence between three microphones because I placed three at the same place, tied together. However, there are no coherence at lines between them. 
7708

Tue Nov 13 21:05:35 2012 
Den  Update  Adaptive Filtering  online and simulation  For a last few days I've been working on oaf and simulink model to simulate it. First I did online subtraction from MC when MC_L path was enabled. Inside my code I've added a sum of squares of filter coefficients so we can monitor convergence of the filter.
To to this I've measured path from OAF output to input without AA and AI filters. Then made a vectfit using 2 poles and zeros. Foton command
zpk( [2.491928e+03;5.650511e02], [4.979872e+01;3.278776e+00], 6.011323e+00)
My simulink model consists of 3 parts:
 cavity with seismic noise at low frequencies, 1/f^2 noise at medium frequencies and white noise at high frequencies
 this cavity is locked using feedback compensation filters that we use to lock arms
 locked cavity with adaptive filter
Adaptive filter in the model uses online ccode. It is connected to simulink block through an Sfunction. Sampling frequency of the model is 10 kHz. It works fairly fast  1 sec of simulation time is computed in 1 sec.
I've tested FxLMS algorithm and MFxLMS algorithm that is faster. I plan to test 2 iir adaptive algorithms that are already coded. 
7764

Fri Nov 30 02:40:44 2012 
Den  Update  Adaptive Filtering  YARM  I've applied FIR adaptive filter to YARM control. Feedback signal of the closed loop was used as adaptive filter error signal and OAF OUT > IN transfer function I assumed to be flat because of the loop high gain at low frequencies. At 100 Hz deviation was 5 dB so I've ignored it.
I've added a filter bank YARM_OAF to C1LSC model to account for downsampling from 16 kHz to 2 kHz and put lowpass filter inside.
I've used GUR 1&2 XYZ channels as witnesses. Bandpass filters 0.410 Hz we applied to each of them. Error signal was filters using the same bandpass filter and 16 Hz 40 dB Q=10 notch filter. As an AI filter I used 32 Hz butterworth 4 order lowpass filter. Consequently, AI, bandpass and notch filters were added to adaptive path of witness signals.
I've used an FIR filter with 4000 taps, downsampling = 16, delay = 1, tau = 0, mu = 0.01  0.1. Convergence time was ~3 mins.

7767

Fri Nov 30 11:49:24 2012 
Koji  Update  Adaptive Filtering  YARM  This is interesting. I suppose you are acting on the ETMY.
Can you construct the compensation filter with actuation on the MC length?
Also can you see how the X arm is stabilized?
This may stabilize or even unstabilize the MC length, but we don't care as the MC locking is easy.
If we can help to reduce the arm motion with the MCL feedforward trained with an arm sometime before,
this means the lock acquisition will become easier. And this may still be compatible with the ALS.
Why did you notched out the 16Hz peak? It is the dominant component for the RMS and we want to eliminate it. 
7769

Fri Nov 30 22:11:50 2012 
Den  Update  Adaptive Filtering  ARMS 
Quote: 
This is interesting. I suppose you are acting on the ETMY.
Can you construct the compensation filter with actuation on the MC length?
Also can you see how the X arm is stabilized?
This may stabilize or even unstabilize the MC length, but we don't care as the MC locking is easy.
If we can help to reduce the arm motion with the MCL feedforward trained with an arm sometime before,
this means the lock acquisition will become easier. And this may still be compatible with the ALS.
Why did you notched out the 16Hz peak? It is the dominant component for the RMS and we want to eliminate it.

I actuate on ETMY for YARM and ETMX for XARM. For now I did adaptive filtering for both arms at the same time. I used the same parameters for xarm as for yarm.
I've notched 16 Hz resonance because it has high Q and I need to think more how to subtract it using FIR filter or apply IIR.
I'll try MC stabilazation method. 
7771

Sat Dec 1 00:13:16 2012 
Den  Update  Adaptive Filtering  ARMS and MC 
Quote: 
I actuate on ETMY for YARM and ETMX for XARM. For now I did adaptive filtering for both arms at the same time. I used the same parameters for xarm as for yarm.
I've notched 16 Hz resonance because it has high Q and I need to think more how to subtract it using FIR filter or apply IIR.
I'll try MC stabilazation method.

Adaptive filtering was applied to MC and X,Y arms at the same time. I used a very aggressive (8 order) butterworth filter at 6 Hz as an AI filter for MC not to inject noise to ARMS as was done before
Mu for MC was 0.2, downsample = 16, delay = 1. I was able to subtract 1 Hz. Stack subraction is not that good as for arms but this is because I used only one seismometer for MC that is under the BS. I might install accelerometers under MC2.
EDIT, JCD, 18Feb2013: Den remembers using mu for the arms in the range of 0.01 to 0.1, although using 0.1 will give extra noise. He said he usually starts with something small, then ramps it up to 0.04, and after it has converged brings it back down to 0.01. 
7591

Tue Oct 23 00:41:55 2012 
Jenne  Update  Alignment  MC locked, spots centered  [Jenne, Raji]
We replaced the MC Refl path BS with the Y1, as usual, so that the full ~100mW goes to the REFL PD, so we don't have WFS or MC refl camera.
The MC spots were all outside of 1mm, and some were beyond 2mm (for MC1,3, P,Y....MC2 is of course free since we have more DoFs than we need), so we touched (very, very slightly) the zigzag mirrors on the PSL table. We realigned the MC, and now the spots are centered to my satisfaction.
MC1,2,3 Pit, MC1,2,3 Yaw (in mm):
[0.46444020918749457, 8.2634316545130009, 0.41417975237831089, 0.89401481457980592, 0.9323196976382162, 1.543145765853893]
MC2 is way off in pitch, according to this measurement, and it's been consistently going down as we move the MC2 spot in the same direction (up on the monitor), but since we started at +15mm and are now at +8, and we've gone quite a ways, I'm not sure that we really want to go all the way to 0. Anyhow, MC1 and MC3 are the ones which define our input pointing, so we're quitting for tonight.
We will turn on the PZTs and begin with the official vent list for dummies tomorrow. 
