Clearly this "riff raff" is referring to me. It won't help today I guess but there is one each on the carts holding the SR785 (currently both in the office/electronics bench area), and the only other unit available in the lab is connected to a Prologix box on the Marconi inside the PSL enclosure.
The Prologix GPIB-ethernet dongle needs +8-13 V to run. Some riff raff has removed the adapter and I was thunderstruck to see that it had not been returned.
The electronics chain used to drive the three elements of the PI PZT on which a mirror is mounted with the intention of controlling the LO phase has been changed, to now use the Trek Mode603 power amplifier instead of the OMC high voltage driver. Attachment #1 shows the new configuration.
The text of Attachment #1 contains most of the details. The main requirement was to map the DAC output voltage range, to something appropriate for the Trek amplifier. The latter applies a 50V/V gain to the signal received on its input pin, and also provides a voltage monitor output which I hooked up to an ADC channel in c1ioo. The gain of the interfacing electronics was chosen to map the full output range of the DAC (-5 to +5 V for a single-ended receiving config in which one pin is always grounded) to 0-2.5 V at the input of the Trek amplifier, so that the effective high voltage drive range is 0-125 V. I don't know what the damage threshold is for the PI PZT, maybe we can go higher. The only recommendation given in the Trek manual is to not exceed +/-12 V on the input jack, so I have configured D2000396 to have a supply voltage of 11.5 V, so that in the event of electronics failure, we still don't exceed this number.
On the electronics bench, I tested the drive chain, and also measured the transfer function, see Attachment #2. Seems reasonable (the Trek amplifier was driving a 3uF capacitive load used to protect the SR785 measurement device from any high voltage, hence the roll-off). The gain of D2000396 was changed from 1/8 to 1/4 after I realized that the DAC full range is only +/- 5 V when the receiving device is single-ended at both input and output. Maybe the next iteration of this curcuit should have differential sending, to preserve the range.
To test the chain, I used the single bounce beam from the ITM, and interfered it with the LO. Clear fringing due to the seismic motion of the ITM (and also LO phase noise) is visible. In this configuration, I drove the PZT mirror in the LO path at a higher frequency, hoping to see the phase modulation in the DCPD output. However, I saw no signal, even when driving the PZT with 50% of the full DAC range. The voltage monitor ADC channel is reporting that the voltage is faithfully being sent to the PZTs, and I measured the capacitance of the PZTs (looked okay), so not sure what is going on here. Needs more investigation.
Update Aug 30 5pm: Turns out the problem here was a flaky elbow connector I used to pipe the high voltage to the PI PZT, it had some kind of flaky contact in it which meant the HV wasn't actually making it to the PZT. I rectified this and was immediately able to see the signal. Played around with the dark fringe Michelson for a while, trying to lock the homodyne phase by generating a dither line, but had no success with a simple loop shape. Probably needs more tuning of the servo shape (some boosts, notches etc) and also the dither/demod settings themselves (frequency, amplitude, post mixer LPF etc). At least the setup can now be worked on interferometrically.
Increasing the compensation capacitance (470 pF now instead of 33 pF) seems to have fixed the oscillation issues associated with this circuit. However, the measured noise is in excess of the model at almost any frequency of relevance. I believe the problem is due to the way the measurement is done, and that we should re-do the measurement once the unit is packaged in a shielded environment.
Attachment #1 shows (schematically) the measurement setup. Main differences from the way I did the last round of testing are:
Attachment #2 shows the measurement results:
I didn't capture the data, but viewing the high voltage output on an Oscilloscope threw up no red flags - the oscillations which were previously so evident were nowhere to be seen, so I think the capacitor switch did the trick as far as stability is concerned.
There is a large excess between measurement and model out to a few kHz, if this is really what ends up going to the suspension then this circuit is useless. However, I suspect at least part of the problem is due to close proximity to switching power supplies, judging by the comb of ~10 Hz spaced peaks. This is a frequent problem in coil driver noise measurements - previously, the culprit was a switching power supply to the Prologix GPIB box, but now a Linear AC-DC converter is used (besides, disconnecting it had no visible effect). The bench supplies providing power to the board, however, is a switching supply, maybe that is to blame? I think the KEPCO supplies providing +/-250 V are linear. I tried the usual voodoo of twisting the wires used to receive the signal, moving the SR785 away from the circuit board etc, but these measures had no visible effect either.
The real requirement of this circuit is that the current noise above 100 Hz be <1pA/rtHz. This measurement suggests a level that is 5x too high. But the problem is likely measurement related. I think we can only make a more informed conclusion after shielding the circuit better and conducting the test in a more electromagnetically quiet environment.
Using a heterodyne measurement setup to track both quadratures, I estimated the relative phase fluctuation between the LO field and the interferometer output field. It may be that a single PZT to control the homodyne phase provides insufficient actuation range. I'll also need to think about a good sensing scheme for controlling the homodyne phase, given that it goes through ~3 fringes/sec - I didn't have any success with the double demodulation scheme in my (admittedly limited) trials.
For everything in this elog, the system under study was a simple Michelson (PRM, SRM and ETMs misaligned) locked on the dark fringe.
This work was mainly motivated by my observation of rapid fringing on the BHD photodiodes with MICH locked on the dark fringe. The seismic-y appearance of these fringes reminded me that there are two tip-tilt suspensions (SR2, SR3), one SOS (SRM) + various steering optics on seismic stacks (6+ steering mirrors) between the dark port of the beamsplitter and the AS table, where the BHD readout resides. These suspensions modulate the phase of the output field of course. So even though the Michelson phase is tightly controlled by our LSC feedback loop, the field seen by the homodyne readout has additional phase noise due to these optics (this will be a problem for in vacuum BHD too, the question is whether we have sufficient actuator range to compensate).
To get a feel for how much relative phase noise there is between the LO field and the interferometer output field (this is the metric of interest), I decided to set up a heterodyne readout so that I can simultaneously monitor two orthogonal quadratures.
Attachment #1 shows the detailed measurement setup. I hijacked the ADC channels normally used by the DCPDs (along with the front-end whitening) to record these time-series.
Attachments #2, #3 shows the results in the time domain. The demodulated signal isn't very strong despite my pre-amplification of the PDA10CF output by a ZFL-500-HLN, but I think for the purposes of this measurement, there is sufficient SNR.
This would suggest that there are pretty huge (~200um) relative phase excursions between the LO and IFO fields. I suppose, over minutes, it is reasonable that the fiber length changes by 100um or so? If true, we'd need some actuator that has much more range to control the homodyne phasethan the single PZT we have available right now. Maybe some kind of thermal actuator on the fiber length? If there is some pre-packaged product available, that'd be best, making one from scratch may be a whole project in itself. Attachment #3 is just a zoomed-in version of the time series, showing the fringing more clearly.
Attachment #4 has the same information as Attachment #2, except it is in the frequency domain. The FFT length was 30 seconds. The features between ~1-3 Hz support my hypothesis that the SR2/SR3 suspensions are a dominant source of relative phase noise between LO and IFO fields at those frequencies. I guess we could infer something about the acoustic pickup in the fibers from the other peaks.
To be continued tomorrow. I think it's a good idea to let the newly installed fiber relax into some sort of stable configuration overnight.
After replacement of the fiber delivering the LO beam to the airBHD setup (some photos here), I repeated the measurement outlined here. There may be some improvement, but overall, conclusions don't change much.
The main addition I made was to implement a digital phase tracker servo (a la ALS), to make sure my arctan2 usage wasn't completely bonkers (the CDS block can be deleted later, or maybe it's useful to keep it, we will see). I didn't measure it today, but the UGF of said servo should be >100 Hz so the attached spectrum should be valid below that (loop has not been done, so above the UGF, the control signal isn't a valid representative of the free running noise). Attachment #1 shows the result. The 1 Hz and 3 Hz suspension resonances are well resolved. Anyways, what this means is that the earlier result was not crazy. I don't know what to make of the high frequency lines, but my guess is that they are electronic pickup from the Sorensens - I'm using clip-mini-grabbers to digitize these signals, and other electronics in that rack (e.g. ALS signals) also show these lines.
It is pretty easy to keep the simple Michelson locked for several minutes. Attachment #2 shows the phase-tracker servo output over several minutes. The y-axis units are degrees. If this is to be believed, the relative phase between the two fields is drifting by 12um ove an hour. This is significantly lower than my previous measurement, while the noise in the ~0.5-10 Hz band is similar, so maybe the shorter fiber patch cable did some good?
I think there is also correlation between the PSL table temperature, but of course, the evidence is weak, and there are certainly other effects at play. At first, I thought the abrupt jumps are artefacts, but they don't actually represent jumps >360 degrees over successive samples, so maybe they are indicative of some real jump in the relative phase? Either fiber slippage or TT suspension jumps? I'll double check with the offline data to make sure it's not some artefact of the phase tracker servo. If you disagree with these conclusions and think there is some meaurement/analysis/interpretation error, I'd love to hear about it.
I have left the heterodyne electronics setup at the LSC rack, but it is not powered (because there are some exposed wires). Please leave it as is.
Over the last couple of days, I've been working towards getting the infrastructure ready to test out the scheme of sensing (and eventually, controlling) the homodyne phase using the so-called RF44 scheme. More details will be populated, just quick notes for now before I forget.
I edited /diskless/root.jessie/home/controls/.bashrc so that I don't have to keep doing this every time I do a model recompile.
Where is this variable set and how can I add the new paths to it?
Attachment #1 shows the optical setup currently being used to send the LO field with RF sidebands on it to the air BHD setup.
Attachment #2 shows spectra of the relative phase drift between LO and IFO output field (from the Dark Michelson).
Attachment #3 shows the signal magntiude of the signals used to make the spectra in Attachment #2, during the observation time (10 minutes) with which the spectra were computed. The dashed vertical lines denote the 1%, 50% and 99% quantiles.
Attempts to close a feeddback loop to control the homodyne phase:
The PMC has been unlocked since September 11 sometime (summary pages are flaky again). I re-locked it just now. I didn't mess with the HEPA settings for now as I'm not using the IFO at the moment, so everything should be running in the configuration reported here. The particulate count numbers (both 0.3um and 0.5um) reported is ~x5-8 of what was reported on Thursday, September 10, after the HEPA filters were turned on. We don't have logging of this information in any automated way so it's hard to correlate things with the conditions in Pasadena. We also don't have a working gauge of the pressure of the vacuum envelope.
The RGA scanning was NOT enabled for whatever reason after the vacuum work. I re-enabled it, and opened VM1 to expose the RGA to the main volume. The unit may still be warming up but this initial scan doesn't look completely crazy compared to the reference trace which is supposedly from a normal time based on my elog scanning (the timestamp is inherited from the c0rga machine whose clock is a bit off).
Update 1500: I checked the particle count on the PSL table and it barely registers on the unit (between 0-20 between multiple trials) so I don't know if we need a better particle coutner or if there is negligible danger of damage to optics due to particulate matter.
The unit was repaired and returned to the 40m. Now, with a DMM, I measure a DC offset value that is ~1% of the AC signal amplitude. I measured the TF of a simple 1/20 voltage divider and it looks fine. In FFT mode, the high frequency noise floor levels out around 5-7nV/rtHz when the input is terminated in 50 ohms.
I will upload the repair documents to the wiki.
The "source" output of the SR785 has a DC offset of -6.66 V. I couldn't make this up.
These were delivered to the 40m today and are on Rana's desk
I'll order a couple of these (5 ordered for delivery on Wednesday) in case there's a hot demand for the jack / plug combo that this one has.
There was an abrupt change in the MC_F spectrum between August 4 and August 5, judging by the summary pages - the 1 and 3 Hz resonances are no longer visible in the spectrum. Possibly, this indicates some electronics failure on the MC servo board / whitening board, the CDS settings don't seem to have changed. There is no record of any activity in the elog around those dates that would explain such a change. I'll poke around at 1X2 to see if anything looks different.
Update 1740: I found that the MCL / MCF cables were disconnected. So since August 5, these channels were NOT recording any physical quantity. Because their inputs weren't terminated, I guess this isn't a clean measurement of the whitening + AA noise, but particularly for MC_F, I guess we could use more whitening (see Attachment #1). Probably also means that the wandering ~10-30Hz line in the spectrogram is a electronics feature. The connections have now been restored and things look nominal again.
After more trials, I think the phase tracker part used to provide the error signal for this scheme needs some modification for this servo to work.
Attachment #1 shows a block diagram of the control scheme.
I was using the "standard" phase tracker part used in our ALS model - but unlike the ALS case, the magnitude of the RF signal is squished to (nearly) zero by the servo. But the phase tracker, which is responsible for keeping the error signal in one (demodulated) quadrature (since our servo is a SISO system) has a UGF that is dependent on the magnitude of the RF signal. So, I think what is happening here is that the "plant" we are trying to control is substantially different in the acquisition phase (where the RF signal magnitude is large) and once the lock is enabled (where the RF signal magnitude becomes comparitively tiny).
I believe this can be fixed by dynamically normalizing the gain of the digital phase tracking loop by the magnitude of the signal = sqrt(I^2 + Q^2). I have made a modified CDS block that I think will do the job but am opting against a model reboot tonight - I will try this in the daytime tomorrow.
I'm also wondering how to confirm that the loop is doing something good - any ideas for an out-of-loop monitor? I suppose I could use the DCPD - once the homodyne phase loop is successfully engaged, I should be able to drive a line in MICH and check for drift by comparing line heights in the DCPD signal and RF signal. This will requrie some modification of the wiring arrangement at 1Y2 but shouldn't be too difficult...
The HEPAs, on the PSL table and near ITMY, were dialled down / turned off respectively, at ~8pm at the start of this work. They will be returned to their previous states before I leave the lab tonight.
This elog suggests that there is uniformly 1 stage engaged across all channels. I didn't look at the board to see what the jumper situation is, but only 1 stage of whitening is compensated digitally for both _F and _L. The Pomona box attached to the NPRO PZT input is also compensated digitally to convert counts to frequency.
I tried the gain re-allocation between VCO gain and FSS COMM (and also compensated for the cts to Hz conversion in MCF), but it doesn't seem to have the desired effect on the MCF SNR in the 5-50Hz band. Since the IMC stays locked, and I had already made the changes to mcup, I'll keep these gains for now. We can revert to the old settings if the IMC locking duty cycle is affected. Explicitly, the changes made were:
VCO gain: +7dB ---> +13 dB
FSS COMM: +6 ddB ---> +0 dB
The mcdown script wasn't modified, so the lock acquisition gains are the same as they've been.
the "Pentek" whitening board that carries the MC channels has jumpers to enable either 1 or 2 stages of 15:150 whitening. Looks lik MC_F has 2 and MC_L has 1.
I had to make a CDS change to the c1lsc model in an effort to get a few more signals into the models. Rather than risk requiring hard reboots (typcially my experience if I try to restart a model), I opted for the more deterministic scripted reboot, at the expense of spending ~20mins to get everything back up and running.
Update 2230: this was more complicated than expected - a nuclear reboot was necessary but now everything is back online and functioning as expected. While all the CDS indicators were green when I wrote this up at ~1800, the c1sus model was having frequent CPU overflows (execution time > 60 us). Not sure why this happened, or why a hard power reboot of everything fixed it, but I'm not delving into this.
The point of all this was that I can now simultaneously digitize 4 channels - 2 DCPDs, and 2 demodulated quadratures of an RF signal.
I don't think the proposed scheme for sensing and controlling the homodyne phase will work without some re-thinking of the scheme. I'll try and explain my thinking here and someone can correct me if I've made a fatal flaw in the reasoning somewhere.
Field spectrum cartoon:
Attachment #1 shows a cartoon of the various field components.
So is there a 90 degree relative shift between the signal quadrature in the simple Michelson vs the DRFPMI? But wait, there are more problems...
Closing a feedback loop using the 44 MHz signal:
We still need to sense the 44 MHz signal with a photodiode, acquire the signal into our CDS system, and close a feedback loop.
I don't have any bright ideas at the moment - anyone has any suggestions?🤔
I wanted to check what kind of signal the photodiode sees when only the LO field is incident on the photodiode. So with the IFO field blocked, I connected the PDA10CF to the Agilent analyzer in "Spectrum" mode, through a DC block. The result is shown in Attachment #2. To calculate the PM/AM ratio, I assumed a modulation depth of 0.2. The RIN was calculated by dividing the spectrum by the DC value of the PDA10CF output, which was ~1V DC. The frequencies are a little bit off from the true modulation frequencies because (i) I didn't sync the AG4395 to a Rb 10 MHz signal, and (ii) the span/BW ratio was set rather coarsely at 3kHz.
I would expect only 44 MHz and 66 MHz peaks, from the interference between the 11 MHz and 55 MHz sideband fields, all other field products are supposed to cancel out (or are in orthogonal quadratures). This is most definitely not what I see - is this level of RIN normal and consistent with past characterization? I've got no history in this particular measurement.
While I stopped by the lab this morning to pick up some things, I took the opportunity to continue the recovery.
At some point, we should run the suspension eigenmode routine (kick optics, let them ringdown, measure peak locations and Qs) to confirm that the remaining suspensions are okay, will also help in actuation re-allocation efforts on ETMY. But I didn't do this today.
Leaving the lab at 1150.
See Attachment #1.
The beam spot on ETMY looks weird (looks almost like a TEM10 mode), but the one on ITMY seems fine, see Attachment #2. Wonder what's going on there, maybe just a focusing problem?
We need a vent to fix the suspension, but until then what we can do is to redistribute the POS/PIT/YAW actuations to the three coils.
The HEPA filters on the PSL enclosure are no longer running. I tried cycling the power switch on the NW corner of the enclosure, and also turned the dial on the Variac down to zero and back up to maximum, no effect. Judging by the indicator LED on it, the power is making it to the Variac - bypassing the Variac and directly connecting the HEPA to the AC power seems to have no effect either.
I can't be sure, but I'm almost certain I heard them running at max speed an hour ago while I was walking around inside the VEA. Probably any damage that can happen has already been done, but I dialled down the Innolight injection current, and closed its shutter till the issue can be resolved.
I got some feedback from Koji who pointed out that the phase tracker is not required here. This situation is similar to the phase locking of two lasers together, which we frequently do, except in that case, we usually we offset the absolute frequencies of the two lasers by some RF frequency, and we demodulate the resulting RF beatnote to use as an error signal. We can usually acquire the lock by simply engaging an integrator (ignoring the fact that if we actuate on the laser PZT, which is a frequency actuator, just a proportional feedback will be sufficient because of the phase->frequency conversion), the idea being that the error signal is frequently going through a zero-crossing (around which the sinusoidal error signal is approximately linear) and we can just "catch" one of these zero crossings, provided we don't run of actuation range.
So the question here becomes, is the RF44 signal a suitable error signal such that we can close a feedback loop in a similar way? To try and get more insight, I tried to work out the situation analytically. I've attached my thinking as a PDF note. I get some pretty messy complicated expressions for the RF44 signal contributions, so it's likely I've made a mistake (though Mathematica did most of the heavy lifting), it'll benefit from a second set of eyes.
Anyways, I definitely think there is some additional complications than my simple field cartoon from the preceeding elog would imply - the relative phases of the sidebands seem to have an effect, and I still think the lack of the PRC/SRC make the situation different from what Hang/Teng et. al. outlined for the A+ homodyne phase control analysis. Before the HEPA failed, I had tried closing the feedback loop using one quadrature of the demodulated RF44 signal, but had no success with even a simple integrator as the loop (which the experience with the PLL locking says should be sufficient, and pretty easily closed once we see a sinusoidally oscillating demodulated error signal). But maybe I'm overlooking something basic conceptually?
The interlocks tripped at ~630am local time. Jordan reported that TP2 was supposedly running at 52 C (!).
V1 was already closed, but TP2 was still running. With him standing by the rack, I remotely exectued the following sequence:
Jordan confirmed (by hand) that TP2 was indeed hot and this is not just some serial readback issue. I'll do the forensics later.
I was in the lab from 1630-1830. I have located and visually inspected all the parts required for the magnet regluing / optic cleaning parts of the planned vent, except the fresh batches of scpectroscopic grade solvents. I was in the cleanroom part of the clean and bake lab from 1630-1700.
The HEPAs work again. After running the HEPAs for ~1 hour, I checked the particle count on the PSL table - the meter registered 0 for both 0.3 um and 0.5 um. So I decided to turn the NPRO back on, at ~1730 local time. The PMC and IMC were readily locke, so the basic interferometer functionality is returned, and we can now go ahead with (i) vent prep (ii) air BHD tests and (iii) IMC debuggin as was discussed on the call today. The earth is shaking, but nothing serious so far, I will resume alignment of the interferometer later in the evening when hopefully things have calmed down a bit more...
Note that the many other issues Koji noted in the preceeding elog (e.g. flaky wiring) have not been addressed.
Chub kindly provided us with an electronic anemometer. With the meter held directly against the HEPA filter inside the enclosure, we measured ~700 cfm of airflow on each of the two HEPAs, with the Variac set to 100% and the HEPAs themselves set to "High". With the Variac at 50%, the flow drops to ~160 cfm. At the nominal setting of 33%, the meter didn't register any flow. I don't know what the spec'd flow rate is for this combination of blower + filter, but Jordan says similar units in Downs register ~1500 cfm at the "High" setting. The two protable (similarly sized) HEPA units in the 40m, one at ITMY and one at ETMY, register ~900 cfm and ~1100 cfm respectively, when set to high. So we may want to revisit what the "nominal" HEPA setting should be, in case the filters have become clogged over time.
Some photos of the HEPA blowers with the pre-filters off and the capacitors switched out may be found here.
The optomechanics stock in the lab was in a sad state. We have obtained the following from Thorlabs in the last two months:
I have used some of these for the ari BHD setup. The unused items are stored in the shelves that house the optomechanics ~halfway down the east arm. I'm wondering what's a good setup to document the stock of this stuff so we can always have a healthy stock of optomechanics (at least the non speciality ones like posts, spacers etc). It sucks to realize at 0000hrs that you're missing a 3mm shim or 250mm converging lens or something.
Attachment #1 shows that the ITMX, ETMY and beamsplitter Oplev light levels have decayed significantly from their values when installed. In particular, the ETMY and ITMX sum channels are now only 50% of the values when a new HeNe was installed. ELOG search revealed that ITMY and ETMX HeNes were replaced with newly acquired units in March and September of last year respectively. The ITMX oplev was also replaced in March 2019, but the replacement was a unit that was being used to illuminate our tourist attraction glass fiber at EX.
We should replace these before any vent as they are a useful diagnostic for the DC alignement reference.
2x500 ml bottles of spectroscopic grade isopropanol were delivered. I marked them with today's date and placed them in the solvent cabinet. In the process, my shoulder bumped the laser interlock switch by the door to the VEA in the drill press area, which turned the PSL NPRO off. I turned it back on just now. The other NPROs are not connected to the interlock and so were unaffected.
Attachment #1 shows that the c1rfm model isn't able to receive any signals from the front end machines at EX and EY. Attachment #2 shows that the problem appears to have started at ~430am today morning - I certainly wasn't doing anything with the IFO at that time.
I don't know what kind of error this is - what does it mean that the receiving model shows errors but the sender shows no errors? It is not a new kind of error, and the solution in the past has been a series of model reboots, but it'd be nice if we could fix such issues because it eats up a lot of time to reboot all the vertex machines. There is no diagnostic information available in all the places I looked. I'll ask the CDS group for help, but I'm not sure if they'll have anything useful since this RFM technology has been retired at the sites (?).
In the meantime, arm cavity locking in the usual way isn't possible since we don't have the trigger signals from the arm cavity transmission.
Update 1500 4 Oct: soft reboots of models didn't do the trick so I had to resort to hard reboots of all FEs/expansion chassis. Now the signals seem to be okay.
After the earthquake on September 19 2020, it looks to me like the only lasting damage to suspensions in vacuum is the ETMY UR magnet being knocked off.
Suspension ringdown tests:
I did the usual suspension kicking/ringdown test:
Attachment #2 also includes info about the matrix diagonalization, and the condition numbers of the resulting matrices are as large as ~30 for some suspensions, but I think this isn't a new feature.
The simple interferometer, composed of a single bounce reflection from ITMY and the LO beam deilvered via fiber to the AS table, can be locked - i.e. the phase of the LO beam can be controlled such that the DC light level on the DCPDs after the two beams are interfered can be stabilized. This test allows us to confirm that various parts of the sensing and actuation chain (e.g. PI PZT for homodyne phase control, Trek amplifier etc etc) are working.
I will post more quantitative analysis tomorrow.
Attachment #2 shows the servo topology.
We received 20pcs of stuffed demodulator boards from Screaming Circuits today. Some caveats:
I removed 1 from the group to stuff some components that weren't sent to Screaming Circuits and test the functionality on the benchtop, the remaining have been stored in a plastic box for now as shown in Attachment #1. The box has been delivered to Chub who will stuff the remaining 19 boards once I've tested the one piece.
This looks cool, we should have something similar, can be really useful.
These boxes were moved from the 40m hallway to the inside of the VEA so that we have some space to walk around. You can find some pictures here.
Attachment #1: spectra of the phase noise between LO and IFO output fields sensed using the RF44 signal.
Closing a feedback loop:
We cleared up some space in the 1Y1 electronics rack to install the 3 new FE machines. I removed the current driver and laser from 1Y1, they are now stored in the E10 cabinet. I will upload some photos to gPhotos soon.
Per Koji's suggestion, I turned the PSL HEPA Variac to 50% just now, so that the power load through the burnt electrical cable is reduced by 75%.
Summary of discussion between Koji and gautam on 14 Oct:
I tried all of these last night / overnight, here are my findings.
Analog locking of the homodyne phase:
See Attachment #1.
Relative stability of two IFR2023s synchronized to the same FS725 Rb standard:
The electrical LO signal for demodulation of the 44 MHz photocurrent is provided by an IFR2023 signal generator. To maintain a fixed phase relation between this signal, and the phase modulation sidebands imprinted on the interferometer light via a separate IFO2023 signal generator, I synchronize both to the same Rb timing standard (a 10 MHz signal from the FS725 is sent to the rear panel frequency standard input on the IFR). We don't have a direct 44 MHz electrical signal available from the main IFO Marconi at the LSC rack (or anywhere else for that matter). So I decided to do this test at 55 MHz.
A look at the time domain signal:
With the Michelson locked on the dark fringe, the RF44 I and Q signals in the time domain are shown in Attachment #3 for a 1 minute stretch.
I packaged the HV coil driver into a 2U chassis, hoping for better shielding from pickup. There is still considerable excess noise in measurement vs model around 100 Hz, see Attachment #1. The projected displacement noise from this noise contribution is shown in Attachment #2 - I've also plotted the contribution from the 4.5kohm (planned value for fast path series resistance) for comparison. Attachment #3 has some photos of the measurement setup so if someone sees some red flags, please let me know.
I've run out of ideas to try and make the measurement cleaner - the presence of the rather prominent power line harmonics suggests that this is still not perfect, but what more shielding can we implement? I have to make the measurement on the circuit side of the 25 kohm series resistor, so I am using some Pomona minigrabbers to clip onto the leg of the wirewound resistor (see photos in Attachment #3), so that's not great maybe, but what's the alternative?
So if this is truly the noise of the circuit, then while it's an improvement on the current situaiton, it's unsatisfying that such a simple circuit can't match the design expectations. But how do we want to proceed?
It's not so easy to directly measure this I think, because the filtering is rather aggressive. Attachment #1 shows the measured transfer function (dots) vs the model and Attachment #2 shows the noise. I think this checks out - but I can't definitively rule out some excess noise at 100 Hz from this stage. Because the gain of the HV stage is x31, we'd need a preamp with better than 1nV/rtHz to directly measure the noise I guess. The Acromag noise model in Attachment #2 is based on a measurement I describe here.
what is the noise level before the HV stage? i.e. how well is the acromag noise being filtered?
Apart from the questionable wiring on the Acromags, one other important difference is in the way the connections were made between the old VME crates to the Eurocrate backplanes, and how we do it now. The thick cables had their sheilds connected to the eurocrate ground (or at least, there was a dedicated ground lug on those cables which we screwed on to the ground terminals on the Eurocrate backplanes). However, in our current configuration, we interface the Acromag ADCs and DACs to the backplane via these adaptor boards. The shields of the DSUB cables are presumably NOT connected to the Eurocrate grounds. This should also be investigated as one potential cause of the grounding issue - while on some of the Eurocrate modules, the P1/P2 connectors may have either the "A" or "C" row of connectors shorted to ground, some may not, and the TTFSS may suffer from such an issue?
Note that we have this problem in all of the slow machines that were upgraded to Acromag (if this turns out to be the issue).
In fact, the problem was the grounding issue (presumably on the IOO racks).
I wanted to look into the ISS situation. Some weeks ago, I found the PD that was previously used as the in-loop photodiode. I wanted to use this and measure the open-loop RIN at a few places (to see if there's any variation and also to check its functionality). However, I didn't get very far tonight - for a start, the PD height is 3" (while our beam height is 4" everywhere outside the vacuum), and I needed to put together a circuit to supply the 5V bias and +/- 15 V since the transimpedance is done on the head. I was only able to do a low-level functionality test tonight, checked that the DC voltage output varied linearly with the incident power (calibrated against an NF1611 photodiode, data will be put up later). I didn't get to measuring any noise performance - is an incandescent light bulb still shot noise limited at ~10 Hz < f < 10kHz? Some notes:
Unconnected to this work - this problem reared its ugly head again (i noticed it yesterday morning already actually). I don't have the energy to embark on a fix tonight, Koji is going to be in the lab all day tomorrow and so he will fix it.
Ok I was using the PD in the black mount because Rana recommended it a few weeks ago.
Regarding the M2ISS, I acquired the hardware from QIL some months ago, including a circuit board, and 2 PDs. These had LEMO outputs though (not BNC), and the mounts are not 4". These photodiodes are what I'm using as the airBHD DCPDs right now, and some photos are here - are these the photodiodes you mentioned? Or are there yet more M2ISS photodiodes? I remember Johannes had some custom mounts extruded to make them 4" high, do you mean those? Can I retrieve them his Cryo setup?
BTW, my elog scraping shows only one spectra from Stefan in the ATF elog, and the performance there is more like 1e-7/rtHz @ 100 Hz, and that’s using a dedicated high BW servo circuit, not the SR560. Am I just missing the measurement of 2e-8/rtHz?
that little PD in the black mount was never very good. The AD829 is not a good opamp for transimpedance and especially not good for low frequencies. Stefan Ballmer and I were able to get 2e-8 out of these (@100 Hz) many years ago.
I wonder if we have some of Zach's M2ISS photodetectors around, perhaps in QIL or Cryo. I doubt that any of them are in use now. Those had good performance nad BNC output.
If everything else looks good, I'll start letting the dry N2 into the main volume after lunch.
Now the green transmissions are visible by the green PDs. Attachment 9 shows the trans and ref of each green beams with and without locking to TEM00. The questionable green TRY was ~0.3. If we compare this with the histrical data (Attachment 10), it is about 1/4 of the value in the past. It's not too crazy but still quite low.
BTW, nice video! @ Koji, How difficult was it to edit it into this form?
We are now ready to take the doors off. I've already done the basic prep work (loosened bolts, cleaned chamber, carts for tools, fresh ameristat on portable HEPAs etc).
[koji, rana, gautam]
This morning, we did the following;
The OSEMs remain in the EY vacuum chamber. The next set of steps are:
We will most likely work on this tomorrow. At ~1615, I briefly opened the PSL shutter and tweaked the IMC alignment. We will almost certainly change the pointing into the IMC when we remove the old OMC and rebalance that table, so care should be taken when working on that...
To be a bit more clear about what we are going to do in the OMC chamber, I marked-up some photos, see Attachments #1 and #2.
I anticipate that after this work, the only components on the table will be
Are we in agreement with this plan?
See #15656 for the updated photo
Good point - looking back, I also see that I already removed the mirror at the SW corner of the table in 2016. Revised photo in Attachment #1. There is an optic on the east edge of this table whose purpose I'm not sure of, but I'm pretty sure it isn't essential to the main functionality and so can be removed.
I believe the mirror next to IM1 is for the green beams to be delivered to the PSL table. I think we still want to keep it. Otherwise, the plan looks fine.
I got a call from Calum ~830am today saying some facilities people entered the lab, opened the south entrance door, and tripped the alarm in the process. I came to the lab shortly after and was able to reset the alarm by flipping the switch on the alarm box at the south end entrance to "Alarm OFF". Then, I double checked that the door is closed, and re-enabled the alarm. The particle count at the SP table is not unusually high and the lasers (Oplev HeNe and AUX X) were still on, so doesn't look like any lasting damage was done. The facilities people were apparently wearing laser safety goggles.
The IMC isn't resonant for a TEM00 mode at the time of writing - we are waiting for the stack to relax, at which point if the IMC isn't resonant for a TEM00 mode, we will tweak the input pointing into the IMC (we want to use the suspended cavity as the reference, since it is presumably more reliable than the table from which we removed ~50 kgs of weight and shifted the balance.
I am working on the setup of a CDS FE, so please do not attempt any remote login to the IPMI interface of c1bhd until I'm done.