Furthermore, I believe we are losing more than 10% of the light due to this BS. The ASDC (which is derived from AS55 PD) level is down at ~110cts as the Michelson is fringing, while it used to be ~200 cts. I will update with a power measurement shortly. But I think we should move ahead with the plan to combine the beam into the IFO's AS mode as discussed at the meeting last week.
Is the 10% specified for P-Pol or for UNP? I contacted CVI about beamsplitters, since their website doesn't list a BS1-1064-90-... option on the website. They say a R=90% beamsplitter would be a custom job. The closest stock item they got is BS1-1064-95-2025-45UNP specified at R=95% for UNPolarized beams. They were kind enough to sent me the measured transmission curves for a recent lot of these, which is attached was uploaded to the wiki [Elog Police K: NO PROPRIETARY DOCUMENTS ON THE ELOG, which is public. Put it on our wiki and put the link here]. The figure is not labeled, but according to the contact Red is S-Pol and Blue is P-Pol, which means that this one actually has R=~90% for P, pretty much what we want. We'll need to buy two of these to make the swap in the setup.
Back to your original point: There's only a BS1-1064-10-2025-45UNP on the website, so unless we got these as custom items, the R for P-Pol is probably NOT actually 10%, just somewhere between 0% and 20%
4 std cataloge item fused silica BS1-1064-95-2025-45UNP
ordered today. They will arrive no later than July 13, 2018
The fardest I can go back on channel C1: Vac_N2pres is 320 days
C1:Vac-CC1_Hornet Presuure gauge started logging Feb. 23, 2018
Did you update the " low N2 message" email addresses?
I moved the N2 check script and the disk usage checking script from the (sudo) crontab of nodus to the controls user crontab on megatron .
Shruti and Sandrine received 40m specific basic safety training this morning.
Pooja and Keirthana received 40m specific basic safety training.
This bad connection is coming back
We may lost the UL magnet or LED
We are getting ready to vent.
Steve and Aaron,
6 hrs vent is reaching equlibrium to room air. It took 3 and a half instrument grade air cilynders [ AI UZ300 as labelled ] at 10 psi pressure. Average vent speed ~ 2 Torr/min
Valve configuration: IFO at atm and RGA is pumped through VM2 by TP1 maglev.
Precondition: 4 days at atm. Atm5
HEPA tent used during the vent at ETMY It reduced partical count 10 fold of 0.5 and 0.3 micron particals. Atm6
New items in vacuum: Clean manual gate valve [Cetec made] from John Worden with 6" id....as it came from Hanford... [ Throttle able gate valve- TGV ] Atm3
( note: we have 3 more identical in the lab. The original intention was to use them for purging gates )
Optiform Au plated reflector , Induceramics heating elements, similar as existing Cooner cables and related lenses, hardwear. see 14078
OMC related item : none......... 14,110
The pumpdown is at 510 mTorr with RP1 & RP3 still pumping. Koji will shut it down the roughing later tonight. Tomorrow morning I will start the pumping by switching over to TP1 maglev.
Thanks for Koji and Gautam' help of the installation of the manual gate valve. Atm4 This will allow us to control the load on our Varian foreline D70 turbo TP3
The manual gate valve scan was clean. Atm1 TP1 was pumping on it overnight.
Pumpdown continued to hand over the pumping to TP1 maglev turbo
V1 was opened at P1 400 mTorr with manual gate at 3/4 turn open position as Magev at 560 Hz rotation.
Two aux fans on to hold tubo temps TP1 & TP3 . Atm3
This is the first time we pumping down from atm with ONE small "beer can" turbo and throttled gate valve to control load on small turbo forepump
The 70 l/s turbo is operating at 50k RPM, 0.7 A and 31 C, pumping speed ~ 44 mTorr/h at 200-400 mTorr range with aux drypump in the foreline of TP3
Watching foreline pressures and current one can keep opening gate valve little by little the so the load is optimized. It is working but not fast.
Let's keep small turbo at 0.8 Amp and 32 C max at this pumpdown.
The P1 pressure is 380mTorr. I allowed Gautam to use the full PSL power (~1W).
IFO P1 6e-4 Torr, manual gate valve is fully open
The annuloses will be pumped down tomorrow.
Valve configuration: vacuum normal, RGA and annuloses are not pumped
This is the first time we pumping down from atm with one small "beer can" turbo and throttled gate to control load on small turbo forepump
The 70 l/s turbo is operating at 50k RPM, 0.7 A and 31 C, pumping speed ~ 44 mTorr/h at 200-400 mTorr range.
Cold cathode gauge just turned on.
Roughing down the annuloses required closing V1 for 13 minutes
IFO is 2.2e-5 Torr
New optical quality BK-7 windows in 2001 [4 substrates ] AR coated R<0.75 % for 630-1064nm " Azury BLue" broadband : TRX, TRY, ITMY-Oplev & ITMX-Oplev viewports.
The BS-Oplev and PRM-Oplev 10" CF with 5.38" diameter view was coated the same way. The window here is Corning 7056 Borosilicate
5 more BK-7 substrates were coated R <0.1% of 1064 nm "Golden Orange" Their location: IMC-IN, IFO-REF and OMC The next vent we have to confirm optical quality window locations.
All other conflat flange viewports are 7056 Kovar sealed .
Technical notes of 2001 40m upgrade can be seen at LIGO-T010115- 00- R ....page 14
Yesterday I inspected this BS oplev viewport. The heavy connector tube was shorting to table so It was moved back towards the chamber. The connection is air tight with kapton tape temporarly.
The beam paths are well centered. The viewport is dusty on the inside.
The motivation was to improve the oplev noise.
I 've just found this time capsule note from Nov. 26, 2000 by Kip Thorne: LIGO will discover gravitational waves by Dec.31, 2007
what is next?
Atm 3, Ron Drever could not celebrate with us because of health issues.
Here is an other big one
A brief follow-up on this since we discussed this at the meeting yesterday: the attached DV screenshot shows the full 2k data for a period of 2 seconds starting just before the watchdog tripped. It is clear that the timescale of the glitch in the UL channel is much faster (~50 ms) compared to the (presumably mechanical) timescale seen in the other channels of ~250 ms, with the step also being much smaller (a few counts as opposed to the few thousand counts seen in the UL channel, and I guess 1 OSEM count ~ 1 um). All this supports the hypothesis that the problem is electrical and not mechanical (i.e. I think we can rule out the Acromag sending a glitchy signal to the coil and kicking the optic). The watchdog itself gets tripped because the tripping condition is the RMS of the shadow sensor outputs, which presumably exceeds the set threshold when UL glitches by a few thousand counts.
The manufacturer of a vacuum pump supplies a chart for each pump showing pumping speed (volume in unit time) vs pressure. The example, for a fictitious pump, shows the pumping speed is substantially constant over a large pressure range.
By multiplying pumping speed by pressure at which that pumping speed occurs, we get a measure called pump throughput. We can tabulate those results, as shown in the table below, or plot them as a graph of pressure vs pump throughput. As is clear from the chart, pump throughput (which might also be called mass flow) decreases proportionally with PRESSURE, at least over the pressure range where pumping speed is constant.
The roughing pump speed actually will reach 0 l/s at it's ultimate pressure performance.
Our roughing pump pumping speed will slowly drop as chamber pressure drops. Below 10 Torr this decrease is accelerated and bottoms out. This where the Root pump can help. See NASA evaluation of dry rough pumps...What is a root pump
We have been operating succsessfully with a narrow margin. The danger is that the Maglev forline peaks at 4 Torr. This puts load on the small turbo TP2, TP3 & large TP1
The temperature of these TP2 & 3 70 l/s drag turbos go up to 38 C and their rotation speed slow to 45K rpm from 50K rpm because of the large volume 33,000 liters
Either high temp or low rotation speed of drag turbo or long time of overloading can shut down the small turbo pumps......meaning: stop pumping, wait till they cool down
The manual gate valve installed helped to lower peak temp to 32C It just took too long.
We have been running with 2 external fans [one on TP1 & one on TP3] for cooling and one aux drypump to help lowering the foreline pressure of TP2 & 3
The vacuum control upgrade should include adding root pump into the zero pumping speed range.
Atm1, Pump speed chart: TP1 turbo -red, root pump -blue and mechanical pump green. Note green color here representing an oily rotory pump. Our small drypumps [SH-100] typically run above 100 mTorr
They are the forepump of TP2 & 3 Our pumpdown procedure: Oily Leybold rotory pumps ( with safety orifice 350 mT to atm ) rough to 500 mTorr
Here we switch over to TP2 & 3 running at 50k RPM with drypumps SH-100 plus Aux Triscroll
TP1- Maglev rotating full speed when V1 is opened at full volume at 500 mTorr
History: the original design of the early 1990s had no dry scroll pumps. Oil free dry scrools replaced the oily forepumps of TP2 & TP3 in ~2002 at the cost of degrading the forline pressure somewhat.
We had 2 temperature related Maglev failers in 2005 Aug 8 and 2006 April 5 Osaka advised us to use AUX fan to cool TP1 This helped.
Atm2, Wanted Root pump - Leybold EcoDry 65 plus
Atm3, Typical 8 hrs pumpdown from 2007 with TP2 & 3
Atm4, Last pumpdown zoomed in from 400 mT to 1mT with throttled gate valve took 9 hrs The foreline pressure of TP1 peaked at 290 mT, TP3 temperature peaked at 32C
This technic is workable, but 9 hrs is too long.
Atm5, The lowest pressure achived in the 40m Vacuum Envelope 5e-7 Torr with pumps Maglev ~300 l/s, Cryo 1500 l/s and 3 ion pumps of 500 l/s [ in April 2002 at pumpdown 53 day 7 ] with annuloses at ~ 10 mTorr
Atm6, Osaka TG390MCAB Throughput with screen ~300 L/s at 12 cfm backing pump
6.2M Bandon, OR did not trip any sus
Glitch, small amplitude, 350 counts & no trip.
The second big glich trips ETMX sus. There were small earth quakes around the glitches. It's damping recovered.
Small earth quakes and suspensions. Which one is the most free and most sensitive: ITMX
The rat is cut by mechanical trap and it was removed from ITMX south west location.
A nagy kover patkanyt a fogo elkapta es megolte.
All suspension tripped. Their damping restored. The MC is locked.
ITMX-UL & side magnets are stuck.
TP-1 Osaka maglev controller [ model TCO10M, ser V3F04J07 ] needs maintenance. Alarm led on indicating that we need Lv2 service.
The turbo and the controller are in good working order.
Our maintenance level 2 service price is $...... It consists of a complete disassembly of the controller for internal cleaning of all ICB’s, replacement of all main board capacitors, replacement of all internal cooling units, ROM battery replacement, re-assembly, and mandatory final testing to make sure it meets our factory specifications. Turnaround time is approximately 3 weeks.
RMA 5686 has been assigned to Caltech’s returning TC010M controller. Attached please find our RMA forms. Complete and return them to us via email, along with your PO, prior to shipping the cont
Osaka Vacuum USA, Inc.
510-770-0100 x 109
our TP-1 TG390MCAB is 9 years old. What is the life expectancy of this turbo?
The Osaka maglev turbopumps are designed with a 100,000 hours(or ~ 10 operating years) life span but as you know most of our end-users are
running their Osaka maglev turbopumps in excess of 10+, 15+ years continuously. The 100,000 hours design value is based upon the AL material being rotated at
the given speed. But the design fudge factor have somehow elongated the practical life span.
We should have the cost of new maglev & controller in next year budget. I put the quote into the wiki.
Electrician is coming to fix one of the fluorenent light fixture holder in the east arm tomorrow morning at 8am. He will be out by 9am.
The job did not get done. There was no scaffolding or ladder to reach troubled areas.
Yuki Miyazaki received 40m specific basic safety training.
M3.4 Colton shake did not trip sus.
Precondition: c1vac1 & c1vac2 all LED warning lights green [ atm3 ], the only error message is in the gauge readings NO COMM, dataviewer will plot zero [ atm1 ], valves are operational
When our vacuum gauges read " NO COMM " than our INTERLOCKS do NOT communicate either.
So V1 gate valve and PSL output shutter can not be triggered to close if the the IFO pressure goes up.
[ only CC1_HORNET_PRESSURE reading is working in this condition because it goes to a different compuer ]
Following the procedure in this elog, we effected a reset of the vacuum slow machines. Usually, I just turn the key on these crates to do a power cycle, but Steve pointed out that for the vacuum machines, we should only push the "reset" button.
While TP1 was spun down, we took the opportunity to replace the TP1 controller with a spare unit the company has sent us for use while our unit is sent to them for maintenance. The procedure was in principle simple (I only list the additional ones, for the various valve closures, see the slow machine reset procedure elog):
However, we were foiled by a Philips screw on the DB37 connector labelled "MAG BRG", which had all its head worn out. We had to make a cut in this screw using a saw blade, and use a "-" screwdriver to get this troublesome screw out. Steve suspects this is a metric gauge screw, and will request the company to send us a new one, we will replace it when re-installing the maintaiend controller.
Attachments #1 and #2 show the Vacuum MEDM screen before and after the reboot respectively - evidently, the fields that were reading "NO COMM" now read numbers. Attachment #3 shows the main volume pressure during this work.
The problem will be revisited on Monday.
Gautam and Steve,
Our TP3 drypump seal is at 360 mT [0.25A load on small turbo] after one year. We tried to swap in old spare drypump with new tip seal. It was blowing it's fuse, so we could not do it.
Noisy aux drypump turned on and opened to TP3 foreline [ two drypumps are in the foreline now ] The pressure is 48 mT and 0.17A load on small turbo.
We want to measure the pressure gradient in the 40m IFO
Our old MKS cold cathodes are out of order. The existing working gauge at the pumpspool is InstruTech CCM501
The plan is to purchase 3 new gauges for ETMY, BS and MC2 location.
Basic cold cathode or Bayard-Alpert Pirani
Steve & Bob,
Bob removed the head cover from the housing to inspect the condition of the the tip seal. The tip seal was fine but the viton cover seal had a bad hump. This misaligned the tip seal and it did not allow it to rotate.
It was repositioned an carefully tithened. It worked. It's starting current transiant measured 28 A and operational mode 3.5 A
This load is normal with an old pump. See the brand new DIP7 drypump as spare was 25 A at start and 3.1 A in operational mode. It is amazing how much punishment a slow blow ceramic 10A fuse can take [ 0215010.HXP ]
In the future one should measure the current pick up [ transient <100ms ] after the the seal change with Fluke 330 Series Current Clamp
It was swapped in and the foreline pressure dropped to 24 mTorr after 4 hours. It is very good. TP3 rotational drive current 0.15 A at 50K rpm 24C
The main laser went off when PSL doors were opened-closed. It was turned back on and the PSL is locked.
Gautam & Steve,
Our controller is back with Osaka maintenace completed. We swapped it in this morning.
Chub Osthelder received 40m specific basic safety traning today.
It took at least ten years to rust away.
Physical plan is cleaning our roof and gutters today.
We have no coffee machine.
We are dreaming about it
We still do not have it.
The Contec test board with Dsub37Fs was on the top shelf of E7
I tried to plot a long trend MC Transmitted today. I could not get farther than 2017 Aug 4
The mode cleaner was misaligned probably due to the earthquake (the drop in the MC transmitted value slightly after utc 7:38:52 as seen in the second plot). The plots show PMC transmitted and MC sum signals from 10th june 07:10:08 UTC over a duration of 17 hrs. The PMC was realigned at about 4-4:15 pm today by rana. This can be seen in the first plot.
The vacuum and MC are OK
Jon and I stuck a extender card into the eurocrate at 1X8 earlier today (~5pm PT), to see if the box was getting +24V DC from the Sorensen or not. Upon sticking the card in, the FAIL LEDs on all the VME cards came on. We immediately removed the extender card. Without any intervention from us, after ~1 minute, the FAIL LEDs went off again. Judging by the main volume pressure (Attachment #1) and the Vacuum MEDM screen (Attachment #2), this did not create any issues and the c1vac1 computer is still responsive.
But Steve can perhaps run a check in the AM to confirm that this activity didn't break anything.
Is there a reason why extender cards shouldn't be stuck into eurocrates?
It is posted at the 40m wiki with Gautam' help. Printed copies posted around doors also.