40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log, Page 141 of 344 Not logged in
ID Date Author Type Category Subject
12729   Tue Jan 17 21:31:57 2017 gautamUpdateGeneralETMX suspension electronics problems?

Last night, I plugged the ETMX suspension coils back into the satellite box. Tonight, we turned on the damping loops for ETMX. Rana centered the Oplev so we can use that as an additional diagnostic to see if the optic gets kicked around overnight. We will re-assess the situation tomorrow.

Sometime earlier today, Lydia noticed that the +/- 5V Sorensens at the X end were not displaying their nominal voltage/current values (as per the stickers on them). She corrected this.

12730   Wed Jan 18 10:41:14 2017 gautamUpdateGeneralETMX suspension electronics problems?

Summary pages show no kicking in the ETMX watchdogs from midnight to 6 AM (0800 - 1400 UTC):

https://nodus.ligo.caltech.edu:30889/detcharsummary/day/20170118/sus/watchdogs/

12738   Thu Jan 19 10:21:54 2017 AshleyUpdateGeneralPreliminary Microphone Data

Brief Summary: I am currently looking at the acoustic noise around both arms to see if there are any frequencies from machinery around the lab that stand out and to see what we can remove/change. I am using a Bluebird microphone suspended with surgical tubing from the cable trays to isolate it from vibrations. I am also using a preamp and the SR875 spectrum analyzer taking 6 sets of data every 1.5 meters (0 to 200Hz, 200Hz to 400Hz, 400z to 800Hz, 800Hz to 3200Hz, 3.2kHz to 12kHz, 12kHz to 100kHz).

·                Attachment 1 is a PSD of the first 3 measurements (from 0 to 12kHz) that I took every 1.5 meters along the x arm with the preamp and spectrum analyzer

·                Attachment 2 is a blrms color map of the first 6 sets of data I took (from 2.4m to 9.9m)

·                Attachmetn 3 is a picture of the microphone set up with the surgical tubing

Problems that occurred: settings on the preamp made the first set of data I took significantly smaller than the data I took with the 0dB button off and the last problem I had was the spectrum analyzer reading only from -50 to -50 dBVpk

12788   Thu Feb 2 12:17:48 2017 SteveUpdateGeneral USB microscope

This AmScope microscope would have 3.5x-180x magnification, calibratable measurement function, 5MP picture and good working distance to work on printed circuit boards.

12808   Tue Feb 7 16:23:49 2017 SteveUpdateGeneralpower interruption tomorrow

received this note: at 4:11pm Tuesday, Feb 7, 2017

Building:         Campus

Date:             Wednesday, February 8, 2017

Time:             7:30 AM – 8:30 AM

Contact:          Rick Rodriguez x-2576

Pasadena Water and Power (PWP) will be performing a switching operation of the

Caltech Electrical Distribution System that is expected to be transparent to Caltech,

but could result in a minor power anomaly that might affect very sensitive equipment.

IMPACT: Negligible impact......?

There may be temporary  power interruption tomorrow!

PS:we did not see any effect

12813   Thu Feb 9 08:03:08 2017 SteveUpdateGeneral USB microscope ordered

http://www.amscope.com/3-5x-180x-boom-stand-trinocular-zoom-stereo-microscope-with-144-led-ring-light-and-10mp-camera.html will be ordered today.

The actual unit we are getting has lockable zoom for better repeatability after calibration: SM-3NTPZZ-144

Quote: CWQ6-020817

12814   Thu Feb 9 11:22:56 2017 gautamUpdateGeneralSorensens and DIN connections at 1X1

I'd like to fix a few things at 1X1 when we plug in the new amplifier for the 29.5MHz modulation signal.

1. Split off separate +24 and ground wires to the green BBPD RF amplifiers and the AOM driver (they are sharing a single fuse at the moment)
2. Tap a new +24 GND -24V set for the FSS Fast summing box - this is currently running with a bench power supply underneath the PSL table set to +/-18V, but I checked the 7815/7915 datasheets and they accept up to 35V input for a 15V output, so it should be fine to use 24V
3. Hook up the ZHL-2A for the IMC modulation.

Steve has ordered rolls of pre-twisted wire to run from 1X1 to the PSL table, so that part can be handled later.

But at 1X1, we need to tap new paths from +/- 24V to the DIN connectors. I think it's probably fine to turn off the two Sorensens, do the wiring, and then turn them back on, but is there any procedure for how this should be done?

12826   Mon Feb 13 17:39:45 2017 AshleyUpdateGeneralPreliminary Microphone Data Update
• Problems that have occurred since my last post: All of the sudden, I was getting very strange data that was very quiet and did not match the previous input range of my last locations (see attachment). After resoldering the custom bnc connection cables with Lydia, which were in disrepair, and checking almost everything we could think of, we found that the gain dial on the preamp was turned all the down. Immediately after it was fixed, the data returned to expected values (based on neighboring locations and data taken at the last location before the problem occurred).
• Updates: Since my last post, I have created a normalized blrms color map in addition to the one I already have. Additionally, I have started working on plotting the color maps next to a labeled, to-scale drawing of the lab, but have yet to complete it.
• Attachment 1: comparison of the psds
• Attachment 2: blrms color map
• Attachment 3: normalized color map
 Quote: Brief Summary: I am currently looking at the acoustic noise around both arms to see if there are any frequencies from machinery around the lab that stand out and to see what we can remove/change. I am using a Bluebird microphone suspended with surgical tubing from the cable trays to isolate it from vibrations. I am also using a preamp and the SR875 spectrum analyzer taking 6 sets of data every 1.5 meters (0 to 200Hz, 200Hz to 400Hz, 400z to 800Hz, 800Hz to 3200Hz, 3.2kHz to 12kHz, 12kHz to 100kHz).   ·                Attachment 1 is a PSD of the first 3 measurements (from 0 to 12kHz) that I took every 1.5 meters along the x arm with the preamp and spectrum analyzer ·                Attachment 2 is a blrms color map of the first 6 sets of data I took (from 2.4m to 9.9m)  ·                Attachmetn 3 is a picture of the microphone set up with the surgical tubing  Problems that occurred: settings on the preamp made the first set of data I took significantly smaller than the data I took with the 0dB button off and the last problem I had was the spectrum analyzer reading only from -50 to -50 dBVpk

12834   Thu Feb 16 13:29:38 2017 gautamSummaryGeneralAlternative Calibration Scheme

Summary:

Craig and I have been trying to put together a Simulink diagram of the proposed alternative calibration scheme. Each time I talk the idea over with someone, I convince myself it makes sense, but then I try and explain it to someone else and get more confused. Probably I am not even thinking about this in the right way. So I am putting what I have here for comments/suggestions.

What's the general idea?

Suppose the PSL is locked to the MC cavity, and the AUX laser is locked to the arm cavity (with sufficiently high BW). Then by driving a line in the arm cavity length, and beating the PSL and AUX lasers, we can determine how much we are modulating the arm cavity length in metres by reading out the beat frequency between the two lasers, provided the arm cavity length is precisely known.

So we need:

1. Both lasers to be stabilized to be able to sense the line we are driving
2. A high bandwidth PDH loop for locking the AUX laser to the arm cavity such that the AUX laser frequency is able to track the line we are driving
3. An accurate and precise way to read out the beat frequency (the proposal here is to use an FPGA based readout)
4. An accurate measurement of the arm length (I think we know the arm lengths to <0.1% so this shouldn't dominate any systematic error).

To be able to sense a 1kHz line being driven at 1e-16 m amplitude, I estimate we need a beat note stability of ~1mHz/rtHz at 1kHz.

Requirements and what we have currently:

• The PSL is locked to the mode-cleaner, and the arm cavity is locked to the PSL. The former PDH loop is high BW, and so we expect the stabilized PSL to have frequency noise of ~1mHz/rtHz at about 1kHz (to be measured and confirmed)
• The AUX laser is locked to the arm cavity with a medium-BW (~10kHz UGF) PDH servo. From past out-of-loop ALS beat measurements, I estimate the expected frequency noise of the AUX laser at 1kHz to be ~1Hz/rtHz with the current PDH setup
• Rana suggested we "borrow" the stability of the PSL by locking the AUX laser and PSL in a high bandwidth PLL - if we want this loop to have ~300kHz BW, then we need to use an EOM as an actuator. The attached Simulink diagram (schematic representation only, though I think I have measurements of many of those transfer functions/gains anyways) shows the topology I had in mind. Perhaps I did not understand this correctly, but if we have such a loop with high gain at 1kHz, and the error signal being the beat between PSL and AUX, won't it squish the modulation we are applying @1kHz?
• Is it feasible to instead add a parallel path to the end PDH loop with an EOM as an actuator (similar to what we do for the IMC locking)? Ideally, what we want is an end PDH loop which squishes the free-running NPRO noise to ~1mHz/rtHz at 1kHz instead of the 1Hz/rtHz we have currently. This loop would then also have negligible tracking error at 1kHz. Then, we could have a low bandwidth PLL offloading onto the temperature of the crystal to keep the beat between the two lasers hovering around the PSL frequency.

Hardware:

On the hardware side of things, we need:

• FSS box to drive the EOM (Rana mentioned there is a spare available in the Cryo lab)

Koji and I briefly looked through the fiber inventory we have yesterday. We have some couplers (one mounted) and short (5m) patch fibers. But I think the fiber infrastructure we have in place currently is adequate - we have the AUX light brought to the PSL table, and there is a spare fiber running the other way if we want to bring the PSL IR to the end as well.

I need to also think about where we can stick the EOM in given physical constraints on the EX table and the beam diameter/aperture of EOM...

12835   Thu Feb 16 21:55:47 2017 ranaSummaryGeneralAlternative Calibration Scheme

Question for Craig: What does the SNR of our lines have to be? IF we're only trying to calibrate the actuator in the audio band over long time scales, it seems we could get by with more frequency noise. Assuming we want a 1% calibration at 50-500 Hz, what is the requirement on the frequency noise PSD curve?

12837   Fri Feb 17 20:04:43 2017 KojiUpdateGeneralProjector not functional / Zita partially working

Koji, Gautam, Johannes

We quickly checked the situation of the projector in the control room.

- We found that the proejctor was indicating "lamp error".
==> Steve, could you remove the projector from the ceiling and check if it still does not work?
If it still does not work, send it back to the vender. It should be covered by the previous service.

- Zita seemed happy with the DVI output. We tried the dual display configration and  VGA and DVI are active right now.
The DVI output (from RADEON something video card) is somewhat strange. We probably need to look into the video display situation.

12842   Tue Feb 21 13:51:35 2017 CraigSummaryGeneralAlternative Calibration Scheme

We get SNR in two ways: the amplitude of applied force and the integration time.  So we are limited in two ways: stability of the lock to applied forces and time of locklosses / calibration fluctuations.

At the sites, you probably know that we blow our spectrum out of the water with the calibration lines, with SNRs of about 100 on the scale of about 10 seconds.  For us this might be impossible, since we aren't as quiet.

If we want 1% calibration on our sweeps, we'll need  0.01 = Uncertainty = sqrt( (1 - COH^2)/(2 * Navg * COH^2) ), where COH is the coherence of the transfer function measurement and Navg is the number of measurements at a specific frequency.  This equation comes from Bendat and Piersol, and is subject to a bunch of assumptions which may not be true for us (particularly, that the plant is stationary in time).

If we let Navg = 10, then COH ~ 0.999.

Coherence = Gxy^2/(Gxx * Gyy), where x(t) and y(t) are the input signal and output signal of the transfer function measurement, Gxx and Gyy are the spectral densities of x and y, and Gxy is the cross-spectral density.

Usually SNR = P_signal / P_noise, but for us SNR = A_signal / A_noise.

Eric Q and Evan H helped me find the relationship between Coherence and SNR:

P = Pn + Pc, Pn = P * (1 - Coh), Pc = P * Coh

==> SNR = sqrt( Pc / Pn ) = sqrt( Coh / 1 - Coh )

From Coh ~ 0.999, SNR ~ 30.

 Quote: Question for Craig: What does the SNR of our lines have to be? IF we're only trying to calibrate the actuator in the audio band over long time scales, it seems we could get by with more frequency noise. Assuming we want a 1% calibration at 50-500 Hz, what is the requirement on the frequency noise PSD curve?

12843   Tue Feb 21 17:05:14 2017 SteveUpdateGeneralProjector lamp replaced

This bulb was blown out on Feb 4, 2017 after 2 months of operation.

12845   Wed Feb 22 10:16:54 2017 ranaSummaryGeneralAlternative Calibration Scheme

OK, but the questions still stands: "Assuming we want a 1% calibration at 50-500 Hz, what is the requirement on the frequency noise PSD curve?"

 Quote: We get SNR in two ways: the amplitude of applied force and the integration time.  So we are limited in two ways: stability of the lock to applied forces and time of locklosses / calibration fluctuations.
12848   Thu Feb 23 14:50:26 2017 SteveUpdateGeneral USB microscope returned

The microscope shipped back to the vendor for credit yesterday.

 Quote: http://www.amscope.com/3-5x-180x-boom-stand-trinocular-zoom-stereo-microscope-with-144-led-ring-light-and-10mp-camera.html will be ordered today. The actual unit we are getting has lockable zoom for better repeatability after calibration: SM-3NTPZZ-144 Quote: CWQ6-020817

12871   Mon Mar 6 16:32:36 2017 SteveUpdateGeneralold NPRO

16 years old Lightwave NPRO M126-1064-700, sn 415 power output is tripping continously to zero.

The Lightwave Controller 125/126-OPN-POS sn516 was used in this test. Settings were lowered to close to nominal values without any success.

One can not determine what is broken: head or controller. This NPRO head was under Manasa's desk.

12872   Tue Mar 7 15:17:19 2017 SteveBureaucracyGeneralproperty tag

Property tag found.

12875   Thu Mar 9 15:25:12 2017 KojiUpdateGeneralIMC/XYarms aligned/locked

As per Steve's request, I've checked the alignment of the IMC and the arms. These three cavities are locked and aligned.

12876   Thu Mar 9 17:26:43 2017 SteveUpdateGeneralattempted ETMY picture taking

I removed the video monitoring can and replaced it with Olympus SP-570UZ camera. It has no IR blocker. The OSEM light are dominant because I can not zoom in more.

I left the camera in place so you can try it. Leave the LEXAN plate on the glass window so no accident can happen. The illuminator is on and you can turn it off-on with the manual switch, close to the camera. Camera manual is on my desk.

12877   Thu Mar 9 20:11:04 2017 KojiUpdateGeneralattempted ETMY picture taking

The attached is the ETMY image with the single arm locked. This was the best I could do. Here is the recipe

• Turn on SP570UZ
• Switch to "M" mode (Manual aperture and exposure)
• Set the aperture to be the widest (smallest F number) and the exposure to be maximum (15 second).
• Switch to AF mode by the lens side switch
• Use the lens dial to adjust the zoom until the OSEMs fill the central 1/3 box (i.e. 1/9 area of the field of view). If you zoom more, you can't focus the spot later.
• Use menu button to switch to ISO1600 (You are now capable to see the beam spot)
• Switch to MF mode by the lens side switch
• Use the lens dial to adjust the focus to have the sharpest image of the spot. This can be achieved at the focal distance of ~1m
• Use menu button to switch back to ISO64
• Push the shutter (I didn't use it, but you should be able to use 2sec timer)
12881   Fri Mar 10 18:00:22 2017 SteveUpdateGeneralattempted ETMY picture taking

12929   Wed Apr 5 16:05:47 2017 gautamUpdateGeneralNB code checkout

[evan, gautam]

We spent some time trying to get the noise-budgeting code running today. I guess eventually we want this to be usable on the workstations so we cloned the git repo into /ligo/svncommon. The main objective was to see if we had all the dependencies for getting this code running already installed. The way Evan has set the code up is with a bunch of dictionaries for each of the noise curves we are interested in - so we just commented out everything that required real IFO data. We also commented out all the gwpy stuff, since (if I remember right) we want to be using nds2 to get the data.

Running the code with just the gwinc curves produces the plots it is supposed to, so it looks like we have all the dependencies required. It now remains to integrate actual IFO data, I will try and set up the infrastructure for this using the archived frame data from the 2016 DRFPMI locks..

12948   Wed Apr 19 15:46:24 2017 gautamUpdateGeneral1611/1811 inventory check

I looked through the lab area to do a fast photodiode inventory check, as we may need to buy some for the higher order mode spectroscopy SURF project. I looked on the following optical tables: ETMY, ITMY, BS, AS, PSL, SP, ITMX, Jenne laser table, and ETMX, as well as the photodiode cabinet, and could only find two 1611s. Here is a summary of the inventory:

• Power supply 0901: 2x in photodiode cabinet (E6 along the Y arm), 1x on Jenne laser table
• Newfocus 1611 S/N 7284-WX, labelled "REF DET" on ITMY optical table, currently unused
• Newfocus 1611 S/N 57109 on Jenne laser table

I have not yet checked if these photodiodes are in working order.

12950   Tue Apr 25 19:35:41 2017 gautamUpdateGeneralIPCS -q

Dataviewer wouldn't launch on pianosa - it seemed to work fine on Donatella though. Rana suggested using the ipcs -q command. The complete fix can be found in this elog. This did the trick, dataviewer runs fine on Pianosa now...

12951   Wed Apr 26 01:00:23 2017 gautamUpdateGeneralDRMI locking

Since we'd like to get back to DRSE locking, I tried locking the DRMI tonight. I did the following:

• First, I aligned the arms, and ran the dither alignment scripts to maximize the arm transmission
• Next, I misaligned the ETMs, and tried to lock the PRC resonant for the carrier (i.e. PRCL on REFL11I, MICH on AS55Q). I got brief lock stretches of a few seconds but not longer. Turns out the AS55 beam was barely hitting the photodiode. I guess this wasn't looked at since Johannes modified the AS path for the loss measurements. Anyways, it just required a minor tweak to center the beam on the AS55 photodiode.
• Once the PRC was locked, I ran the PRC and MICH dither align scripts. The way these are set up right now, the error signals to these servos are REFLDC and ASDC respectively (demodulated at the respective dither frequencies). But looking at the spots on the ITM cameras with the PRC resonant, the spots seem shifted (in both PIT and YAW) relative to the spots when the arm cavity is resonant. Shouldn't they be the same mode? Or maybe I am missing something.

• Next, I tried to lock the DRMI with the 1f error signals: i.e. PRCL on REFL11 I, SRCL on REFL55 I, and MICH on AS55 Q. After some demod phase tweaking, I was able to get some locks going. Turning on the PRC angular feedforward seemed to help the locking, but I have no idea if the installed filters are still the correct ones. I believe the POP QPD channels are the witnesses used to train this filter, I will look at the predicted vs achieved subtraction.
• At this point, I was able to get locks lasting a few minutes - see the attachment. I ran the UGF servos and tweaked the loop gains a little, but before I could start a loop measurement, I lost the lock. I am calling it for the night.

GV 26 April 2017, 3pm: Forgot to note yesterday that I re-connected the suspect Satellite box, which has been connected to the SRM signal chain, back to the SRM suspension. I did not see any instances of glitching during my work last night. Also added pictures showing shifted spots on ITMs when PRC is locked relative to when arms are locked...

12954   Fri Apr 28 02:04:36 2017 gautamUpdateGeneralDRMI locking

I got a couple of ~30min long DRMI lock stretches today. The settings I used are essentially the same as what I had back in November. Though we have since made some changes to the IMC RF signal chain, I guess it is not unreasonable that the LSC Demod phases that worked then work now as well.

In the lock stretches, I did the following:

• Took loop measurements for MICH, PRCL, SRCL
• Turned on the sensing oscillator lines for error signal calibration
• Tried turning on the analog whitening on AS55, REFL11 and REFL55. The latter two worked fine, but everytime I turned the REFL55 whitening on, I broke the lock. I'm also unable to acquire lock if I leave the whitening turned on all the time. The ADC overflow indicators also indicate frequent overflows when I turn the whitening on. Oddly, this seems to happen even if I turn the analog whitening gain to 0dB - the signals look well within the ADC range on dataviewer and DTT timeseries mode. Not sure what's going on here, I will investigate further tomorrow.
• We should have some stretches where we can look at the possibility of seismic feedforward for some DRMI length DOFs.

On the side, I'm also looking at whether the PRC angular feedforward filters, last trained in October 2016, remain valid. Even post midnight, I am unable to lock the DRMI without turning on the FF, and looking at the POP QPD PIT and YAW signal spectra with the FF on vs FF off, there is definitely some improvement in the 1-4Hz band (plot to follow), question is whether we can do better and hence improve the DRMI duty cycle/ make the lock acquisition easier. To this end, I centered the beam on the POP QPD after locking and dither aligning the PRC on carrier, and have taken some data to look at.

So, much data analysis to follow - the idea is to put together a DRMI noise budget with Evan's NB code. For now, here are the uncalibrated control signal spectra.

12955   Fri Apr 28 13:56:26 2017 ranaUpdateGeneralDRMI locking

one of these signals does not look like the others: explanation?

12957   Fri Apr 28 19:32:06 2017 gautamUpdateGeneralDRMI locking - PRCL angular FF

I took a closer look at the POP QPD/ PRC angular feedforward situation yesterday. I thought it would be useful to have a POP QPD MEDM screen. Looking at the PIT and YAW channel filter modules, the anti-whitening filters seemed different from what we have for other channels that are connected to the Pentek interface board (e.g. MCL). So I copied over the 150:15 (z:p) filter, and also turned on a 60Hz comb. The LSC offsets script does not set the dark offsets for this QPD, so I manually put in the dark offsets for the PIT, YAW and SUM channels as well. For the locking, I first locked the arms on IR an dither aligned them. Then I locked the PRMI on carrier, ran the PRC dither alignment, and went over to the ITMX pickoff table and centered the beam on the QPD by making the PIT and YAW channel timeseries oscillate around approximately zero.

After these tweaks, I collected ~40mins of data with the angular FF OFF/ON. I did not DC couple the ITM Oplev servos, but Eric tells me that this did not make a difference to the achievable subtraction in the past. Here is the frequency domain multicoherence analysis - I used the BS_X and BS_Y seismometer channels as witnesses. I've also put a plot with what the raw FF filter coefficients look like (no fitting yet).

Looks like we can do better for both DOFs - it even seems like we are injecting noise with the current FF filters in some bands, perhaps we can do a better job of rolling off the filters outside the band of interest. Eric and I were discussing MATLAB's "reduce" routine for this purpose, I will play around with it and see if I get a better fit.

Unfortunately, I encountered a strange error when trying to pull data with nds2 today, it kept complaining RuntimeError: Too many channels or too much data requestedeven though I have pulled longer stretches of data for more channels with 16k sampling rate as recently as last week. Shorter duration requests (<600 seconds) seemed to work fine though... So I had to use cds.getdata to pull the data, and they're much too large to attach. Has anyone else encountered a similar error?

The mystery of the spots on the ITMs when the PRC is locked on carrier remains - after talking this over with Koji, we figured that even with the carrier resonant, the spot will be much dimmer than the spots when the arms are locked, but what I see on the cameras is still a pretty beefy spot. The real cavity mode is actually visible where it should be (I marked the locations of the spots with arms well-aligned with a marker on the monitors), as given away by some twinkling that is visible only when the cavity is locked. But what ghost beam is so intense it looks almost as bright as when the arm is locked?

GV 10pm 28 April 2017: Turns out this is the spot from the single bounce off the ETM transmitting back through the ITM and hitting the suspension cage (hence the bright spot). Johannes and I confirmed by moving the ETM, the spot moved with it. I just never paid attention to this spot before.

12960   Mon May 1 16:29:51 2017 gautamUpdateGeneralDRMI locking

For the traces I posted, I had not turned on the whitening for the SRCL sensing PD (REFL55). However, I took a spectrum on a subsequent lock, with the analog whitening + digital dewhitening turned on for all 3 PDs (AS55, REFL11 and REFL55), and the HF part of the SRCL spectrum still looked anomalous. I'm putting together the detailed NB, but here's a comparison between the signals from the 3 RFPDs with the PSL shutter closed (but whitening engaged, and with the analog gains at the same values as used during the locking).

To convert the y-axis into m/rtHz, I used data from a sensing matrix measurement I took yesterday night during a DRMI lock - I turned on lines between 300 Hz and 325 Hz for the 3DOFs for ~5 minutes, downloaded the RFPD error signal data and did the demodulation. I used numbers from this elog to convert the actuator drive from cts to m. The final numbers I used were:

MICH (AS55_Q):   8.706 * 10^11 cts/m

PRCL (REFL11_I): 2.757 * 10^12 cts/m

SRCL (REFL55_I): 1.995 * 10^10 cts/m

So it looks like there may be something weird going on with the REFL55 signal chain. Looking at the LSC rack (and also suggested by an elog search), it looks like the demodulation is done by a demod board labelled "POP55" - moreover, the demodulated outputs are taken not from the regular output ports on this board, but from the "MON" ports on the front panel.

 Quote: one of these signals does not look like the others: explanation?

12962   Mon May 1 21:45:54 2017 ericqUpdateGeneralDRMI locking
Comparing counts doesn't get you anywhere; each PD has different whitening gain which may vary from measurement to measurement. The better thing to compare is Volts coming out of the demod board, since this (hopefully) only changes when we touch the PD or analog signal chain; this is what I used for the most recent DRMI sensing measurements. (ELOG 11589) We have calibrated actuator channels in the CAL model, which will give you the control signal in m for the DRMI lengths. Perhaps you can convert your sensing matrix measurement to demod board output volts per meter to compare with the last measurement.

Also, the monitor ports are the LEMO ports to the left; the SMA ports where the signal is coming from are from a daughter board that has a better output opamp that the nominal output; we're using the same output on the REFL11 and AS55 demod boards.
12963   Wed May 3 16:00:00 2017 gautamSummaryGeneralNetwork Topology Check

[johannes, gautam]

I forgot we had done this last year already, but we updated the control room network switch labels and double checked all the connections. Here is the status of the connections and labels as of today:

There are a few minor changes w.r.t. labeling and port numbers compared to the Dec 2015 entry. But it looks like there was no IP clash between Rossa and anything (which was one of the motivations behind embarking on this cleanup). We confirmed by detatching the cable at the PC end of Rossa, and noticed the break in the ping signals. Plugging the cable back in returned the pings. Because Rossa is currently un-bootable, I couldn't check the MAC address.

We also confirmed all of this by using the web browser interface for the switch (IP = 192.168.113.249).

12964   Wed May 3 16:02:36 2017 SteveUpdateGeneralPI pzt inventory check

One is broken, two are ready to steer green and 3 available in un known condition

12967   Wed May 3 16:47:45 2017 KojiUpdateGeneralPI pzt inventory check

I also have a functional one on my desk, which has one of the wires repaired.

 Quote: One is broken, two are ready to steer green and 3 available in un known condition

12969   Wed May 3 18:45:45 2017 ranaUpdateGeneralDRMI locking

 Quote: Comparing counts doesn't get you anywhere; each PD has different whitening gain which may vary from measurement to measurement. The better thing to compare is Volts coming out of the demod board, since this (hopefully) only changes when we touch the PD or analog signal chain; this is what I used for the most recent DRMI sensing measurements. (ELOG 11589) We have calibrated actuator channels in the CAL model, which will give you the control signal in m for the DRMI lengths. Perhaps you can convert your sensing matrix measurement to demod board output volts per meter to compare with the last measurement. Also, the monitor ports are the LEMO ports to the left; the SMA ports where the signal is coming from are from a daughter board that has a better output opamp that the nominal output; we're using the same output on the REFL11 and AS55 demod boards.

Wrong! RTFS.

SMA outputs are the bare, passive outputs of the mixer/lowpass.
TNC outputs are the low-noise, acti amplified outputs via the daughter board.
LEMO outputs are the high noise, G=2, LT1125 buffered outputs
12972   Thu May 4 19:03:15 2017 gautamUpdateGeneralDRMI locking - preliminary MICH NB

Summary:

I've been playing around with Evan's NB code trying to put together a noise budget for the data collected during the DRMI locks last week. Here is what I have so far.

Attachment #1: Sensing matrix measurement.

• This is basically to show that the MICH error signal is mostly in AS55Q.
• The whitening gain used was 0dB, and the demod phase was -82 degrees.
• The MICH sensing response was 5.31*10^8 V/m, where V is the demod board output. The 40m wiki RFPD page for AS55 says the RF transimpedance is ~550ohms, and I measured the Demod Board puts out 5.1V of IF signal (measured at after the Preamp, which is what goes to the ADC) for 1V of RF signal at the PD input. Using these numbers, and assuming a PD responsivity of 0.8 A/W at 1064nm, the sensing response is 2.37*10^5 W/m. I don't have a feeling yet for whether this is a reasonable number, but it would be a number to compare to what my Finesse model tells me to expect, for example.
• Actuator calibration used to arrive at these numbers was taken from this elog

Attachment #2: MICH OLTF measurement vs model

• In order to build the MICH OLTF model, I used MATLAB to put together the following transfer functions:
• BS pendulum
• Digital servo filters from LSC_MICH
• Violin mode filters
• Analog/Digital AA and AI filters. For the digital AA/AI filters, I took the coefficients from /opt/rtcds/rtscore/release/src/fe/controller.c
• The loop measurement was taken with digital filter modules FM1, FM2, FM3, FM7, FM9 engaged.
• In order to fit the model to the measurement, I tried finding the best-fit values for an overall loop gain and delay.
• The agreement between model and measurement isn't stellar, but I decided to push ahead for a first attempt. This loop TF was used to convert various noises into displacement noise for plotting.

Attachment #3: Noise budget

• It took me a while to get Evan's code going, the main changes I made were to use nds2 to grab data instead of GWPy, and also to replace reading in .txt files with importing .mat files. This is a work in progress.
• Noises plotted:
• Measured - I took the in loop error signal and estimated the free-running displacement noise with the model OLTF, and calibrated it into metres using the sensing response measurement. This looks consistent with what was measured back in Dec 2015.
• Shot noise - I used the measured DC power incident on the PD, 13mW, RF transimpedance of 550 V/A, and the V/m calibration factor mentioned above, to calculate this (labelled "Quantum Noise").
• Dark noise - measured with PSL shutter closed.
• Seismic noise, thermal noise, gas noise - calculated with GWINC

I think I did the various conversions/calibrations/loop algebra correctly, but I may have overlooked something. Now that the framework for doing this is somewhat set up, I will try and put together analogous NBs for PRCL and SRCL.

GV 22 August 2017: Attachment #4 is the summary of my demod board efficiency investigations, useful for converting sensing measurement numbers from cts/m to W/m.

12974   Fri May 5 10:13:02 2017 ericqUpdateGeneralMICH NB questions
Is suspension thermal noise missing? I take it "Thermal" refers just to thermal things going on in the optic, since I don't see any peaks at the bounce/roll modes as I would expect from suspension thermal noise.

What goes into the GWINC calculation of seismic noise? Does it include real 40m ground motion data and our seismic stacks?

I'm surprised to see such a sharp corner in the "Dark Noise" trace, did you apply the OLG correction to a measured dark noise ASD? (The OLG correction only needs to be applied to the in-lock error signals to recover open loop behavior, there is no closed loop when you're measuring the dark noise so nothing to correct for.)
12975   Fri May 5 12:10:53 2017 gautamUpdateGeneralMICH NB questions

 Quote: Is suspension thermal noise missing? I take it "Thermal" refers just to thermal things going on in the optic, since I don't see any peaks at the bounce/roll modes as I would expect from suspension thermal noise. What goes into the GWINC calculation of seismic noise? Does it include real 40m ground motion data and our seismic stacks? I'm surprised to see such a sharp corner in the "Dark Noise" trace, did you apply the OLG correction to a measured dark noise ASD? (The OLG correction only needs to be applied to the in-lock error signals to recover open loop behavior, there is no closed loop when you're measuring the dark noise so nothing to correct for.)

I've included the suspension thermal noise in the "Thermal" trace, but I guess the GWINC file I've been using to generate this trace only computes the thermal noise for the displacement DoF. I think this paper has the formulas to account for them, I will look into including these.

For the seismic noise, I've just been using the seis40.mat file from the 40m SVN. I think it includes a model of our stacks, but I did not re-calculate anything with current seismometer spectra. In the NB I updated yesterday, however, I think I was off by a factor of sqrt(3) as I had only included the seismic noise from 1 suspended optic. I've corrected this in the attached plot.

For the dark noise, you are right, I had it grouped in the wrong dictionary in the code so it was applying the OLG inversion. I've fixed this in the attached plot.
12976   Sat May 6 21:52:11 2017 ranaUpdateGeneralMICH NB questions

I think the most important next two items to budget are the optical lever noise, and the coil driver noise. The coil driver noise is dominated at the moment by the DAC noise since we're operating with the dewhitening filters turned off.

12979   Wed May 10 01:56:06 2017 gautamUpdateGeneralMICH NB - OL coupling

### Last night, I tried to estimate the contribution of OL feedback signal to the MICH length error signal.

In order to do so, I took a swept sine measurement with a few points between 50 Hz and 500 Hz. The transfer function between C1:LSC-MICH_OUT_DQ and the Oplev Servo Output point (e.g. C1:SUS-BS_OL_PIT_OUT etc) was measured. I played around with the excitation amplitude till I got coherence > 0.9 for the TF measurement, while making sure I wasn't driving the Oplev error point too hard that side-lobes began to show up in the MICH control signal spectrum.

#### The Oplev control signal is not DQ-ed. So I locked the DRMI again and downloaded the 16k data "live" for ~5min stretch using cdsutils.getdata on the workstation. The Oplev error point is DQ-ed at 2k, but I found that the excitation amplitude needed for good SNR at the error point drove the servo to the limiter value of 2000cts - so I decided to use the control signal instead. Knowing the transfer function from the Oplev *_OUT* channel to C1:LSC-MICH_IN1_DQ, I backed out the coupling - the transfer function was only measured between 50 Hz and 500 Hz, and no extrapolation is done, so the estimation is only really valid in this range, which looks like where it is important anyways (see Attachment #2, contributions from ITMX, ITMY and BS PIT and YAW servos added in quadrature).

I was also looking at the Oplev servo shapes and noticed that they are different for the ITMs and the BS (Attachment #1). Specifically, for the ITM Oplevs, an "ELP15" is used to do the roll-off while an "ELP35" is employed in the BS servo (though an ELP35 also exists in the ITM Oplev filter banks). I got lost in an elog search for when these were tuned, but I guess the principles outlined in this elog still hold and can serve as a guideline for Oplev loop tweaking.

Coil driver noise estimation to follow

 Quote: I think the most important next two items to budget are the optical lever noise, and the coil driver noise. The coil driver noise is dominated at the moment by the DAC noise since we're operating with the dewhitening filters turned off.

GV 10 May 12:30pm: I've uploaded another copy of the NB (Attachment #3) with the contributions from the ITMs and BS separated. Looks like below 100Hz, the BS coupling dominates, while the hump/plateau around 350Hz is coming from ITMX.

12981   Wed May 10 16:53:38 2017 ranaUpdateGeneralMICH NB - OL coupling

That's a good find.

1. The OL control signal can be gotten from the DQ error signal. You just need to multiply it by the digital filters and the gain. The state of the filters and the gain can be gotten using matlab tools like getFotonFilt.m. For python ChrisW wrote a tool called foton.py which is in the GDS SVN. You should ask him for it. It requires access to some ROOT libraries to run.
2. We should have sub budgets for everything like OL and thermal, etc. They should be automatically produced each time you run the main budget and should be separate pages in the same PDF file. Jamie / Chris may have something going along these lines so check to see if they are already on it.
12983   Wed May 10 17:17:05 2017 gautamUpdateGeneralDAC / Coil Driver noise

# Suspension Actuator noise:

### There are 3 main sources of electronics noise which come in through the coil driver:

1. Voltage noise of the coil driver.
1. The input referred noise is ~5 nV/rHz, so not a big issue.
2. The Johnson noise of the output resistor which is in series with the coil is sqrt(4*k*T*R) ~ 3 nV/rHz. We probably want to increase this resistor from 200 to 1000 Ohms once Gautam convinces us that we don't need that range for lock acquisition.
2. Voltage noise of the dewhitening board.
1. In order to reduce DAC noise, we have a "dewhitening" filter which provides some low passing. There is an "antiDW" filter in the digital part which is the inverse of this, so that when they are both turned on, the result is that the main signal path has a flat transfer function, but the DAC noise gets attenuated.
2. In particular, ours have 2 second order filters (each with 2 poles at 15 Hz and 2 zeros at 100 Hz).
3. We also have a passive pole:zero network at the output which has z=130, 530 Hz and p = 14, 3185 Hz.
4. The dewhitening board has an overall gain of 3 at DC to account for our old DACs having a range of +/-5 V and our coil drivers having +/- 15 V power supplies. We should get rid of this gain of 3.
5. The dewhitening board (and probably the coil driver) use thick film resistors and so their noise is much worse than expected at low frequencies.
3. DAC voltage noise.
1. The General Standards 16-bit DACs have a noise of ~5 uV/rHz.
4. the satellite box is passive and not a significant source of noise; its just a flaky construction and so its problematic.
12984   Wed May 10 17:46:44 2017 gautamUpdateGeneralDAC / Coil Driver noise - SRM coil driver + dewhite board removed

## I've removed the SOS coil driver (D010001-B, S/N B151, labelled "SRM") + Universal Dewhitening Board (D000183 Rev C, S/N B5172, labelled "B5") combo for SRM from 1X4, for photo taking + inspection.

I first shutdown the SRM watchdog, noted cabling between these boards and also the AI board as well as output to Sat. Box. I also needed to shutdown the MC2 watchdog as I had to remove the DAC output to MC2 in order to remove the SRM Dewhitening board from the rack. This connection has been restored, MC locks fine now.

12985   Thu May 11 09:45:46 2017 ranaUpdateGeneralDAC / Coil Driver noise - SRM coil driver + dewhite board removed

I believe the ETMs and ITMs are different from the others.

12986   Thu May 11 18:59:22 2017 gautamUpdateGeneralSRM coil driver + dewhite board initial survey

I've added marked-up schematics + high-res photographs of the SRM coil driver board and dewhitening board to the 40m DCC Document tree (D1700217 and D1700218).

In the attached marked-up schematics, I've also added the proposed changes which Rana and I discussed earlier today. For the thick-film -> thin-film resistor switching, I will try and make a quick LISO model to see if we can get away with replacing just a few rather than re-stuff the whole board.

### Since I have the board out, should I implement some of these changes (like AD797 removal) before sticking it back in and pulling out one of the ITM boards? I need to look at the locking transients and current digital limit-values for the various DoFs before deciding on what is an appropriate value for the output resistance in series with the coil.

Another change I think should be made, but I forgot to include on the markups: On the dewhitening board, we should probably replace the decoupling capacitors C41 and C52 with equivalent value electrolytic caps (they are currently tantalum caps which I think are susceptible to fail by shorting input to output).

12987   Fri May 12 01:36:04 2017 gautamUpdateGeneralSRM coil driver + dewhite board LISO modeling

I've made the LISO models for the dewhitening board and coil driver boards I pulled out.

Attached is a plot of the current noise in the current configuration (i.e. dewhitening board just has a gain x3 stage, and then propagated through the coil driver path), with the top 3 noise contributions: The op-amps (op3 and op5) are the LT1125s on the coil driver board in the bias path, while "R12" is the Johnson noise from the 1k input resistace to the OP27 in the signal path.

Assuming the OSEMs have an actuation gain of 0.016 N/A (so 0.064 N/A for 4 OSEMs), the current noise of ~1e-10 A/rtHz translates to a displacement noise of ~3e-15m/rtHz at ~100Hz (assuming a mirror mass of 0.25kg).

I have NOT included the noise from the LM6321 current buffers as I couldn't find anything about their noise characteristics in the datasheet. LISO files used to generate this plot are attached.

Quote:

I've added marked-up schematics + high-res photographs of the SRM coil driver board and dewhitening board to the 40m DCC Document tree (D1700217 and D1700218).

In the attached marked-up schematics, I've also added the proposed changes which Rana and I discussed earlier today. For the thick-film -> thin-film resistor switching, I will try and make a quick LISO model to see if we can get away with replacing just a few rather than re-stuff the whole board.

### Since I have the board out, should I implement some of these changes (like AD797 removal) before sticking it back in and pulling out one of the ITM boards? I need to look at the locking transients and current digital limit-values for the various DoFs before deciding on what is an appropriate value for the output resistance in series with the coil.

Another change I think should be made, but I forgot to include on the markups: On the dewhitening board, we should probably replace the decoupling capacitors C41 and C52 with equivalent value electrolytic caps (they are currently tantalum caps which I think are susceptible to fail by shorting input to output).

12988   Fri May 12 12:34:55 2017 gautamUpdateGeneralITM and BS coil driver + dewhite board pulled out

I first set the bias sliders to 0 on the MEDM screen (after checking that the nominal values were stored), then shut down the watchdogs, and then pulled out the boards for inspection + photo-taking.

12990   Fri May 12 18:50:08 2017 gautamUpdateGeneralITM and BS coil driver + dewhite board pulled out

I've uploaded high-res photos + marked up schematics to the same DCC page linked in the previous page. I've noted the S/Ns of the ITM, BS and SRM boards on the page, I think it makes sense to collect everything on one page, and I guess eventually we will unify everything to a one or two versions.

To take the photos, I tried to reproduce the "LED light painting" technique reported here. I mounted the Canon EOS Rebel T3i on a tripod, and used some A3 sheets of paper to make a white background against which the board to be photographed was placed. I also used the new Macro lens we recently got. I then played around with the aperture and exposure time till I got what I judged to be good photos. The room lights were turned off, and I used the LED on my phone to do the "painting", from ~a metre away. I think the photos have turned out pretty well, the component values are readable.

 Quote: I first set the bias sliders to 0 on the MEDM screen (after checking that the nominal values were stored), then shut down the watchdogs, and then pulled out the boards for inspection + photo-taking.

12999   Fri May 19 19:18:53 2017 KaustubhSummaryGeneralTesting of the new Photo Detectors ET-3010 and ET-3040

Motivation:

I got some hands-on-experience on using RF photodetectors and the Network Analyzer from Koji. There were newly purchased RF photodetectors from Electro-Optics Technology, Inc.. These were InGaAs Photodetectors with model no.: 120-10050-0001(ET-3010) and 120-10056-0001(ET-3040). The User Guide for the two detectors can be found here. This is the first time we bought the ET-3010 model PD for the 40m lab. It has an operation bandwith >1.5GHz(not tested yet), much higher than other PDs of its kind. This can be used for detecting the output as we 'sweep' the laser frequency for getting data on the optical cavities and the resonating modes inside the cavity. We just tested out the ET-3040 model today but will test out the ET-3010 next week.

Tools and Machines Used:

We worked on the optical bench right in front of the main entrance to the lab. We put the cables, power chords, etc. to their respective places. We used screws, poles, T's, I's, multimeter, Network/Spectrum Analyzer(along with the moving table), a lab computer, Oscilloscope, power supply and the aforementioned PDs for our testing. We took these items from the stack of tools at the Y-arm and the boxes of various different labelled palced near the X-arm. We moved the Network Analyzer(along with the bench) from near the Y-arm to our workplace.

Procedure:

I will include a rough schematic of the setup later.

We alligned the reference PD(High Speed Photoreceiver model 1611) and the test PD(ET-3040 in this case) to get optimal power output. We had set the pump current for the laser at 19.5mA which produced a power of 1.00mW at the output of the fiber couple. At the reference detector the measured voltage was about 1.8V and at the DUT it was about 15mV. The DC transimpedance for the reference detector is 10kOhm and its responsivity to 1064 nm is around 0.75A/W. Using this we calculate the power at the reference detector to be 0.24mW. The DC transimpedance for the DUT is 50Ohm and the responsivity of about 0.9A/W. This amounts to a power of about 0.33mW. After measuring the DC voltages, we connected the laser input to the Network Analyzer and gave in an RF signal with -10dBm and frequency modulation from 100 kHz to 500 MHz. The RF output from the Analyzer is coupled to the Reference Channel(CHR) of the analyzer via a 20dB directional coupler. The AC output of the reference detector is given at Channel A(CHA) and the output from the DUT is given to Channel B(CHB). We got plots of the ratios between the reference detector, DUT and the coupled refernce for the Transfer Function and the Phase. We found that the cut-off frequency for the ET3040 model was at arounf 55 MHz(stated as >50MHz in the data sheet). We have stored the data using the lab PC in the directory .../scripts/general/netgpibdata/data.

Result:

The bandwidth of the ET-3040 PD is as stated in the data sheet, >50 MHz.

Precaution:

These PDs have an internal power supply of 3V for ET-3040 and 6V for ET-3010. Do not leave these connected to any instruments after the experiments have been performed or else the batteries will get drained if there is any photocurrent on the PDs.

To Do:

A similar procedure has to be followed in order to test the ET-3010 PD. I will be doing this tentatively on Monday.

13003   Mon May 22 13:37:01 2017 gautamUpdateGeneralDAC noise estimate

### Summary:

I've spent the last week investigating various parts of the DAC -> OSEM coil signal chain in order to add these noises to the MICH NB. Here is what I have thus far.

### Current situation:

• Coils are operated with no DAC whitening
• So we expect the DAC noise will dominate any contribution from the electronics noise of the analog De-Whitening and Coil Driver boards
• There is a factor of 3 gain in the analog De-Whitening board

### DAC noise measurement:

• I essentially followed the prescription in G1401335 and G1401399
• So far, I only measured one DAC channel (ITMX UL)
• The noise shaping filter in the above documents was adapted for this measurement. The noise used was uniform between DC and 1kHz for this test.
• For the >50Hz bandstops, I used 1 complex pole pair at 5Hz, and 1 compelx zero pair at 50Hz to level off the noise.
• For <50Hz bandstops, I used 1 compelx pole pair at 1Hz and 1 complex zero pair at 5Hz to push the RMS to lower frequencies
• I set the amplitude ("gain" = 10,000 in awggui) to roughly match the Vpp when the ITM local damping loops are on - this is ~300mVpp (measured with a scope).
• The elliptic bandstops were 6th order, with 50dB stopband attenuation.
• The SR785 input auto-ranging was disabled to allow a fair comparison of the various bandstops - this was fixed to -20 dBVpk for all measurements, and the SR785 noise floor shown is also for this value of the input range. Input was also AC coupled, and since I was using the front-panel LEMO for this test, the signal was effectively single-ended (but the ground of the SR785 was set to "floating" in order to get the differential signal from the DAC)
• Attachment #1 shows the results of this measurement - I've subtracted the SR785 noise from the other curves. The noise model was motivated by G1401399, but I use an f^-1/2 model rather than an f^-1 model. It seems to fit the measurement alright (though the "fit" is just done by eye and not by systematic optimization of the parameters of the model function).

### Noise budget:

• I then tried to translate this result into the noise budget
• The noises for the 4 face coils are added in quadrature, and then the contribution from 3 optics (2 ITMs and BS) are added in quadrature
• To calibrate into metres, I converted the DAC noise spectral density into cts/rtHz, and used the numbers from this elog. I thought I had missed out on the factor of 3 gain in the de-white board, but the cts-to-meters number from the referenced elog already takes into account this factor.
• Just to be clear, the black line for DAC noise in Attachment #2 is computed from the single-channel measurement of Attachment #1 according to the following relation: $\script{n}_{\mathrm{DAC}} ~ (m/\sqrt{Hz}) = n_{1-ch} (V/\sqrt{Hz}) \times (2^{15}/20) (cts/V) \times G_{act} \times 2 \times \sqrt{6}$, where G_act is the coil transfer function from the referenced elog, taken as 5nm/f^2 on average for the 2 ITMs and BS, the factor of 2 comes from adding the noise from 4 coils in quadrature, and the factor of sqrt(6) comes from adding the noise from 3 optics in quadrature (and since the BS has 4 times the noise of the ITMs)
• Using the 0.016N/A number for each coil gave me an answer than was off by more than an order of magnitude - I am not sure what to make of this. But since the other curves in the NB are made using numbers from the referenced elog, I think the answer I get isn't too crazy...
• Attachment #2 shows the noise budget in its current form, with DAC noise added. Except for the 30-70Hz region, it looks like the measured noise is accounted for.

• I have made a number of assumptions:
• All DAC channels have similar noise levels
• Tried to account for asymmetry between BS and ITMs (BS has 100 ohm resistance in series with the coil driver while the ITMs have 400 ohms) but the individual noises haven't been measured yet
• This noise estimate holds for the BS, which is the MICH actuator (I didn't attempt to simulate the in-lock MICH control signal and then measure the DAC noise)
• But this seems sensible as a first estimate
• The dmesg logs for C1SUS don't tell me what DACs we are using, but I believe they are 16-bit DACs (I'll have to restart the machine to make sure)
• In the NB, the flattening out of some curves beyond 1kHz is just an artefact of the fact that I don't have data to interpolate in that region, and isn't physical.
• I had a brief chat with ChrisW who told me that the modified EEPROM/Auto-Cal procedure was only required for 18-bit DACs. So if it is true that our DACs are 16-bit, then he advised that apart from the DAC noise measurement above, the next most important thing to be characterized is the quantization noise (by subtracting the calculated digital control signal from the actual analog signal sent to the coils in lock)
• More details of my coil driver electronics investigations to follow...
13005   Mon May 22 18:20:27 2017 KaustubhSummaryGeneralTesting of the new Photo Detectors ET-3010 and ET-3040

I am adding the text files with the data readings and paramater settings along with the Bode Plot of the data. I plotted these graphs using matplotlib module with python 2.7.

 Quote: Motivation: I got some hands-on-experience on using RF photodetectors and the Network Analyzer from Koji. There were newly purchased RF photodetectors from Electro-Optics Technology, Inc.. These were InGaAs Photodetectors with model no.: 120-10050-0001(ET-3010) and 120-10056-0001(ET-3040). The User Guide for the two detectors can be found here. This is the first time we bought the ET-3010 model PD for the 40m lab. It has an operation bandwith >1.5GHz(not tested yet), much higher than other PDs of its kind. This can be used for detecting the output as we 'sweep' the laser frequency for getting data on the optical cavities and the resonating modes inside the cavity. We just tested out the ET-3040 model today but will test out the ET-3010 next week...

ELOG V3.1.3-