40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log, Page 174 of 344 Not logged in
ID Date Author Type Category Subject
14014   Mon Jun 25 19:14:02 2018 UditSummaryGeneralRe: A summary of the Tip-TIlt Mirror Holder design changes

2. Weighted screw rod at the bottom for tilting the mirror-holder:

The screw length selected here (2") is not interfering with any part of the assembly.

The 'weights' I have here are just thumb nuts from Mcmaster, so their weight is fixed (1.65g each, btw).

Problem I'd like to solve: Find an assortment of weighted, symmetric nuts with caps on one end to fix position on shaft.

3. Set-screws on both side of wire clamp to adjust its horizontal position:

Thanks for pointing out the mismatch in travel distance of protrusion and clamp screws. To match them, the clamp screw slot now sticks out of the profile (by 1.5mm). The range of the clamp motion is +/- 3 mm.

Also, here's a screenshot of the slot in the mirror holder:

--

- Excluding the weighted screw rod assembly, the height gap between assembly COM and wire release point is 3.1 mm.

 Quote: > 2. Weighted screw rod at the bottom for tilting the mirror-holder: Too long. The design of the holder should be check with the entire assembly. We should be able to make it compact if we heavier weights. How are these weights fixed on the shaft? Also can we have options for smaller weights for the case we don't need such a range? Note the mass of the weights. > 3. Set-screws on both side of wire clamp to adjust its horizontal position: How much is the range of the clamp motion limited by the slot for the side screws and the slot for the protrusion? Are they matched? Can you show us the design of the slot made on the mirror holder? >> Where is the center of mass (CoM) for the entire mirror holder assy and how much is the height gap between the CoM and the wire release points. Can you do this with 3/8" and 1/2" fused silica mirrors?

13429   Thu Nov 16 00:14:47 2017 Udit KhandelwalUpdateSUSSOS Sapphire Prism design

Summary:

• SOS solidworks model is nearly complete
• Having trouble with the design of the sensor/actuator head assembly and the lower clamps
• After Gautam's suggestion, installed Abaqus on computer. Teaching it to myself to eventually do FEM analysis and find resonant frequency of the system
• Goal is to replicate frequency listed in the SOS documents to confirm accuracy of computer model, then replace guide rods with sapphire prisms and change geometry to get same results

Questions:

• How accurate do the details (like fillet, chamfer, placement of little vent holes), and material of the different SOS parts need to be in the model?
• If I could get pictures of the lower mirror clamp (document D960008), it would be helpful in making solidworks model. Document is unclear. Same for sensor/actuator head assembly.
13460   Fri Dec 1 17:09:29 2017 Udit KhandelwalSummaryGeneral

Current objectives and statuses:

• CAD Model of 40m lab (facility, chambers, invacuum components etc)
• Status: On hold since I'm unable to acquire general dimensions of
13472   Wed Dec 13 17:46:08 2017 Udit KhandelwalSummaryGeneralSummary of Current Tasks

1. 40m_bldg.dwg has 2D drawing of the 40m building

• After importing file as a 2D sketch into solidworks, make sure to retrace all the lines before performing any 3D extrusion stuff.

2. 40m_VE.dwg has the Vaccuum Envelope.

• Divided the file into individual sketches for the tubes, test mass, and beam splitter chambers (so they can be individually modified later if required).

3. 40melev.dwg has the relative positioning between (1) and (2).

• Using this file to position objects inside building cad.

4. All files can be found in Dropbox folder [40m SOS Modeling], which should be renamed to [40m CAD].

5. Next step would be to add the optical table, mirrors.

Tip-Tilt Suspension

1. Current objective: (refer to D070172) - Increase the length of the side arms (so it matches the dimensions of D960001), while keeping the test mass subassembly at the same height.
2. Future objective: Resonant frequency FEM of the frame (sans the test mass), and then change height to get the desired frequency.

Past Work

• Completed solidworks model of SOS (D960001). I understand this is not the focus right now so this is for reference that the model is ready to be used.

• I will be in India from 16th December until 6th January so this is my final visit for this year. I have enough material to work from home, and will correspond with Koji over email regarding Lab CAD and tip-tilt suspension.
13544   Fri Jan 12 20:35:34 2018 Udit KhandelwalSummaryGeneral2018/01/12 Summary
1. Worked further on positioning vacuum tubes and chambers in the building.
2. Next step would be to find some drawings for optical table positions and vibration isolation stack. Need help with this!
2. Tip Tilt Suspension (D070172)
1. Increased the length of side arms. The overall height of D070172 assembly matches that of D960001.
2. The files are present in dropbox in [40mShare] > [40m_cad_models] > [TT - Tip Tilt Suspension]
13560   Fri Jan 19 15:22:19 2018 Udit KhandelwalSummaryGeneral40m CAD update 2018/01/19

1. All parts will be now named according to the numbering system in this excel sheet: LIGO 40m Parts List in dropbox folder [40mShare] > [40m_cad_models] > [40m Lab CAD]
2. I've placed optical tables in the chambers at 34.82" from the bottom for now. This was chosen by aligning the centre of test mass of SOS assembly (D960001) with that of vacuum tube (Steve however pointed out last week they might not necessarily be concentric).

13561   Fri Jan 19 20:59:07 2018 Udit KhandelwalUpdateGeneralSolidworks Rendering

Rendered the SOS assembly (D960001) with correct materials and all and it looks very nice. Will extend this to the building cad later.

13601   Fri Feb 2 21:12:46 2018 Udit KhandelwalSummaryGeneralSummary - 2018/02/02

Discussed with Koji about motivation to simplify the design of this assembly, which has many unnecessary over-constraints. I have started to cad alternate parts with the aim of removing these over-constraints.

Acquired a stack of original engineering drawings of the vacuum chambers from Steve which I will take home, get scanned, and then use as reference for the cad i'm working on.

Other:
Started paperwork at west bridge office to get paid as an "occasional employee". Hopefully I receive old money.

13638   Fri Feb 16 21:03:17 2018 Udit KhandelwalSummaryGeneralSummary 2018/02/16

Updated the dimensions of and fleshed out the chambers in greater detail, by referring to the engineering drawings that Steve gave to me. I have scanned and uploaded most of these drawings to Dropbox in [40mShare]>[40m_cad_models]>[Vacuum Chamber Drawing Scans]. The excel file "LIGO 40m Parts List" in the [40m Lab CAD] folder also lists the Steve drawings I referenced for dimensions of each part.

Next steps:
1. Finish details of all chambers.
2. Start placing representative blocks on the optical table.

13654   Fri Feb 23 20:46:04 2018 Udit KhandelwalSummaryGeneralCAD Summary 2018/02/23

I have more or less finished cadding the test mass chamber by referring to the drawings Steve gave me. Finer details like lugs and bolts and window flaps can be left for later. Here's a quick render:

13677   Fri Mar 9 20:35:41 2018 Udit KhandelwalSummaryGeneralSummary 2018/03/09

1. Optical Table Layout

I had discussed with Koji a way to record coordinates of optical table equipments in a text file, and load to solidworks. The goal is to make it easier to move things around on the table in the CAD. While I have succeeded in importing coordinates through txt files, there is still a lot of tediousness in converting these points into sketches. Furthermore, the task has to be redone everytime a coordinate is added to or changed in the txt file. Koji and I think that this can all be automated through solidworks macros, so I will explore that option for the next two weeks.

Steve will help find manufacturing drawings of the BS chamber. I have completed the ETM chambers, while the ITM ones are identical to them so I will reuse parts for the CAD.

13865   Fri May 18 18:14:18 2018 Udit KhandelwalSummaryGeneralSummary 05/18/2018

Tip-Tilt Suspension Design:

Designed a new ECD plate and changed dimensions of the side arms after discussing with Koji. After getting feedback on the changes, I will finish the assembly and send it to him to get approved for manufacturing.

13884   Wed May 23 19:24:37 2018 Udit KhandelwalSummaryGeneralSummary 05/23/2018

Tip-Tilt Redesign Project with Koji:

Did further itirations to the ECD backplate. Going to determine minimum thickness between magnet hole and plus sign for eddy current damping.

Chamber optical table layouts

Finished the positioning of optics and instruments in SolidWorks for the Vertex chambers. The reference for positioning is "40m_upgrade_layout_Dec2012.dwg", and solidworks files I created are in the main 40m CAD folder.

13996   Thu Jun 21 14:23:22 2018 Udit KhandelwalSummaryGeneralA summary of the Tip-TIlt Mirror Holder design changes

Here’s a quick summary of the Tip-Tilt Design updates (all files are in the dropbox in [TipTiltSus>TT_New]) that I have been working on with Koji and Steve's help.

1. Plate on top to hold mirror in place:

The plate is 0.5 mm thick. I did a rough FEA with 10 N force on the point of pressure on it, and it bent easily.

2. Weighted screw rod at the bottom for tilting the mirror-holder:

I did a very simplified free body analysis to calculate the required length of the rod to achieve a +/- 15 mRad tilt, and got around 1.5 inches.

3. Set-screws on both side of wire clamp to adjust its horizontal position:

• Front view (showing set screws on either side of the clamp to push it into the desired position, and the clamp in the middle with screws on top and bottom to fix its position):

• Exploded view showing protrusion in clamp that sits in the mirror holder inset:

• Exploded view showing inset in the mirror holder to slide protrusion in:

1. Used the same screw size in most places to reduce complexity.

2. The mirror holder I have worked on is a little different from the actual piece I have on my table. Which one do you prefer (Koji)?

14042   Fri Jul 6 19:39:37 2018 Udit KhandelwalSummaryGeneralCAD drawings of cantilever suspension required

Request to Koji to acquire the drawings or 3D CAD of the cantilever suspensions of the Tip-Tilt Assembly!

14047   Mon Jul 9 17:29:28 2018 Udit KhandelwalSummaryTip-TIltTipTilt mirror holder final changes

Final Summary of changes to mirror holder in Tip-Tilt holder.

Determining minimum range for Side Clamp:

1. The initial distance b/w wire-release point and mirror assembly COM = 0.265 mm

2. But this distance is assuming that wire-release point is at mid-point of clamp. So I'm settling on a range of +/- 1mm. The screenshots below confirm range of ~1mm between (1) side screw & protrusion and (2) clamp screw and clamp.

Determining length of tilt-weight assembly rod at the bottom to get $\pm$ 20mRad range

The tilt-weight assembly is made from following Mcmaster parts:
Rod   - 95412A864 18-8 SS  #2-56 Threaded Rod
Nuts  - 91855A103 18-8 SS #2-56 Acorn Cap Nut

Since the weights are fixed, only rod length can be changed to get the angle range.

$tan \theta =\frac{d}{h}$

$d= h \times tan\theta = 34.25\text{mm} \times tan(20 \text{mRad}) = 0.69 \text{mm}$
So a range of 1 mm between nut's inner face and mirror-holder face should be enough. Since holder is 12 mm thick, rod length = 12mm + 2 x 1mm + 2 x (nut length) = 12 + 2 + 9.6 = 23.6 mm = 0.93 inch. So a 1" rod from Mcmaster will be fine.

6161   Tue Jan 3 17:46:38 2012 Updated Stewart platform design requirementsUpdateGeneralUpdated design requirements for a Stewart platform

I updated the design requirements for the Stewart platform.  I weighed the unloaded "dirty" SOS that was sitting on the workbench in the control room; its mass is 11 kg.  Steve suggested that the OSEMs (not installed on this model) would add another 0.5 kg.  From the specs in the final SOS design document, LIGO-T970135, I added 0.25 kg for the optic itself; I am therefore taking the total payload mass to be 11.75 kg.  (Now, the upper stage of the Stewart platform itself will likely add a nontrivial amount, but I am not worrying about this yet.)

I have e-mailed Janeen Romie to obtain the actual center of mass and principal moments of inertia of the platform.  I also cooked up a simple scheme to measure both quantities, should this information not be available.  It would involve rigidly mounting the dirty SOS to a rigid bar hung from a pivot.  By translating the mount point in two dimensions and measuring the period of the pendulum, I ought to be able to find the center of mass and moments of inertia by multilinear regression.  However, this elaborate scheme is not necessary to just compute some ballpark figures; it could wait until a later stage in the design.  For the time being, I just rescaled the moment of inertia proportional to the increase in mass, such that the torque-to-force ratio is unchanged.

As such, the design requirements are now

• linear travel: 40 microns peak to peak (based on SOS design requirements in LIGO-T950011)
• angular travel: 3 mrad peak to peak (based on SOS design requirements in LIGO-T950011)
• payload mass: 11.75 kg (measured unloaded mass, plus educated guesses about combined mass of OSEMs and optic)
• payload moment of inertia: 0.0232 kg m^2 (wild guess)
• bandwidth: 500 Hz (suggestion of Rana and Koji: ~kHz)

From these assumptions, the revised actuator requirements and dimensions are:

• peak actuator force: 2.04 kN
• minimum radius of top platform: 15 cm
• minimum radius of bottom platform: 30 cm
• minimum height: 26 cm

See the attached PDF document.

It appears that the actuator that I had originally nominated, PI's model P-225.80, would very nearly meet the actuator force requirement.  Steve also pointed out the following single-axis shakers that are already in use in the 40m:

• Brüel & Kjær type 4809
• Brüel & Kjær type 4810

I want to find out if either of these would meet the present need, but I'm waiting on a response from the manufacturer to get access to the data sheets.

397   Sun Mar 23 10:42:54 2008 ValeraSummaryElectronicsRFAM of the RF stabilization box is measured
I reconstructed Tobin's setup to measure the RFAM after the RF stabilization box in the 166 MHz modulation path.
The setup consisted of the splitter and the mixer followed by the RF low pass filter and the SR560 (gain x100).
The RF level into splitter was 20 dBm. The Mini-Circuits ZLW-3H (17 dBm LO) mixer was used. The LO was taken
straight out of the splitter and the RF path was attenuated by 11 dBm, The DC out of the mixer was 700 mV.
The noise floor was measured with the RF input of the mixer terminated on 50 Ohm. The 45 MHz measurement
in broad band setting looks better than the noise floor at high frequencies. I am not sure what was wrong with
one or both of those measurements. The 9 MHz measurements are above the noise floor.

The RFAM meets the AdvLIGO requirements in the detection band (f > 10 Hz).

The attached zipped files are:
SRS003 9 MHz DC-200 Hz
SRS004 9 MHz DC-26 kHz
SRS006 45 MHz DC-200 Hz
SRS005 45 MHz DC-26 kHz
SRS007 Noise floor DC-200 Hz
SRS008 Noise floor DC-26 kHz
12121   Wed May 18 17:42:52 2016 VarunUpdateGeneralSURF 2016

Finished writing the code on whitening. I have to still test it. uploaded on github noise cancellation repo. @eric could you give me some data of noise power spectral density for testing the code?

-Varun

 Quote: Hello, I am Varun Kelkar. I will be working at the 40m lab as a SURF student this summer with Eric Quintero on Audio processing for real time control system signals. This week I will mostly be working on implementing basic DSP C-code offline. Currently I am trying to write a code for noise whitening. -Varun

12123   Fri May 20 00:06:19 2016 VarunUpdateGeneralSURF 2016

I have written a basic version of AGC, and have done some tests with a data file. will do tests on whitening and agc today. Also, today I have to go to the SSN office. Hence will be late.

-Varun

Quote:

Finished writing the code on whitening. I have to still test it. uploaded on github noise cancellation repo. @eric could you give me some data of noise power spectral density for testing the code?

-Varun

 Quote: Hello, I am Varun Kelkar. I will be working at the 40m lab as a SURF student this summer with Eric Quintero on Audio processing for real time control system signals. This week I will mostly be working on implementing basic DSP C-code offline. Currently I am trying to write a code for noise whitening. -Varun

12127   Mon May 23 17:47:51 2016 VarunUpdateGeneralSURF 2016

Tested the AGC today with LSC cavity transmission signal and error signal. Not in real time still.

Key to attachments:

cav_tr-eps-converted-to.pdf: LSC cavity transmission signal input

cav_tr_out-eps-converted-to.pdf: LSC cavity transmission signal, output of the AGC.

12129   Tue May 24 17:55:17 2016 VarunUpdateElectronicsUsing Altium

Contacted Charles regarding use of Altium. Got to know that Altium is installed on cit40m iMac in Win7 on VirtualBox. Had to update Virtualbox to get it working. Altium now works for sometime, but then fails, saying that it is unlicensed.

12136   Wed May 25 14:29:31 2016 VarunUpdateGeneralSURF 2016

Edited the AGC to include overlapping frames yesterday. forgot to put an elog on it!

 Quote: Tested the AGC today with LSC cavity transmission signal and error signal. Not in real time still. Key to attachments: cav_tr-eps-converted-to.pdf: LSC cavity transmission signal input cav_tr_out-eps-converted-to.pdf: LSC cavity transmission signal, output of the AGC.

12137   Thu May 26 18:10:48 2016 VarunUpdateGeneralSURF 2016

Wrote and tested a function for downconversion. It contains a mixer with a sinusoidal input for modulation with the desired frequency and a 2nd order butterworth low pass filter to remove the higher frequency-shifted part of the modulated signal. I have tested this with input of 2kHz giving a good output of 200 Hz on the speaker. Codes are uploaded on github, will update the real time document tomorrow.

-Varun

Quote:

Edited the AGC to include overlapping frames yesterday. forgot to put an elog on it!

 Quote: Tested the AGC today with LSC cavity transmission signal and error signal. Not in real time still. Key to attachments: cav_tr-eps-converted-to.pdf: LSC cavity transmission signal input cav_tr_out-eps-converted-to.pdf: LSC cavity transmission signal, output of the AGC.

12139   Fri May 27 11:54:22 2016 VarunUpdateGeneralPackage delivery

A package labelled 'UPS Ground' has arrived.

-Varun

12143   Wed Jun 1 11:19:14 2016 VarunUpdateGeneralUpdate of work till now

Completed:

Wrote and tested a code for AGC using cavity transmission signal and length error signal.

Wrote and tested a code for frequency shifting (downconversion) using mixing and LPF

Wrote a code for whitening using FFT.

Altium working on cit40m iMac

Plans:

Writing codes for Frequency warping and whitening in time domain.

Implement AGC and frequency shifting on the real time control system.

Calculate requirements for Anti-aliasing filter.

12150   Fri Jun 3 17:56:14 2016 VarunUpdateGeneralDAFI update

Wrote and tested a phase vocoder, with two of its applications:

1) Time scaling: This enables change of time duration without affecting the pitch.

2) Frequency warping: This changes the pitch of the sound without affecting the time duration.

1 & 2 tested offline with cavity transmission signal. 1) gives speedup of 2, and 2 gives frequency warping (pitch lowering by a factor of 2)

12154   Tue Jun 7 18:20:18 2016 VarunUpdateGeneralDAFI update

Tried to implement AGC on FE. Had some trouble bringing the code into the correct form. It looks okay now. However, this agc code as well as idenntity code (input = output) doesnt seem to build on the c1lsc FE. Have not tried too many debugging steps yet, will come and check the problem tomorrow.

-Varun

 Quote: Wrote and tested a phase vocoder, with two of its applications: 1) Time scaling: This enables change of time duration without affecting the pitch. 2) Frequency warping: This changes the pitch of the sound without affecting the time duration. 1 & 2 tested offline with cavity transmission signal. 1) gives speedup of 2, and 2 gives frequency warping (pitch lowering by a factor of 2) codes uploaded on github repo

12159   Wed Jun 8 16:12:38 2016 VarunUpdateGeneralDAFI update

Summary: I am implementing digital audio filtering on various interferometer signals in order to listen to the processed audio which will help in characterizing and noise reduction in the interferometer. Following is an implementation of an Automatic Gain control (AGC) block on an LSC input signal.

Details of AGC: Currently, the AGC code implemented on FE takes input to fill a frame, then calculates the power in each frame and gives an appropriate gain to it, so that the new power content is to the required level. It is then written to the output, frame by frame. The frame is currently a rectangular window. The frame length and hop size can be adjusted. Current values are as follows:

frame length is 512 samples

hop length is 128 samples.

The input and output are delayed by 1 frame.

Details of testing: Attachment 1 shows a simulink diagram of the DAF system. Eric made this and I modified it later on. Testing was done using signal from the "LSC1" channel. Attachments 2 and 3 show aquired input and output of the AGC respectively. Gain of the preamp of the LSC input signal was varied over a total time span of 200 s. Each gain value was kept for a duration of about 20 seconds. The varying power levels can be seen in the input plot.

The output shows a uniform power level throughout.

Quote:

Tried to implement AGC on FE. Had some trouble bringing the code into the correct form. It looks okay now. However, this agc code as well as idenntity code (input = output) doesnt seem to build on the c1lsc FE. Have not tried too many debugging steps yet, will come and check the problem tomorrow.

-Varun

 Quote: Wrote and tested a phase vocoder, with two of its applications: 1) Time scaling: This enables change of time duration without affecting the pitch. 2) Frequency warping: This changes the pitch of the sound without affecting the time duration. 1 & 2 tested offline with cavity transmission signal. 1) gives speedup of 2, and 2 gives frequency warping (pitch lowering by a factor of 2) codes uploaded on github repo

12164   Thu Jun 9 19:08:58 2016 VarunUpdateElectronicsAnti-Aliasing Filter update

Eric gave me a psd plot of a signal which would be the input of a channel of the AA filter. the Nyquist freq. is about 32.8kHz.

Following are plots depicting the ratio of the aliased downconverted signal and the signal below 32.8 kHz. The first plot is for (to-be) aliased signal frequencies from 32.8 to 65.5k, and the second plot is for (to-be) aliased signals from 65.5k to 98.3k. In case of the first plot, the 36kHz peak will alias to 29kHz, and is about 30 times (29.5dB) greater than the signal there. Hence, the filter should give about 70dB attenuation there. Since this attenuation is not required by most other frequencies up to 65.5k, an option could be to use a notch filter to remove the frequency peak at 36k, and put a requirement of 45-50 dB attenuation on other frequencies.

In case of the second plot, the frequencies between 90 to 100k again need to be attenuated by more than 70 dB. However, if there is a -20dB/decade slope in stop band, we already have about 10 dB attenuation here as compared to around 32k.

The X axis of both plots is in Hz.

12203   Mon Jun 20 16:33:09 2016 VarunUpdateElectronicsAnti-Aliasing Filter circuit schematic

Summary: The aim is to design an analog anti-aliasing (AA) filter placed before the ADC, whose function is to filter out components of the input spectrum that have frequencies higher than the Nyquist frequency. This needs to be done so that there is no contamination of aliased downconverted high-frequency signals into the ADC output. I have put down and simulated a circuit to do this, based on the spectra of a few interferometer signals that eric Provided. Attachment 1 shows such an input PSD, treated with whitening filter, before the AA. The sampling rate is 65536 Hz and hence the Nyquist freq. is 32768 Hz.

Motivation: Attachments 2 and 3 show the plot of required attenuation for various frequencies above the Nyquist. We can see a peak at 36 kHz, which will alias to about 29kHz. It will require about 70 dB attenuation here. This indicates that use of a notch filter combined with a low pass filter can be used.

Details of Schematic: Attachment 4 shows the schematic of a Boctor low pass notch filter, cascaded by a 2nd order LPF. The stopband frequency of the boctor filter can be tuned to around 36 kHz. Its main advantage for the boctor is better insensitivity to component value tolerances, use of a single op amp, and relatively independent tuning of parameters.  The various component values are calculated from here. The transfer functions for the circuit shown in attachment 4 were simulated using TINA - a spice based simulation software. The transfer function is shown in attachment 5.

A few more calculations: Attachment 6 shows the output psd after the signal has been treated with AA. Attachments 7 and 8 show the ratio of aliased downconverted signal and the unaliased signal of the output. Here, we can see that above about 13 kHz, the ratios go above -40dB, which is apparently undesirable. However, we also see from the transfer function of the filter that the gain falls to less than -20dB after about this frequency, and the aliased signals are atleast 20 dB lower than this, atleast upto about 29 kHz in attachment 7 and about 25 kHz in attachment 8. This means that the aliased signals are negligible as compared to the low frequencies even if they are not negligible as compared to the higher frequencies (above 13 kHz) into which they would get downconverted due to sampling. But these higher frequencies (above 13 kHz) themselves are small.

The filter overall, is 4th order. Considering this and the above discussion, I need to decide what changes to make in the existing schematic. For now, I could discuss with eric to finalize the opamp and start building the pcb board design.

12212   Wed Jun 22 14:03:42 2016 VarunUpdateElectronicsAnti-Aliasing Filter circuit schematic

I found an anti-aliasing circuit on the 40m wiki. It consists of A differential LPF made using THS4131 low noise differential op-amp (one of the main applications of which is preprocessing before the ADC), and a notch. I modified it to arrange for the desired bandwidth (about 8 kHz) and notch after the Nyquist frequency at 36 kHz. I simulated it to get the attached results:

Attachment 1: It shows the input PSD (same as the one posted in the previous elog), the filter transfer function, and The resulting output.

Attachment 2: The circuit schematic. The initial part using THS4131 is a differential LPF and the subsequent RC network is the notch.

Attachment 3: This shows the ratio of the aliased downconverted signal to the the in-band signal, representative of the contamination in each bin. Here too, the aliased signals are negligible as compared to the low frequencies but they are not negligible as compared to the higher frequencies (above 10 kHz) into which they would get downconverted due to sampling. However, here, the attenuation at 8kHz is less than 6 dB while in the previous circuit, it was about 12 dB. One problem with this circuit is at about 6kHz, there is aliased signal from the 65k to 98kHz band, but this can be taken care of by adding an LPF later.

12251   Wed Jul 6 10:58:29 2016 VarunUpdateCDSDAFI review

The DAFI block was reviewed by Rana yesterday. The following changes/improvements were suggested: (Updated on 20th July 2016 with tasks taat remain in red)

1) include all the various channels like PEM, LSC, ASC, SUS, SEI, etc. as the inputs. Currently the inputs are only the LSC.
2) include all the control signals.
3) create a very detailed diagram of the entire signal flow and plan tasks accordingly.
4) Enable cascading of various DSP processes.
5) Adjusting the gain of the AGC such that the amplitude of the output signal comes to about half the peak amplitude offered by the ADC. This will help taking advantage of the entire dynamic range of the ADC.
6) change the enable button styles from a text input based controller to a button controller.
7) Currently, disabling a particular signal terminates the signal. Instead, it should turn into a unity gain block on disabling.
8) Check if the Fibox does AC coupling or not. If not, add an AC coupling arrangement in the DAFI.
9) Check the nature of the ADC1 and ADC2 inputs to the DAFI. I checked them yesterday, and they are channels 25 and 26 of ADC0, which are empty.

12118   Tue May 17 05:50:43 2016 Varun KelkarUpdateGeneralSURF 2016

Hello, I am Varun Kelkar. I will be working at the 40m lab as a SURF student this summer with Eric Quintero on Audio processing for real time control system signals. This week I will mostly be working on implementing basic DSP C-code offline. Currently I am trying to write a code for noise whitening.

-Varun

7558   Tue Oct 16 16:46:20 2012 Vent Czar and CzarinaSummaryGeneralvent stuff

We're discussing the plan for the next vent.  Now that we have all the tip-tilt stuff in hand, we should get ready for the big TT installation vent ASAP.  The question is what remains to be done, and when can we be ready

Stuff that needs doing:

* characterize TTs, check electrical connection of quadrupus cables, bake cables and base plates - a week or so (not including baking)

* phase map of LaserOptik mirrors (decide on needed resolution) - a week or so?

* get ready to swap G&H mirrors with LaserOptik mirrors (PR2, PR3, SR2, SR3)

* assembly of black glass beam dumps - one afternoon

* green periscope moving - day or so to make sure greens are resonant in arms, so we have something to align to

* redraw of REFL path?? (rotate 2nd refl mirror to reflect beam to the east, then add mirror to get beam out of IOO chamber viewport), cut hole in BS oplev table's box, install black beam tube

* FI camera mount - make sure it's here, baked

* active TT pitch damping plan

* passive TT weight addition and pitch alignment plan

* camera plan for taking in-chamber photos without touching the tables

* look at layout of ITMY table.  POY pick-off too close to main beam.  Can we move POY pick-off to BS table?

* remount black glass baffles on SOS cages.

Stuff to bake:

* TT base plates

* FI mirror mount for FI camera

* stuff to add to active TTs

* stuff to add to passive TTs

6117   Wed Dec 14 12:22:00 2011 VladimirHowToComputersligo_viewer installed on pianosa

I made a test installation of ligo_viewer in /users/volodya/ligo_viewer-0.5.0c . It runs on pianosa (the Ubuntu machine) and needs Tcl/Tk 8.5.

To try it out run the following command on pianosa:

cd /users/volodya/ligo_viewer-0.5.0c/

./ligo_viewer.no_install

Press "CONNECT" to connect to the NDS server and explore. There are slides describing ligo_viewer at http://volodya-project.sourceforge.net/Ligo_viewer.pdf

Installation notes:

Use /users/volodya/ligo_viewer-0.5.0c.tgz or later version - it has been updated to work with 64 bit machines.

Make sure Tcl and Tk development packages are installed. You can find required packages by running

apt-file search tclConfig.sh

apt-file search tkConfig.sh

If apt-file returns empty output run apt-file update

Unpack ligo_viewer-0.5.0c.tgz, change into the created directory.

Run the following command to configure:

export CFLAGS=-I/usr/include/tcl8.5
./configure --with-tcl=/usr/lib/tcl8.5/ --with-tk=/usr/lib/tk8.5/

This works on Ubuntu machines. --with-tcl and --with-tk should point to the directories containing tclConfig.sh and tkConfig.sh correspondingly.

Run "make".

You can test the compilation with ./ligo_viewer.no_install

If everything works install with make install

If Tcl/Tk 8.5 is unavailable it should work with Tcl/Tk 8.3 or 8.4

6061   Thu Dec 1 18:30:39 2011 Vladimir, DenUpdatedigital noiseFoton error

Foton Matlab Error

We investigated some more the discrepancy between Matlab and Foton numbers. The comparison of cheby1(k, 1, 2*12/16384) was done between versions implemented in Matlab, R and Octave. Filters created by R and Octave agree with Foton.

Also, we found that Matlab has gross precision errors for cutoff frequencies just smaller than used in our fitler, for example cheby1(6, 2*3/16384) fails miserably.

6083   Wed Dec 7 20:55:44 2011 Vladimir, DenUpdatedigital noiseMatlab error

 Quote: It would be useful to see some plots so we could figure out exactly what magnitude and phase error correspond to "gross" and "miserable".

To show why Matlab is bad in filtering at small cut-off frequencies we did the same thing in Matab, Octave and R: we've taken the low-pass chebyshev filter of the type 1, order 6, ripple 1 dB, the sampling frequency was 16384 Hz and cut-off frequency varied from 1 to 1000 Hz. Here is the plot for the gain of the zpk model versus to cut-off frequency.

We can see that Matlab's gain shows surprising gains for low cut-off frequencies through for > 100 Hz it is fine. In the next table we compare gain from Foton, Matlab, R and Octave for the same filter. So Foton is also good

 freq R_gain matlab_gain octave_gain Foton_gain 1 3.05186270655452e-24 4.8824152e-22 3.05186271e-24 3.05186e-24 10 3.04698335947093e-18 1.8268825e-16 3.04698336e-18 3.04698e-18 100 2.99910841393644e-12 2.9991084e-12 2.99910841e-12 2.99911e-12 1000 2.60247212721439e-06 2.6024721e-06 2.60247213e-06 2.60247e-06
16942   Thu Jun 23 15:05:01 2022 Water MonitorUpdateUpgradeWater Bottle Refill

22:05:02 UTC Jordan refilled his water bottle at the water dispenser in the control room.

6869   Mon Jun 25 15:19:07 2012 YaakovUpdatePEMAdded microphone channels, moved accelorometer channels

Jenne and I renamed the mic channels Den created (elog 6664) to MIC_1, MIC_2, etc from the original accelerometer names to keep things clear. We then added 6 new channels (22-27) for the accelerometers, named ACC_MC1_X, Y, Z, ACC_MC2_X, Y, Z, etc. (See the screenshot below). We also added a DAQ channel block and listed out the IN1 channel of all the sensors. We compiled and started the model, and checked that all the channels were there in DataViewer.

6905   Mon Jul 2 23:08:38 2012 YaakovUpdateSTACISTurning on STACIS

This past Friday I swapped out a burnt resistor on the spare STACIS unit I'm working with and powered it up. Here's the setup:

And here's what happened:

X an Y directions: When I switched from open to closed loop (making the internal geophones provide feedback), the STACIS started making a loud noise- it seemed like it was oscillating uncontrollably.

Z direction: The same thing happened in z until I added some weight to the top of the STACIS- then it quieted down, and seemed to work okay. The geophone signal dropped considerably compared to the open loop signal, which is expected if the feedback is working.

Then I tried driving the PZTs with a signal from the SR785 network analyzer. With an amplitude of tens of mV and frequencies from around .1 to 200 Hz, I could see the accelerometers I mounted on top of the STACIS definitely register motion, which means I was successfully driving the PZTs.

Below are transfer functions of the STACIS as I drove the PZTs from .1 to 100 Hz at 10 mV. The top graph is open loop, the second is closed loop. These were measured with the internal geophones.

In the bottom graph, "A" is closed loop and "B" is open loop, where the transfer functions were taken with the accelerometers instead of the geophones.

6908   Tue Jul 3 18:58:14 2012 YaakovUpdateSTACISMore transfer functions and netGPIB status

I'm still having issues with the STACIS oscillating uncontrollably with the slightest extra vibration, but with some more added weight both x and z direction are stable if you don't disturb the setup.

I took more transfer functions of the STACIS. In the last data I took Jenne pointed out that the geophone signals were not correlated well with the driving signal, so I increased the amplitude of the driving signal and am looking in x and y too instead of just z.

Details of the driving signal: 25 mV, swept sine from 0.1 to 100 Hz from the SR785.

NOTE: The data below was all transferred from the SR785 using netGPIB, which works fine, if anyone was interested in using it.

Open loop in the y direction, taken with the y geophone (magnitude on top, phase on bottom):

Open loop in the x direction, taken with the x geophone (with some extra weight to try to make the closed loop more stable):

Open loop in the x direction, taken with accelerometer instead of geophone:

6963   Wed Jul 11 14:27:29 2012 YaakovSummarySTACISCurrent STACIS Status

The X and Y directions in the STACIS still both oscillate uncontrollably in closed loop, so I'll be doing my testing in Z for now. If I need to use the other axes I'll lower their gain with the pots and add weight to the STACIS platform to try to make it more stable.

Measurements I've taken for Z:

--Open loop gain, taken by driving the PZTs with a swept sine signal and measuring with both internal geophones and external accelerometers. These measurements look a lot like the plots supplied by the STACIS manufacturer, with a resonance at 15-16 Hz (X and Y also look good). Figure below was taken with geophones:

--Open loop gain, where the input is ambient seismic noise measured by one set of accelerometers on the floor and one set on top of the STACIS:

--Closed loop gain, where the input is ambient seismic noise, and feedback is supplied by the geophones (like normal STACIS operation). There's a definite drop in the transfer function, as expected:

--Open and closed loop transfer functions superimposed (the higher one is open):

I am currently working on using the less-noisy accelerometers to provide feedback instead of the geophones. I have found the right point before the extension board to input the accelerometer signal which is NOT the same as the Signal IN/OUT cables- those are at the end of the board, after amplifying and filtering. I want the accelerometer signal to go through the same circuitry as the geophone signal so that the noise of the sensors themselves can be compared.

Problem: Coherence isn't great between the accelerometer sets at low frequencies, which leads to a not very smooth transfer function. I might try using the shaker, because the larger motion may lead to better coherence between the accelerometers on top of the STACIS and at its base.

6969   Thu Jul 12 15:43:07 2012 YaakovUpdateSTACISNew input point, using accelerometers for feedback

Here's a picture of where I am now inputting signals into the STACIS with the accelerometers (the orange and blue wires):

I know this is the right point because I could see the geophone signal from these points . By inputting a swept sine signal into this point, I was able to take a transfer function of this first amplifier/filtering circuit board, which will be useful if I need to make my own filter for the STACIS:

I have unplugged the geophones and am inputting a signal from an accelerometer into this point. The accelerometers output a different signal than the geophones, so I am trying to modify the accelerometer signal to be closer to the geophone one. I've lowered the gain on all the pots for the z axis and put in several BNC attenuators to lower the accelerometer signal amplitude.

At the moment, using the accelerometers as feedback makes the platform vibration worse, which will hopefully be solved by some more attenuation or filtering of the accelerometer signal.

6988   Wed Jul 18 13:53:34 2012 YaakovUpdateSTACISWeekly update

I have been working on substituting the internal geophones in the STACIS with accelerometers, and this week specifically I have been trying to modify the accelerometer signal so the STACIS PZTs respond properly.

The major problem was that the high signal amplitude caused the STACIS to oscillate uncontrollably, so I lowered all of the pots (for the z direction) and placed several BNC attenuators before the accelerometer signal enters the first amplifier board. The accelerometers now successfully provide feedback without making the STACIS unstable, as shown by this transfer function (the higher and flatter line is open loop, the lower is closed loop with accelerometers providing feedback):

The next step is to optimize the accelerometer feedback so it provides good isolation from 0.1 to 3 Hz, a span that the geophones introduced a lot of noise into. The accelerometers definitely don't introduce as much noise in that region, but don't seem to be doing much isolation either. I will also make some more quantitative plots of the platform motion (using the calibration value for the Wilcoxon accelerometers in the velocity setting with a gain of 1).

Some random discoveries I made this week which are relevant for STACIS testing:

1) Placing weight on the STACIS platform improves stability, but NOT if several blocks are placed on top of each other (they rub against each other, causing lots of vibrations).

2) The accelerometer that is providing feedback must be VERY securely fastened to the STACIS platform; even with three clamps there was extra motion that caused instability. Luckily, there's a convenient steel flange Steve showed me which has a hole that perfectly accommodates the accelerometer and doubles as a weight for the platform. Here is said flange, clamped to the STACIS platform with the accelerometer sitting in the center:

3) Using the shaker next to the STACIS (all on one platform) improves coherence between the base and platform accelerometers above around 10 Hz, but does nothing lower than that, which unfortunately is the region I'm most concerned with.

7020   Tue Jul 24 18:33:12 2012 YaakovUpdateVIDEOCentering the MCR camera

Jenne and I centered the MCR camera on the AP table.

We moved the camera as far as it could go on its mount without shifting its screws, and adjusted the optic that the camera looks at for the rest of the way.

7026   Wed Jul 25 11:41:00 2012 YaakovUpdateVIDEOCentering the MCR camera

 Quote: Jenne and I centered the MCR camera on the AP table. We moved the camera as far as it could go on its mount without shifting its screws, and adjusted the optic that the camera looks at for the rest of the way.

The spot was still off center on the quad display in the control room, so I re-recentered it today.

7027   Wed Jul 25 12:00:21 2012 YaakovUpdateSTACISWeekly update

The past few days I've been working on making a noise budget for the STACIS that incorporates all the different noises that might be contributing.

The noises I am concentrating on are accelerometer noise, geophone noise, electrical noise, and ground noise. Noise from the PZT stacks themselves should be tiny according to various PZT spec sheets, but I haven't actually found a value for it (they all just say it's negligible), so I'll keep it in mind as a potential contributor.

Here's how I am determining accelerometer noise: I take two vertical accelerometers side by side, sitting on a granite block and covered in a foam box, and take the time series of both. Then I take the difference of the time series and calculate the PSD of that, which with the calibration factor of the accelerometers (the V/m/s sensitivity) I am able to find the noise in m/s/rtHz. The noise agreed with the accelerometer specs I found at low frequencies but was higher than expected at high frequencies, so I'm still investigating. If I can't find an obvious problem with my measurements I'll try the three-corner hat method (as per Jenne's recommendation), which would allow me to determine the noises of the independent accelereometers.

I tried a similar method for the geophone noise, but the value I came up with was actually higher than the accelerometer noise, which seems very fishy. I realized that the geophones were still connected to the STACIS circuitry when I took the measurements ,which was probably part of the problem. So this morning I disconnected the STACIS entirely and am looking at just the geophone signals which should give a more accurate noise estimate.

Once I have all the noises characterized, the next step is seeing how those noises affect the closed loop performance of the STACIS. I've been working on a block model that incorporates the different noises and transfer functions involved, and when I have the noises characterized I can test a prediction about how a certain noise affects the platform motion.

7040   Thu Jul 26 16:08:59 2012 YaakovUpdateSTACISNoise plot update

I have a tentative noise plot for the STACIS that includes accelerometer noise, geophone noise, and platform motion with the STACIS off. (Accelerometer noise was measured for the VEL and NONE setting, which are settings on the accelerometer box which make the accelerometer signal correspond to velocity and acceleration, respectively. ) I'm focusing on sensor noise because this is the variable I am looking at changing, and knowing how the sensor noise translates into STACIS platform motion is therefore important.

The accelerometer and geophone noise I determined as described in my last eLog (http://nodus.ligo.caltech.edu:8080/40m/7027) Along the way I found out several things of importance:

1) Horizontal geophones are ONLY horizontal geophones. This is obvious in retrospect, because the springs supporting the magnet inside must be oriented based on vertical/horizontal operation.

2) The geophones in the STACIS are GS-11D (geospace), with a sensitivity of 32 V/m/s (compared to about 3.9 V/m/s for the accelerometers in VEL setting).

3) The accelerometers have different V/m/s sensitivities. I noticed the voltage output of one was consistently higher than the other, leading to very high noise estimates, but then Jenne showed me the actual calibration factors of the individual accelerometers which differed by as much as 0.4 V/m/s (a few percent difference). Taking this into account made the noise plots much more reasonable, but variations in calibration could still create some error.

The accelerometer noise agrees fairly well with the specs on the Wilcoxon page (http://www.wilcoxon.com/prodpdf/731A-P31%2098079a1.pdf). The geophone noise seems surprisingly low; it is even better than the geophone below about 4 Hz.

To see how this noise translates into actual platform motion, I took PSDs of the STACIS while it was off, on with accelerometer feedback, and on with geophone feedback (the "off" PSD is in the above noise plot). Using this data I'm working on estimating a transfer function that shows how the sensor noise translates to motion so I can come up with a sensor noise budget.

This shows that the geophones are actually doing a better job of isolating than the accelerometers, which is not surprising if the noise plot is accurate and the geophones are actually lower noise. It must be noted, though, that the noise plot was for the horizontal geophones whereas the plot above is for the vertical axis which may have a different noise level. Also, the vertical have some extra isolation by being enclosed in a metal stack with rubber padding at its base.

The problem with the STACIS in the past was the differential motion it introduced. I think this might be because the horizontal isolation was not uniform for each chamber. This means that even what would be symmetric motion (no differential length change) would be translated to differential motion because one end is more fixed than the other. Having accelerometers or better-padded geophones (maybe like the vertical geophones) in the STACIS ought to help with this by making the horizontal isolation more consistent and thus reducing differential motion. So the key may not be the geophone noise as much as varying geophone sensitivities or variation in how well they're mounted in the STACIS. I can test this by swapping out the horizontal geophones with other spares, changing the tightness of the mount, and seeing if either of these changes the horizontal isolation significantly, since these are factors that may differ from unit to unit.

I will also compare horizontal closed loop response with geophone vs. accelerometer feedback to see if the geophones are only doing a good job in the above plot because of their extra padding (the vertical stack).

7053   Mon Jul 30 17:24:45 2012 YaakovUpdateSTACISRevised sensor noise plot; dead PZT

The geophone noise in my last eLog was taken before any amplification of the signal, but what really matters is the noise after amplification, since it is this signal that the PZTs are driven by. The noise goes on to be amplified about 1000x before the geophone signal gets to the PZTs.

To obtain a more relevant noise plot, I multiplied the geophone noise by 1,000, the approximate gain of the amplification stage for the geophones (called the "compensator board", the semicircular board that sits toward the top of the STACIS). Below is a plot () that shows the noise for the geophones after amplification and the accelerometer noise (with accelerometers set with a gain of 100x, their highest).

The actual signal from both these sensors has the right magnitude to drive the PZTs (whereas it was much too small in my last plot, where I looked at the sensors before any gain)- this means that for these sensors, both of which are outputting signals that are ready to provide feedback to the STACIS, the accelerometer noise is significantly lower than the geophone noise. This is good news, because it means that there could be a real advantage to using the accelerometers instead of the geophones.

In the process of investigating further advantages of the accelerometers, I believe I killed one of the horizontal PZTs in the spare STACIS (the eBay one). The story: I had that axis in closed loop, and I saw the STACIS shudder, heard a noise, and there was a faint acrid smell. I shut the STACIS off and took out the high voltage card at the base but couldn't find any visible signs of damage (like the current-limiting resistors which burn when a PZT shorts, acc. to old STACIS records). I then tried driving the PZTs with a sine wave, and there was no response in that axis (the other axes looked fine), which leads me to believe I either did unseen damage to the high voltage amplifier (for the y-axis) or killed the PZT itself.

ELOG V3.1.3-