40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log, Page 75 of 339 Not logged in
ID Date Author Type Category Subject
136   Wed Nov 28 19:44:18 2007 tobinUpdatePSLHEPA
I found the HEPA turned off completely. I turned it on.
163   Tue Dec 4 23:16:35 2007 tobinUpdatePSLISS
I was confused to find that I could increase the ISS gain slider all the way from 15dB to 30dB without seeing much of any increase in gain in the measured open-loop transfer function. While making these swept-sine measurements, the saturation indicator almost never tripped, indicating it was seemingly happy. But then I noticed an odd thing: if I disable the test ("analog excitation") input, the saturation indicator trips immediately. I hooked up a scope to the current shunt test point (TP12). With the test input enabled, the loop closed, and the analog excitation port connected to the SR785, I see a a 5 Vpkpk, 2.55 MHz triangle wave there. It is there even if I set the SR785 excitation amplitude to zero, but it disappears if I disconnect the cable from the SR785.

I found oscillations at TP20, TP30, TP36, TP41, and TP42. Many of these are in the (unused) "outer loop" circuitry and currently lack compensation capacitors.
165   Wed Dec 5 13:49:08 2007 albertoUpdateElectronicsRF AM PD lines monitor
In the last weeks I’ve been working on the design of an electronic board to measure directly the power of the main spectral lines on of the RF-AM photodiode from as many independent outputs. The idea is to have eventually a monitor channel in the CDS network for the power of each line.
Looking at at the spectrum from the RF-AM PD (see attached plot), there are 5 main lines:
Frequency
3 fsr = 33 195 439 Hz
4 fsr = 66 390 878 Hz
12 fsr = 132 781 756 Hz
15 fsr = 165 977 195 Hz
18 fsr = 199 172 634 Hz

Two main approaches have been proposed for the circuit depending on the way followed to isolate the lines:
1) Filters: the frequencies are separated by narrow notch filters, then a diode bridge rectifies and a low pass filter extracts the DC component.
2) Mixers: for each frequency there is a mixer driven by a copy of the correspondent modulation frequency provided by the function generators (the Marconi). The mixers automatically give the DC component of the rectified signals.
Because of the phase lags that we should compensate if we used mixers, we would prefer the first approach, if it works.
Starting with a tolerance of about 10% between the channels, the spectrum (see attachment) sets the constraint to the filter’s suppression:
Filter central frequency [MHz]******Suppression within 30 Mhz [dB]
33*********************************-7-20 = -27
66**********************************7-20 = -13
133*********************************12-20 = -8
166********************************-12-20 = -32
199*********************************10-20 = 10

So far I’ve tried two kinds of designs for the filters, Butterworth (see attachment) and LC and I'm measuring transfer functions tuning the components to match the central frequency and the bandwdth of the filters with the requirements.

The frequencies we’re dealing with are rather high and several adjustments had to be done to the measurement system in order to shield the circuit from the impedance of the input and the output line (i.e., amplifier turned out to be necessary). Also, an the mixer had to be replaced to an RF one.
It seems I'm now measuring new transfer functions (which look quite different from what I've got with no amplifiers).
To be posted soon.
Attachment 1: alberto.spectrum2.png
Attachment 2: Butterworth.PNG
167   Wed Dec 5 17:49:57 2007 tobinUpdatePSLISS
Attached is a plot of the ISS RIN with a variety of gain settings.

Unfortunately the dark noise is huge now--a result of the new cables & wiring?
Attachment 1: rin.pdf
168   Wed Dec 5 18:08:36 2007 AndreyUpdateASCOptical Lever laser for ETMX is installed

A new laser with \lambda=633nm has been intalled and the mirror adjusted so that the signal hits the center of the photodetector.

Output power level of that laser is 3.45 +- 0.05 mW.

Only about 0.29mW hits the photodetector.

Cable clips have been used to firmly fix the power supply cable for the laser.

See attached photopicture of the ETMX - "oplev" - optical - table.
Attachment 1: DSC_0199.JPG
169   Wed Dec 5 18:22:03 2007 tobinUpdatePSLISS dark noise
Attached is a plot of the dark noise spectrum of the ISS photodiodes (1) before fooling with them, (2) after replacing the 4151's with 4131's (improvement!), and (3) after replacing the cables and changing the wiring (disaster!).
Attachment 1: sense_noise.pdf
171   Wed Dec 5 20:32:51 2007 tobinUpdatePSLISS dark noise
The ISS dark noise is not coming from the PD heads; the spectrum is essentially unchanged when the PD is unplugged from the ISS. Did the input opamps both get semi-fried in the same way? (They worked so well when they were first installed.) What else changed? I'm baffled.
177   Thu Dec 6 19:30:43 2007 tobinUpdatePSLISS dark noise - 60 Hz!
A higher resolution spectrum [attached] shows that nearly all of the excess dark noise on the ISS is in 60 Hz harmonics (with some 256 Hz harmonics too--are these from the DAQ?).

With the loop closed and the slider at 5dB, the laser light coming out has a noise floor of 10^-7 RIN or better from 40 Hz to 8 kHz.

Now to figure out why all this 60 Hz is getting in... (I tried turning off all the lights and the HEPA, and moving the SR785 further away, none of which did anything.)
Attachment 1: iss.pdf
181   Fri Dec 7 18:28:30 2007 tobinUpdateComputer Scripts / Programscompiled matlab hoses itself
Andrey pointed out to me that some matlab functions in the Signal Processing Toolbox were dying with errors. Looking into the .m file (identified using the "which" command), I was surprised to see binary garbage rather than glistening, clear Matlab prose. Then I noticed the directory in which it was finding the .m file:
>> which decimate
/cvs/cds/caltech/apps/mDV/extra/linetrack_c_mcr/toolbox/signal/signal/decimate.m
See that "linetrack_c_mcr" directory? This is what is generated when a "compiled" (grumble) Matlab program is run--it decompresses itself into a subdirectory containing weird semi-compiled binary .m files. Unfortunately this is somehow getting incorporated into the matlab path. (I assume there is something in mDV that says "put all subdirectories into the path.")

I hate the Matlab compiler.
182   Fri Dec 7 18:31:30 2007 tobinUpdateComputer Scripts / Programscompiled matlab hoses itself
Addendnum. The reason the linemon_mcr command was in the path was because of the user issuing the command "addpath(genpath(pwd))" where genpath(D) "returns a path string starting in D, plus, recursively, all the subdirectories of D."

The Matlab compiler is still bad, however.
183   Fri Dec 7 19:14:30 2007 tobinUpdatePSLISS dark noise - ground loop enlightenment
My alleged 60 Hz harmonics were all from a ground loop created by connecting the SR785 ground to the ISS circuit ground; they disappeared when I set the SR785 input to "floating ground." doh!

I modified the ISS PD's to have a 100 ohm resistor in series with the output (in place of 20 ohms). The diodes are again in place on the table, ready for action.
185   Mon Dec 10 18:42:20 2007 tobinUpdatePSLISS RIN script
I wrote a script to measure the ISS RIN. The script uses the "labca" interface (described in an earlier entry) to read and twiddle EPICS settings and mDV to get DAQ data. The script measures open loop RIN, closed loop RIN at each of several gain slider settings, and dark noise. The dark noise is obtained by misaligning (unlocking) the PMC. The script also compares the whitened and unwhitened spectra for an open loop measurement and performs a fit of a simple pole to find the dewhitening filter.

This is all very exciting, but I don't quite believe the results, since the closed loop RIN seems to bottom out at 2e-7/rtHz regardless of the gain slider setting.

Sample output attached. The script may be found at scripts/PSL/ISS/rin.m.
Attachment 1: rin-20071210-1831.pdf
192   Sun Dec 16 16:52:40 2007 dmassUpdateComputersComputer on the end Fixed
I had Mike Pedraza look at the computer on the end (tag c21256). It was running funny, and turns out it was a bad HD.

I backed up the SURF files as attachments to their wiki entries. Nothing else seemed important so the drive was (presumably) swapped, and a clean copy of xp pro was installed. The username/login is the standard one.

Also - that small corner of desk space is now clean, and it would be lovely if it stayed that way.
193   Mon Dec 17 11:47:13 2007 albertoUpdateElectronicsan alternative design for the RFAM monitor's filter at 33Mhz
Since the Butterworth turned out o be rather wide-band, I tried an other configuration for the 33 MHz filter. Attached are the simulated transfer function and the measured. As one can see, the measured peak is much broader than expected.
Attachment 1: RFSim99-33MHz.png
Attachment 2: RF99-SimmButterworthPrototype.png
Attachment 3: RFSim99-33MHz-TFplot.png
195   Tue Dec 18 00:51:39 2007 AndreyUpdateComputer Scripts / ProgramsResults of Saturday overnight measurements

As I indicated in the previous e-log entry (#194), I made overnight measurements in XARM in the night from Saturday to Sunday.

Root-mean-square values of the peaks in calibrated spectra were calculated, and I plotted them as functions of suspension gains in ITMX and ETMX "position" degrees of freedom.
More specifically, Q_ITMX means the value in the channel "C1:SUS-ITMX_SUSPOS_GAIN", while Q_ETMX means the value in the channel "C1:SUS-ETMX_SUSPOS_GAIN".

Root-mean-square values (RMS) were calculated during that night in three intervals:

1) around 0.8 HZ in the interval (0.6 Hz <-> 1.0 Hz);

2) around 3.0 Hz in the interval (2.0 Hz <-> 3.6 Hz);

3) in the broad interval from 0.6Hz to 3.6Hz.

I plotted three results for RMS in the abovementioned three intervals in three different ways:

1) view from the top in the axes (Q_{ITMX}+Q_{ETMX})/2 and (Q_{ITMX}-Q_{ETMX}) -> first three graphs (attachments 1 -3);

2) view from the side in the same sum- and difference-axes -> next three graphs (attachments 4-6);

3) view from the side in Q_{ITMX} and Q_{ETMX} axes -> next three graphs (attachments 7-9), above accelerometer spectra (attachments 10-11).

Also, I compared the ground noise level by comparing spectra of accelerometer signals at different times during that night. As a reminder, before my disease I installed one accelerometer near ITMX and another accelerometer near ETMX (see entries 161 and 172 in ELOG). The plots of ratios of accelerometer signals at different times (pairs of times that were used: 12AM and 3AM, 12AM and 6AM, 12AM and 9AM) are given below, see attachments 10-11.

Tomorrow I will try to compare the results with the second measurements that are being taken tonight.
Attachment 1: RMS_08Hz_top_view.png
Attachment 2: RMS_3Hz_top_view.png
Attachment 4: RMS_08Hz_Qsum-Qdiff-axes.png
Attachment 5: RMS_3Hz_Qsum-Qdiff-axes.png
Attachment 7: RMS_08Hz_Qaxes.png
Attachment 8: RMS_3Hz_Qaxes.png
Attachment 10: Accel_ITMX.png
Attachment 11: Accel_ETMX.png
196   Tue Dec 18 16:50:35 2007 tobinUpdateSAFETYuvex laser safety glasses defective
A few days ago we noticed what appeared to be a blotched, speckled fracturing of the coating of the "UVEX" laser safety glasses. These are the glasses with "transparent" (reflective to 1064nm) lenses and white frames that we keep in a box on top of a filing cabinet in the control room. Today Steve measured the transmission of these glasses and found 80% transmission of 1064nm in several cases.

Do not use the white, transparent "uvex" laser safety glasses until further notice. Steve has hidden them away so that you won't be tempted.

Below is attached a photo of a bad lens.
197   Tue Dec 18 21:31:31 2007 tobinUpdatePSLISS RIN
My measurements of the ISS RIN via the SR785 and via the DAQ disagree considerably. The spectral shapes are very similar, however, so I expect that a constant factor is creeping in somewhere. Measurements taken at the PD DC monitor points using the SR785 attached. There is a lot of excess noise in the 300 Hz - 1 kHz region.
Attachment 1: iss-rin.pdf
201   Wed Dec 19 15:51:00 2007 AndreyUpdateComputer Scripts / ProgramsDaytime measurements in XARM and their results

I was making measurements in XARM for three different nights. All the results agree with each other (I will put the results from the last night soon).

Steve Vass recommended to me to compare those results with the daytime data, in order to see if there is a real necessity to run the scripts overnight or if daytime results will yield similar results.

XARM has been locked, and I am taking measurements today from 3.30PM till 11.30PM.

I will be changing the suspension damping gains in ETMX and ITMX "position" degrees of freedom in the interval from 1.0 to 3.75 with the step 0.25.

BELOW: RESULTS OF MEASUREMENTS WERE ADDED ON THURSDAY, DEC. 20.

All the meaning of the attachments 1-3, 4-6, 7-9, 10-11 is the same as in previous ELOG entries # 195, # 199, # 202, see in those entries which graph corresponds to which coordinate axes orientation.
Attachment 1: RMS-08Hz-Top_View.png
Attachment 2: RMS-3Hz-Top_View.png
Attachment 4: RMS-08Hz-Side-View.png
Attachment 5: RMS-3Hz-Side_View.png
Attachment 7: RMS-08Hz-Q_I-Q_E-Axes.png
Attachment 8: RMS-3Hz-Side_View.png
Attachment 10: Accelerometer_ETMX.png
Attachment 11: Accelerometer_ITMX.png
203   Wed Dec 19 16:40:12 2007 steveUpdateSAFETYlaser safety glasses measured
I measured the coarse transission at 1064nm of the 40m safety glasses today.

12 pieces of UVEX # LOTG-YAG/CO2 light green, all plastic construction, ADSORBANT

3 pieces of 6KG5, Scott colored filter glass type,

individual prescription glasses: alan, bob, ben, jay and steve

7 pieces of dual waveleght glasses

These glasses showed 0.00mW transmission out of 170mW Crysta Laser 1064
205   Thu Dec 20 02:04:09 2007 AndreyUpdateComputer Scripts / ProgramsNew overnight measurements in XARM and their results

I ran in the daytime/evening time my program, changing the damping gains in suspension "position" degree of freedom for ETMX and ITMX
in the interval from 1.00 to 3.75 with the step 0.25 (see entry # 201).

Now I am running overnight (from 2AM till 9AM) the program changing the gains in the interval from 1.3 to 3.5 with the step 0.20,
12 X 12 = 144 experimental points. I started so late because I fell asleep after my Wednesday evening dinner, then woke up half an hour ago and hurried to the lab.

BELOW: RESULTS OF MEASUREMENTS WERE ADDED ON THURSDAY EVENING, DEC. 20.

All the meaning of the attachments 1-3, 4-6, 7-9, 10-11 is the same as in previous ELOG entries # 195, # 199, # 202, see in those entries which graph corresponds to which coordinate axes orientation.
Attachment 1: RMS-08Hz-Top-View.png
Attachment 2: RMS-3Hz-Top-View.png
Attachment 4: RMS-08Hz-Side_View.png
Attachment 5: RMS-3Hz-Side_View.png
Attachment 7: RMS-08Hz-Q_I-Q_E-Axes.png
Attachment 8: RMS-3Hz-Q_I-Q_E-Axes.png
Attachment 10: Accelerometer-ETMX.png
Attachment 11: Accelerometer-ITMX.png
207   Thu Dec 20 19:10:03 2007 waldmanUpdateOMCStressful reattachment of heater
Photos may follow eventually, but for now here's the rundown. I scraped the heater clean of the thermal epoxy using a clean razor blade. Then I stuffed a small piece of lint free cloth in the OTAS bore and wrapped the OMC in tin foil. With a vacuum sucking directly from the face of the OTAS, I gently scraped the glue off the OTAS aluminum. I wiped both the OTAS and the heater down with an isoproponal soaked lint-free cloth. I put a thin sheen of VacSeal on the face of the heater, wiping off the excess from the edges with a cloth. Then I clamped the heater to the OTAS using 2" c-clamps from the tombstone back to the heater front, making sure the alignment of the OTAS was correct (connector on the absolute bottom, concentric with the OTAS outer diameter). I added a second clamp, then beaded the outside of the joint with a little bit extra VacSeal, just for kicks. I'll leave it covered at least overnight, and maybe for a day or two.

sam
208   Thu Dec 20 21:57:34 2007 AndreyUpdateComputer Scripts / ProgramsMeasurements in XARM today

Today at 2PM I started a program, it should change the suspension gains in the interval from 1.0 to 3.8 with the step 0.2. Estimated running time is till 3.30AM coming night.

Results will be reported on Friday.

Due to some unforeseen circumstances, I was unable to add results on Friday. I have so far accelerometer spectra only, which I add to this ELOG entry.

I have files with the measurement results, and I will process them after Christmas and add to this ELOG entry. I might not be in the lab on Dec. 24 and 25.
Attachment 1: Accelerom_ETMX.png
Attachment 2: Accelerom_ITMX.png
210   Fri Dec 21 20:32:25 2007 tobinUpdatePhotosGigE camera
I couldn't resist any longer: I plugged in the Prosilica GC 750 GigE camera and took it for a spin. This is the little CMOS camera which sends out video over gigabit ethernet.

There were no difficulties at all in getting it running. I just plugged in the power, plugged in ethernet, and put on a lens from Steve's collection. I downloaded the "Sample Viewer" from the Prosilica website and it worked immediately.

It turns out that "Kirk's" computer has not only a gigabit ethernet card, but a little gigabit ethernet switch. I plugged the camera into this switch. The frame rate is amazing. With the camera under fluorescent lights I thought I saw some wacky automatic gain control, but I think this ~10Hz flicker is aliasing of the 60 Hz room lighting.

I put the camera on the PSL table briefly and tried viewing the image from a laptop over the (54mbs) wireless network. This didn't work so well: you could get a couple frames out of the camera, but then the client software would complain that it had lost communications. It appeared that scattered 1064nm light did show up brightly on the camera image. There is a green ethernet cable currently stashed on the roof of the PSL that appears unused. We can try mounting the gigE CMOS cable in place of one of the CCD video cameras.

I did not try the Linux software.

The camera is currently set up at Kirk's desk, using the cool little tripod Rana got from CyberGuys.

This camera looks very promising! Also, in the test image attached below, a very unusual condition has been documented.
Attachment 1: robs_desk.png
213   Wed Dec 26 15:00:06 2007 ranaUpdateSUSETMY tripping
Steve mentioned to me that ETMY is still tripping more than ETMX. The attached DV plot
shows the trend of the watchdog sensors; essentially the RMS fluctuations of the shadow
sensors. (note** DV can make PNG format plots directly which are much better than JPG
when making plots and much smaller than PS or PDF when plotting lots of points).
Attachment 1: etm.png
214   Wed Dec 26 15:12:48 2007 ranaUpdateSUSETMY tripping
It turned out that the ETMY POS damping gain was set to 1.0 while the ETMX had 3.8.

I put both ETMs to a POS gain of 4 and then also set the PIT, YAW and SIDE gains for
ETMY. Let's see if its more stable now.

In the next week or so Andrey should have perfected his damping gain setting technique
and the numbers should be set more scientifically.
215   Thu Dec 27 12:12:02 2007 pkpUpdate Update on GigE Camera
So I finally got the linux software to compile on mafalda. I got the software to dump all the information regarding the camera onto a file. I tried to take a tiff snap and came up empty. So I looked at the configuration file and realized that the camera thinks that the frame-rate is a nan. Am reading up the manual to fix the frame-rate manually and then will attempt to take another snap.

All the files are in a folder called Prosilica in /home/controls/ on mafalda. All the executables are in /home/controls/Prosilica/bin-pc/x86/* . On another note, I am looking for a name for the camera. Any suggestions are welcome.
216   Thu Dec 27 13:08:04 2007 ranaUpdateSUSETMY tripping
Here's a trend from the last 2 days of ETMX and ETMY. You can see that the damping gain increase
has made them now act much more alike. Problem fixed.
Attachment 1: Untitled.png
217   Thu Dec 27 18:18:56 2007 ranaUpdateComputersUpdate on GigE Camera

 Quote: So I finally got the linux software to compile on mafalda. I got the software to dump all the information regarding the camera onto a file. I tried to take a tiff snap and came up empty. So I looked at the configuration file and realized that the camera thinks that the frame-rate is a nan. Am reading up the manual to fix the frame-rate manually and then will attempt to take another snap. All the files are in a folder called Prosilica in /home/controls/ on mafalda. All the executables are in /home/controls/Prosilica/bin-pc/x86/* . On another note, I am looking for a name for the camera. Any suggestions are welcome.

Suggestion #1: put this in the target area in a directory called /prosilica/. /home/controls is not backed up.

Suggestion #2: put a readme file in there on any work that was necessary to get it to compile.

Suggestion #3: make a wiki page for the camera with all the info that camera code developers will need
218   Sun Dec 30 02:36:35 2007 pkpUpdateGeneralAnother update
So I followed suggestions 1 and 3 so far and have started writing up what all needs to be done in order to compile and use the camera. I wrote a program to ping the camera and get its properties and am working on a program to get an image. The reason why I want to write my own programs to do this, is that it will be easier to reuse and also to compile/use in the first place. The programs currently rest in /cvs/cds/caltech/target/Prosilica/ . Unfortunately I will be away for the next couple of days and will have another update on the 2nd.
220   Thu Jan 3 08:53:55 2008 steveUpdateSUSetmy vs etmx
Rana have corrected sus gain damping setting of ETMY 8 days ago

gain settings: pos, pit, yaw & sd
etmx: 4,2,2,& -16
etmy: 4,2,2,& 50
Attachment 1: sus.jpg
221   Thu Jan 3 09:12:59 2008 steveUpdatePSLMZ servo
Here is MZ trend for one year and 40 days.
Now days it runs out of range on the low side.
This is the weakest link in the psl today.
Attachment 1: mz1y.jpg
Attachment 2: mz40d.jpg
222   Thu Jan 3 09:55:11 2008 steveUpdateSUSetmy sus damping restored
ETMY watch dog was lost at midnight
Attachment 1: etmy12h.jpg
225   Fri Jan 4 08:42:03 2008 steveUpdateSUSetmy trips again
ETMY sus damping tripped at 6am this morning
It was reset. We should put an accelerometer to the south end to see
the garbage dumping effect.
Attachment 1: etmy20m.jpg
Attachment 2: etmy120s.jpg
Attachment 3: etmysenV.jpg
226   Mon Jan 7 09:01:39 2008 steveUpdateSUSBS sus damping restored
The BS sus damping was lost at 8am Sunday morning.
Attachment 1: bssdl.jpg
227   Tue Jan 8 15:20:17 2008 PkpUpdateCamerasGigE update
[Tobin , Pinkesh]

Finally we got the camera doing something (other than giving out its attributes). The only thing that seems to work so far is a program called AAviewer, which converts the image into an ASCII format and displays it on the screen. If you want to play around with it, log into mafalda (131.215.113.23) via rana.ligo.caltech.edu. Access /cvs/cds/caltech/target/Prosilica/bin-pc/x86/ and there should be a few programs in there, one of which is AAviewer, which requires you to get an IP address (which is 131.215.113.103) for the camera right now. (You can also get the IP information via the ListCameras program). The camera is physically in the 40m near the network rack.

Other programs dont seem to be working and its probably due to the network/packetsize issues. Since linux2 can change its packetsize to a higher number, I will get it to compile on linux2 for now and then give it a shot.
228   Wed Jan 9 10:47:15 2008 frickeUpdateComputersethernet wiring in office/control room
The electrical people have been here for three days, installing ethernet cables and drops in the 40m office area, in the old conduits where there was power and 10base2. Soon we will have reliable ethernet connections, instead of relying on switches hanging from the ceiling, etc!
230   Wed Jan 9 20:36:42 2008 GoUpdateTreasureMoney in lab
Go's Desk.
Attachment 1: DSC_0370.JPG
232   Thu Jan 10 10:38:02 2008 steveUpdateSUSetmy damping restored
The IST building onstruction has really started yesterday and continuing today with big heavy ground breaking
machinary. The MC is holding lock and the suspentions are hanging on.

ETMY does not like this.

SUS-MC2_LLPD_VAR monitor is a good indicator of seismic activity on this 12 days plot
Attachment 1: etmysus.jpg
Attachment 2: sustrend16d.jpg
233   Thu Jan 10 12:08:23 2008 steveUpdateSUSwhy did the BS move?
Attachment 1: bshopped.jpg
234   Thu Jan 10 13:45:52 2008 PkpUpdateCamerasGLIBC Error
So, I have tried to compile the camera files which are in /cvs/cds/caltech/target/Prosilica/examples for the past 2 days now and have been unable to get rid of the following error. (specifically ListCameras.cpp, as it doesnt have any other libraries required, which unnecessarily complicates things)

../../bin-pc/x86/libPvAPI.so: undefined reference to `__stack_chk_fail@GLIBC_2.4'
collect2: ld returned 1 exit status
make: *** [sample] Error 1

I used to get this error on mafalda too, but I had fixed it by installing the latest version of the glibc libraries. Inspite of doing so on linux2, the error still persists. I suspect it had something to do with it being a FC3 machine. My own laptop, which also runs Ubuntu works fine too. The problem with these Ubuntu machines is that they dont let me set the packet sizes to 9 kb which is required by the camera. Linux2 does.

If anyone has any idea how to resolve this issue, please let me know.

Thanks
Pinkesh.
235   Thu Jan 10 15:04:04 2008 steveUpdateSUSilluminator light effect on BS position
The bs chamber illuminator light was turned on this morning and left on.
Earlier on Rana noticed that the bs moved.
I follwed up to see what happened. I turned off oplev servo and tried to recenter on oplev pd
by adjusting pitch and yaw biases. It did not move. I looked at suspention and realized that the
illuminator was still on. I turned it off and to my amazement the the AP spot started flashing
Attachment 1: bssusilum.jpg
236   Fri Jan 11 17:01:51 2008 pkpUpdateCamerasGigE again
So, here I detail all the efforts on the GigE so far

(1) The GiGE camera requires a minimum of 9 kb packet size, which is not available on mafalda or on my laptop ( both of which run Ubuntu and the Camera programs compile there). The programs which require smaller sizes work perfectly fine on these machines. I tried to statically compile the files on these machines so that I could then port them to the other machines. But that fails because the static libraries given by the company dont work.

(2) On Linux2, which lets me set a packet size as high as 9 Kb, it doesnt compile because of a GLIBC error. I tried updating the glibc and it tells me that the version already existing is the latest ( which it clearly is not). So I tried to uninstall GLIBC and reinstall it, but it wont let me uninstall (it == rpm) glibc, since there are a lot of dependencies. A dead end in essence.

Steps being taken

(1) Locally installing the whole library suites on linux2. Essentially install another version of gcc and g++ and see if that helps.
(2) IF this doesnt work, then the only course of action I can take is to cannibalize linux2's GigE card and put it on mafalda. ( I need permission for this ).

Once again any suggestions welcome.
237   Mon Jan 14 14:41:09 2008 steveUpdateSUSetmy sus damping restored & mz relocked
Tree days trend of MZ HV drift is typical these days.
So as the etmy sus inability to hold damping for longer then 2-3 days.
Attachment 1: etmysus&mzhvtrend.jpg
239   Tue Jan 15 13:15:27 2008 tobinUpdateEnvironmentlots of noise
They're throwing concrete around at the construction site.
240   Wed Jan 16 14:06:24 2008 robUpdateLSCmonday's locking
rob, tobin, johnnie

We did some locking work monday night, with decent progress. Working in the PRFPMI style, we managed to get through the part of the script that hands off the offset-CARM DOF to the MCL, but were not successful in engaging the AO path.

We also confirmed the problem with tdsread which prevent it from reading from multiple TLS (Three Lettered Subsystems) at the same time. Tobin traced this to a problem with the ezca library which tds uses, but it's not clear how to fix it. For now we just split the tdsread calls so that there are no multiple TLS calls. Tobin will report further on this.
241   Wed Jan 16 14:09:45 2008 robUpdateLSCtuesday's locking

I got a little further with the locking (PRFPMI) last night, after discovering that the cable going from the CM board to the MC board was unplugged at the MC side. This explains why we weren't able to engage the AO path last night. Tonight, I got up to the point where DARM is handed off to OMC transmission, a step which repeatedly failed.
Eventually I realized that although all the lights are the green, the OMC Trans signal was not being updated in the LSC's memory. I suspect this is because the c1ass machine was powered down. Work continues.
242   Wed Jan 16 18:24:41 2008 ranaUpdateSUSETMY Watchdog
Because Steve keeps complaining about ETMY, I looked at some minute trend to see if there was something exotic happening at that time. It looks like there is some tremendous seismic activity to make it happen.

The trend shows that there is nothing special happening on the ETMX accelerometer or the ETMX suspension. At the same time, however, there is a huge jump in the ETMY sensors and therefore the watchdog signal. Whenever the watchdog value goes above 140, it trips.

After Andrey moves some accelerometers over to the Y end we can see the effect more directly.
Attachment 1: A.pdf
244   Thu Jan 17 14:13:20 2008 robUpdateLSCWednesday's locking
Incremental progress on locking yet again. This time the handoff of DARM to the OMC worked, and progress halted at handing off control of the common mode to REFL166.
245   Thu Jan 17 15:11:13 2008 josephbUpdateCamerasWorking on Malfalda
1) I can statically compile the ListCamera code (which basically just goes out and finds what cameras are connected to the network) on Malfalda and use that compiled code to run on Linux2 without a problem. Simply needed to add explicit links to libpthread.a and librt.a.
(i.e. -Bstatic -L /usr/lib/ -lpthread -Bstatic -L /usr/lib -lrt)

With appropriate static libraries, it should be possible to port this code to other linux machines even if we can't get it to compile on the target machine itself.

2)I've modified the Snap.cpp file so that it uses a packet size of 1000 or less. This simply involves setting the "PacketSize" attribute with the built in functions they provide in their library. After un-commenting some lines in that code, I was able to save tiff type images from the camera of up to 400x240 pixels on Malfalda. The claimed maximum resolution for the camera is 752x480, but it doesn't seem to work with the current setup. The max number of pixels seems to about 100 times the packet size. I.e. packet size of 1000 will allow up to 400x240 (96000) but not 500x240 (120,000). Not sure if this is an issue just with snap code or the general libraries used.

3)Will be working towards getting video running over the next day or so.
246   Thu Jan 17 18:22:14 2008 AlbertoUpdateElectronicsRF Monitor Band-pass Filter
After we finalized the schematic for the RF monitor board based on buffered LC resonators, on Richard Abbott's suggestion to avoid the complication brought in by the fast op-amps, we gave another chance to the a passive configuration of the band-pass filter based on a Chebyshev topology. Rich and Ben gave me an old but very powerful software tool to design that kind of filters and showed me the way to circumvent many hassles in making RF test boards.

I made a test circuit for the 166MHz line (see attached schematic), using tunable inductors. The TF are also attached.
We get more than 20 dB of isolation after 33MHz (with a loss of only few dB at the resonance - it could be less), which is enough for all the other frequencies (33,133,199 MHz) but we would like more for the 166. We are going to add one or two extra orders to the filter.

We also have to understand the spike at about 320Mhz and eventually somehow get rid of it.

Alberto
Attachment 1: RF166Mhz.png
Attachment 2: Chebyshevb.png
Attachment 3: Chebyshev2b.png
ELOG V3.1.3-