Aaron complained to me earlier that the PMC could not be locked. Turned out to be a classic sticky slider problem, I keyed the c1psl VME crate, and did the usual burtrestore trick. After that, I could immediately lock the PMC and IMC with the nominal gain settings.
I also looked at the wiring at the rack. An SLP250 was installed at the mixer IF output, in parallel with a 50 ohm terminator to ground. I removed this, because as Aaron pointed out, the PMC servo card "FP1 TEST" input is already 50 ohm, and has two cascaded LC filter stages immediately after to filter out the 2f component, so the extra low-pass filtering is superfluous (in any case, 250 MHz is much too high a cutoff to be using for cutting out the 2f component which will be at ~70 MHz).
Finally, in the last ~2 weeks, we have been running with the PMC servo gain of +17 (as opposed to +18 from before). The old gain is too high, as noted by Milind. But the MEDM field for this gain goes RED unless the gain is +18. I adjusted the value of the C1:PSL-STAT_PMC_NOM_GAIN channel to +17, so that this is no longer the case. I also edited the PMC MEDM screen to get rid of my comment that the "SLOW ADC IS DEAD" for the PMC TRANS field, since I have now hooked up the PMC trans photodiode to our temporary Acromag box.
For the beam spot position tracking, I am wondering if there is any benefit to going for a wider field of view and getting the OSEMs in the frame? It may provide some "anchor points" against which whatever algorithm can calibrate the spot position against. But there are also several point scatterers visible in the current view, and perhaps the Gaussiam beam profile moving over them and tracking the scattered intensity from these point scatterers serves the same function? I don't know of a good solution to have a "switchable" field of view configuration in the already cramped camera enclosure though.
Also, I think it may be useful to have a cron job take a picture of MC2 and archive it (once a week? or daily?) to have some long term diagnostic of how the scattered light received by the camera changes over several months.
The GigE is focused now and I have closed the lid. I'm attaching a picture of the MC2 beam spot, captured using GigE at an exposure time of 400µs
Now that the IMC is remaining locked for extended periods of time, the next problem to attack is the ASS dither alignment system. For a start, I decided to try and get the POX and POY locking working again, as we have not fully recovered the interferometer alignment after the most recent pumpdown. I spent a couple of hours tweaking the alignment of the arm cavity mirrors, BS, and TTs to try and recover the maximum possible TRX and TRY - however, my best efforts only yielded TRX~0.8, TRY~0.75. Moreover, the beam axis is such that the spot is significantly off in YAW on both ETMs, as evidenced by the camera views (also true but less obvious on the ITMs). However, trying to bring the beam back to the center of the optics yields TRY and TRX values lower than the above reported maxima. The EX green beam is currently unavailable to verify the arm cavity alignment because of my hijacking the EX NPROs PZT control for PLL investigations, but with the Y arm, I'm able to lock a TEM00 mode. Probably just needs more careful systematic alignment, but I'm not pursuing this tonight.
After a more systematic alignment effort, I was able to get the spots better centered on the optics (judged by eye from the analog camera views). TRY ~0.7, TRX~1.15. The X-arm dither alignment system seems to work out-of-the-box with the existing settings, I was able to run it and maximize the X-arm transmission.
Other work: I also cleaned up the area around MC2 a litte - laptop from on top of the vacuum chamber was removed and a rogue ethernet cable was also removed. The resulted in some misalignment of the IMC, which I corrected by manual alignment. Now the IMC is locked again with nominal transmission levels.
On the PSL table, I re-routed the RF output from the BeatMouth to the regular IR-ALS electronics chain (it was hijacked for PLL investigations). At EX, I disconnected the cable running from the LB1005 to the EX NPRO laser PZT (again was being used for PLL locking), and re-connected the output from the Green uPDH box to allow for some ALS tests to be done. I could then lock the EX green beam to the X-arm, and achieved GTRY ~ 0.35 using the ASX system. More to follow on ALS tests later today.
To practise the dither alignment servo tuning, I decided to make the ASX system work again (mainly because it has fewer DoFs and so I thought it'd be easier to manage). Setup is: dither PZT mirrors on EX table-->demodulate green transmission at the dither frequencies-->Servo the error signals to 0 by an integrator.
The adjusted demod phases, servo gains were saved to the .snap file which gets called when we run the "DITHER ON" script. Also updated the StripTool template.
I plan to repeat similar characterization on the IR dither alignment servos. I think the tuning of the ASS settings can be done independently of figuring out the mystery of why the TRY level is so low.
We crossed off another couple of bullets today.
It took me ~1 hour to realize that c1susaux requries the running of sudo /sbin/ifup eth0 to be run in order to see the martian network - why???
I don't understand the exact chain of causation, but during this work, the fast c1sus model crashed. I had to go through a few iterations of the scripted vertex machine rebooting, but things seem to be back in a normal state now, see Attachment #2. Should probably run the IFO test suite to make sure everything is a-okay, but for now, I am able to lock the IMC so I'm moving on.
The main task remaining here is to take new pictures of everything and upload to the wiki. Also, need to update the Sorensen labels to reflect their current values, some of them are outdated.
Since the work earlier this morning, the fast c1sus model has crashed ~5 times. Tried rebooting vertex FEs using the reboot script a few times, but the problem is persisting. I'm opting to do the full hard reboot of the 3 vertex FEs to resolve this problem.
Judging by Attachment #1, the processes have been stable overnight.
In preparation for the ASS debugging, I decided to check out the beam path on the EY table. In order to be able to do this, I had to setup the POY locking to trigger on AS110 instead of TRY (as is usual for this kind of debugging). Then I could poke an IR card in the beam path without destroying the lock.
There are two irides in the beam path immediately between the vacuum window and the harmonic separator that splits off the IR and green beams. I found that the beam was in fact getting clipped on both of them. It was also somewhat off center on a 2" beamsplitter that sends half of the light to the QPD (currently decommissioned). The purpose of these irides are (I think) to eliminate some ghost reflections of the green beam and also the Oplev beam. I opened up the irides until I felt that there wasn't any more clipping of the IR beam, but the appropriate ghost beams were still getting caught.
I also re-aligned the beam onto the TRY Thorlabs PD so as to better center it on the active area. In summary, the result of this work was that the TRY level went from ~0.6 to ~0.93. There may still be some scope for optimizing this - I tried running the Y-arm ASS scripts, and already, the loops don't run away any more. I'll do the systematic analysis of the servo anyways. But given that the IMC Trans lev el used to be ~15,500 counts and is now ~14,500 counts, I think ~7% drop in TRY level is in line with what we "expect" (assuming the pre-power-degradation TRY level was 1.000).
Note that these irides were installed (I think) by Yuki, and so cannot explain the ASS anomalies of July 2018 (i.e. it does not exonerate in-vacuum clipping of the beam, as Koji had already verified that the in-air path was clean back then).
A question was raised as to how much passive filtering we benefit from if we pick off the local oscillator beam for BHD from the PRC. I did some simplified modeling of this. For the expected range of arm cavity round trip losses (20-50 ppm), I think that the 40m CARM pole will be between 75-85 Hz. The corresponding recycling gain will be 40-50, with the current PRM. I assumed 1000 ppm loss inside the PRC. The net result is that, assuming the single pole coupled cavity response, we will get ~8-9 dB of filtering at ~200 Hz of the intensity noise of the input laser field to the interferometer if we pick the LO beam off from the PRC (e.g. PR2 transmission), instead of picking it off before.
The next questions are: (i) can we do a sufficiently good job of achieving the required RIN stability on the LO field for BHD without relying on the passive filtering action of the PRC? and (ii) is the benefit of the PRC filtering ruined in the process of routing the LO field from wherever the pickoff happens to the BHD setup?
After this activity, the DC bias voltage required on ETMX to restore good X arm cavity alignment has changed by ~1.3 V. Assuming a full actuation range of 30 mrad for +/- 10 V, this implies that the pitch alignment of the stack has changed by ~2 mrad? Or maybe the suspension wires shifted in the standoff grooves by a small amount? This is ~x10 larger than the typical change imparted while working on the table, e.g. during a vent.
Main point is that this kind of range requirement should probably be factored in when thinking about the high-voltage coil driver actuation.
We unstuck ETMX by shaking the stack. Most effective was to apply large periodic human sized force to the north STACIS mounts.
The Y-arm ASS was tuned to be in a workable state. Basically, I followed Koji's recipe.
The SNR of the dither lines in the TRY and YARM control signals were checked - Attachment #1. The dither frequencies are marked with vertical dashed lines (can't figure out how to add 4 cursors in DTT so there's two in each row for a total of 4). A couple of days ago, when I was doing some preliminary checks, I found that the oscillator at 24.91 Hz caused a broadband increase in the TRY noise between DC and ~100 Hz. But today I saw no evidence of such behaviour. So I decided against changing the frequency.
The linearity of the demodulated error signals around the quadratic maxima of the TRY level was checked. I did not, however, investigate in detail the frequency-dependent offset Koji has reported in his elog.
After this work, the TRY level is at 0.95. This is commensurate with the MC trans level being lower by ~7% relative to July 2018. Furthermore, the ASS servo is able to return to TRY~0.95 with a time-constant of ~5 seconds in response to misalignment of the cavity optics. After I investigate the X-arm ASS, I will reset the normalization for TRX and TRY.
Update 645pm: In the spirit of general IFO recovery, I re-centered the ITM and ETM oplev spots, and also the IR beam on the IPPOS QPD to mark the new input pointing alignment (the spot is slightly lower on the AS camera than what I remember). I then tweaked the XARM transmission to maximize it, and re-set the TransMon normalization. I edited the normalization script to comment out the normalizing of the TransMon QPD gains as the QPDs are in some kind of indeterminate state now. Attachment #2 shows the current status, you can also see the normalization being reset. LSC mode disabled for overnight.
Once the XARM ASS is also checked out, I propose moving back to locking the DRMI / PRFPMI configs.
Last documented replacement in Nov 2018, so ~7 months, which I believe is par for the course. I am disconnecting its power supply cable.
There was no green light even though the EX NPRO was on. I checked the doubling oven temperature controller and found that its power cable was loose on the rear. I reconnected it, and now there is green light again.
Arnaud and I moved one of the two spare TT suspensions from the south clean cabinet to the bake lab clean room. The main purpose was to inspect the contents of the packaging. According to the label, this suspension was cleaned to Class A standards, so we tried to be clean while handling it (frocks, gloves, masks etc). We found that the foil wrapping contained one suspension cage, with what looked like all the parts in a semi-assembled state. There were no OSEMs or electronics together with the suspension cage. Pictures were taken and uploaded to gPhoto. Arnaud is going to plan his tests, so in the meantime, this unit has been stored in Cabinet #6 in the bake lab cleanroom.
We noticed that the PRM watchdog was tripping frequently. This is a period of enhanced seismic activity. The reason PRM in particular trips often is because the SIDE OSEM has 5x increased transimpedance. We implemented a workaround by modifying the watchdog tripping condition to scale the SD channel RMS by a factor of 0.2 (relative to the UL and LL channels). We restarted the modbus process on c1susaux and tested that the new logic works. Here is the relevant snippet of code:
# PRM Side is special, see elog 14745
field(DESC,"Tests whether RMS too high")
field(INPA,"C1:SUS-PRM_ULPD_VAR NPP NMS")
field(INPB,"C1:SUS-PRM_PD_MAX_VAR NPP NMS")
field(INPC,"C1:SUS-PRM_LLPD_VAR NPP NMS")
field(INPD,"C1:SUS-PRM_SDPD_VAR NPP NMS")
The db file has a note about this as well so that future debuggers aren't mystified by a factor of 0.2.
I looked into the design of the P2 interface board. The main difficulty here is geometric - we have to somehow accommodate sufficient number of D-sub connectors in the tight space between the two P-type connectors.
I think the least painful option is to stick with Johannes' design for the P1 connector. For the CM board, the P2 connector only uses 6 pairs of conductors for signals. So we can use a D-15 connector instead of 2 D-37 connectors. Then we can change the PCB shape such that the P1 connector can be accommodated (see Attachment #1). The other alternative would be to have 2 P-type connectors and 3 D-subs on the same PCB, but then we have to be extra careful about the relative positioning of the P-type connectors (otherwise they wont fit onto the Eurocrate). So I opted to still have two separate PCBs.
I took a first pass at the design, the files may be found here. I just auto-routed the connections, this is just an electrical feedthrough so I don't think we need to be too concerned about the PCB trace routing? If this looks okay, we should send out the piece for fab ASAP.
I will work on putting together the EPICS server machine (SuperMicro) this afternoon.
2. D040180 / D1500308 Common Mode Board
CM servo board itself doesn't need any modification. The CM board uses P1 and P2. So we need to manufacture a special connector for CM Board P2. (cf The adapter board for P1 T1800260). See also D1700058.
Arnaud has taken 1 TT suspension from the 40m clean lab to Downs for modal testing. Estimated time of return is tomorrow evening.
I looked at the PSL/IOO racks to check for which boards, if any, require an additional P2 interface, so that we can try and design a generic one for the IMC/CM boards and whatever else may require it. While searching the elog, I saw that Koji and Johannes had already done this, see Koji's elog in this thread. Some remarks:
Conclusion: Only the IMC Servo and CM boards need their P2 connectors connected to Acromag.It would be helpful to remove the TTFSS Interface board and figure out what exactly the pin-mapping for the backplane connectors are, but I didn't do this today because there is a "High Voltage" line going to the Interface Board and I'm not actually sure of the signal chain for the FSS servo.
All suspension watchdogs were tripped ~90mins ago. I restored the damping. IMC is locked.
ITMX was stuck. I set it free. But notice that the UL Sensor RMS is higher than the other 4? I thought ITMY UL was problematic, but maybe ITMX has also failed, or maybe it's coincidence? Something for IFOtest to figure out I guess. I don't think there is a cable switch between ITMX/ITMY as when I move the ITMX actuators, the ITMX sensors respond and I can also see the optic moving on the camera.
Took me a while to figure out what's going on because we don't have the seis BLRMS - i moved the usual projector striptool traces to the TV screen for better diagnostic ability.
Update 16 July 1515: Even though the RMS is computed from the slow readback channels, for diagnosis, I looked at the spectra of the fast PD monitoring channels (i.e. *_SENSOR_*) for ITMX - looks like the increased UL RMS is coming from enhanced BR-mode coupling and not of any issues with the whitening switching (which seems to work as advertised, see Attachment #3, where the LL traces are meant to be representative of LL, LR, SD and UR channels).
In addition to c1psl needing a reboot, megatron was un-ssh-able (although it was responding to ping). Clue was that the NPRO PZT control voltage was drifting a lot on the StripTool trace. Koji hard-rebooted the machine. Now IMC is locked, and FSS slow servo is also running.
There were several small/medium earthquakes in Ridgecrest and one medium one in Blackhawk CA at about 2000 UTC (i.e. ~ 2 hours ago), one of which caused BS, ITMY, and ETM watchdogs to trip. I restored the damping just now.
I worked on preparing for the c1iscaux upgrade a bit today.
All the required additional parts should be here by the end of the week - I'd like to aim for Wednesday 7/24 for the installation in 1Y3 and in-situ testing. While talking to Rana, I realized that we should also factor in the c1aux slow channels into this acromag crate - there is no need for a separate machine to handle the shutters and illuminators. But let's not worry about that for now, those channels can simply be added later.
Koji pointed out an important subtlety pertaining to the "LATCH ENABLE" signal line on the CM board. The purpose of this line is to smoothly facilitate the transition of a change in the "multi-bit-binary-outputs", a.k.a. "mbbo", that are controlled by MEDM gain sliders, to the analog electronics on the CM board. Why is this necessary? Imagine changing the gain from 7dB (=0111 in mbbo representation) to 8dB (=1000 in mbbo representation). In order to realize this change, all 4 bits have to change their state. But this almost certainly doesn't happen synchronously, because our EPICS interface isn't synchronous. So at some intermediate times, the mbbo representation could be 0100 (=4dB), or 1111 (=15dB), or many other possible values, which are all significantly different from either the initial value or the desired final state. This is clearly undesirable.
In order to protect against this kind of error, a Latched output part, 74ALS573, is used to buffer the physical digital logic levels from the switches in the analog gain stages. So in the default state, the "LATCH ENABLE" signal line is held "LOW". When a change happens in the EPICS value corresponding to a gain slider, the "LATCH ENABLE" state is quickly toggled to "HIGH", so as to enable the appropriate analog gain stages to be switched, and then again to "LOW", at which point the latch holds its output state. This logic is currently implemented by a piece of code called "latch.o", which is the compiled version of "latch.st", which may be found in /cvs/cds/caltech/target/c1iool0 where it presumably was written for the IMC servo board, but not in /cvs/cds/caltech/target/c1iool0 , which is where the CM board database files reside. The only elog reference I can find pertaining to this particular piece of code is from Alan, and doesn't say anything about the actual logic.
For the new c1iscaux, we need to implement this logic somehow. After discussion between Koji and me, we feel that a piece of python code is sufficient. This would continuously run in the background on the supermicro server machine. The channel hierarchy for each gain channes is as follows (I've taken the example of C1:LSC-CM_REFL1_GAIN):
So the logic will be that it continuously scans the EPICS channel C1:LSC-CM_REFL1_GAIN for a change in set value. When a change is detected, it has to update the C1:LSC-CM_REFL1_SET channel. In the next EPICS refresh cycle, this would result in the mbbo bits, C1:LSC-CM_REFL1_BITS , all changing to the appropriate values. After these changes have happened, we need to toggle the LATCH ENABLE in order to allow the changes to propagate to the analog gain stage switches. Need to think about what's the best way to do this.
I completed the translation of the .db files for the EPICS database records from the VME notation to the Acromag/Modbus/Asyn notation. The channels are now organized into 5 database files, located in /cvs/cds/caltech/target/c1iscaux3/, for convenience:
For reasons unknown to me, the database files in the other Acromag system target directories (e.g. c1susaux, c1auxex) all had 755 level access permission - maybe this is required for systemctl to handle the EPICS serving? Anyways, I upgraded the permission level of the above 5 files using chmod.
There are almost certainly typos / other errors, and I may have missed copying over some soft/calibrated channels, but I hope that this way of grouping by subsystem will make the debugging less painful. Once Chub connects up the power lines to the Acromags, I will run the soft tests. For this purpose, I've also made a C1_ISC-AUX.cmd file and a C1_ISC-AUX.env file in the above target directory, and also made the modbusIOC.service file in /etc/systemd/system on the supermicro.
Now that the .db files were prepared, I wanted to test for errors. So I did the following:
All the Acromags are seen on the 192.168.114 subnet on c1iscaux3 - however, when I run the modbusIOC process, I see various errors in the logfile , so more debugging is required. Nevertheless, progress.
Update 2245: Turns out the errors were indeed due to a copy/paste error - I had changed the IP addresses for the ADCs from the .115 subnet c1susaux was using, but forgot to do so for the DACs and BIOs. Now, if I turn off the existing c1iscaux so that there aren't any EPICS clashes, the EPICS server initializes correctly. There are still some errors in the log file - these pertain to (i) the mbbo notation, which I have to figure out, and (ii) the fact that this version of EPICS, 7.0.1, does not support channel descriptions longer than 28 characters (we have several that exceed this threshold). I think the latter isn't a serious problem.
Getting closer... Note that I turned off the c1iscaux VME crate to prevent any EPICS server clashes. I will turn it back on tomorrow.
Rana and I talked about some (genius) options for the large range DC bias actuation on the SOS, which do not require us to supply high-voltage to the OSEMs from outside the vacuum.
What we came up with (these are pretty vague ideas at the moment):
For the thermal option, I remembered that (exactly a year ago to the day!) when we were doing cavity mode scans, once the heaters were turned on, I needed to apply significant correction to the DC bias voltage to bring the cavity alignment back to normal. The mechanism of this wasn't exactly clear to me - furthermore, we don't have a FLIRcam picture of where the heater radiation patter was centered prior to my re-centering of it on the optic earlier this year, so we don't know what exactly we were heating. Nevertheless, I decided to look at the trend data from that night's work - see Attachment #1. This is a minute trend of some ETMY channels from 0000 UTC on 18 July 2018, for 24 hours. Some remarks:
Also see this elog by Terra.
Attachment #2 shows the results from today's heating. I did 4 steps, which are obvious in the data - I=0.6A, I=0.76A, I=0.9A, and I=1.05A.
In science, one usually tries to implement some kind of interpretation. so as to translate the natural world into meaning.
Bulb replaced. Projector is back on.
The control room UPS started making a beeping noise saying batteries need replacement. I hit the "Test" button and the beeping went away. According to the label on it, the batteries were last repalced in March 2016, so maybe it is time for a replacement, @Chub, please look into this.
For some reason, rossa's Xdisplay won't start up anymore. This happened right after the UPS reset. Koji and I tried ~1.5 hours of debugging, got nowhere.
The database files for C1ISCAUX seem to work file - the exception being the mbbo channels for the CM board.
This was just a software test - the actual functionality of the channels will have to be tested once the Acromag crate has been installed in the rack. One change I had to make on the MEDM screen for the LSC PD whitening gains was to get rid of the "NMS" suffix on the EPICS channel names for whitening gain sliders/drop-down-menus. I suspect this has to do with the EPICS version we are using, 7.0.1. Furthermore, AS165 and POP55 no longer exist - I hold off removing them from the MEDM screen for the moment.
From the software point of view, the major steps are:
I am stopping the EPICS server on the new machine and restarting the old VME crate over the weekend.
What channels are you trying to read?
I'm not able to get trends of the TM adjustment test that Rana had asked us to perform, from the dataviewer. It's throwing the following error:
Connecting to NDS Server fb (TCP port 8088)
Server error 7: connect() failed
datasrv: DataWrite failed: daq_send: Resource temporarily unavailable
T0=19-07-20-01-27-39; Length=600 (s)
No data output.
SnapPy scripts made to work on Pianosa.
Of course rossa was the only machine in the lab that could run the python scripts to interface with the GigE camera. And it is totally bricked now. Lame.
So I installed several packages. The key was to install pypylon - if you go to the basler webpage, pypylon1.4.0 does not offer python2.7 support for x86_64 architecture, so I installed pypylon1.3.0. Here are the relevant lines from the changelog:
gstreamer-plugins-bad-0.10.23-5.el7.x86_64 Sat 20 Jul 2019 11:22:21 AM PDT
gstreamer-plugins-good-0.10.31-13.el7.x86_64 Sat 20 Jul 2019 11:22:11 AM PDT
gstreamer-plugins-ugly-0.10.19-31.el7.x86_64 Sat 20 Jul 2019 11:20:08 AM PDT
gstreamer-python-devel-0.10.22-6.el7.x86_64 Sat 20 Jul 2019 10:34:35 AM PDT
pygtk2-devel-2.24.0-9.el7.x86_64 Sat 20 Jul 2019 10:34:34 AM PDT
pygobject2-devel-2.28.6-11.el7.x86_64 Sat 20 Jul 2019 10:34:33 AM PDT
pygobject2-codegen-2.28.6-11.el7.x86_64 Sat 20 Jul 2019 10:34:33 AM PDT
gstreamer-devel-0.10.36-7.el7.x86_64 Sat 20 Jul 2019 10:34:32 AM PDT
gstreamer-python-0.10.22-6.el7.x86_64 Sat 20 Jul 2019 10:34:31 AM PDT
gtk2-devel-2.24.31-1.el7.x86_64 Sat 20 Jul 2019 10:34:30 AM PDT
libXrandr-devel-1.5.1-2.el7.x86_64 Sat 20 Jul 2019 10:34:28 AM PDT
pango-devel-1.42.4-1.el7.x86_64 Sat 20 Jul 2019 10:34:27 AM PDT
harfbuzz-devel-1.7.5-2.el7.x86_64 Sat 20 Jul 2019 10:34:26 AM PDT
graphite2-devel-1.3.10-1.el7_3.x86_64 Sat 20 Jul 2019 10:34:26 AM PDT
pycairo-devel-1.8.10-8.el7.x86_64 Sat 20 Jul 2019 10:34:25 AM PDT
cairo-devel-1.15.12-3.el7.x86_64 Sat 20 Jul 2019 10:34:25 AM PDT
mesa-libEGL-devel-18.0.5-3.el7.x86_64 Sat 20 Jul 2019 10:34:24 AM PDT
libXi-devel-1.7.9-1.el7.x86_64 Sat 20 Jul 2019 10:34:24 AM PDT
pygtk2-doc-2.24.0-9.el7.noarch Sat 20 Jul 2019 10:34:23 AM PDT
atk-devel-2.28.1-1.el7.x86_64 Sat 20 Jul 2019 10:34:21 AM PDT
libXcursor-devel-1.1.15-1.el7.x86_64 Sat 20 Jul 2019 10:34:20 AM PDT
fribidi-devel-1.0.2-1.el7.x86_64 Sat 20 Jul 2019 10:34:20 AM PDT
pixman-devel-0.34.0-1.el7.x86_64 Sat 20 Jul 2019 10:34:19 AM PDT
libXinerama-devel-1.1.3-2.1.el7.x86_64 Sat 20 Jul 2019 10:34:19 AM PDT
libXcomposite-devel-0.4.4-4.1.el7.x86_64 Sat 20 Jul 2019 10:34:19 AM PDT
libicu-devel-50.1.2-15.el7.x86_64 Sat 20 Jul 2019 10:34:18 AM PDT
gdk-pixbuf2-devel-2.36.12-3.el7.x86_64 Sat 20 Jul 2019 10:34:17 AM PDT
pygobject2-doc-2.28.6-11.el7.x86_64 Sat 20 Jul 2019 10:34:16 AM PDT
pygtk2-codegen-2.24.0-9.el7.x86_64 Sat 20 Jul 2019 10:34:15 AM PDT
Camera server is running on a tmux session on pianosa. But it keeps throwing up some gstreamer warnings/errors, and periodically (~every 20 mins) crashes. Kruthi tells me that this behavior was seen on Rossa as well, so whatever the problem is, doesn't seem to be because I missed out on installing some packages on pianosa. Moreover, if the server is in fact running, I am able to take a snapshot - but the camera client does not run.
The boards arrived. I soldered on a DIN96 connector, and tested that the goemetry will work. It does . The only constraint is that the P2 interface board has to be installed before the P1 interface is installed. Next step is to confirm that the pin-mapping is correct. The pin mapping from the DIN96 connector to the DB15 was also verified.
*Maybe it isn't obvious from the picture, but there shouldn't be any space constraint even with the DB37/DB15 cables connected to the respective adapter boards.
Can you please be more specific about what the error is? Is this the usual instability with the camera server code? Or was it something new?
The camera server is throwing an error and is not grabbing snapshots :(
DATED, SEE ELOG14941 for the most up-to-date info on latch.py.
I wrote (/cvs/cds/caltech/target/c1iscaux3/latch.py) and tested the logic illustrated in Attachment #1. Results of a test are shown in Attachment #2, the various channels change as expected. Note that for negative values of the gain channel, the corresponding "BITS" channel will take on values like 65536 - this is because the mbboDirect data type is a 16 bit data type, and presumably the MSB is the sign bit. A bit mask is applied to this channel before the actual BIO unit bits are set - we should verify that the correct behavior happens, but I don't immediately see any problems.
To me, this is a robust logic, but it will benefit from more sets of eyes giving it a look over. The idea is to run this continuously on the Supermicro machine.
Apart from this, I also fixed some errors in the mbboDirect record syntax - so now I am able to start up the EPICS server without it throwing any error messages. It remains to verify that changing an EPICS gain slider results in the appropriate gain bits being flipped in the correct way (on the hardware side, I think the correct behavior is happening on the software end). For this testing, I turned off the old c1iscaux crate at ~10am, and started up the server on c1iscaux3. I am reverting to the nominal config now (~1pm).
Further testing will require the wiring inside the Acromag chassis to be completed. This should be the priority task for next week.
*Update 1130 22 July 2019: I've now installed the required dependencies on c1iscaux3 and setup the latch.py script to run as a systemctl process dependent on modbusIOC.service.
Somebody changed the settings on painosa without elogging anything about it. Why does this keep happening? I thought the point of the elog was to communicate. I think there are sufficient number of problems in the lab without me having to manually reset the control room workstation settings every week. Please make an elog if you change something.
My goal tonight was to lock the PSL frequency to be resonant in the XARM cavity, using the PSL+EX beat as the error signal. I was not successful - mainly, I was plagued by huge BR mode coupling in the error signal, and I could not enable the BR notch filter in the control loop without breaking the lock. Need to think about next steps.
Anyway, now that I have a workable set of settings that gets me close to the ALS lock of the XARM, I expect debugging to proceed faster.
Update 2019 July 23: I looked at the control loop shape today, see Attachment #3. I'm not sure I understand why the "BounceRoll" filter in this filter bank looks like a resonant gain rather than a notch, as it does for the Oplev or SUSPOS loops for example - don't we want to not actuate at these frequencies because the reason the signal exists is because of the imperfect OSEM/magnet positioning? This does not explain the spectrum shown in Attachment #2 however, as that filter was disabled.
I succeeded in locking the PSL frequency to the XARM cavity length, with 9 pm RMS (Attachment #1) motion below 1 kHz, by actuating on MC2 to change the IMC length. The locks were pretty stable (~20 minutes) - the dominant cause of lockloss was the infamous ETMX drifting problem.
My main motivation here is to make some measurements and investigate the SoCal idea using a toy system, i.e. a single arm cavity controlled using ALS, so that's what Craig and I will attempt next.
Since there are multiple SURF projects that rely on the cameras:
My changes were necessary because the grabHDR.py script was throwing python exceptions, whereas it was running just fine before Jon's changes. We can move the "new_*" dirs to the default once the SURFs are gone.
Let's freeze the camera software config in this state until next week.
These spectra were taken with the arm cavity length locked to the PSL frequency using POX as an error signal, and the EX laser frequency locked to the XARM cavity length by the analog PDH servo at EX, so there is no feedback control with the ALS beat signal as an error signal.
I think rana did some more changes to this workstation to make it useful for commissioning activities - but the MEDM screens were still white blanks. The problem was that the firewalld wasn't disabled (last two steps of the KThorne setup wiki). I disabled it. Now donatella can run MEDM, ndscope and StripTool. DTT doesn't work to get online data because of a "Synchronization Error", I'm not bothering with this for now. I think Kruthi successfully demonstrated the fetching of offline data with DTT.
I brought one CPU (Dell T3500) and one 28" monitor from Mike Pedraza's office in Downs to the 40m. It is on Steve's desk right now, pending setup. The machine already has Solidworks and Altium installed on it, so we can set it up at our leisure. The login credentials are pasted on the CPU with a post-it should anyone wish to set it up.
We need to determine the geometry (= round-trip length and RoC of curved mirrors) of the OMC cavities for the 40m BHD experiment. Sticking to the aLIGO design of a 4 mirror bowite cavity with 2 flat mirrors and 2 curved mirrors, with a ~4deg angle of incidence, we need to modify the parameters for the 40m slightly on account of our different modulation frequencies. I've setup some infrastructure to do this analytically - even if we end up doing this with Finesse, it is useful to have an analytic calculation to validate against (also not sure if Finesse can calculate HOMs up to order 20 in a reasonable time, I've only seen maxtem 8).
Attachment #1: Heatmap of the OMC transmission for the following fields:
The code used for the ABCD matrix calcs have been uploaded to the BHD modeling GIT (but not the one for making this plot, yet, I need to clean it up a bit). Some design considerations have also been added to our laundry list on the 40m wiki.
Chub brought the replacement Supermicro we ordered to the 40m today. I stored it at the SW entrance to the VEA, along with the other Supermicro. At the time of writing, we have, in hand, two (unused) Supermicro machines. One is meant for EY and the other is meant for c1psl/c1iool0. DDR3 RAM and 120 GB SSD drives have also been ordered, but have not yet arrived (I think, Chub, please correct me if I'm wrong).
Update 20190802: The DDR3 RAM and 120 GB SSD drives arrived, and are stored in the FE hardware cabinet along the east arm. So at the time of writing, we have 2 sets of (Supermicro + 120GB HD + 4GB RAM).
We should ask Chub to reorder several more SuperMicro rackmount machines, SSD drives, and DRAM cards. Gautam has the list of parts from Johannes' last order.
I'd like to confirm that the IR ALS scheme will work for locking. The X-arm performance so far has been encouraging. I want to repeat the characterization for the Y arm. So I inspected the layout on the EY table, and made a list of characterization tasks. The current EY beam routing is difficult to work with, and it will definitely benefit from a re-do. However, I don't know how much time I want to spend re-doing it, so for a start, I will just try and couple some amount of light into a fiber and bring it to the PSL table, and see what noise performance I get.
Attachment #1: Photo of the current beam layout. The powers indicated were measured with the Ophir power meter.
Attachment #2: A candidate mode-matching solution, given the constraints outlined above. It isn't great, with only 85% modematching even theoretically possible. The lenses required are also pretty fast lenses. But I think it's the best possible without a complete overhaul of the EY layout. I'm still waiting for the lens kit to arrive, but as soon as they get here, I will start this work.
I want to collect some data with the arms locked to investigate the possibility/usefullness of having seismic feedforward implemented for the arms (it is already known to help the IMC length and PRC angular stability at low frequencies). To facilitate diagnostics I modified the file /users/Templates/Seismic/Seismic_vs_TRXTRYandMC.xml to have the correct channel names in light of Lydia's channel name changes in 2016. Looking at the coherence data, the alignment of the cartesian coordinate system of the Seismometers at the ends and the global interferometer coordinate system can be improved.
I don't know if for the MISO filter design if there is any difference in using TRX/TRY as the target, or the arm length control signal.
Data collection started at 1249018179. I've setup a script running in a tmux shell to turn off the LSC enable in 2 hours.
The VEA laptop asia was configured to be able to connect to too many WiFi networks - it was getting conflicted in its default position at the vertex and trying to hop between networks, for some reason trying to connect to networks that had poor signal strength. I deleted all options from the known networks except 40MARS. Now the network connection seems much more stable and reliable.