40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log, Page 192 of 344 Not logged in
ID Date Author Type Category Subject
7957   Tue Jan 29 19:50:49 2013 JenneUpdateLockingBetter POP layout, no extra PRM motion with locked cavity

[Jenne, Jamie, Manasa]

Today's activities focused on getting the POP layout improved, so that we could get clean data for the mode scan measurement.

As Jamie and Koji pointed out yesterday, the beam was still a little too big on the POP DC PD, and was falling off the diode when the beam moved a small amount.  We have fixed things so that the PD is now at the focus of the lens, and the camera is at a place where the beam takes up most of the area on the TVs.  The beam no longer falls off the PD with cavity fluctuations.  A key point of this work was also to use an extra 2" optic to steer the beam down the length of the POP table, and then do the 50/50 beam splitting later with a 1" optic.  The 1" BS that we had been using (including with the "real" POP beam) is too small.  We could not find a 2" 50/50 BS, so we opted to do the splitting closer to the focal point.  Also, the BS that was splitting the beam between the PD and the camera was a 33% reflector, but now is a 50/50 BS. When we put back the 'real' POP path, we need to consider using larger optics, or a faster lens. The POP path is now good, hopefully for the duration of the half cavity test.

After getting the POP path taken care of, and tweaking up the cavity alignment a little bit, the transmitted power on POP DC is ~22,000 counts, with occasional fluctuations as high as 25,000 counts.

Jamie looked at the REFL path, and things look sensible there.  The unlocked REFL power is ~36 counts, and the locked power is ~20 counts.  I'm not sure what the 160 counts that Koji mentioned in his edits to elog 7949 is about.

I looked at the PRM oplev with the cavity locked and unlocked, and with today's alignment, there seems to be no difference in the amount of PRM motion when the cavity is locked vs unlocked.

It still looks like we might be seeing some clipping in the in-vac POP steering mirrors - we haven't gotten to them yet.

Jamie is currently modifying Yuta's mode scan analysis script to look at the data that we have of the cavity.

We need more 2" optics.  There are no mounted 2" spares in the various optic "graveyards" (which, PS, we should consolidate all into the cabinet with doors near the optics bench), and the options for boxes in the drawers is slim pickin's.  We have some S-pol stuff, but no Y1s or BS-50s for P-pol.  Since POP, POX, POY, IPANG, TRX and TRY all come out of the vacuum with large beams, we should have some options for these laying around for this kind occasional temporary thing.  We also need to choose, then purchase better 2" lenses for the pickoffs.

7958   Tue Jan 29 20:28:11 2013 ericqUpdateGeneralEarly work on Mirror Mounts

[Q, Chloe]

Chloe has been to the lab twice to start up her investigations in acoustic noise coupling to mirrors. The general idea for the setup is a HeNe laser bouncing off a mirror and onto a QPD, whose signal provides a measure of beam displacement noise. The mirror will be mounted and excited in various ways to make quantitative conclusions about the quality of different mounting schemes.

We have set up the laser+mirror+QPD on the SP table, and collected data via SR560s->SR785, with the main aim of evaluating the suitability of this setup. The data we collected is not calibrated to any meaningful units (yet). For now, we are just using QPD volts.

Chloe collected data of vertical displacement noise for the following schemes: Terminated SR785 input, Terminated SR560 inputs, Laser centered directly onto the QPD, Laser shining on mirror centered on QPD, laser/mirror/qpd with some small desktop speakers producing white noise from http://www.simplynoise.com. Data shown below.

7959   Tue Jan 29 21:07:48 2013 JenneUpdateLockingPRM coils need diagonalizing

[Jenne, Jamie]

We tried actuating on PRM so that we go through fringes in a known, linear way.  We used C1:SUS-PRM_LSC_EXC and awggui.  It seems that we get a lot of angular motion when we actuate....we need to look into this tomorrow.

EDIT/UPDATE:  Last night we tried several combinations of frequency and amplitude, but just for an idea,  we were using 2Hz, 1000cts.  Using Kiwamu's calibration in elog 5583 for the PRM actuator of 2e-8/f^2 m/cts, this means that we were pushing ~5nm.  But when we pushed much harder (larger amplitude) than that, we saw angular fringing.

7960   Wed Jan 30 03:01:55 2013 KojiUpdateGeneralEarly work on Mirror Mounts

I can't believe that SR785 can have such a low input noise level (<1nV/rtHz). Review your calibration again.

It is also described in the manual that SR560 typically has the input noise level of 4nV/rtHz, although this number depends on which gain you use.

7961   Wed Jan 30 11:16:32 2013 JenneUpdateLockingMode spacing calc

[Jenne, Jamie]

We did a few pen and paper calculations yesterday to confirm for ourselves that the half PRC should have nicely separated modes.  The half cavity is L=4.34m long, assuming flat mirror is 3.5 inches in front of BS.  That 3.5" is a guess, not a measurement.

Finesse

F = ( pi * sqrt(r1 * r2) ) / (1 - r1*r2) = 111.

Full width at half max

FWHM = c / (2 * L * F) = 311 kHz

FWHM in meters = FWHM * L/f = L*1064nm/c = 4.8 nm

Free spectral range

nu_fsr = F * FWHM = 34.5 MHz

Mode Spacing (eq 19.23 from Siegman)

omega = (n + m) * arccos(\pm sqrt(g1*g2)) / pi     *   (2*pi*c)/(2L)

For our half cavity, g1*g2 = 0.96

For the 01 or 10 modes, n+m = 1

mode spacing between 00 and 01 = 2.2 MHz

Thus, the modes should be well separated

=>  spacing is 2.2 MHz while FWHM is 0.311 MHz  (cavity fsr = 34.5 MHz)

EDIT JCD 31Jan2013:  Fixed mode spacing eqn to be diff between TEM00 mode and HOM, not plane wave and HOM.  Then fixed the factor of 2 error in the mode spacing numbers.

7962   Wed Jan 30 11:18:31 2013 ManasaUpdateScatteringScattering setup

 Quote: [Jan, Manasa] We are trying to get some scattering measurements in the Y-arm cavity. We have removed one of  the viewport windows window covers of ETMY chamber and have installed cameras on a ring that clamps to the window. The window along with the ring attachment is covered with aluminium foil when not in use.

[Jan, Manasa]

To align the camera to see small angle scattering from the ITMY, we tried shooting a green laser pointer at the pickoff mirror that was installed in the ETMY chamber such that we hit the face of ITMY. But we concluded that to be a very bad way to align the camera because we have no means to reconfirm that the camera was exactly looking at the scattering from ITMY.

Since we are in air, we came up with a plan B. The plan is to temporarily install a mirror in the ITMY chamber to steer the beam from the laser pointer (installed on the POY table) through ITMY to the pickoff mirror at the ETMY end. This way, we can install the camera at the ETMY window and be sure we are looking at ITMY scattered light.

7963   Wed Jan 30 13:50:27 2013 JenneUpdateComputersc1iscex still down

[Koji, Jenne]

We noticed that the iscex computer is still down, but the IOP is (was) running.  When we sat down to look at it, c1x01 was 'breathing', had a non-zero CPU_METER time, and the error was 0x4000, which I've never seen before.  The fb connection was still red though.  Also, it is claiming that its sync source is 1pps, not TDS like it usually is.

Since things were different, Koji restarted the 2 other models running on iscex, with no resulting change.  We then did a 'rtcds restart all', and the IOP is no longer breathing, and the error message has changed to 0xbad.  The sync source is still 1pps.

Moral of the story:  c1iscex is still down, but temporarily showed signs of life that we wanted to record.

7964   Wed Jan 30 14:00:02 2013 CharlesUpdateISSISS Design and Prototyping

Attached are both the circuit diagram and the liso formatted *.fil for the main branch of the ISS, as well as the resulting transfer function when analyzed. Unfortunately, as noted in the file, not all of the elements are possible to analyze in liso, such as any type of op-amp with more than two inputs and one output (AD602 used in this design has 16 pins with two distinct amplifiers contained within).

I have begun prototyping this circuit on a breadboard.

7965   Wed Jan 30 14:37:01 2013 ZachUpdateISSISS Design and Prototyping

 Quote: Unfortunately, as noted in the file, not all of the elements are possible to analyze in liso, such as any type of op-amp with more than two inputs and one output (AD602 used in this design has 16 pins with two distinct amplifiers contained within).

Typically, you can still find a way to model the important parts of the stages that are not as simply added. In the case of the differential input stage, in particular, it is important to include it because it will usually set the input noise level of the circuit. In this case, the noise is the same as the second stage (U5) and it has a gain of 1, so there is essentially no difference (up to factors of sqrt(2) or 2).

You can edit the opamp.lib file and add in custom components. For the input stage, you can just pretend it is a simple non-inverting amplifier with the specified noise characteristics from the datasheet: un = 1.3n, uc = 50 Hz (see below).

For dual op amps, you can usually just model each part separately. For example, the OPA2604 is a dual op amp that is included in the opamp.lib and can be treated as a single one in a model.

7966   Wed Jan 30 15:45:09 2013 ZachUpdateLockingMode spacing calc

 Quote: Thus, the modes should be well separated =>  spacing is 4.3 MHz while FWHM is 0.311 MHz  (cavity fsr = 34.5 MHz)

Something looks fishy. I calculate a transverse mode spacing of 2.21 MHz---is there a factor of two missing somewhere in your analytical calculation?

delta_f = (1/2/pi) * w01 - w00 = (1/2/pi) * acos(±sqrt(0.96)) /pi *2 * pi * c /2 /L = 2.21 MHz

I guess that's still OK, but if you are using 11-MHz sidebands, there is a n+m=5 mode within one linewidth of resonance. Can you use 55?

-------------

May I suggest my arbcav() tool for things like this? I think it's pretty handy for just this sort of calculations. I'm actually hoping to revamp the I/O to make it much cleaner and more intuitive.

>> T = [0.055 20e-6];

>> L = [4.34 4.34];

>> RoC = [115.5 1e10];

>> theta = [0 0];

>> fmod = 11e6;

>> lambda = 1064e-9;

>> num_pts = 1000;

>> loss = 50e-6;

>> [fin,coefs,df] = arbcav(T,L,RoC,theta,fmod,loss,lambda,num_pts);

>> fmod = 55e6;

>> [fin,coefs,df] = arbcav(T,L,RoC,theta,fmod,loss,lambda,num_pts);

7968   Wed Jan 30 19:30:21 2013 JenneUpdateLockingPOP in-vac improved

 Quote: It still looks like we might be seeing some clipping in the in-vac POP steering mirrors - we haven't gotten to them yet.

[Jenne, Jamie]

We fixed up, as best we can, the in-vac POP alignment.  We are entirely limited in yaw by the aperture size of the 2" 45deg mirror launching the beam out of the vacuum.  The main centroid of the beam is well centered, but the inflated weird part of the beam is totally clipped.  There's nothing we can do about it except use a much larger mirror, install a fast lens inside the chamber, or just fix the damn PRC.  I vote for the third option there.

How did we work our magic?

We put a green laser pointer where the POP DC PD was, and injected it into the vacuum, just like we normally do.  However, this time, we made sure the green laser was centered on all of the out of vacuum mirrors, so that there was no real work to do once we turned off the laser pointer. We locked the cavity, and confirmed that we are well centered on all of the in and out of vacuum mirrors, and discovered our aperture problem with the last in-vac mirror.

Here is a snapshot of the POP camera:

7969   Wed Jan 30 19:34:17 2013 JenneUpdateLockingPRM coils need diagonalizing

[Koji, Jamie, Jenne]

Koji did this, while we actuated on PRM in pos, and watched the oplev.  Empirically, he found the following values for the POS column of the output matrix:

UL = 1.020

UR = 0.990

LL = 1.000

LR = 0.970

SD = 0.000

(The nominal values are all +1, except for Side, which is 0).

Actuation of PRM was through C1:SUS-PRM_LSC_EXC, f=0.1Hz, A=100 counts.

Ed by KA:
This means UL and UR are increased by 2% and UR and LR are decreased by 3%. More precisely UR should be 1.02*0.97.
This is just a quick hack which works only for the DC.

7970   Thu Jan 31 10:23:39 2013 JamieUpdateComputersc1iscex still down

 Quote: [Koji, Jenne] We noticed that the iscex computer is still down, but the IOP is (was) running.  When we sat down to look at it, c1x01 was 'breathing', had a non-zero CPU_METER time, and the error was 0x4000, which I've never seen before.  The fb connection was still red though.  Also, it is claiming that its sync source is 1pps, not TDS like it usually is.  Since things were different, Koji restarted the 2 other models running on iscex, with no resulting change.  We then did a 'rtcds restart all', and the IOP is no longer breathing, and the error message has changed to 0xbad.  The sync source is still 1pps. Moral of the story:  c1iscex is still down, but temporarily showed signs of life that we wanted to record.

There's definitely a timing issue with this machine.  I looked at it a bit yesterday.  I'll try to get to it by the end of the week.

7971   Thu Jan 31 11:53:31 2013 ManasaUpdateScatteringScattering setup

 Since we are in air, we came up with a plan B. The plan is to temporarily install a mirror in the ITMY chamber to steer the beam from the laser pointer (installed on the POY table) through ITMY to the pickoff mirror at the ETMY end. This way, we can install the camera at the ETMY window and be sure we are looking at ITMY scattered light.

[Jan,Manasa]

We executed plan B. We installed the green laser pointer on POY table and steered the beam  through ITMY to hit the pick off mirror at the ETM end by installing *temporary mirrors. The pick off mirror was adjusted in pitch and yaw to center the reflected beam on the viewport window. We have installed irides on the ring attached to the viewport window to direct the beam to the camera.

*Temporary mirrors were removed from the ITMY chamber after this alignment.

7972   Thu Jan 31 12:44:42 2013 RijuUpdate Photodiode transimpedance

Today I collected the data for shot noise intercept current for MC REFL PD. I didn't get many data points at higher DC voltage of the photodiode, cause the incandescent bulbs get burnt at that level; two bulbs I have burnt today. I will process the data and report.

7973   Thu Jan 31 14:30:50 2013 JenneUpdateLockingMode spacing calc

I have calculated (using Zach's sweet software) the expected mode content for the various possible PRCs that we can make.

Also, Zach was right about the factor of 2.  I see now that I was calculating the mode spacing between a plane wave and a HOM, so the guoy phase had a factor of (n+m+1).  The right thing to do is to get the spacing between the 00 mode and HOMs, so the guoy phase just has (n+m).  Switching from n+m+1=2 to n+m=1, that fixes the factor of 2 problem.

I attach my results as a pdf, since I'm listing out 5 configurations.  Each config has a cartoon, with a small (hard to read) HOM plot, and then at the end, each HOM plot is shown again, but larger.  Also, "TM" is the "test mirror", the flat G&H that we're using as the cavity end mirror.

7974   Thu Jan 31 14:46:05 2013 JenneUpdateRF SystemPhotodiode transimpedance

 Quote: Today I collected the data for shot noise intercept current for MC REFL PD. I didn't get many data points at higher DC voltage of the photodiode, cause the incandescent bulbs get burnt at that level; two bulbs I have burnt today. I will process the data and report.

This work was done in-situ, so no optics on the AS table were moved.  The PSL shutter was blocked since the IR beam was not necessary, and would scatter off the bulb Riju put in front of the PD.

7975   Thu Jan 31 15:20:46 2013 ZachUpdateLockingMode spacing calc

I should mention that I just found a bug in how it treats odd-mirror-number cavities. For such cavities, HG modes with odd horizontal indices should receive an extra roundtrip phase of pi/2 (due to the rotation by the cavity). Because of a numbering convention issue, arbcav actually used to apply this phase shift to even-order modes. Essentially, the only difference is that the fundamental mode was shifted to anti-resonance. Everywhere else, there are modes at both corresponding locations in frequency space, and so it does not back a big difference in terms of cavity design.

Thanks to this IMC modeling we are doing at the workshop, I caught it! It has been fixed in the SVN.

 Quote: I have calculated (using Zach's sweet software) the expected mode content for the various possible PRCs that we can make.  Also, Zach was right about the factor of 2.  I see now that I was calculating the mode spacing between a plane wave and a HOM, so the guoy phase had a factor of (n+m+1).  The right thing to do is to get the spacing between the 00 mode and HOMs, so the guoy phase just has (n+m).  Switching from n+m+1=2 to n+m=1, that fixes the factor of 2 problem.  I attach my results as a pdf, since I'm listing out 5 configurations.  Each config has a cartoon, with a small (hard to read) HOM plot, and then at the end, each HOM plot is shown again, but larger.  Also, "TM" is the "test mirror", the flat G&H that we're using as the cavity end mirror.

7976   Thu Jan 31 15:34:22 2013 RijuUpdateRF SystemPhotodiode transimpedance

Quote:

 Quote: Today I collected the data for shot noise intercept current for MC REFL PD. I didn't get many data points at higher DC voltage of the photodiode, cause the incandescent bulbs get burnt at that level; two bulbs I have burnt today. I will process the data and report.

This work was done in-situ, so no optics on the AS table were moved.  The PSL shutter was blocked since the IR beam was not necessary, and would scatter off the bulb Riju put in front of the PD.

Thanks Jenne.

7977   Thu Jan 31 15:56:38 2013 RijuUpdate Photodiode transimpedance

Summary: Measurement and plot of shot-noise-intercept-current for PDA10CF.

Motivation:It is to measure the shot noise intercept current for PDA10CF.

Result: The final plot is attached here. The plot suggests that the value of shot-noise-intercept current is 0.21mA

Discussion:

The plot is for the measured data of Noise voltage (V/sqrt(Hz)) vs DCcurrent(A). The fitted plot to this measured data follows the noise equation

Vnoise = gdet* sqrt[ 2e (iDC+idet)] ,  where gdet= transimpedance of the PD in RF region as described in manual of PDA255 (i.e. 5e3 when it is not in High-impedance region).

To get an approximate idea of the shot noise intercept current, we may follow the same procedure described in 7946

In the present case dark-noise is 4.3e-08 V/sqrt(Hz)

Therefore dark current(in2) ~dark noise voltage/RF transimpedance = 8.6pA/sqrt(Hz)

Therefore the approximate shot noise intercept current ~ (8.6/18)^2=0.22mA

This value matches well with the fitted data.

From PDA10CF manual, NEP=1.2e-11W/sqrt(Hz) and responsivity~0.9A/W. Therefore the noise current level will be ~10pA.

7978   Thu Jan 31 20:06:22 2013 EvanUpdateLockingPRM/PR2 cavity

[Jenne, Evan]

Tonight we made a non-folded cavity between the PRM and PR2 as follows. I put down two dog clamps to constrain the original position of the PR2 mount. I then loosened the dog clamps holding the mount to the table and nudged the mount until we saw a few reasonably well-aligned bounces in the cavity. I then dogged down the mount.

We played with the PRM and TT2 steering until we saw flashes of TEM00. However, the resonance is not clean so we couldn't lock.

Since we changed the PRM alignment, we had to redo the last bit of steering for the PRM oplev into the photodiode. We also put a few ND filters on the POP camera.

7980   Thu Jan 31 23:48:45 2013 KojiUpdateLockingPRM/PR2 cavity

Wow! What's happened?

As the video showed good quality of resonances, I stopped by at the 40m on the way back home.

I looked at the error signals and found that they indicate high finesse and clear resonance of the sidebands.
The lock was immediate once the gain is set to be -0.004 (previous 0.05ish). This implies the optical gain is ~10 times larger than the previous configration.
The alignment was not easy as POPDC was saturated at ~27000I leave this as a daytime job.

As I misaligned the PRM, I could see that the lock hopped into the next higher order. i.e .from TEM00 to TEM01, from TEM01 to TEM02, etc
This means that the modes are closely located each other, but sufficiently separated to sustain each mode.

I definitely certify that cavity scans will give us meaningful information about the cavity.

7981   Fri Feb 1 09:33:11 2013 JamieUpdateLockingPRM/PR2 cavity

I replaced the BS1 between the POPDC PD and the camera with a 98 reflector, and moved the 50 up before the BS to dump half the light.  Still saturating POPDC, but hopefully the ratio between POPDC and the camera should be better.  We just need to dump more of the power before we get there.  I'll come back to this after C&D if no one else has already gotten to it.

I don't know why I didn't pay more attention last night, but things look way WAY better.  The beams are much cleaner and the power level is much much higher.

7983   Fri Feb 1 12:34:55 2013 JenneUpdatePSLToo much power injected into vacuum

I noticed (while relocking the MC after Jamie and I zeroed the LSC offsets) that the MC refl power was 4.8 V.  Usually we should be ~4.2, so I closed the PSL shutter and went in to measure the power.  We were injecting ~125mW or a little more.  I had adjusted the power the other day, and through yesterday, it looked fine, but overnight it looks like it drifted up.

7984   Fri Feb 1 14:47:17 2013 RijuUpdate Photodiode transimpedance

Summary: Measurement and plot of shot-noise-intercept-current for MC REFL PD.

Motivation:It is to measure the shot noise intercept current for MC REFL PD.

Result: The final plot is attached here. The plot suggests that the value of shot-noise-intercept current is 0.041mA

Discussion:

The plot is for the measured data of Noise voltage (V/sqrt(Hz)) vs DCcurrent(A). The fitted plot to this measured data follows the noise equation

Vnoise = gdet* sqrt[ 2e (iDC+idet)] ,  where gdet= transimpedance of the PD in RF region as described in manual of PDA255 (i.e. 5e3 when it is not in High-impedance region).

To get an approximate idea of the shot noise intercept current, we may follow the same procedure described in 7946

In the present case minimum noise value is 2.03e-08 V/sqrt(Hz)

Therefore dark current(in2) ~dark noise voltage/RF transimpedance = 4.06pA/sqrt(Hz)

Therefore the approximate shot noise intercept current value is (4/18)^2 ~ 0.049mA, which is close to the fitted value.

... hard to believe these numbers. Wrong DC transimpedance? (KA)

7985   Fri Feb 1 15:12:53 2013 JenneUpdateLockingPRM/PR2 cavity

After Jamie did all the work this morning on the POP table, I was able to get the cavity to lock.  It's not very stable until I engage the boost filters in the PRCL loop.  After locking, I tuned up the alignment a bit more.  Now we're taking mode scan data.  Look for results hopefully shortly after Journal Club!

7986   Fri Feb 1 19:55:33 2013 JenneUpdateLockingPRM/PR2 cavity

[Jamie, Koji, Jenne]

We are looking at the mode scan data, and have some preliminary results!  We have data from when the cavity was aligned, when it was slightly misaligned in pitch, and slightly misaligned in yaw.

Inverting the equation for transverse mode spacing, we infer (for pitch misalignment) a cavity g-factor of 0.99, and from there (assuming the G&H mirror is flat and so has a g-factor of 1), we infer a PRM radius of curvature of 168 meters which is ~50% longer than we expected.

More results to come over the weekend from Jamie.

7987   Fri Feb 1 23:12:42 2013 KojiUpdateLockingPRM/PR2 cavity

During the scanning we were riddled by the fact the PDH error and the transmission peaks do not happen simultaneously.

After a little investigation, it was found that "LP100^2" filter is left on in the POPDC filter.

Moreover, it was also found that the whitening filter switches for the POPDC does not switch the analog counterpart.

These were the culprit why we never saw accidental hitting of the max transmission by the peaks when the cavity was not locked.

I know that the most of the whitening filter in the RF paths were checked before (by Keiko?), but the similar failure still exists in the POX path.
We should check for the whitening filters in the DC path as well and fix everything at once. I can offer assistance on the fixing part.

7988   Fri Feb 1 23:52:59 2013 ranaUpdateLockingPRM/PR2 cavity

Very exciting result, if true. I suppose we should try to reconfirm this result by doing another phase map of PRM03.

Is it possible that PR2 is not flat? How would we test to see if the tip-tilt frame screw gives it a curvature? Perhaps we can check with COMSOL.

7993   Mon Feb 4 15:26:10 2013 JamieUpdateComputer Scripts / ProgramsNew "getdata" program to pull NDS channel data, including test points

I've added a new program called getdata (to scripts/general/getdata) that will conveniently pull arbitrary data from an NDS server, either DQ or online (ie. testpoints).

Start times and durations may be specified.  If past data is requested, you must of course be requesting DQ channels.  If no start time is specified, data will be pulled "online", in which case you can specify testpoints.

If an output directory is specified, the retrieved data will be stored in that directory in files named after the channels.  If an output directory is not specified, no output will be

Help usage:

controls@pianosa:~ 0$/opt/rtcds/caltech/c1/scripts/general/getdata --help usage: getdata [-h] [-s START] [-d DURATION] [-o OUTDIR] channel [channel ...] Pull online or DQ data from an NDS server. Use NDSSERVER environment variable to specify host:port. positional arguments: channel Acquisition channel. Multiple channels may be specified acquired at once. optional arguments: -h, --help show this help message and exit -s START, --start START GPS start time. If omitted, online data will be fetched. When specified must also specify duration. -d DURATION, --duration DURATION Length of data to acquire. -o OUTDIR, --outdir OUTDIR Output directory. Data from each channel stored as '.txt'. Any existing data files will be automatically overwritten. controls@pianosa:~ 0$


7999   Tue Feb 5 09:08:11 2013 SteveUpdatePEM exterier doors to be painted

 Quote: The wood exteier walls, gutters and doors were painted at CES-Annex building #69

The east and south end of the 40m emergency exit doors are sealed- tapped off temporarily.  They will be painted on the out side only.  This job will be done by tomorrow noon

Do not open chamber if you smell the paint !

8001   Tue Feb 5 10:18:54 2013 SteveUpdatePEMhigh particle count ALART

Quote:

 Quote: The BS camber is open only. We should close ASAP Outside air quality is 1.7- 2.2  million particles  / cf min of 0.5 micron

Air is still bad and the chambers are closed. Before lunch  Jamie repointed the PRM oplev. Manasa and I reset oplevs: BS and ITMX.

ETMX and ETMY are fine.

SRM and ITMY oplevs needs more work.

The bad outside air quality is pushing up the inside counts.

The outside air is 5 million counts / cf min for 0.3 micron and 2 million counts / cf min for 0.5 micron particles

Do not open chamber over 10,000 counts / cf min of 0.5 micron

8003   Tue Feb 5 12:08:43 2013 Max HortonUpdateSummary PagesUpdating summary pages

Getting started:  Worked on understanding the functionality of summary_page.py.  The problem with the code is that it was written in one 8000 line python script, with sparse documentation.  This makes it difficult to understand and tedious to edit, because it's hard to tell what the precise order of execution is without tracing through the code line by line.  In other words, it's difficult to get an overview of what the code generally does, without literally reading all of it.  I commented several functions / added docstrings to improve clarity and start fixing this problem.

Crontab:  I believe I may have discovered the cause of the 6PM stop on data processing.  I am told that the script that runs the summary_pages.py is called every 6 hours.  I believe that at midnight, the script is processing the next day's data (which is essentially empty) and thus not updating the data from 6PM to midnight for any of the days.

Git:  Finally, created git repository called public_html/__max_40m-summary_testing to use for testing the functionality of my changes to the code (without risking crashing the summary_pages).

8004   Tue Feb 5 15:31:03 2013 SteveUpdateGeneralclean assembly room benches cleaned up

Manasa, Jamie and Steve,

Tip-Tilts and parts moved into the most north " 40m "  cabinet  in the assembly room.

Green-black glass and related components were moved to the 40m E0 cabinet in plastic boxes.

The north flow bench has a few items that belong to us: HE/Ne laser, qpd on translation stages, an iris and one red mirror.  These were moved to the north edge of this bench.

However this leveled table is still full with other people's stuff

8007   Wed Feb 6 11:59:12 2013 JenneUpdateLockingPRC cavity gains

EDIT:  These numbers are for a perfect, non-lossy arm cavity.  So, a half real, half imaginary world.

Carrier uses arm cavity reflectivity for perfectly resonant case.

PRC carrier gain, flipped PR2, PR3 = 61

PRC carrier gain, regular PR2, PR3 = 68  (same value, within errors, for no folding at all).

Carrier gain loss = (68-61)/68 = 10%

SB uses arm cavity reflectivity for perfectly anti-resonant case.

PRC SB gain, flipped PR2, PR3 = 21

PRC SB gain, regular PR2, PR3 = 22 (same value, within errors, for no folding at all). <--- yes, this this "regular PR2, PR3 = 22..."

SB % gain loss = (22-21)/22 = 4.5%

I claim that we will be fine, recycling gain-wise, if we flip the folding mirrors.  If we do as Yuta suggests and flip only one folding mirror, we'll fall somewhere in the middle.

8008   Wed Feb 6 14:51:25 2013 JenneUpdateElectronics1 power supply replaced!

 Quote: Currently, DC power for amplifiers ZHL-1000LN+ is supplied by Aligent E3620A. I tried to use power supply from the side of 1X1 rack, but fuse plug(Phoenix Contact ST-SI-UK-4) showed red LED, so I couldn't use it.

Yuta, Jenne

We fixed things so that we are now using regular fused rack power for these amplifiers.  The fuse no longer had a red LED, but it measured open when we checked the resistance.  Although, somehow (magic?) 13.73V were getting to the other side of the fuse.

Anyhow, replacing the fuse with a new one fixed the problem right up.

8009   Wed Feb 6 15:05:18 2013 SteveUpdateSAFETYcameras must be anchord

Cameras must be immediately anchord to avoid a possible collusion with the view port !

8010   Wed Feb 6 15:10:22 2013 SteveUpdatePEMnew safety signs on exterier doors

Quote:

 Quote: The wood exteier walls, gutters and doors were painted at CES-Annex building #69

The east and south end of the 40m emergency exit doors are sealed- tapped off temporarily.  They will be painted on the out side only.  This job will be done by tomorrow noon

Do not open chamber if you smell the paint !

The east and south end of the 40m emergency exit doors received new safety signs.

8011   Wed Feb 6 15:11:21 2013 JenneUpdateElectronics"Temporary" power supply situation

[Jenne, Yuta, Rana, Steve, Manasa]

We have taken stock of the lab "temporary" power supply situation, and things look much better.

This morning, I removed 2 unused power supplies and a function generator from the PSL table - these had been used for MC ringdown things.

I also removed the non-connected cables that had been used for the RAMMON setup, and the EOM temperature controller circuit.

This afternoon, Yuta removed the 2 HV power supplies that were used to keep PZT2 working near the end of its life.  Since we now have the active TTs, these have been turned off for a while, and just needed to be removed.

Manasa removed the power supply under the POP/POX table that was powering the amplifier for POP22.  If we are going to continue using a Thorlabs PD for POP22, then we need to make a twisted pair of wires (~20 feet) to get power from 1X1.  If we are going to (finally) install a gold RF PD, then that will not be necessary.

I removed the power supply sitting near the bottom of the LSC rack, for another amplifier for POP22 (with minicircuits filters attached).  Again, if we get a gold RF PD, we can remove the filters and amplifier.  If not, we can use the existing twisted pair of wires, and plug them into the rack rather than a power supply.

The power supply under the NE corner of the PSL table was no longer in use.  It was most recently used for amplifiers for the green beat PDs, as Yuta mentioned in elog 6862, those were moved over to 1X2.  In elog 8008 I mentioned that Yuta and I moved those amplifiers over to rack power.

The HV supply, the function generator and the OSA controllers that were on top of the short OMC rack next to the AS table have all been removed.  We need to come up with a better place for the OSAs, since we need to re-install them.  The power supply and the function generator (which was used just for a voltage offset) were formerly used for the output steering PZTs, but lately we have just been using those mirrors as fixed mirrors, since we don't need to steer into the OMC.  Some day, we will replace those mirrors with the output steering active tip tilts, and re-commission the OMC....someday.

The power supply for the amplifier set (that goes with the set of minicircuits filters) for the RAMMON PD (which took light from the IPPOS path) has been removed.  If we determine that we need RAMMON back, we will have to make a twisted pair to power those amplifiers.

SUMMARY:

* If we don't install a gold RF PD for POP22, we need a 20ft twisted pair for +15/GND.

* Also, if we don't install a gold RF PD for POP22, we need to plug the amplifier at the LSC rack into the rack power (twisted pair already exists).

* If we need RAMMON back, we will need a twisted pair to power those amplifiers.

* All other power supplies have been removed, and put away.  We currently have 0 "temporary" power supplies in use in the lab!

8013   Wed Feb 6 15:39:19 2013 SteveUpdateElectronicsDC power supplies in cabinets

East arm cabinet E9 and E10

8014   Wed Feb 6 18:39:08 2013 JenneUpdateLockingPRC cavity gains

[Yuta, Jenne]

We have both calculated, and agree on the numbers for, the PRC gain for carrier and sideband.

We are using the measured arm cavity (power) loss of 150ppm....see elog 5359.

We get a PRC gain for the CARRIER (non-flipped folding) of 21, and PRC gain (flipped folding) of 20This is a 4.7% loss of carrier buildup.

We get a PRC gain for the SIDEBANDS (non-flipped folding) of 69, and PRC gain (flipped folding) of 62This is an 8.8% loss of sideband buildup.

The only difference between the "flipped" and "non-flipped" cases are the L_PR# values - for "non-flipped", I assume no loss of PR2 or PR3, but for the "flipped" case, I assume 1500ppm, as in Rana's email.  Also, all of these cases assume perfect mode matching.  We should see what the effect of poor mode matching is, once Jamie finishes up his calculation.

Why, one might ask, are we getting cavity buildup of ~20, when Kiwamu always quoted ~40?  Good question!  The answer seems, as far as Yuta and I can tell, to be that Kiwamu was always using the reflectivity of the ITM, not the reflectivity of the arm cavity.  The other alternative that makes the math work out is that he's assuming a loss of 25ppm, which we have never measured our arms to be so good.

For those interested in making sure we haven't done anything dumb:

ppm = 1e-6; % ||      |      |        ||            || % PRM    PR2    PR3      ITM           ETM T_PRM = 0.05637; t_PRM = sqrt(T_PRM); L_PRM = 0 *ppm; R_PRM = 1 - T_PRM - L_PRM; r_PRM = sqrt(R_PRM); T_PR2 = 20 *ppm; t_PR2 = sqrt(T_PR2); L_PR2 = 1500 *ppm; R_PR2 = 1 - T_PR2 - L_PR2; r_PR2 = sqrt(R_PR2); T_PR3 = 47 *ppm; t_PR3 = sqrt(T_PR3); L_PR3 = 1500 *ppm; R_PR3 = 1 - T_PR3 - L_PR3; r_PR3 = sqrt(R_PR3); T_ITM = 0.01384; t_ITM = sqrt(T_ITM); L_ITM = 0;%100 *ppm; R_ITM = 1 - T_ITM - L_ITM; r_ITM = sqrt(R_ITM); T_ETM = 15 *ppm; t_ETM = sqrt(T_ETM); L_ETM = 0 *ppm; R_ETM = 1 - T_ETM - L_ETM; r_ETM = sqrt(R_ETM); rtl = 150*ppm;  % measured POWER round trip loss of arm cavities. rtl = rtl/2;     %    because we need the sqrt of the exp() for ampl loss....see Siegman pg414. eIkx_r = exp(-1i*2*pi); r_cav_res = -r_ITM + (t_ITM^2 * r_ETM * eIkx_r * exp(-rtl)) / (1 - r_ITM*r_ETM * eIkx_r * exp(-rtl) ); eIkx_ar = exp(-1i*pi); r_cav_antires = -r_ITM + (t_ITM^2 * r_ETM * eIkx_ar * exp(-rtl)) / (1 - r_ITM*r_ETM * eIkx_ar * exp(-rtl) ); %% PRC buildup gain g_antires = t_PRM*eIkx_ar / (1-r_PRM*r_PR2*r_PR3*r_cav_antires*eIkx_ar); G_ar = g_antires^2; G_ar = abs(G_ar)  % Just to get rid of the imag part that matlab is keeping around. g_res = t_PRM*eIkx_r / (1-r_PRM*r_PR2*r_PR3*r_cav_res*eIkx_r); G_r = g_res^2; G_r = abs(G_r) 

8015   Wed Feb 6 19:59:35 2013 ranaUpdateLockingPRC cavity gains

Getting closer, but need to use the real measured AR reflectivity values, not the 1500 ppm guess. These should be measured at the correct angles and pol, using an NPRO.

8016   Wed Feb 6 20:00:06 2013 ManasaUpdateElectronicsBNC cables piled up at every corner

[Yuta, Steve, Manasa]

There are cables piled up around the access connector area which have been victims of stampedes all the time. I have heard these cables were somehow Den's responsibility.

Now that he is not around here:

I found piled up bnc's open at one end and with no labels lying on the floor near the access connector and PSL area. Yuta, Steve and I tried to trace them and found them connected to data channels. We could not totally get rid of the pile even after almost an hour of struggle, but we tied them together and put them away on the other side of the arm where we rarely walk.

There are more piles around the access connector...we should have a next cleanup session and get rid of these orphaned cables or atleast move them to where they will not be walked on.

8017   Wed Feb 6 20:03:50 2013 ManasaUpdateLockingPRC cavity gains

 Quote: Getting closer, but need to use the real measured AR reflectivity values, not the 1500 ppm guess. These should be measured at the correct angles and pol, using an NPRO.

I'm still on that!

8018   Wed Feb 6 20:19:52 2013 ManasaUpdateOpticsG&H and LaserOptik mirrors

[Koji, Manasa]

We measured the wedge angle of the G&H and LaserOptik mirrors at the OMC lab using an autocollimator and rotation stage.

The wedge angles:

G&H : 18 arc seconds (rough measurement)

LaserOptik : 1.887 deg

8019   Wed Feb 6 22:39:23 2013 JamieUpdateGeneralPRC/arm mode matching with flipped PR2/PR3: coming soon

I intended to post a long analysis of the PRC/arm mode matching for the various TT situations using Nic's a la mode mode matching program, but I seem to have encountered what I think might be a bug.  I'll talk to Nic about it first thing in the AM.  Once the issue is resolved I should be able to post the full analysis fairly quickly.  Sorry about the delay.

8020   Thu Feb 7 09:03:54 2013 ManasaUpdateGeneralStore optics in respective cabinets

@Yuta

The ITMX table has been left open since yesterday. I am disconnecting your oscilloscope and closing the table.

To whomsoever it may concern...

I found about half a dozen new cvi optics (beam splitters, waveplates and lenses) lying around on the SP table.

Please store optics back in their respective cabinets if you are not using them immediately. Somebody might be looking around to use them.

8021   Thu Feb 7 10:35:35 2013 yutaUpdateGeneralStore optics in respective cabinets

I'm not the one who opened the ITMX table yesterday, but thanks for reminding me.
I put POP DC oscilloscope and its cables back.

Also, I relocked PMC and MC. It was unlocked since last night.

8023   Thu Feb 7 14:10:25 2013 ManasaUpdateOpticsLaserOptik - AR Reflectivity - Bad data

Reflectivity of AR surface of LaserOptik (SN6)

The first step measurements of R for AR surface. I am not convinced with the data....because the power meter is a lame detector for this measurement.

I'm repeating the measurements again with PDs. But below is the log R plot for AR surface.

R percentage

6000ppm @ 42 deg
3560ppm @ 44 deg
7880ppm @ 46 deg
4690ppm @ 48 deg

8024   Thu Feb 7 15:46:42 2013 JenneUpdateLockingPR2 flipped

More correctly, a different G&H mirror (which we have a phase map for) was put into the PR2 TT, backwards.

Order of operations:

* Retrieve flat test G&H from BS chamber.  Put 4th dog clamp back on BS optic's base.

* Remove flat G&H from the DLC mount, put the original BS that was in that mount back.  Notes:  That BS had been stored in the G&H's clean optic box.  The DLC mount is engraved with the info for that BS, so it makes sense to put it back.  The DLC mount with BS is now back in a clean storage box.

* Remove PR2 TT from ITMX chamber.

* Remove suspension mounting block from TT frame, lay out flat, magnets up, on lint-free cloth on top of foil.

* Remove former PR2 G&H optic.

* Put what was the flat G&H test optic into the PR2 optic holder, with AR surface at the front.

* Put PR2 suspension block back onto TT frame.

* Put PR2 assembly back in the chamber, solidly against the placement reference blocks that Evan put in last Thursday.

* Close up, clean up, put labels on all the boxes so we know what optic is where.

Why the switcho-changeo?  We have a phase map for the G&H that is the new PR2, and a measured RoC of -706m, surface rms of 8.7nm.  Now, we can measure the former PR2 and see how it compares to our estimate of the RoC from the cavity measurements we've taken recently.

ELOG V3.1.3-