Added zita to the martian host table, and configured his network accordingly:
zita A 192.168.113.217
I installed 'nfs-client' on zita (the StripTool terminal). It now has mounted all the shared disks, but still can't do StripTool since its a 32-bit machine and our StripTool is 64.
I'll finish the rest of the config to turn it into a full-fledged workstation tomorrow.
Here is a comparison of the response of various DoFs in our various RFPD sensors for two different CARM offsets. Even in the case of the smaller CARM offset of ~1kHz, we are several linewidths away from the resonance. Need to do some finesse modeling to make any meaningful statement about this - why is the CARM response in REFL11 apparently smaller for the smaller CARM offset?
If you mistrust my signal processing, the GPS times for which I ran the sensing lines are:
CARM offset = ~30kHz (arm transmission <0.02) --- 1257064777+5min
CARM offset = ~1kHz (arm transmission ~5) --- 1257065566+5min
There seems to be stronger-than-expected coupling between CARM and the 3f sensors.