The results of the AM/PM measurements:
Both the AM and PM TFs were scaled to make them have the same average value. Manually adjusting the delay line offset for each measurement using the oscilloscope was probably not accurate enough and therefore resulted in different scaling which this should somewhat compensate.
The new calibration factor used: 5 MHz/V at the output of the mixer to obtain the frequency modulation and then division by the mod. freq. to obtain PM.
5 MHz/V because changing the PZT voltage by 0.01 V=> change in beat frequency by 0.1 MHz, which was seen as a 20 mV change in the delay line mixer output.
Again, the calibration is not very precise and I will probably repeat this experiment at some point more precisely.
A few years ago, Koji and I setup a delay line phase shifter, which can be used to impart a (switchable) delay to a signal path. Since we talked about reviving the fast (= high bandwidth) ALS control scheme at the meeting, I reminded myself of the infrastructure available.
For a beat note in the regime 10-100 MHz, we should have plenty of range in this module to add a delay such that we zero one quadrature of the ALS DFD output (for a linear error signal).
I then proceeded to connect the single-ended front panel BNC corresponding to the ALS_X_I DFD channel to the IN2 input of the CM board (this would be what we use for high bandwidth ALS feedback). The conventional ALS system uses the differential output from a rear-panel D-sub, so in principle, both systems could run in parallel. I confirmed that I could see a signal when the IN2 path on the CM board was engaged (monitored using ndscope at the CM_Slow output), and that this signal stabilized when the green laser was locked to the X-arm length, which itself was slaved to the PSL frequency using the usual POX locking scheme. I have not yet routed the LO leg of the ALS_X beat through the delay line phase shifter - see next elog for details.
Update about the ALS MEDM screen slider: the trick was to change the OMSL field of the C1:LSC-BO_1_0 channel to "closed_loop" instead of "supervisory". Once this is done, the DOL value of the same channel can be set to the soft channel C1:ALS-DelayCalc, which sets the 16 bit binary string that controls the delay. Because arbitrary delays are not possible, I think it's more natural for the user to interact with this 16-bit binary string rather than the actual delay itself. So the MEDM screen has been slightly modified from what is shown in Attachment #1.
[Meenakshi, Gautam, Shruti]
- We initially aligned the arm cavities to get the green lasers locked to them. For the X arm cavity, we tweaked the ITMX and ETMX pitch and yaw and toggled the X green shutter until we saw something like a TEM00 mode on the monitor and a reasonable transmitted power.
- With the LSC servo enabled, the IR light also became resonant with the cavities.
- Then we measured the noise in different configurations. Attachment 1 shows the the ALS OOL (in the IR beat signal) noise with the arms locked inidividually via PDH.
The script for plotting the ALS beat frequency noise is:
In the process of setting up some cabling at 1Y2, I must've bumped a cable to the c1lsc expansion chassis. Anyways, the c1lsc models crashed. I ran the reboot script around 530pm PDT. Usual locking behavior was recovered after this. The work at 1Y2 was:
The IN2 to CM board was already connected to I single ended output of the ALS X demodulator. The ~100 Hz UGF digital locking using the CM_SLOW path is straightforward but I didn't have any success with the AO path tonight. I wonder how high BW this lock can be made without injecting a ton of noise into the IMC loop, given that the EX uPDH only has ~ 10 kHz UGF.
Attachment #1 shows the spectra of the ALS signal
Attachment #2 is an OLTF measurement.
Even though we were not able to lock the the IR beat (by enabling LSC) during the day possibly because of increased seismic activity, we tried to the measure the ALS beat frequency noise by changing the PDH side-band frequency to different values.
I tried choosing values that corresponded to the peaks in the PM/AM as found in elog:15206 but for some reason unknown to us the cavity did not lock between 700-800 kHz.
The three attachments have data for different sideband frequencies:
Attachment 1: 819.472 kHz (peak in PM/AM, measured around noon)
Attachment 2: 225.642 kHz (peak in PM/AM, measured earlier in the morning)
Attachment 3: 693.500 kHz (not a peak in PM/AM)
We don't think these plots mean much and will do the measurement at some quieter time more systematically.
While doing the experiment, the ITMY pitch actuation was changed from -2.302 to -2.3172V because it locked better.
The ITMX, ETMX alignment was also tweaked to try to lock with different sideband frequencies and then restored to the values that were found earlier in the morning.
Unlikely, the alignment was probably just not good. I restored the alignment and now the arms can be locked to IR frequency.
Even though we were not able to lock the the IR beat (by enabling LSC) during the day possibly because of increased seismic activity
We proceeded with the trying to measure the ALS out-of-loop noise of the X arm when the X arm cavity is green-locked using different PDH sideband frequencies.
Before doing the experiment, Koji helped us with getting the arm cavities locked in IR using LSC (length) and ASC (angular).
With the arms locked in IR and green, we repeated the same measurements as before at different sideband frequencies (Refer Attachment 1 - label in Hz). We did not optimize the phase nor did we look at the PDH error signal today which is possibvly why we did not see an improvement in the noise. We will look into this possibly tomorrow.
Could you please put physical units on the Y-axis and also put labels in the legend which give a physical description of what each trace is?
It would also be good to a separate plot which has the IR locking signal and the green locking signal along with this out of loop noise, all in the same units so that w can see what the ratio is.
Once we adjust the phase we may be able to increase the servo gain for optimal locking. Unless it may be a good idea to increase the gain without optimizing the phase?
Check out this elog: ELOG 4354
If this summing box is still used as is, it is probably giving the demod phase adjustment.
In order to adjust the relative phase for PDH locking, we used the Siglent SDG 1032X function generator which has two outputs whose relative phase can be adjusted.
This Siglent function generator was borrowed from Yehonathan's setup near the PSL table and can be found at the X end disconnected from our setup after our use.
Initially, we used the Siglent at 231.250 kHz and 5 Vpp from each output with zero relative phase to lock the green arm cavity. By moving the phase at intervals of 5deg and looking at the PDH error signals when the cavity was unlocked we concluded that 0deg probably looked like it had the largest linear region (~1.9 V on the yaxis. Refer elog 15218 for more information) as expected.
Then we tried the same for 225.642 kHz, 5 Vpp, and found the optimal demod phase to be -55deg, with linear region of ~3 V (Ref. Attachment 2). A 'bad' frequency 180 kHz was optimized to 10deg and linear region of ~1.5 V.
The error signals at higher frequencies appeared to be quite low (not sure why at the moment) and tuning the phase did not seem to help this much.
For the noise measurement, the IFO arms were locked to IR and green, but even after optimizing the transmission with dither, we couldn't achieve best locking (green transmission was around ~0.2). Further, the IMC went out of lock during the experiment after which Koji helped us by adjusting the gains a locking point of the PMC servo. Attachment 1 contains some noise curves for the 3 frequencies with a reference from an earlier 'good' time.
to make the comparisons meaningfully
one needs to correct for the feedback changes
There was some UNELOGGED work at EX today. The DFD outputs were also hijacked for loss measurement. Unclear who the culprit was, but there is now a broad noise bump centered around ~180 Hz in the ALS X noise curve, which certainly wasn't there yesterday. Maybe let's keep the few working systems working, it is annoying to have to deal with these auxiliary issues every night. I'll push ahead with locking, hopefully the ALS noise is "good enough".
It appears that the EY green steering PZTs have somehow lost their bipolar actuation range. I will check on them the next time I go to the lab for an N2 switch.
Could be that the power outage busted something in the drive electronics.
I went to EY and saw that the HV power supply was only putting out 50 V and had hit the current limit of 10 mA (nominally, it should be 100 V, drawing ~7mA). This is definitely a problem that has come up after the power shutdown event, as when I re-energized the HV power supply at EY, I had confirmed that it was putting out the nominal values (the supply was not labelled with these nominal numbers so I had to label it). Or maybe I broke it while running the dither alignment tests yesterday, even though I never drove the PZTs above 50 Hz with more than 1000cts (= 300 mV * gain 5 in the HV amplifier = 1.5 V ) amplitude.
The problem was confirmed to be with the M2 PZT (YAW channel) and not the electronics by driving the M2 PZT with the M1 channels. Separately, the M1 PZT could be driven by the M2 channels. I also measured the capacitance of the YAW channels and found it to be nearly twice (~7 uF) of the expected 3 uF - this particular PZT is different from the three others in use by the ASX and ASY system, it is an older vintage, so maybe it just failed? 😔
I don't want to leave 100 V on in this state, so the HV supply at EY was turned off. Good GTRY was recovered by manual alignment of the mirror mounts. If someone has a spare PZT, we can replace it, but for now, we just have to live with manually aligning the green beam often.
Yes, we are supposed to have a few spare PI PZTs.
The apparent increase in the ALS noise (witnessed in-loop, e.g. Attachment #2 here) during the CARM offset reduction may have an optomechanical origin.
Update 415pm 5/6: Per the discussion at the meeting, I have now uploaded as Attachment #2 the force-->displacement (i.e. m/N) transfer functions. I now think these are appropriate units. For the ALS case, we could convert the m/N to Hz/N of extra frequency noise imprinted on the AUX laser due to the increased cavity motion. Is W/N really better here, since the mechanism is extra frequency noise on a beatnote, and there isn't really a PDH or DC error signal?
I started my attempt on noise budgeting of ALS by going back to how Kiwamu did it and adding as many sources as I could find up till now. This calculation is present in ALS_Noise_Budget notebook. I intend to collect data for noise sources and all future work on ALS in the ALS repo.
The noise budget runs simulink through matlab.engine inside python and remaining calculations including the pygwinc ones are done in python. Please point out any errors that I might have done here. I still need to add noise due to DFD and the ADC after it. For the residual frequency noise of AUX laser, I have currently used an upper limit of 1kHz/rt Hz at 10 Hz free-running frequency noise of an NPRO laser.
I've added 4 proposed schemes for implementing ALS in voyager. Major thing to figure out is what AUX laser would be and how we would compare the different PSL and AUX lasers to create an error signal for ALS. Everywhere below, 2um would mean wavelengths near 2 um including the proposed 2128nm. Since it is not fixed, I'm using a categorical name. Same is the case for 1um which actually would mean half of whatever 2 um carries.
I finally managed to install a differential-receiving whitening board in 1Y2 - 4 channels are available at the moment. As I claimed, one stage of 15:150 Hz z:p whitening does improve the ALS noise a little, see Attachment #1. While the RMS (from 1kHz-0.5 Hz) does go down by ~10 Hz, this isn't really going to make any dramatic improvement to the 40m lock acquisiton. Now we're really sitting on the unsuppressed EX laser noise above ~30 Hz. This measurement was taken with the arm cavities locked with POX/POY, and end lasers locked to the arm cavities with uPDH boxes as usual. This was just a test to confirm my suspicion, the whitening board is to be used for the air BHD channels, but when we get a few more stuffed, we can install it for the ALS channels too.
No, there should be no unscheduled visits from any inspector, marshal, tech, or vendor. They all have to be escorted or they don't get in. If they have a problem with that, please give them my cell #.
For the ALS, in addition to the beat note spectrum, I think we need to know the loop gain use to feedback to the ETM to determine the true cavity length fluctuation. w/o ALS, the noise would be only due to the seismic noise, OSEM damping noise, and the IR-PDH residual. Those are all suppressed by the ALS loop, but then the ALS loop puts its sensing noise onto the cavity. So, if I'm thinking about this right, the ALS beat noise > 200 Hz doesn't matter so much to the CARM RMS. So the whitening seems to be doing good in the right spot, but we would like to have another boost in the green PDH to up the gain below ~300 Hz?
I found out an error I did in copying some control model values from Kiwamu's matlab code. On fixing those, we get a considerably reduced amount of total noise. However, there was still an unstable region around the unity gain frequency because of a very small phase margin. Attachment 3 shows the noise budget, ALS open-loop transfer function, and AUX PDH open-loop transfer function with ALS disengaged. Attachment 4 is the yaml file containing all required zpk values for the control model used. Note that the noise budget shows out-of-loop residual arm length fluctuations with respect to PSL frequency. The RMS curve on this plot is integrated for the shown frequency region.
Adding two more poles at 100 Hz in the ALS digital filter seems to work in making the ALS loop stable everywhere and additionally provides a steeper roll-off after 100 Hz. Attachment 1 shows the noise budget, ALS open-loop transfer function, and AUX PDH open-loop transfer function with ALS disengaged. Attachment 2 is the yaml file containing all required zpk values for the control model used. Note that the noise budget shows out-of-loop residual arm length fluctuations with respect to PSL frequency. The RMS curve on this plot is integrated for the shown frequency region.
But is it really more stable?
For that, we'll have to take present noise source estimates but Gautum vaguely confirmed that this looked more realistic now 'shape-wise'. If I remember correctly, he mentioned that we currently can achieve 8 pm of residual rms motion in the arm cavity with respect to the PSL frequency. So we might be overestimating our loop's capability or underestimating some noise source. More feedback on this welcome and required.
The code used to calculate the transfer functions and plot them is in the repo 40m/ALS/noiseBudget
Attachment 5 here shows a block diagram for the control loop model used. Output port 'Res_Disp' is used for referring all the noise sources at the residual arm length fluctuation in the noise budget. The open-loop transfer function for ALS is calculated by -(ALS_DAC->ALS_Out1 / ALS_DAC->ALS_Out2) (removing the -1 negative feedback by putting in the negative sign.) While the AUX PDH open-loop transfer function is calculated by python controls package with simple series cascading of all the loop elements.
## Cavity Pole
I think the digital loop in the ALS budget is too optimistic. You have to include all the digital delays and anti-aliasing filters to get the real response.
aslo, I recommend grabbing some of the actual spectra from the in-lock times with nds and using the calibrated spectra as inputs to this mode. Although we don't have good models of the stack, you can sort of infer it by using the calibrated seismometer data and the calibrated MC_F or MC_L channels (for IMC) or XARM/YARM signals for those.
This is not a reply to comments given to the last post; Still working on incorporating those suggestions.
Rana suggested looking first at what needs to be suppressed and then create a filter suited for the noise from scratch. So I discarded all earlier poles and zeros and just kept the resonant gains in the digital filter. With that, I found that all we need is three poles at 1 Hz and a gain of 8.1e5 gives the lowest RMS noise value I could get.
Now there can be some practical reasons unknown to me because of which this filter is not possible, but I just wanted to put it here as I'll add the actual noise spectra into this model now.
This ALS loop is not stable. Its one of those traps that comes from using only the Bode plot to estimate the loop stability. You have to also look at the time domain response - you can look at my feedback lecture for the SURF students for some functions.
Yes, that loop was unstable. I started using the time domain response to check for the stability of loops now. I have been able to improve the filter slightly with more suppression below 20 Hz but still poor phase margin as before. This removes the lower frequency region bump due to seismic noise. The RMS noise improved only slightly with the bump near UGF still the main contributor to the noise.
For inclusion of real spectra, time delays and the anti-aliasing filters, I still need some more information.
Related Elog post with more details: 40m/15587
I used D1400293 to get the latest logged details about the universal PDH box used to lock the green laser at X end. The uPDH_X_boost.fil file present there was used to obtain the control model for this box. See attachment one for the code used. Since there is a variable gain stage in the box, I tuned the gain of the filter model F_AUX in ALS_controls.yml to get the maximum phase margin in the PDH lock of the green laser. Unity gain frequency of 8.3 kHz can be achieved in this loop and as Gautam pointed out earlier, it can't be increased much further without changes in the box.
The ALS control model remains stable with a reduction in total estimate noise because of the above update. There are few things to change though:
For all the loops where we drive the NPRO PZT, there is some notch/resonance feature due to the PZT mechanical resonance. In the IMC loop this limits the PZT/EOM crossove to be less than 25 kHz. I don't have a model for this, btu it should be included.
If you hunt through the elogs, people have measured the TF of ALS NPRO PZT to phase/frequency. Probably there's also a measured ALS PDH loop somewhere that you could use to verify your model.
The only two PZT Phase modulation transfer function measurements I could find are 40m/15206 and 40m/12077. Both these measurements were made to find a good modulation frequency and do not go below 50 kHz. So I don't think these will help us. We'll have to do a frequency transfer function measurement at lower frequencies.
I'm still looking for ALS PDH loop measurements to verify the model. I found this 40m/15059 but it is only near the UGF. The UGF measured here though looks very similar to the model prediction. A bit older measurement in 2017 was this 40m/13238 where I assume by ALS OLTF gautum meant the green laser PDH OLTF. It had similar UGF but the model I have has more phase lag, probably because of a 31.5 kHz pole which comes at U7 through the input low pass coupling through R28, C20 and R29 (See D1400293)
If the green laser is not being used, can I go and take some of these measurements myself?
Koji recommended that I can add whitening filters to suppress ADC noise easily. I added a filter before ADC in ALS loop with 4 zeros at 1.5 Hz and 4 poles at 100 Hz and added a reversed filter in the digital filter of ALS. This did not change the performance of the loop but significantly reduced the contribution of ADC noise above 1 Hz. One can see ALS_controls.yaml for the filter description. Please let me know if this does not make sense or there is something that I have overlooked.
Now, the dominant noise source is DFD noise below 100 Hz and green laser frequency noise above that. For DFD noise, I used data dating back to Kiwamu's paper. The noise contribution from DFD in the model is lower than the latest measured ALS noise budget post on elog. I'll look further into design details and noise of DFD.
Code, data, and schematics
I want to get back to locking the interferometer so I can test out the newly installed AS WFS. However, the ALS noise is far too high, at least the transition of arm length control from IR to ALS fails reliably with the same settings that worked so reliably previously. I worked on investigating it a bit today.
In the latter half of last year, I was focused on the air-BHD setup, so I wasn't checking in on the ALS noise as regularly.
All tests are done with the arm cavity length locked to the PSL frequency using POX. Then, the EX laser is locked to the arm cavity length using the AUX PDH servo. The fluctuations in the beatnote between the two lasers is what is monitored as a diagnostic. See Attachment #1. The reference traces in the top pane are from a known good time. The large excess noise between ~80-200 Hz is what I'm concerned about.
A separate issue that can improve the noise is to track down the noise in the 20-80 Hz band - probably some IMC frequency noise issue.
See Attachment #2.
So what could it be? The only things I can think of are (i) the beat mouth photodiode (NF1611) or (ii) excess noise in the fiber carrying the light from EX to the PSL table (but only on this fiber, and not on the EY fiber). Both seem remote to me - I'll test the former by switching the EX and EY fiber inputs to the beat mouth, but apart from this, I'm out of ideas...
To avoid this kind of issue, we should really have scripted locks of all the basic IFO configs and record the data to summary pages or something - maybe something to do once Guardian is installed, it'd be pretty hacky to do cleanly with shell scripts.
I'm also wondering why the error monitors for the X and Y loops report such wildly different spectra for the suppressed frequency noise of the AUX laser relative to the cavity length, see Attachment #1. The y-axis should be approximately Hz/rtHz. In both cases, the servo's error point monitor is connected to the DAQ system via a G=10 SR560. With the SR785, I measure for EX a nice bucket-shaped spectrum, bottoming out at ~10 uV/rtHz around 40 Hz, see Attachment #2. The SR560 should have an input-referred noise much less than this at the G=10 setting. The ADC noise level is only ~5 uV/rtHz, and indeed, the EY spectrum shows the expected shape. So what's up with the EX error mon? Tried swapping out the SR560 for a different unit, no change. And both the SR560 noise, and the ADC noise, check out when everything is checked individually. So some kind of interaction once everything is connected together, but it's only present at EX...
Today, I tried switching the EX and EY fibers going into the beat mouth, but I preserved the channel mapping after the beat mouth by switching the electrical outputs as well (the goal was to make sure that the beat photodiodes weren't the issue here, I think the electronics are already exonerated since driving the channel with a Marconi doesn't produce these noisy features). The EX spectrum remains noisy. I've switched everything back to the nominal configuration now to avoid further confusion. So it would appeat that this is real frequency noise that gets added in the EX fiber somehow. What can I do to fix this? The source of coupling isn't at the PSL table, else the EY channel would also show similar features. Visually, nothing seems wrong to me at EX either. So the problem is somehow in the cable tray along which the 40m of fiber is routed? This is already inside some nice foam/tubing setup, what can be done to improve it? Still doesn't explain why it suddenly became noisy...
How about resurrecting the PSL table green beat for the X arm to see if the non-fiber setup shows the same level of the freq noise (e.g. the PDH locking became super noisy due to misalignment etc).
I thought about it, but wouldn't that show up at the AUX PDH error point? Or because the loop gain is so high there we wouldn't see a small excess? I suppose there could be some clipping on the Faraday or something like that. But the GTRX level and the green REFL DC level when locked are nominal.
If the sensing noise level of the end PDH degraded for some reason, it'd make the out of loop stability worse without making the end pdh error level degraded.
It's just speculation.
I did this test today. The excess noise around 100 Hz doesn't show up in the green beat.
See Attachment #1. The setup was as usual:
So, this excess appears to truly be excess phase noise on the fiber (though I have no idea what the actual mechanicsm could be or what changed between Aug and Oct 2020 that could explain it. Maybe the HEPA?
For this work, I had to spend some time aligning the two green beams onto the beat photodiode. During this time, I shuttered the PSL, disabled feedback via the FSS servo, turned the HEPA high, and kept the EX green locked to the arm so as to have a somewhat stable beat signal I could maximize. Everything has been returned to nominal settings now (obviously, since I locked the arms to get the data).
You may ask, why do we care. In terms of RMS frequency noise, it would appear that this excess shouldn't matter. But in all my trials so far, I've been unable to transition control of the arm cavity lengths from POX/POY to ALS. I suppose we could try using the green beat, but that has excess low frequency noise (which was the whole point of the fiber coupled setup).
I see no evidence of anything radically different from my PSL table optical characterization in the IMC transmitted beam, see Attachment #1. The lines are just a quick indicator of what's what and no sophisticated peak fitting has been done yet (so the apparent offset between the transmission peaks and some of the vertical lines are just artefacts of my rough calibration I believe). The modulation depths recovered from this scan are in good agreement with what I report in the linked elog, ~0.19 for f1 and ~0.24 for f2. On the bright side, the ALS just worked and didn't require any electronics fudgery from me. So the mystery continues.
Here is our first attempt at a single-arm noise budget for ALS.
Attachment 1 shows the loop diagram we used to calculate the contribution of different noises.
Attachment 2 shows the measured noise at C1:ALS-BEATX_PHASE_FINE_OUT_HZ when XARM was locked to the main laser and Xend Green laser was locked to XARM.
Here's an updated X ARM ALS noise budget.
I attempted a single arm actuation calibration using IR beatnote (in the directions of soCal idea for DARM calibration)
Rana suggested in today's meeting to put in a notch filter in the XARM IR PDH loop to avoid suppressing the excitation line. We tried this today first with just one notch at 1069 Hz and then with an additional notch at 619 Hz and sent two simultaneous excitations.
We attempted to simulate "oscillator based realtime calibration noise monitoring" in offline analysis with python. This helped us in finding about a factor of sqrt(2) that we were missing earlier in 16171. we measured C1:ALS-BEATX_FINE_PHASE_OUT_HZ_DQ when X-ARM was locked to main laser and Xend green laser was locked to XARM. An excitation signal of amplitude 600 was setn at 619 hz at C1:ITMX_LSC_EXC.
We got a value of . The calibration factor in use is from nm/cts from 13984.
Next steps could be to budget this noise while we setup some way of having this calibration factor generated in realitime using oscillators on a FE model. Calibrating actuation of a single optic in a single arm is easy, so this is a good test setup for getting a noise budget of this calibration method.
We did a quick check to make sure there is no saturation in the C1:SUS-ITMX_LSC_EXC analog path. For this, we looked at the SOS driver output monitors from the 1X4 chassis near MC2 on a scope. We found that even with 600 x 10 = 6000 counts of our 619 Hz excitation these outputs in particular are not saturating (highest mon signal was UL coil with 5.2 Vpp). In comparison, the calibration trials we have done before had 600 counts of amplitude, so we can safely increase our oscillator strength by that much
Things that remain to be investigated -->
I did this analysis again by just doing demodulation go 5s time segments of the 60s excitation signal. The major difference is that I was not summing up the sine-cosine multiplied signals, so the error associated was a lot more. If I simply multpy the whole beatnote signal with digital LO created at excitation frequency, divide it up in 12 segments of 5 s each, sum them up individually, then take the mean and standard deviation, I get the answer as:
as opposed to that was calculated using MICH signal earlier by gautum in 13984.
Attachment 1 shows the scatter plot for the complex calibration factors found for the 12 segments.
My aim in the previous post was however to get a time series of the complex calibration factor from which I can take a noise spectral density measurement of the calibration. I'll still look into how I can do that. I'll have to add a low pass filter to integrate the signal. Then the noise spectrum up to the low pass pole frequency would be available. But what would this noise spectrum really mean? I still have to think a bit about it. I'll put another post soon.
It was known that the Y end ALS PZTs are not working. But Anchal reported in the meeting that the X end PZTs are not working too.
We went down to the X arm in the afternoon and checked the status. The HV (KEPCO) was off from the mechanical switch. I don't know this KEPCO has the function to shutdown the switch at the power glitch or not.
But anyway the power switch was engaged. We also saw a large amount of misalignment of the X end green. The alignment was manually adjusted. Anchal was able to reach ~0.4 Green TRX, but no more. He claimed that it was ~0.8.
We tried to tweak the SHG temp from 36.4. We found that the TRX had the (local) maximum of ~0.48 at 37.1 degC. This is the new setpoint right now.
We replaced a shutter driver for GRY since it stopped working this morning.
We replaced it with a free driver which was sitting on the ITMY table.
The reverse polarity issue of C1:AUX-GREEN_Y_Shutter was fixed by switching one of the switches of the driver from N.O. to N.C.
Also, "Toggle" button was added to IFO_ALIGN.adl so that we can toggle shutters easily to find TEM00. It runs /home/controls/Git/40m/scripts/ALS/ShutterToggler.py.
The green Y shutter now works but in reverese, meaning that sending 1 to C1:AUX-GREEN_Y_Shutter closes the shutter and vice versa. This needs to be fixed.
We took the long BNC cable that ran from ETMX to ETMY and ran it from ETMX into the control room instead. This way Cici and Deeksha can send small voltage signals to the AUX PZT and read back using the beatnote pickoff that is usually connected to the spectrum analyzer.
I took ~ 7 minutes of XALS beatnote data with the XAUX laser locked to the XARM cavity, and the XARM locked to PSL to develop an allan deviation estimator. The resulting timeseries for the channel C1:ALS-BEATX_FINE_PHASE_OUT_HZ_DQ (decimated timeseries in Attachment #1) was turned into an allan variance using the "overlapped variable tau estimator":
Where represents the k-th data point in the raw timeseries, and are the variable integration intervals under which two point variances are computed (the allan variance is a special case of M-point variance, where M=2). Then, the allan deviation is just the square root of that. Attachment #2 shows the fractional deviation (normalized by the mean beat frequency ~ 3 MHz for this measurement) for 100 integration times spanning the full duration (~ 7 min = 420 s).
The code used for this lives in Git/40m/labutils/measuremens/ALS/
If this estimate is any good, wherever the fractional beatnote deviation reaches a minimum value can be used as a proxy for the longest averaging time that give a statistical increase in SNR. After this timescale, the frequency comparison is usually taken over by "environmental instabilities" which I don't think I can comment further on. In our particular estimate, the 100 second integration gives a fractional deviation of ~ 0.44 %, or absolute deviation of 12.925 kHz.
what's the reasoning behind using df/f_beat instead of df/f_laser ?
I took ~ 7 minutes of XALS beatnote data with the XAUX laser locked to the XARM cavity, and the XARM locked to PSL to develop an allan deviation estimator.
I guess it didn't make sense since f_beat can be arbitrarily moved, but the beat is taken around the PSL freq ~ 281.73 THz. Attachment #1 shows the overlapping tau allan deviation for the exact same dataset but using the python package allantools, where this time I used the PSL freq as the base frequency. This time, I can see the minimum fractional deviation of 1.33e-13 happening at ~ 20 seconds.
The allan variance is related to the beatnote spectral density as a mean-square integral (the deviation is then like the rms) with a sinc window.