Larry W said that some security issues were flagged on nodus. So I ran
on nodus. The exclude flag is because there were some conflicts related to that particular package. Hopefully this has fixed the problem. It's been a while since the last update, which was in January of this year.
I've attached the diagram of what I mean.
There are a couple caveats and changes that would have to be made that are not included in this diagram, because they would be made on different tables.
Gautam also had some questions about the BHD/OMC timeline and plan. I feel somewhat on shaky ground with the answers, but figured I'd post them so I can be corrected once and for all.
That seems fine, I wasn't thinking of that beam. in that case could we just have a PBS directly behind MMT2 and send both beams to the same OMMT?
Alternatively we can move OM5 and the beam path OMPO-OMMTSM towards -y, then put the LO-OMMT parallel to the existing OMMT but displaced in +x... we'd have to move the existing OMC and BHD towards +x as well.
Can we use the leakage beam from MMT2 on the OMC table as the LO beam? I can't find the spec for this optic, but the leakage beam was clearly visible on an IR card even with the IMC locked with 100 mW input power so presumably there's enough light there, and this is a cavity transmission beam which presumably has some HOM content filtered out.
We finished up making the new c1omc model (screenshot attached).
The new channels are only four DAC for ASC into the OMC, and one DAC for the OMC length:
In preparation for attempting some DRMI locking, I did the following:
Not related to this work, but I turned the Agilent NA off since we aren't using it immediately.
While working on the single arm alignment, I noticed that today, i was able to get the X arm transmission back to ~1.22, and the GTRX to 0.52. These are closer to the values I remember from prior to the vent. Running the dither alignment promptly degrades both the green and IR transmissions. Since the pianosa SL7 upgrade, I can't use the sensoray to capture images, but to me, the spot looks a little off-center in Yaw on ETMX in this configuration, I've tried to show this in the phone grab (Atm #2). Maybe indicative of clipping somewhere upstream of ITMX?
Anyways, I'm pushing onwards for now, something to check out in the daytime.
After I effected the series resistance change for ETMX, the X arm ASS didn't work (i.e. IR transmission would degrade if the servo was run). Today, we succeeded in recovering a functional ASS servo .
We then tried to maximize GTRX using the PZT mirrors, but were only successful in reaching a maximum of 0.41. The value I remember from before the vent was 0.5, and indeed, with the IR alignment not quite optimized before we began this work, I saw GTRX of 0.48. But the IR dither servo signals indicate that the cavity axis may have shifted (spot position on the ITM, which is uncontrolled, seems to have drifred significantly, the Pitch signal doesn't stay on the StripTool scale anymore). So we may have to double check that the transmitted beam isn't falling off the GTRX DC PD.
After tweaking the AS55 demod phase, SRM alignment, triggering settings, I got a few brief DRMI locks in tonight, I'm calling it a success (though this isn't really robust yet). The main things to do now are:
I think the main IFO characterization remaining to be done to determine the status of the IFO post vent is to measure the losses of the arm cavities. IMO, we will need to certainly fix the clipping at ETMY before we attempt some serious locking.
I made a script to scan the OMC length at each setpoint for the two TTs steering into the OMC. It is currently located on nodus at /users/aaron/OMC/scripts/OMC_lockScan.py.
I haven't tested it and used some ez.write syntax that I hadn't used before, so I'll have to double check it.
My other qualm is that I start with all PZTs set at 0, and step around alternative +/- values on each PZT at the same magnitude (for example, at some value of PZT1_PIT, PZT1_YAW, PZT2_PIT, I'll scan PZT2_YAW=1, then PZT2_YAW=-1, then PZT2_YAW=2). If there's strong hysteresis in the PZTs, this might be a problem.
It looks like we can have a stable SRC of length 4.044 m without getting any new mirrors, so this is an option to consider in the short-term.
gautam 245pm: Koji pointed out that the G&H mirrors are coated for normal incidence, but looking at the measurement, it looks like the optic has T~75ppm at 45 degree incidence, which is maybe still okay. Alternatively, we could use the -600m SR3 as the single folding mirror in the SRC, at the expense of slightly reduced mode-matching between the arm cavity and SRC.
I took another pass at this. Here is what I have now:
Attachment #1: Composite amplifier design to suppress voltage noise of PA91 at low frequencies.
Attachment #2: Transfer function from input to output.
Attachment #3: Top 5 voltage noise contributions for this topology.
Attachment #4: Current noises for this topology, comparison to current noise from fast path and slow DAC noise.
Attachment #5: LISO file for this topology.
Looks like this will do the job. I'm going to run this by Rich and get his input on whether this will work (this design has a few differences from Rich's design), and also on how to best protect from HV incidents.
Starting c1cal now, let's see if the other c1lsc FE models are affected at all... Moreover, since MC1 seems to be well-behaved, I'm going to restore the nominal eurocrate configuration (sans extender board) tomorrow.
The manufacturer of a vacuum pump supplies a chart for each pump showing pumping speed (volume in unit time) vs pressure. The example, for a fictitious pump, shows the pumping speed is substantially constant over a large pressure range.
By multiplying pumping speed by pressure at which that pumping speed occurs, we get a measure called pump throughput. We can tabulate those results, as shown in the table below, or plot them as a graph of pressure vs pump throughput. As is clear from the chart, pump throughput (which might also be called mass flow) decreases proportionally with PRESSURE, at least over the pressure range where pumping speed is constant.
The roughing pump speed actually will reach 0 l/s at it's ultimate pressure performance.
Our roughing pump pumping speed will slowly drop as chamber pressure drops. Below 10 Torr this decrease is accelerated and bottoms out. This where the Root pump can help. See NASA evaluation of dry rough pumps...What is a root pump
We have been operating succsessfully with a narrow margin. The danger is that the Maglev forline peaks at 4 Torr. This puts load on the small turbo TP2, TP3 & large TP1
The temperature of these TP2 & 3 70 l/s drag turbos go up to 38 C and their rotation speed slow to 45K rpm from 50K rpm because of the large volume 33,000 liters
Either high temp or low rotation speed of drag turbo or long time of overloading can shut down the small turbo pumps......meaning: stop pumping, wait till they cool down
The manual gate valve installed helped to lower peak temp to 32C It just took too long.
We have been running with 2 external fans [one on TP1 & one on TP3] for cooling and one aux drypump to help lowering the foreline pressure of TP2 & 3
The vacuum control upgrade should include adding root pump into the zero pumping speed range.
Atm1, Pump speed chart: TP1 turbo -red, root pump -blue and mechanical pump green. Note green color here representing an oily rotory pump. Our small drypumps [SH-100] typically run above 100 mTorr
They are the forepump of TP2 & 3 Our pumpdown procedure: Oily Leybold rotory pumps ( with safety orifice 350 mT to atm ) rough to 500 mTorr
Here we switch over to TP2 & 3 running at 50k RPM with drypumps SH-100 plus Aux Triscroll
TP1- Maglev rotating full speed when V1 is opened at full volume at 500 mTorr
History: the original design of the early 1990s had no dry scroll pumps. Oil free dry scrools replaced the oily forepumps of TP2 & TP3 in ~2002 at the cost of degrading the forline pressure somewhat.
We had 2 temperature related Maglev failers in 2005 Aug 8 and 2006 April 5 Osaka advised us to use AUX fan to cool TP1 This helped.
Atm2, Wanted Root pump - Leybold EcoDry 65 plus
Atm3, Typical 8 hrs pumpdown from 2007 with TP2 & 3
Atm4, Last pumpdown zoomed in from 400 mT to 1mT with throttled gate valve took 9 hrs The foreline pressure of TP1 peaked at 290 mT, TP3 temperature peaked at 32C
This technic is workable, but 9 hrs is too long.
Atm5, The lowest pressure achived in the 40m Vacuum Envelope 5e-7 Torr with pumps Maglev ~300 l/s, Cryo 1500 l/s and 3 ion pumps of 500 l/s [ in April 2002 at pumpdown 53 day 7 ] with annuloses at ~ 10 mTorr
Atm6, Osaka TG390MCAB Throughput with screen ~300 L/s at 12 cfm backing pump
I had a very fruitful discussion with Rich about this circuit today. He agreed with the overall architecture, but made the following suggestions (Attachment #1 shows the circuit with these suggestions incorporated):
If all this sounds okay, I'd like to start making the PCB layout (with 5 such channels) so we can get a couple of trial boards and try this out in a couple of weeks. Per the current threat matrix and noises calculated, coil driver noise is still projected to be the main technical noise contribution in the 40m PonderSqueeze NB (more on this in a separate elog).
EDIT: After discussing with Koji and checking the existing M2ISS PDs I put the two C30642G back and took two C30665GH (active diameter: 3mm) diodes. Only one of this type remains in storage.
I removed two C30642G photodiodes from the stash for the new M2ISS hardware and updated the wiki page accordingly.
When I came in this morning no light was reaching the MC. One fast machine was dead, c1lsc, and a number of the slow machines: c1susaux, c1iool0, c1auxex, c1auxey, c1iscaux. Gautam walked me through reseting the slow machines manually and the fast machines via the reboot script. The computers are all back online and the MC is again able to lock.
I expanded the previous panels to 6U height for the new DAQ chassis we're buying for the upgrade. I figure it's best if we stick to the modular design, so I'm showing a panel for 8 BNC connectors as an example. The front panel has 12 slots, the back has 10 plus power connectors, switches, and the ethernet plug.
I moved the power switch to the rear because it's a waste of space to put it in the front, and it's not like we're power cycling this thing all the time. Note that the unit only requires +24V (for general operation, +20V also does the trick, as is the situation for ETMX) and +15V (excitation field for the binary I/O modules). While these could fit into a single CONEC power connector, it's probably for the better if we don't make a version that supplies a large positive voltage where negative is expected, so I put in two CONEC plugs for +/- 15 and +/- 24.
I want to order 5-6 of these as soon as possible, so if anyone wants anything changed or sees a problem, please do tell!
I've taken a PI Piezo Actuator (P-810.10) from the 40m collection. I forgot to note it on the equipment checklist by the door, will do so when I next drop by.
We went around the LSC, PSL, IOO, and SUS racks to check how many dual backplane interfaces will be required.
Euro card modules are connected to the backplane with two DIN 41612 connectors (as you know). The backplane connectors provide DC supplies and GND connections.
In addition, they are also used for the input and output connections with the fast and slow machines.
According to the past inspection by Johannes, most of the modules just use the upper DIN41612 connector (called P1). But there are some modules exhibited the possibility of the additional use of the other connector (P2).
Tuesday afternoon Johannes and I made the list of the modules with the possible dual use. And I took a time to check the modules with DCC, Jay's schematics, and the visual inspection of the actual modules.
6.2M Bandon, OR did not trip any sus
I think we don't need to keep Crystal Ref: we can change this into a regular Wenzel box with no outside control or monitoring.
Glitch, small amplitude, 350 counts & no trip.
Here is an other big one
A brief follow-up on this since we discussed this at the meeting yesterday: the attached DV screenshot shows the full 2k data for a period of 2 seconds starting just before the watchdog tripped. It is clear that the timescale of the glitch in the UL channel is much faster (~50 ms) compared to the (presumably mechanical) timescale seen in the other channels of ~250 ms, with the step also being much smaller (a few counts as opposed to the few thousand counts seen in the UL channel, and I guess 1 OSEM count ~ 1 um). All this supports the hypothesis that the problem is electrical and not mechanical (i.e. I think we can rule out the Acromag sending a glitchy signal to the coil and kicking the optic). The watchdog itself gets tripped because the tripping condition is the RMS of the shadow sensor outputs, which presumably exceeds the set threshold when UL glitches by a few thousand counts.
I tried unsuccessfully to relock the MC this afternoon.
I came in to find it in a trouble state with a huge amount of noise on C1:PSL-FSS_PCDRIVE visible on the projector monitor. Light was reaching the MC but it was unable to lock.
I don't know what had been wrong, but I could lock the PMC as usual.
The IMC got relocked by AutoLocker. I checked the LSC and confirmed at least Y arm could be locked just by turning on the LSC servos.
I don't know what had been wrong, but I could lock the PMC as usual.
The IMC got relocked by AutoLocker. I checked the LSC and confirmed at least Y arm could be locked just by turning on the LSC servos.
The second big glich trips ETMX sus. There were small earth quakes around the glitches. It's damping recovered.
Small earth quakes and suspensions. Which one is the most free and most sensitive: ITMX
The rat is cut by mechanical trap and it was removed from ITMX south west location.
A nagy kover patkanyt a fogo elkapta es megolte.
I found c1lsc unresponsive again today. Following the procedure in elog #13935, I ran the rebootC1LSC.sh script to perform a soft reboot of c1lsc and restart the epics processes on c1lsc, c1sus, and c1ioo. It worked. I also manually restarted one unresponsive slow machine, c1aux.
After the restarts, the CDS overview page shows the first three models on c1lsc are online (image attached). The above elog references c1oaf having to be restarted manually, so I attempted to do that. I connect via ssh to c1lsc and ran the script startc1oaf. This failed as well, however.
In this state I was able to lock the MICH configuration, which is sufficient for my purposes for now, but I was not able to lock either of the arm cavities. Are some of the still-dead models necessary to lock in resonant configurations?
All suspension tripped. Their damping restored. The MC is locked.
ITMX-UL & side magnets are stuck.
TP-1 Osaka maglev controller [ model TCO10M, ser V3F04J07 ] needs maintenance. Alarm led on indicating that we need Lv2 service.
The turbo and the controller are in good working order.
Our maintenance level 2 service price is $...... It consists of a complete disassembly of the controller for internal cleaning of all ICB’s, replacement of all main board capacitors, replacement of all internal cooling units, ROM battery replacement, re-assembly, and mandatory final testing to make sure it meets our factory specifications. Turnaround time is approximately 3 weeks.
RMA 5686 has been assigned to Caltech’s returning TC010M controller. Attached please find our RMA forms. Complete and return them to us via email, along with your PO, prior to shipping the cont
Osaka Vacuum USA, Inc.
510-770-0100 x 109
our TP-1 TG390MCAB is 9 years old. What is the life expectancy of this turbo?
The Osaka maglev turbopumps are designed with a 100,000 hours(or ~ 10 operating years) life span but as you know most of our end-users are
running their Osaka maglev turbopumps in excess of 10+, 15+ years continuously. The 100,000 hours design value is based upon the AL material being rotated at
the given speed. But the design fudge factor have somehow elongated the practical life span.
We should have the cost of new maglev & controller in next year budget. I put the quote into the wiki.
I freed ITMX and coarsely realigned the IFO using the OPLEVs. All the alignments were a bit off from overnight.
The IFO is still only able to lock in MICH mode currently, which was the situation before the earthquake. This morning I additionally tried restoring the burt state of the four machines that had been rebooted in the last week (c1iscaux, c1aux, c1psl, c1lsc) but that did not solve it.
Electrician is coming to fix one of the fluorenent light fixture holder in the east arm tomorrow morning at 8am. He will be out by 9am.
The job did not get done. There was no scaffolding or ladder to reach troubled areas.
c1lsc crashed again. I've contacted Rolf/JHanks for help since I'm out of ideas on what can be done to fix this problem.
Rolf came by today morning. For now, we've restarted the FE machine and the expansion chassis (note that the correct order in which to do this is: turn off computer--->turn off expansion chassis--->turn on expansion chassis--->turn on computer). The debugging measures Rolf suggested are (i) to replace the old generation ADC card in the expansion chassis which has a red indicator light always on and (ii) to replace the PCIe fiber (2010 make) running from the c1lsc front-end machine in 1X6 to the expansion chassis in 1Y3, as the manufacturer has suggested that pre-2012 versions of the fiber are prone to failure. We will do these opportunistically and see if there is any improvement in the situation.
Another tip from Rolf: if the c1lsc FE is responsive but the models have crashed, then doing sudo reboot by ssh-ing into c1lsc should suffice* (i.e. it shouldn't take down the models on the other vertex FEs, although if the FE is unresponsive and you hard reboot it, this may still be a problem). I'll modify I've modified the c1lsc reboot script accordingly.
* Seems like this can still lead to the other vertex FEs crashing, so I'm leaving the reboot script as is (so all vertex machines are softly rebooted when c1lsc models crash).
Todd E. came by this morning and gave us (i) 1x new ADC card and (ii) 1x roll of 100m (2017 vintage) PCIe fiber. This afternoon, I replaced the old ADC card in the c1lsc expansion chassis, and have returned the old card to Todd. The PCIe fiber replacement is a more involved project (Steve is acquiring some protective tubing to route it from the FE in 1X6 to the expansion chassis in 1Y3), but hopefully the problem was the ADC card with red indicator light, and replacing it has solved the issue. CDS is back to what is now the nominal state (Attachment #1) and Yarm is locked for Jon to work on his IFOcoupling study. We will monitor the stability in the coming days.
(i) to replace the old generation ADC card in the expansion chassis which has a red indicator light always on and (ii) to replace the PCIe fiber (2010 make) running from the c1lsc front-end machine in 1X6 to the expansion chassis in 1Y3, as the manufacturer has suggested that pre-2012 versions of the fiber are prone to failure. We will do these opportunistically and see if there is any improvement in the situation.
Looks like the ADC was not to blame, same symptoms persist.
The PCIe fiber replacement is a more involved project (Steve is acquiring some protective tubing to route it from the FE in 1X6 to the expansion chassis in 1Y3), but hopefully the problem was the ADC card with red indicator light, and replacing it has solved the issue.
Gautam and I restarted the models on c1lsc, c1ioo, and c1sus. The LSC system is functioning again. We found that only restarting c1lsc as Rolf had recommended did actually kill the models running on the other two machines. We simply reverted the rebootC1LSC.sh script to its previous form, since that does work. I'll keep using that as required until the ongoing investigations find the source of the problem.
LIGO GC notified us that nodus had SSL2.0 and SSL3.0 enabled. This has been disabled now.
The details are described on 40m wiki.
The PMC and IMC were unlocked. Both were re-locked, and alignment of both cavities were adjusted so as to maximize MC2 trans (by hand, input alignment to PMC tweaked on PSL table, IMC alignment tweaked using slow bias voltages). I disabled the inputs to the WFS loops, as it looks like they are not able to deal with the glitching IMC suspensions. c1lsc models have crashed again but I am not worrying about that for now.
9pm: The alignment is wandering all over the place so I'm just closing the PSL shutter for now.
Yuki Miyazaki received 40m specific basic safety training.
I restarted the LSC models in the usual way via the c1lsc reboot script. After doing this I was able to lock the YARM configuration for more noise coupling scripting.
M3.4 Colton shake did not trip sus.
[steve, yuki, gautam]
The plastic tubing/housing for the fiber arrived a couple of days ago. We routed ~40m of fiber through roughly that length of the tubing this morning, using some custom implements Steve sourced. To make sure we didn't damage the fiber during this process, I'm now testing the vertex models with the plastic tubing just routed casually (= illegally) along the floor from 1X4 to 1Y3 (NOTE THAT THE WIKI PAGE DIAGRAM IS OUT OF DATE AND NEEDS TO BE UPDATED), and have plugged in the new fiber to the expansion chassis and the c1lsc front end machine. But I'm seeing a DC error (0x4000), which is indicative of some sort of timing error (Attachment #1) **. Needs more investigation...
Pictures + more procedural details + proper routing of the protected fiber along cable trays after lunch. If this doesn't help the stability problem, we are out of ideas again, so fingers crossed...
** In the past, I have been able to fix the 0x4000 error by manually rebooting fb (simply restarting the daqd processes on fb using sudo systemctl restart daqd_* doesn't seem to fix the problem). Sure enough, seems to have done the job this time as well (Attachment #2). So my initial impression is that the new fiber is functioning alright .
The PCIe fiber replacement is a more involved project (Steve is acquiring some protective tubing to route it from the FE in 1X6 to the expansion chassis in 1Y3)
This didn't go as smoothly as planned. While there were no issues with the new fiber over the ~3 hours that I left it plugged in, I didn't realize the fiber has distinct ends for the "HOST" and "TARGET" (-5 points to me I guess). So while we had plugged in the ends correctly (by accident) for the pre-lunch test, while routing the fiber on the overhead cable tray, we switched the ends (because the "HOST" end of the cable is close to the reel and we felt it would be easier to do the routing the other way.
Anyway, we will fix this tomorrow. For now, the old fiber was re-connected, and the models are running. IMC is locked.
Attachments #1 is the current setup of AUX Y Green locking and it has to be improved because:
About the above two:
One of the example for improvement is just adding a new lens (f=10cm) soon after the doubling crystal. That will make mode matching better (100%) and also make separation better (85 deg) (Attachments #4 and #5). I'm checking whether we have the lens and there is space to set it. And I will measure current power of transmitted main laser in order to confirm the improvement of alignment.
About the last:
I am considering what component is needed.
[steve, koji, gautam]
We took another pass at this today, and it seems to have worked - see Attachment #1. I'm leaving CDS in this configuration so that we can investigate stability. IMC could be locked. However, due to the vacuum slow machine having failed, we are going to leave the PSL shutter closed over the weekend.
Steve pointed out that some of the vacuum MEDM screen fields were reporting "NO COMM". Koji confirmed that this is a c1vac1 problem, likely the same as reported here and can be fixed using the same procedure.
However, Steve is worried that the interlock won't kick in in case of a vacuum emergency, so we are leaving the PSL shutter closed over the weekend. The problem will be revisited on Monday.