I mapped out the corresponding pins on both ends of the Guralp seismometer cable. Here is the diagram:
The circular 26-pin end of the cable (that plugs into the seismometer) is labeled as above. The other end (the 39-pin end) is not physically numbered, so I just came up with a numbering system. They are both pictured on the non-cable end of the connector. The colored circles indicate the pin pairs.
FROM JENNE, 30JULY2009: the Dsub end is 37 pin, not 39.
Q. When should we use plano-convex lenses, and when should we use bi-convex?
As I had the same question from Jenne and Dmass in a month,
I just like to introduce a good summary about it.
Lens selection guide (Newport)
At a first order, they have the same function.
Abberation (= non-ideal behavior of the lens) is the matter.
ETMY oplev is still out of order. Hopefully I'll get it under control by tomorrow.
I found two ThorLabs PDA55 Si photodetectors that says detect visible light from DC to 10MHz that I'm going to use from now on. I don't know how low of a frequency they will actually be good to.
I set the MC back to its good alignment (June 21st) using this procedure. The trend of the OSEM values over the last 40 days and 40 nights is attached.
Then I aligned the periscope to that beam. This took some serious periscope knob action. Without WFS, the transmission went to 2.7 V and the reflection down to 0.6V.
Then I re-aligned the MC_REFL path as usual. The beam was far enough off that I had to also re-align onto the MC LSC PD as well as the MC REFL camera (~2 beam radii).
Beams are now close to their historical positions on Faraday and MC2. I then restored the PZT sliders to their April snapshot and the X-arm locked.
Steve - please recenter the iris which is on the periscope. It has been way off for a long time.
So it looks OK now. The main point here is that we can trust the MC OSEMs.
Afterwards I rebooted c1susvme1 and c1susvme2 because they were skewed.
I'm impressed by Rana's simple way to align the MC. IFO arms are locked or flashing. 20 days trend attached.
It is really surprising that we now have again the data from the MC OSEMs since up to two days ago the record looked corrupted (see the attachments in my entry 1774).
The reason I ended up severely misaligning the the MC is exactly that there wasn't anymore a reference position that I could go back to and I had to use the camera looking a the Faraday.
When I turned them on, the control signal in Pitch from WFS2 started going up with no stop. It was like the integrator in the loop was fed with a DC bias. The effect of that was to misalign the MC cavity from the good state in which it was with the only length control on (that is, transmission ~2.7, reflection ~ 0.4).
I don't know why that is happening. To exclude that it was due to a computer problem I first burtrestored C1IOO to July the 18th, but since that did not help, I even restarted it. Also that didn't solve the problem.
At least one problem is the mis-centering of the resonant spot on MC2, which can be viewed with the video monitors. It's very far from the center of the optic, which causes length-to-angle coupling that makes the mulitple servos which actuate on MC2 (MCL, WFS, local damping) fight each other and go unstable.
I played with the MC alignment for the beam centering. After that, I restored the alignment values.
In principle, one can select the MC2 spot as one likes, while the transmitted beam axis to the IFO is not changed
as far as you are at the best alignment. This principle is almost trivial because the beam axis matches
to the input beam axis at the best alignment.
The alignment solution is not unique for a triangle cavity if we don't fix the end spot position.
In practice, this cruising of the MC2 spot is accomplished by the following procedure:
0) Assume that you are initially at the best alignment (=max transmission).
1) Slightly tilt the MC2.
2) Adjust MC1/MC3 so that the best transmission is restored.
I started from the following initial state of the alignment sliders:
After many iterations, the spot was centered in some extent. (See the picture)
The instability looked cured somewhat.
Further adjustment caused a high freq (10Hz at the camera) instability and the IMCR shift issue.
So I returned to the last stable setting.
Of course, if you move MC1, the reflected spot got shifted.
The spot has been apparently off-centered from the IMCR camera. (up and right)
At this stage, I could not determine what is the good state.
So, I restored the alignment of the MC as it was.
But now Alberto can see which mirror do we have to move in which direction and how much.
After speaking with Rana and realizing that it would be better to use smaller inductances in the flying-component circuit (and after a lot of tinkering with the original), I rebuilt the circuit, removing all of the resistors (to simplify it) and making the necessary inductance and capacitance changes. A picture of the circuit is attached, as is a circuit diagram.
A plot of the measured and simulated transfer functions is also attached; the general shape matches between the two, and the resonant frequencies are roughly correct (they should be 11, 29.5, and 55 MHz). The gain at the 55 MHz peak is lower than the other two peaks (I'd like them all to be roughly the same). I currently have no idea what the impedance is doing, but I'm certain it is not 50 Ohms at the resonant peaks, because there are no resistors in the circuit to correct the impedance. Next, I'll have to add the resistors and see what happens.
This is a quite nice measurement. Much better than the previous one.
1) For further steps, I think now you need to connect the real EOM at the end in order to incorporate
the capacitance and the loss (=resistance) of the EOM. Then you have to measure the input impedance
of the circuit. You can measure the impedance of the device at Wilson house.
(I can come with you in order to consult with Rich, if you like)
Before that you may be able to do a preparatory measurement which can be less precise than the Wilson one,
but still useful. You can measure the transfer function of the voltage across the input of this circuit,
and can convert it to the impedance. The calibration will be needed by connecting a 50Ohm resister
on the network analyzer.
2) I wonder why the model transfer function (TF) has slow phase changes at the resonance.
Is there any implicit resistances took into account in the model?
If the circuit model is formed only by reactive devices (Cs and Ls), the whole circuit has no place to dissipate (= no loss).
This means that the impedance goes infinity and zero, at the resonance and the anti-resonance, respectively.
This leads a sharp flip of the phase at these resonances and anti-resonances.
The real circuit has small losses everywhere. So, the slow phase change is reasonable.
The last week I've started setting up the HeNe laser on the PSL table and doing some basic measurements (Beam waist, etc) with the beam scan, shown on the graph. Today I moved a few steering mirrors that steve showed me from at table on the NW corner to the PSL table. The goal setup is shown below, based on the UCSD setup. Also, I found something that confused me in the EUCLID setup, a pair of quarter wave plates in the arm of their interferometer, so I've been working out how they organized that to get the results that they did. I also finished calculating the shot noise levels in the basic and UCSD models, and those are also shown below (at 633nm, 4mw) where the two phase-shifted elements (green/red) are the UCSD outputs, in quadrature (the legend is difficult to read).
0. Probably, you are working on the SP table, not on the PSL table.
1. The profile measurement looks very nice.
2. You can simplify the optical layout if you consider the following issues
A. The matching lenses just after the laser:
You can make a collimated beam only with a single lens, instead of two.
Just put a lens of f0 with distance of f0 from the waist. (Just like Geometrical Optics to make a parallel-going beam.)
Or even you don't need any lens. In this case, whole optical setup should be smaller so that your beam
can be accomodated by the aperture of your optics. But that's adequately possible.
B. The steering mirrors after the laser:
If you have a well elevated beam from the table (3~4 inches), you can omit two steering mirrors.
If you have a laser beam whose tilte can not be corrected by the laser mount, you can add a mirror to fix it.
C. The steering mirrors in the arms:
You don't need the steering mirrors in the arms as all d.o.f. of the Michelson alignment can be adjusted
by the beamsplitter and the mirror at the reflected arm. Also The arm can be much shorter (5~6 inches?)
D. The lenses and the mirrors after the PBS:
You can put one of the lenses before the PBS, instead of two after the lens.
You can omit the mirror at the reflection side of the PBS as the PBS mount should have alignment adjustment.
The simpler, the faster and the easier to work with!
This afternoon I kept working on the alignment of the beam so that it matches at the same the PSL periscope, the Mode Cleaner and the Faraday isolator at the input of the IFO.
The camera looking at the Farady showed a beam quite low from the center of the Faraday's entrance. I wanted to move it up.
After working on the periscope alignment and on the MC mirrors, I think I managed to moved it up a bit. To know whether that was enough or not I wanted to evaluate the alignment to the X arm by checking the value of TRX.
In order for the MC to be finely matched to the input beam from the periscope, the WFS controls have to be on. Before turning them on, I centered the beam on their QPDs and run the WFS_zero_offset script.
Flashes at ETMX show at least that the beam is going through the Farady. How well, I can't tell untill the MC is under full control.
I have to leave the lab now, but I can be back tomorrow to keep working on that.
ETMY oplev is currently a work in progress. The HeNe beam is hitting the photodiode, but the spot size there is pretty much the size of the entire QPD. Thus, the ETMY oplev isn't really useful right now. I'm re-figuring things out (note to self: close to the laser, you have to use Gaussian optics...regular ray tracing doesn't really work), and hopefully will have the oplev back under control by the time Alberto is finished realigning the IFO, so this doesn't keep anyone from doing any exciting locking work.
This morning I found the elog down. I restarted it using the procedure in the wiki.
After yesterday's changes in the MC cavity state today it was necessary to optimize the alignment to the Faraday.
The way I did it was by tuning the PSL periscope in pitch and yaw trying to maximize TRX with the arm locked. After a small change in either one of the two directions I first maximized the MC transmitted power and then I ran the alignment script for the X arm.
I explored the space for both pitch and yaw and the max that I could get from TRX was 0.91. I'm not sure whether the increase in TRX is entirely due to a better alignment to the Farady rather than to a higher MC transmitted power.
Also I'm not sure I'm well interpreting the image from the camera pointing at the Farady. I guess I need someone more familiar with it to tell me if it shows any sign of clipping.
Anyway, last week, even before the MC got misaligned, TRX didn't go above 0.90. So now I wonder whether it's the MC's fault or something else's if we have that value..
Chronicles of periscope and MC alignment
Yesterday morning I started aligning the periscope but it turned out to be trickier than usual. With the ASC (Alignment Sensing Control) off and only the length controls on, the Mode Cleaner didn't lock easily, although I knew I wasn't very far from the sweet spot.
In the afternoon the struggle continued and the matching of the the beam to the MC cavity became just worse. At some point I noticed that the ASC inputs somehow had got on - although the ASC still looked disabled from the MClock MEDM main screen. So I was actually working against the Wave Front Sensors and further worsening the periscope alignment.
That hurled me to the weeds. After hours of rowing across the stormy waters of a four-dimensional universe I got to have occasional TEM00 flashes at the transmission but still, surprisingly, no MC locking. Confused, I kept tuning the periscope but that just kicked me off road again.
Then at about 7pm Koji came to my rescue and suggested a more clever and systematic way to solve the problems. He suggested to keep record of the MC mirrors alignment state and re-align the cavity to the periscope. Then we would gradually bring the cavity back to the original good position changing the periscope alignment at the same time.
That would have worked straight away, if we hadn't been fighting against a subtle and cruel enemy: the 40m computer network. But I (as John Connor), and Koji (as the Terminator) didn't pull back.
Here's a short list of the kinds of weapons that the computers threw to us:
We then proceeded with Koji's plan. In an iterative process, we aligned the MC cavity maximizing the transmission and tuned the periscope in order to match the Faraday input of the interferometer. The last thing we did it by looking at the camera pointing at the Faraday isolator.
We found that we didn't have to tune the periscope much. That means that all afternoon I didn't really go too far, but the autolocker wasn't working properly, or it wasn't working at all.
Then we ran the alignment script for the X arm but it didn't work before we aligned the steering mirrors.
Then we ran it three times but could not get more than 0.87 at TRX. That means that there we still have to work on the alignment to the Faraday. That's job for today in the trenches of the lab.
See Adhikari eLOG entry: http://nodus.ligo.caltech.edu:8080/AdhikariLab/194
I compiled and ran a simple (i.e. empty) front end controller on scipe12 at wilson house. I hooked a signal into the ADC and watched it in the auto-generated medm screens.
There were a couple of gotchas:
1. Add an entry SYS to the file /etc/rc.local, to the /etc/setup_shmem.rtl line, where the system file is SYS.mdl.
2. If necessary, do a BURT restore. Or in the case of a mockup set the BURT Restore bit (in SYS_GDS_TP.adl) to 1.
in the rack next to the printer. It sounds like a fan is hitting something.
This past week, I have mostly been debugging my software. I have tried to use the fluorescent lights to test the camera, but I can't tell for sure if my code is finding the correct amplitude and phase or not. I am currently using Mathematica to double check my calculations in solving for the phase and amplitude.
Also, I have taken dark field images using a lens with a closed shutter. I have found that the dark band across the top of the images only appears after the camera heats up. Also, there is an average electronic noise of 14 with a maximum of 40. However, this electronic noise as well as any consistent ambient noise will be automatically corrected for in the calculations I'm using because I'm taking the differences between the CCD images to calculate relative phases and amplitudes.
I should be able to start setting up optics and performing better tests of my software this week.
All suspentions are kicked up. Sus dampings and oplev servos turned off.
c1iscey and c1lsc are down. c1asc and c1iovme are up-down
The computers and RFM network are up working again. A boot fest was necessary.Then I restored all the parameters with burtgooey.
The mode cleaner alignment is in a bad state. The autolocker can't get it locked. I don't know what caused it to move so far from the good state that it was till this afternoon. I went tuning the periscope but the cavity alignment is so bad that it's taking more time than expected. I'll continue working on that tomorrow morning.
I now suspect that after the reboot the MC mirrors didn't really go back to their original place even if the MC sliders were on the same positions as before.
we diagnosed the problem. It was related with sticky sliders. After a reboot of C1:IOO the actual output of the DAC does not correspond anymore to the values read on the sliders. In order to update the actual output it is necessary to do a change of the values of the sliders, i.e. jiggling a bit with them.
I've updated the slider twiddle script to include the MC alignment biases. We should run this script whenever we reboot all the hardware, and add any new sticky sliders you find to the end of the script. It's at
I have built a version of the circuit with flying components; the completed circuit is shown in the attached picture. I built the circuit in segments and measured the transfer function after each segment to see whether it matched the LTSpice simulation after each step. The segments are shown in the circuit diagram.
After building the first segment, the measured transfer function looked pretty much the same as the simulated transfer function; it appears shifted in the attached plot, but this is because I didn't do a careful job of tuning at this point, and I'm relatively sure that I could have tuned it to match the simulation. After adding the second segment of the circuit, the measured and simulated transfer functions were similar in shape, but I was unable to increase the frequency of the peaks (through tuning) any more than what is shown in the plot (I could move the peaks so that their frequency was lower, but they are shown as high as they will go). When I added the final segment to complete the circuit, the measured and simulated transfer functions no longer had the same shape; two of the peaks were very close together and I was barely able to differentiate one from the other.
In order to understand what was happening, I tried making modifications to the LTSpice model to recreate the transfer function that was measured. I was able to create a transfer function that closely resembles the measured transfer function in both the circuit as of the 2nd segment and the completed circuit by adding extra inductance and capacitance as shown in red in the circuit diagram. The transfer functions simulated with these parasitic components are shown in red in both plots. While I was able to recreate the response of the circuit, the inductance and capacitance needed to do this were much larger than I would expect to occur naturally within the circuit (2.2uH, 12 pF). I'm not sure what's going on with this.
Trying to track the MC positions back for a few days, it seems that the data hasn't been recorded properly for a while. Something happened yesterday after my boot fest and then the record got restored. Attached here are the readbacks showing the event for MC1.
Is anything wrong with the data record?
While Clara was working on her Wiener filtering and optimizing the locations of the accelerometers, she discovered that MC_L and MC_L_256 are totally flatlined. I looked at them, and it looks like they've been dead since ~9:30pm-ish on Sunday night. Bootfest-type activities shall commence shortly.
Under Alberto's tutalage, I rebooted the whole vme set (iovme, sosvme, susvme1, susvme2), and after that MC_L was all good again.
I added a temporary channel, to input 9 on the PEM ADCU. Beware the 30, 31, and 32 inputs. I tried 32 and it only gave noise.
There managed to be just enough 100 kOhm resistors to stuff all the "2" channels (VERT2, N/S2, E/W2) with the fancy low-noise resistors. The first six channels (VERT 1/2, NS 1/2, EW 1/2) are now completely done with the thin-film resistors, taking into account the changes that were made on the circuit diagram. I also replaced the C8 capacitor with the fancy Garrett ones and added capacitors on top of R4 and R13 (after painstakingly making sure that the capacitances are exactly the same for each pair) for the "2" channels. It looks like the capacitors on the "1" channels are the cheaper ones. I will compare the noise measurements later to see if there is any difference - if so, I can replace those as well (although, we're out of the 1 uF capacitors needed for C8).
Speaking of, we are now out of or very low on several types of the Garrett resistors/capacitors: 1 uF, 1kOhm, 100 Ohm, 14.0 Ohm, and 100 kOhm. I left the specifics on Steve's desk so that more can be ordered for the eventual time when the third set of channels needs to be restuffed.
I came into the 40m to sign things out briefly then swiftly return them, and the alarms were going off on op540m at 1am.
The cat and donkey? were making much noise.
I found the alignment biases for the PRM and the SRM in a funny state. It seemed like they had been "saved" in badly misaligned position, so the restore scripts on the IFO configure screen were not working. I've manually put them into a better alignment.
APC Smart-UPS (uninterruptible power supply) batteries RBC12 replaced at 1Y8 vacuum rack.
Their life span were 22 months.
Before heading back to the 40m to check on the computer situation, I thought I'd check the web screenshots page that Kakeru worked on, and it looks like none of the screens have been updated since June 1st. I don't know what the story is on that one, or how to fix it, but it'd be handy if it were fixed.
Apparently I broke this when I added op540m to the webstatus page. It's fixed now.
I've been trying for most of the week to get noise measurements on the output of the Guralp box as well as scross the AD640 chip. The measurements haven't really been making sense, and, being at a loss as to what else I should try, I decided to redo the resistors on the N/S 2 and E/W 2 channels. (I had been comparing the VERT1 and VERT2 channels, as VERT1 has been restuffed and VERT2 has not.) I don't need all three of the second set of channels to do more measurements, so it seemed like a good use of time.
The first thing I noticed was that the VERT2 channel was missing two resistors (R24 and R25). I probably should have noticed this sooner, as they are right by the output points I had been measuring across, but it didn't occur to me that anyone did anything to the VERT2 channel at all. So, probably the measurements on VERT2 are no good.
Note the existence of 100 kOhm resistors on the top channel, and none on the bottom channel (VERT2).
Then, while I was soldering in some 100 Ohm resistors, I happened to notice that the resistors I was using had a different number (1001) on them than the corresponding ones on the already redone channels (1003). I checked the resistance, and the ones on the already redone channels turned out to be 100 kOhm resistors, rather than 100 Ohms. So, I double checked the circuit diagram to make sure that I had read it correctly, and there were a number of resistors that had been relabeled as 100 Ohms and several relabeled as 100 kOhms. On the board, however, they were ALL 100 kOhms. Clearly, one of them is wrong, and I suspect that it is the circuitboard, but I don't know for sure.
The diagram clearly shows that R6 should be a 100k resistor, while R5 and R8 should be 100 Ohm resistors, but they are all the same (100k) on the board. I suspect this may have something to do with larger-than-expected noise measurements. But, it's possible the diagram is wrong, not the board. In any case, I didn't really know what to do, since I wasn't sure which was right, so I just replaced all the resistors I was sure about and removed the 100k and 100 Ohm resistors without replacing them with anything. Incidentally, the box of 100kOhm resistors seems to be missing, so I wouldn't have been able to finish those anyway.
It's railed. This is what halted locking progess on Monday night, as this channel is used for the offloadMCF script, which slowly feeds back a CARM signal to the ETMs to prevent the VCO from saturating.
Attached is a 5 day trend, which shows that the channel went dead a few days ago. All the channels shown are being collected from the same ICS110B (I think), but only some are dead. It looks like they went dead around the time of the "All computers down" from Sunday.
Attached are the channels being recorded from the ICS110B in 1Y2 (the IOO rack). Channels 12, 13, 16, 17, 22, 24, 25 appear to have gone dead after the computer problems on Sunday.
This has been fixed by one of the two most powerful & useful IFO debugging techniques: rebooting. I keyed the crate in 1Y2.
I made and tested a female-to-female TRS(audio)-RNC cable. It only has a single channel, so it won't work for stereo speakers or anything, but I should only need one speaker for testing the microphones. The tip of the plug is the signal, the sleeve is ground, and the ring is null.
The fb40m just went out of order with status indicator number 8
It recovered on its own five minutes later.
Backup script restarted, backup of trend frames and /cvs/cds is up-to-date.
Since lately the alignment of the input beam to the interferometer has changed, I went checking the alignment of the beam on the photodiodea. They were all fine except for pd9, that is AS DD 199. Here the DC is totally null. The beam seems to go right on the diode but the scope on the PD's DC output shows no power. This is really strange and bad.
After inspecting PD9 with the viewer and the cards, the beam looks like it is aligned to the photodiode althought there is no signal at the DC output of the photodetector. So I checked the spectrum for PD9_i and Q (see attachments) and it seems that those channels are actually seeing the beam. I'm going to check the alignemtn again and see the efefct on the spectra to make sure that the beam is really hitting the PD.
I aligned PD9. Here are the spectra confirming that.
Using FET probes, I was able to measure a transfer function that looks a little more like what I expected. There are only two peaks, but I think this can be explained by a short between the two capacitors (and two tunable capacitors) in the LC pairs, as shown (in red) in the circuit diagram attached. The measured transfer function (black), along with the simulated transfer functions with (red) and without (blue) the short are shown in the attached plot. The measured transfer function doesn't look exactly like the simulated transfer function with the short, but I think the difference can be explained by stray impedances.
Koji recommended that we use the optical setup pictured below. Although it uses fewer optics, I can't think of a way to test the phase camera using this configuration because any modulation of the wavefront with a lens or whatever would be automatically corrected for in the PLL so I think I'll have to stick with the old configuration.
I talked with Zach. So this is just a note for the others.
The setup I suggested was totally equivalent with the setup proposed in the entry http://22.214.171.124:8080/40m/1721, except that the PLL PD sees not only 29.501MHz, but also 1kHz and 59.001MHz. These additional beating are excluded by the PD and the PLL servo. In any case the beating at 1kHz is present at the camera. So if you play with the beamsplitter alignment you will see not only the perfect Gaussian picture, but also distorted picture which is resulted by mismatching of the two wave fronts. That's the fun part!
The point is that you can get an equivalent type of the test with fewer optics and fewer efforts. Particularly, I guess the setup would not be the final goal. So, these features would be nice for you.
Yet again, the DAQAWG flipped out for an unknowable reason. In order of restart activities listed on the Wiki, I keyed the crate and nothing really happened, then I hit the physical reset button and nothing happened, and then I did the 'telnet....vmeBusReset', and a couple minutes later, it was all good again.
Lately, I have been able to externally trigger the camera using a signal generator passing through the op-amp circuit that I built. The op-amp circuit stabilizes the jitter in the sine wave from the signal generator and rectifies the wave. I wrote the calculations into the code allowing me to find the phase and amplitude from the images I take. I still need to develop code that will plot these arrays of phase and amplitude.
The mysterious dark band at the top of the ccd images continues to defy explanation. However, I have found that it only appears for short exposure times even when the lens is completly covered. During the next couple of days, I will try to write a routine to correct for this structure in the dark field.
Koji recommended that we use the optical setup pictured below. This configuration would require fewer optics and we would have to rely on slight misalignments between the carrier and reference beams to test the effectiveness of the phase camera instead of a wavefront-deforming lens.