Attached is the plot of relation between the average arm round trip loss and power recycling gain. 2 % loss due to PR3 AR reflection is taken into account.
Can I ask you to make a plot of the power recycling gain as a function of the average arm loss, indicating the current loss value?
I took PR3 AR reflectivity and calculated PRG (PR3 is flipped and so AR surface is inside PRC).
As shown in attached figure, which shows AR specification of the LaserOptik mirror (PR3 is this mirror), AR reflectivity of PR3 is ~0.5 %. Since resonant light in PRC goes through AR surface of PR3 4 times per round trip, round trip loss due to this is ~2 %. Then I got
PRG = 7.8.
To check if the strange behavior of ASDC is caused by SR2/SR3 or not, I did the following measurement:
ASDC measures the power of the light reflected by ITMX. POXDC measures the power of the light reflected by ITMX and SRM successively. Then I varied the angle of ITMX in YAW direction and compared the behaviors of ASDC and POXDC.
The results are shown in Attachments 1-3.
As you can see in these figures, the strange up-and-down behavior appeared ONLY in ASDC. Therefore, the cause of this behavior exists between AS table and SRM (I had confirmed that the angle of SRM did not affect ASDC).
And this behavior is fringe-like, as can be seen in the figures (there seems to be 3 "peaks" and 2 "valleys"), so the cause could be interference between main path and not good AR reflection at a mirror after SRM before AS table (I suspect a mirror is flipped mistakenly).
I did additional tests for the strange behavior of ASCD. ETMY, ETMX and ITMY were misaligned so that only light reflected by ITMX went into AS port. I had done similar measurement before with ITMY YAW varied.
Attachment 1 shows how ASDC level changed when ITMX PIT varied.
Attachment 2 shows how ASDC level changed when ITMX YAW varied.
Attachment 3 shows how the power of light measured by a power meter just after the AS view port varied when ITMX YAW varied.
Comparing 1 & 2, we can say that this behavior is not unique to YAW direction.
From 2 & 3, we can say something strange is happening inside the chamber.
NDS2 and the usual ports so that we can use optimus as a comsol server.
I don't think there are any other ports we need open, but I could be wrong. Let me know if I broke something you need!
I spent this afternoon trying to debug fb1, with very little to show for it. We're back to running from fb.
The first thing I did was to recompile EPICS from source, so that all the libraries needed by daqd were compiled for the system at hand. I compiled epics-3.14-12-2_long from source, and installed it at /opt/rtapps/epics on local disk, not on the /opt/rtapps network mount. I then recompiled daqd against that, and the framecpp, gds, etc from the LSCSoft packages. So everything has been compiled for this version of the OS. The compilation goes smoothly.
There are two things that I see while running this new daqd on fb1:
The mx stream connection between the front ends and the daqd is flaky. Everything will run fine for a while, the spontaneously one or all of the mx_stream processes on the front ends will die. It appears more likely that all mx_stream processes will die at the same time. It's unclear if this is some sort of chain reaction thing, or if something in daqd or in the network itself is causing them all to die at the same time. It is independent of whether or not we're using multiple mx "end points" (i.e. a different one for each front end and separate receiver threads in the daqd) or just a single one (all front ends connecting to a single mx receiver thread in daqd).
Frequently daqd will recover from this. The monit processes on the front ends restart the mx_stream processes and all will be recovered. However occaissionally, possibly if the mx_streams do not recover fast enough (which seems to be related to how frequently the receiver threads in daqd can clear themselves), daqd will start to choke and will start spitting out the "empty blocks" messages that are harbirnger of doom:
Aborted 2 send requests due to remote peer 00:30:48:be:11:5d (c1iscex:0) disconnected
00:30:48:d6:11:17 (c1iscey:0) disconnected
mx_wait failed in rcvr eid=005, reqn=182; wait did not complete; status code is Remote endpoint is closed
disconnected from the sender on endpoint 005
mx_wait failed in rcvr eid=001, reqn=24; wait did not complete; status code is Remote endpoint is closed
disconnected from the sender on endpoint 001
[Wed Dec 9 18:40:14 2015] main profiler warning: 1 empty blocks in the buffer
[Wed Dec 9 18:40:15 2015] main profiler warning: 0 empty blocks in the buffer
[Wed Dec 9 18:40:16 2015] main profiler warning: 0 empty blocks in the buffer
My suspicion is that this time of failure is tied to the mx stream failures, so we should be looking at the mx connections and network to solve this problem.
There's possibly a separate issue associated with writing the second or minute trend files to disk. With fair regularity daqd will die soon after it starts to write out the trend frames, producing the similar "empty blocks" messages.
[Eric Q, Gautam, Koji]
We went through the network connections to produce the mapping of the instruments.
Gautam summarized the notes into a spread sheet. See attachments.
We didn't find any irregular connections except for the connection of NETMGR port of c1ioo to Martian Network.
This cable was removed.
Glitches are gone. Rga scan is good
I measured the PZT actuator gain for the Lightwave NPRO at the Y-end to be 3.6 +/- 0.3 MHz/V. This is somewhat lower than the value of 5 MHz/V reported here, but I think is consistent with that measurement.
In order to calibrate the Y-axis of my Aux PDH loop noise budget plots, I wanted a measurement of the end laser actuator gain. I proceeded to measure this as follows:
The attached plot shows the measured data. The X-axis is shown after the conversion mentioned in the last bullet point. The error bars are the standard deviations of the averaging at each DC offset.
I estimated power recycling gain with the results of arm loss measurement.
From elog 11818 and 11857, round trip losses including transmittivity of ETM of Y arm and X arm (let us call them and ) are 229+13.7=243 ppm and 483+13.7=495 ppm, respectively.
How I calculated:
I used the following formula.
Amplitude reflectivity of an arm cavity :
(see elog 11816)
Amplitude reflectivity of FPMI :
With power transmittivity of PRM and amplitude reflectivity of PRM , power recycling gain is
I assumed , , and , and then I got
PRG = 9.8.
Since both round trip losses have relative error of ~ 4 % and PRG is proportional to inverse square of up to the leading order of it, relative error of PRG can be estimated as ~ 8 %, so PRG = 9.8 +/- 0.8.
According to elog 11691, which says TRX and TRY level was ~125 when DRFPMI was locked, power recycling gain was at the last DRFPMI lock.
Measured PRG is lower than PRG estimated here, but it is natural because various causes such as mode mismatch between PRC mode and arm cavity mode, imperfect contrast of FPMI, and so on could decrease PRG, which Eric suggested to me.
Added on Dec 9
If were as small as , PRG would be 16.0. PRC would be still under coupled.
The noise floor of the Rga scan is glitching less today
I've done a couple things to try and make nodus a little more secure. Some have worried that nodus may be susceptible to being drafted into a botnet, slowing down our operations.
1. I configured the ssh server settings to disallow logins as root. Ubuntu doesn't enable the root account by default anyways, but it doesn't hurt.
2. I installed fail2ban. Function: If some IP address fails to authenticate an ssh connection 3 times, it is banned from trying to connect for 10 minutes. This is mostly for thwarting mass brute force attacks. Looking at /var/log/auth.log doesn't indicate any of this kind of thing going on in the past week, at least.
3. I set up and enabled ufw (uncomplicated firewall) to only allow incoming traffic for:
Here I explain usage of my scripts for loss map measurement. There are 7 script files in a same directory /opt/rtcds/caltech/c1/scripts/lossmap_scripts. With these scripts, round trip loss of an arm cavity with the beam spot on one mirror shifted to 5x5 (option: 3x3) points is measured. You can choose on which cavity you measure, the beam spot on which mirror you shift, and maximum shift of the beam spot in vertical and horizontal direction.
To start measurement from the beginning
Run the following command in an arbitrary directory and you will get several text files including the result of loss map measurement:
> python /opt/rtcds/caltech/c1/scripts/lossmap_scripts/lossmap.py [maximum shift in mm (PIT)] [maximum shift in mm (YAW)] [arm name (XorY)] [mirror name (E or I)]
Optionally, you can add "AUTO" at the end of the above command. Without "AUTO", you will be asked if the dithering has already settled down or not after each shift of the beam spot and you can let the scripts wait until the dithering settles down sufficiently. If you add "AUTO", it will be judged if the dithering has settled down or not according to some criteria, and the measurement will continue without your response to the terminal.
The files to be created in the current directory by the scripts are:
- lossmapETMX1-1.txt # [POX power (locked)] / [POX power (misaligned)]
- lossmapETMX1-2.txt # standard deviation of [POX power (locked)] / [POX power (misaligned)]
- lossmapETMX1-3.txt # TRX
- lossmapETMX1-1_converted.txt # round trip loss (ppm) calculated from lossmapETMX1-1.txt
- lossmapETMX1-1_converted_sigma.txt # standard deviation of round trip loss calculated from 1-1.txt and 1-2.txt
- lossmapETMX_result.txt # round trip loss and its error in a clear form.
The name of the files would be "lossmapITMY1-1.txt" etc. depending on which mirror you have chosen.
To restart measurement from a certain point
Run the following command in a directory containing "lossmap(mirror name)1-1.txt", "lossmap(mirror name)1-2.txt" and "lossmap(mirrorname)1-3.txt" which are created by previous not-completed measurement:
> python /opt/rtcds/caltech/c1/scripts/lossmap_scripts/lossmap.py [maximum shift in mm (PIT)] [maximum shift in mm (YAW)] [arm name (XorY)] [mirror name (E or I)] [restart point (PIT)] [restart point (YAW)]
You can also add "AUTO".
How to designate the restart point:
Matrix elements of output of this measurement procedure are characterized by a pair of two numbers as the following shows.
(-1,-1) -> (-1,-0.5) -> (-1,0) -> (-1,0.5) -> (-1,1)
(-0.5,1) <- (-0.5,0.5) <- (-0.5,0) <- (-0.5,-0.5) <- (0.5,-1)
(0,-1) -> (0,-0.5) -> (0,0) -> (0,0.5) -> (0,1)
(0.5,1) <- (0.5,0.5) <- (0.5,0) <- (0.5,-0.5) <- (0.5,-1)
(1,-1) -> (1,-0.5) -> (1,0) -> (1,0.5) -> (1,1)
Please write the numbers that correspond to the matrix element you want to restart at. Arrows show the order of sequence of measurement. About the correspondence between the matrix elements and real position on the ETMY and ETMX, see elog 11818 and 11857, respectively.
This script will overwrite the files (~1-1.txt etc.) so it is safer to make backup of the files before you run this script.
Some notes on the scripts and measurement
- Calibration has been done only for ETMs, i.e. for ITMs unit of [maximum shift] is not mm, but the values written in [maximum shift] equal to the maximum offsets added just after demodulation of ASS loop (ex. C1:ASS-YARM_ITM_PIT_L_DEMOD_I_OFFSET).
- It should be checked before doing measurement if the following parameters are correct or not.
POXzero (L47 in lossmapx.py and L52 in lossmapx_resume.py: the value of C1:LSC-POXDC_OUTPUT when no light injects into POXPD.)
POYzero (L45 in lossmapy.py and L50 in lossmapy_resume.py: the value of C1:LSC-POYDC_OUTPUT when no light injects into POYPD.)
mmr (L11 in lossmap_convert.py: (mode matching carrier power)/(total power))
Tf (L12 in lossmap_convert.py; transmittivity of ITM)
Tetm (L13 in lossmap_convert.py: transmittivity of ETM in ppm)
- Changing n (L50 in lossmap.py) from 5 to 3, the grid points will be 3x3 changed from the default value of 5x5. If 3x3, the matrix elements are characterized by
(-1,-1) -> (-1,0) -> (-1,1)
(0,1) <- (0,0) <- (0,-1)
(1,-1) -> (1,0) -> (1,1)
similarly to the case of 5x5.
- You can copy the directory lossmap_scripts anywhere in controls and use it. These scripts will work as long as all the 7 scripts exist in a same directory.
I changed the snapshot file for ASS, /opt/rtcds/caltech/c1/scripts/ASS_DITHER_ON.snap as follows:
L124 > C1:ASS-XARM_ETM_PIT_GAIN 1 -5.000000000000000e-02
=> C1:ASS-XARM_ETM_PIT_GAIN 1 -1.500000000000000e-02
L128> C1:ASS-XARM_ETM_YAW_GAIN 1 5.000000000000000e-02
=> C1:ASS-XARM_ETM_YAW_GAIN 1 1.500000000000000e-02
The purpose of this change is to avoid the oscillation when the dithering of X arm is running.
A question to Jamie: although the new framebuilder prototype still had the same problem with trend writing, can it handle this higher testpoint/DQ channel load?
The new fb1 daqd was also crashing even without the trend writing enabled. I'm not sure how much that's affected by the load, though, e.g. it might be able to handle the extra load fine but then die because of some other issue not related to the number of channels being acquired.
We should schedule some time this week to work on fb1 some more.
CC1 and CC2 are working again. Why did they start working again ?
On the day before yesterday and in this morning, I measured loss map of ETMX. I reported the method I used to change the beam spot on ETMX below.
Round trip loss was measured for 5 x 5 points. The result is below.
455.4 +/- 21.1 437.1 +/- 21.8 482.3 +/- 21.8 461.6 +/- 22.5 507.9 +/- 20.1
448.4 +/- 20.7 457.3 +/- 21.2 495.6 +/- 20.2 483.1 +/- 20.8 472.2 +/- 19.8
436.9 +/- 19.3 444.6 +/- 19.7 483.0 +/- 19.5 474.9 +/- 20.9 498.3 +/- 18.7
454.4 +/- 18.7 474.4 +/- 20.6 487.7 +/- 21.4 482.6 +/- 20.7 487.0 +/- 19.9
443.7 +/- 18.6 469.9 +/- 20.2 482.8 +/- 18.7 480.9 +/- 19.5 486.1 +/- 19.2
The correspondence between the loss shown above and the beam spot on ETMX is shown in the attached figure. In the figure, "up" and "right" indicate direction of shift of the beam spot when you watch it via the camera (ex. 455.4 ppm corresponds to the highest and rightest point in the view via the camera).
This result is consistent withe previous result of 561.19 +/- 14.57 ppm ericq got with ASDC and reported in elog 10248 if the discussion I reported in 11819 is taken into account. Elog 11819 says in short that the strange behavior of ASDC could give us 60-70 ppm error.
The reason why the error is larger than that of the measurement for ETMY is that the noise of POX is larger than that of POY. But I am not sure to what extent the statistical error needs to be reduced.
How I shifted the beam spot on ETMX:
Basically, the method was same as one used for Y arm. Different point is: for Y arm we have two steering mirrors TT1&2, but for X arm we have only one steering mirror BS. Then in order to shift incident beam so that the beam spot on ITMX does not change, I ran the dithering of X arm as well as that of Y arm and added offsets to both dither loops that caused same amount of shift on ETMX and ETMX. Thanks to the symmetry between X arm and Y arm, the dithering of Y arm ensured that the beam spot on ITMX was unchanged as well as that of ITMY. The idea of this method is schematically shown in Attachment 2.
The calibration of how much the beam spot shifted is based on the results of elog 11846 . The offset was [-15,-7.5,0,7.5,15]x[-5,-2.5,0,2.5,5] for pitch and yaw, respectively.
Since removing c1pem from the daqd master file, daqd has not crashed. I suppose we're running into the stability issue that motivated us to disable some of the other models (IOPs, RFM, etc.) during the RCG upgrade.
I added 1 line to one of the ASS scripts, UNFREEZE_DITHER.py like this:
L29> ez.cawrite('C1:ASS-'+dof+'_GAIN', 0)
The reason why I added this is: without this line, C1:ASS-'+dof+'_GAIN become larger that 1.0, which is nomial value, if you UNFREEZE DITHER when the dither is already running or C1:ASS-'+dof+'_GAIN is not 0.0.
I glanced at the summary pages and noticed that, since Friday around when we first loaded up the new BLRMS parts, daqd has crashing very frequently (few times per hour).
I'm going to comment out the c1pem lines from the daqd master file for tonight, and see if that helps.
I don't know if just running "modprobe" will work or not, because I didn't try it... When the same problem happens again, we can try just running "modprobe" first.
Do we need "make" everytime? Do you mean just running "modprobe" didn't work?
Today image capture did not work again, though it had worked 3 days before. I also found that red indicator light on the front pannel of SENSORAY was not turned on, which had been turned on 3 days before (you can find SENSORAY on the floor near Pianosa). Possible reason that it did not work again was I restarted Pianosa last night. Anyway, it works now. Here I report what I did to make it work.
I ran thes commands in shell, following the instruction of the manual of SENSORAY 2253 (Page 5; link or you can find the manual in /users/sensoray; I put it there).
> cd /users/sensoray/sdk_2253_1.2.2_linux
> make all
> sudo make install
> modprobe s2253
Then the red light got turned on, and image capture worked.
If you recieve an error like "No such file or directory: /dev/video0" at the beginning of the error message when you run image capture scripts from the medm screen, or if you notice that the red indicator light of SENSORAY is not on, this procedure could help you.
Now, the beam on POX11 PD is well centered and well focused.
We found out why POXDC had behaved as reported in elog 11839. There were a few reasons: the beam was not focused enough, hight of a mirror was not matched to the beam well, path of the light reflected by misaligned SRM was occasionally close to the path of POX beam.
Then, What we did is following:
- changed orientation of SRM slightly
- changed the hight of the mirror whose hight had not matched well, by changing the pedestal (hight of which mirror was changed is shown in Attachment 1.)
- put a lens with f=250 mm (where the lens is located is shown in Attachment 1.)
- refined alignment for the POX beam to hit on the center of POX11 PD.
As a result, POX DC level behaved as shown in Attachment 2&3 when the orientation of ITMX was varied (Attachment 2: POX DC vs ITMX PIT, Attachment 3: POX DC vs ITMX YAW).
You can see broad plateau when varied in both PIT and YAW directions, and the beam is at the center of the plateau if ITMX is aligned ideally.
BLRMS filters have been set up for the coil outputs and shadow sensor signals. The signals are sent to the C1PEM model from C1MCS, where I use the library block mentioned in the previous elog to put the filters in place. Some preliminary observations:
Unrelated to this work: we cleaned up the correspondence between the accelerometer numbers and channels in the C1PEM model. Also, the 3 unused ADC blocks in C1PEM (ADC0, ADC1 and ADC2) are required and cannot be removed as the ADC blocks have to be numbered sequentially and the signals needed in C1PEM come from ADC3 (as we found out when we tried recompiling the model after deleting these blocks).
I had noticed for a while that the c1ioo frontend model had much higher variability than any of the other other 16k models, and would run longer than 60us multiple times an hour. This struck me as odd, since all it does is control the WFS loops. (You can see this on the Nov17 Summary page. (But somehow, the CDS tab seems broken since then, and I'm not sure why...))
This has happily now been solved! While poking around the model, I noticed that the MC2 transmission QPD data streams being sent over from c1mcs were using RFM blocks. This seemed weird to me, since I wasn't even aware that c1ioo had an RFM card. Since the c1sus and c1ioo frontends are both on the Dolphin network, I changed them to Dolphin blocks and voila! The cylce time now holds steady at 21usec.
Update: I think I figured out the problem with the CDS summary pages. Looking at the .err files in /home/40m/public_html/summary/logs on the 40m LDAS account showed that C1:FEC-33_CPU_METER wasn't found in the frame files. Indeed, this channel was commented out in c1/chans/daq/C0EDCU.ini. I enabled it and restarted daqd. Hopefully the CDS tab will come back soon...
To focus POX beam on POX11 PD, I added an iris and a lens before POX11 PD as you can see in Attachment 1.
It seemed that the beam is well focused, but the behavior of POXDC has not changed, as shown in Attachments 2 & 3.
As I did for YARM (elog 11779), I measured the relation between offsets added just after the demodulation of the dithering loop of XARM and beam spot shift on ETMX. Defferent from YARM, the beam spot on ITMX DOES change because only BS is used as a steering mirror (TT1&2 are used for the dithering of YARM). Instead, the beam spot on BS DOES NOT change.
This time, I measured by oplevs the angles of both ETMX and ITMX for each value of offset, and using these angles I calculated the shift of the beam spot on ETMX so that I got two independent estimations (one from ETMX oplev, and the other from ITMX oplev) as shown below. The calibration of the oplevs reported in elog 11831 is taken into account.
The difference of two estimations comes from the error of calibration of oplevs and/or imperfect alignment, I think.
To avoid the strange kicking of ETMX, I locked XARM with ITMX actuated instead of ETMX so that I changed elements of C1LSC_OUTPUT_MTRX; before: XARM=ETMX, after: XARM=ITMX.
And I change C1:LSC-XARM_GAIN from 0.007 to 0.022.
With this setup, I ran dither but then error signals of dithering oscillated as shown in the figure below.
Then I found that if C1:ASS-XARM_ETM_PIT_L_DEMOD_SIG_GAIN / C1:ASS-XARM_ETM_YAW_L_DEMOD_SIG_GAIN in C1ASS_LOCKINS_XARM.adl are changed as 0.200 -> 0.100 and 0.200 -> 0.100, respectively, the dithering works well.
But the burt file that is loaded when you let dithering "ON" is not changed, because now I don't know how to update a burt file. So, if you let dithering "ON", the dithering will run with the condition that C1:ASS-XARM_ETM_PIT_L_DEMOD_SIG_GAIN / C1:ASS-XARM_ETM_YAW_L_DEMOD_SIG_GAIN are not 0.100 but 0.200.
In order to be consistent with the naming conventions for the new BLRMS filters, I made a library block that takes all the input signals of interest (i.e. for a generic optic, the coil signals, the local damping shadow sensors, and the Oplev Pitch and Yaw signals - so a total of 12 signals, unused ones can be grounded). The block is called "sus_single_BLRMS". Inside the block, I've put in 12 BLRMS library blocks, with each input signal going to one of them. All the 7 outputs of the BLRMS block are terminated (I got a compiling error if I did not do this). The idea is to identify the optic using this block, e.g. MC2_BLRMS. The BLRMS filters inside are called UL_COIL, UR_COIL etc, so the BLRMS channels will end up being called C1:SUS-MC2_BLRMS_UL_COIL_0p01_0p03 and so on. I tried implementing this in C1PEM, but immediately after compiling and restarting the model, I noticed some strange behaviour in the seismic rainbow STS strip in the control room - this was right after the model was restarted, before I attempted to make any changes to the C1PEM.txt file and add filters. I then manually opened up the filter bank screens for the RMS_STS1Z bandpass and lowpass filters, and saw that the filter switches were OFF - I wonder if this has something to do with these settings not being updated in the SDF tables? So I manually turned them on and cleared the filter hitsory for all 7 low pass and band pass filter banks, but the traces on the seismic striptool did not return to their nominal levels. I manually checked the filter shapes with Foton and they seem alright. Anyways, for now, I've reverted to the C1PEM model before I made any changes, and the seismic strip looks to be back at its normal level - when I recompiled and restarted the model with the changes I made removed, the STS1Z BLRMS bandpass and lowpass filters were ON by default again! I'm not sure what I'm doing wrong here, I will investigate this further.
I updated the figures. I think it's easier to read now.
Let Loren help you make your Oplev data readable to humans.
While trying to resolve the strange SRCL loop shape seen yesterday (which has been resolved, eric will elog about it later), we got a chance to put in the correct filters to the "CINV" branch in the C1CAL model for MICH, PRCL, and SRCL - so we have some calibrated spectra now (Attachment #1). The procedure followed was as follows:
The final set of gains used were:
MICH: -247 dB
PRCL: -256 dB
SRCL: -212 dB
and the gain-only filters in the CINV filter banks are all called "DRMI1f".
Once we are able to lock the DRFPMI again, we can do the same for CARM and DARM as well...
I've made several changes to the C1MCS model and C1PEM model, and have installed BLRMS filters for the MC mirror coils, which are now running. The main idea behind this test was to see how much CPU time was added as a result of setting up IPC channels to take the signals from C1MCS to C1PEM (where the BLRMS filtering happens) - I checked the average CPU time before and after installing the BLRMS filters, and saw that the increase was about 1 usec for 15 IPC channels installed (it increased from ~27usec to 28usec). A direct scaling would suggest that setting up the BLRMS for the vertex optics might push the c1sus model close to timing out - it is at ~50usec right now, and I would need, per optic, 12 IPC channels, and so for the 5 vertex optics, this would suggest that the CPU timing would be ~55usec. I have not committed either of the changed models to the SVN just yet.
Now that I have a procedure in place to install the BLRMS filters, we can do so for other channels as well, such as for the coils and Oplevs of the vertex optics, and the remaining PEM channels (SEIS, accelerometers, microphones?). For the vertex optics though, I am not sure if we need to do some rearrangement to the c1sus model to make sure it does not time out...
I checked how POXDC level changes when the angle of ITMX is varied. ETMX was misaligned.
Then I found that in YAW direction the POXDC level is maximized but it doesnt have plateau, and in PIT direction it is not maximized so that it is at the slope and it doesnt have plateau, as shown in attached figures. These results indicate that the beam size on POX11 is not small enough compared to the size of the diode and it is not centered well.
I disconnected the cable that was connected to CH6 of the whitening filter in 1Y2, then connected POXDC cable to there (CH6). This channel is where POXDC used to connect.
Then I turned on the whitening filter for POXDC and POYDC (C1:LSC-POXDC FM1, C1:LSC-POYDC FM1) and changed the gain of analog whitening filter for POXDC and POYDC from 0 dB to 45 dB and from 0 dB to 39 dB, respectively (C1:LSC-POXDC_WhiteGain, C1:LSC-POYDC_WhiteGain).
Somehow, the controls user account on donatella lost its membership to the sudoers group, which meant doing anything that needs root authentication was impossible.
I fixed this by booting up from a Linux install USB drive, mounting the HD, and running useradd controls sudo
useradd controls sudo
Since ETMX seems to have been on good behavior lately, we tried to fire the IFO back up.
We had a fair amount of trouble locking the DRMI with the arms held off resonance. For reasons yet to be understood, we discovered that the SRCL OLG looks totally bananas. It isn't possible to hold the DRMI for very long with this shape, obviously.
With the arms misaligned and the DRMI locked on 1F, the loop shape is totally normal. I haven't yet tried 3F locking with the arms misaligned, but this is a logical next step; I just need to look up the old demod angles used for this, since it wasn't quickly possible with the 3F demod angles that are currently set for the DRFPMI.
MC Autolocker got stack somewhere. I had to go to megatron and kill MC Autolocker.
init relaunched the autolocker automatically, and now it started properly.
SLOWDC servo was dead. I followed EricQ's instruction.
We've been talking about putting in BLRMS filters for several channels - it would be a pain to manually copy over the correct bandpass and lowpass filter coefficients into the newly created filter banks, and so I've set up a script (attached) that can do the job. As template filters, I'vm using the filters rana detailed here. Essentially, what the script does is identify the (empty on creation) block of text for a given filter: e.g. RMS_STS1Z_BP_0p01_0p03 for STS1Z), and appends the template filter coefficients. To test my script, I first backed up the original C1PEM.txt file from /opt/rtcds/caltech/c1/chans, removed all the filter coefficients for the STS1Z BLRMS filters, and then replaced it with one generated using my script. I then loaded the coefficients for all the filters in the C1PEM modules, without any obvious error messages being generated. I also checked that foton could read the new file, and checked tmake sure that sensible filter shapes were seen for some channels. Since this seems to be working, I'm going to start putting in BLRMS blocks into the models tomorrow.
Q adjusted the Dataviewer so it is not chopping of data any more. Thanks.
Cold cathode gauge reading of 10 years.
With the same method as reported in elog 11785, I calibrated oplevs for ITMX/ETMX.
According to this measurement, ratio of the calibration factor derived with this measurement (NEW) and the calibration factor for now (OLD), i.e. NEW/OLD was:
The calibration factors of the oplevs for ETMY/ITMY are NOT UPDATED YET. I updated on Dec 11, 2015
Dataviewer x axis end is not there.
On ( 2600 days) longer plots it is missing 8 moths and on (100 days) shorther plot it is missing 1 month.
It wasn't fully mentioned in ELOG 11814.
We checked the PD first and this behavior didn't change after the realignment of the AS55PD.
Yutaro confirmed that this effect is happening in the vacuum chamber.
With captured images of ETMYF, I measured the shift of the beam spot on ETMY.
The conclusions are:
the baffle would have almost no effect on loss map measurement and
the calibration of beam spot shift is confirmed to be not so bad.
What I did:
I captured ETMYF images in the cases that (i)beam spot is centered on ETMY, beam spot is at the rightest and lowest point of my loss map measurement (corresponding to [0,0] component of the matrix shown in elog 11818), and beam spot is at the leftest and highest point of my loss map measurement ([4,4] component). Each captured image is attached.
Then using ImageJ, I measured the shift of the beam spot. I calibrated lengh in horizontal direction and vertical direction with the diameter of the mirror.
The amount of the beam shift was 7.2 mm and 8.0 mm for each case.
These values indicate that clapping loss due to the baffle is less than 10 ppm in a round trip.
Today's results support the previous calibration with oplev, which says the amount of the beam shift is 7.0 mm. Two values derived by different calibrations coincide within ~10 % though they are totally different methods. This also support the calibration of the oplev for ETMY (elog 11785) indirectly.
One possible explanation of this behavior is simply poor centering of the AS beam on AS55 (whose DC level provides ASDC, if memory serves me correctly).
I misaligned ETMY, and moved ITMY through its current nominal alignment while looking at the POYDC and ASDC levels.
In both pitch and yaw, the nominal alignment is fairly close to the "plateau" in which the AS beam is fully within the PD active surface. I.e. it doesn't take much angular motion to start to lose part of the beam, and thus introduce a first order coupling of angle to power. (Look at the plateaus at around -2min and -0.5min, and where the rapidly changing oplev trace crosses zero)
Furthermore, POYDC seems to be in some weird condition where it is actually possible to increase the reported powerwhen misaligning in pitch, but somehow there is more angular coupling in this state.
In any case, I would advise that the POY11 and AS55 RFPDs have their spots recentered with optics in their nominal aligned states. In fact, given how we found REFL11 alingment to be less-than-ideal not so long ago, all of the RFPDs could probably use a checkup.
T1000461 tells us that the nominal LO input is 2dBm although we don't know what's the LO level is at the mixers in the demod boards.
I checked the RF levels at the LSC LO distribution box, with the agilent scope and a handful of couplers. This was all done with the Marconi at +13dBm.
I only checked the channels that are currently in use, since the analyzer only measures 3 channels at a time, and rewiring involves walking back and forth to the IOO rack to make sure unpowered amps aren't driven, and I was getting hungry.
For the most part, the LO levels coming into the LSC demod boards are all around +1.5dBm (i.e. I measured around -18.0dBm out of the ZFDC-20-5 coupler, which has a nominal 19.5dB coupling factor)
The inputs piped over from the IOO rack, labeled as "+6dBm" were found to be 4.7dBm and 2.9dBm for 11Mhz and 55MHz, respectively.
The 2F signals were generally about 40dB lower, with two exceptions:
Here are the raw numbers I measured out of the couplers, all in dBm:
11MHz in: -14.8
55MHz in: -16.6
POP55: -18.8 (this port is used as the REFL55 LO)