C1:SUS-ETMX_QPD is removed and internal SM1 thread adapter epoxied into position as it is at the Y end
This adapter will take FL1064-10 line filter holder
Line filter is attached and qpd needs alignment.
Ophir power meter gets new filter with calibration. This is not cheap. It was the second time we lost it.
Filter leash is attached.
Some one already took off the filter and did not care to put it back on. This is carelessness!
Missing filter found. Labeled drawer OPHIR in control room - behind soldering station- with spare battery and filter
100 and 10 days trends of ETMX and ETMY_SUSPIT. One can see clearly the earthquaks of Dec.30 and 31 on the 10 day plot. You can not see the two shakes M3.0 & M4.3 of Jan. 3
The long term plot looks OK , but the 10 day plot show the problem of ETMX as it was shaken 4 times.
Yesterday morning was dusty. I wonder why?
The PRM sus damping was restored this morning.
ETMX YAW stopped drifting Jan 8, 2015
I made little scripts to go with the sus driftmon buttons, that will servo the alignment sliders until the susyaw and suspit values match the references on the driftmon screen.
I have just put the seismometers back into their nominal positions, on the concreted slabs. The T-240 is in the vertex, and the 2 Guralps are at the end stations.
The vertex location doesn't have a spaghetti pot right now. There is an aluminum support for cable trays that is welded to the supports under the beam tube that is in the way. The pot looks like it will fit barely, if it were slid totally horizontally into place. However we can't do that with the seismometer in place. I'll chat with Steve this afternoon about our options.
Since I don't know that we are planning on ever putting a cable tray on the inside of the beamtube, perhaps we can cut ~6 inches of this piece away.
Aluminum support beam removed and seismometer is covered.
Yesterday afternoon at 4 the dust count peaked 70,000 counts
Manasa's alergy was bad at the X-end yesterday. What is going on?
There was no wind and CES neighbors did not do anything.
Baja 4.9 m earth quake tripped suspentions, except ETMX Sus damping recovered. MC is locking.
Safety audit went soothly. We thank all participients.
1, Bathroom water heater cable to be stress releived and connector replaced by twister lock type.
2, Floor cable bridge at the vacuum rack to be replaced. It is cracked.
3, Sprinkler head to be moved eastward 2 ft in room 101
4, Annual crane inspection is scheduled for 8am Marc 3, 2015
5, Annual safety glasses cleaning and transmission measurement will get done tomorrow morning.
Safety glasses were measured and they are all good. I'd like to measure your personal glass if it is not on this picture.
The temperature of the east and south ends are normal, they are about the same.
Konecranes' Fred inspected and load tested all tree cranes at with 450 lbs
The BS oplev servo was kicking up the BS. It was turned off
Green glass for aLIGO OMC shield is temporarly stored in the inside of the Y-arm.
Polaris mounts ordered.
In the attached plot you can see that the MC REFL fluctuations started getting larger on Feb 24 just after midnight. Its been bad ever since. What happened that night or the afternoon of Feb 23?
The WFS DC spot positions were far off (~0.9), so I unlocked the IMC and aligned the spots on there using the nearby steering mirrors - lets see if this helps.
Also, these mounts should be improved. Steve, can you please prepare 5 mounts with the Thorlabs BA2 or BA3 base, the 3/4" diameter steel posts, and the Polanski steel mirror mounts? We should replace the mirror mounts for the 1" diameter mirrors during the daytime next week to reduce drift.
Are the two visible small srews holding the adapter plate only?
If yes, it is the weakest point of the IOO path.
We run out of N2 for the vacuum system. The pressure peaked at 1.3 mTorr with MC locked. V1 did not closed because the N2 pressure sensor failed.
We are back to vac normal. I will be here tomorrow to check on things.
We run out of N2 for the vacuum system 6 hrs ago. The pressure rose to 1.2 mTorr with V1 closed. The interlock worked! See Nirogen presure reading fixed at http://nodus.ligo.caltech.edu:8080/40m/10968
The vacuum interlock: Nitrogen pressure transducer is reading the pneumatic pressure continously at the pump spool and c1vac1 processing it. When it drops below 60 PSI it closes V1 gate valve and V4 & V5. Gate valve V1 needs minimum 60 PSI to close. It is critical that V1 is closed before you run out of Nitrogen so the IFO pressure is contained.
IFO vacuum is back to Vac Normal. The MC is locked.
cc4 = 2E-6 Torr with VM1 open.
Daily N2 consumption measured to be 530PSI as 3 days on 3-27-2015 but note: it does vary !
I have seen it as high as 900 psi The long term average ~750 psi
To achive the same beam height each components needs their specific post height.
We have 2.625" tall, 3/4" OD SS posts for Polaris K1 mirror mounts: 20 pieces
Ordered Newport LH-1 lens mounts with axis height 1.0
I turned on the HEPA at the south end during the LSC. Sorry I ment to turn it off.
Overall a "meh" night for locking I think. The script to all-RF worked several times earlier in the evening, although it was delicate and failed at least 50% of the time. Later in the evening, we couldn't get even ~10% of the lock attempts all the way to RF-only.
Den looked into angular things tonight. With the HEPA bench at the Xend on (which it was found to be), the ETMX oplevs were injecting almost a factor of 10 noise (around 10ish Hz?) into the cavity axis motion (as seen by the trans QPD) as compared to oplevs off. Turning off the HEPA removed this noise injection.
Den retuned the QPD trans loops so that they only push on the ETMs, so that we can turn off the ETM oplevs, and leave the ITMs and their oplevs alone.
We are worried again about REFL55. There is much more light on REFL55 than there is on REFL11 (a 90/10 beam splitter divides the light between them), and we see this in the DC output of the PDs, but there seems to be very little actual signal in REFL55. Den drove a line (in PRCL?) while we had the PRMI locked with the arms held off resonance, and REFL55 saw the line a factor of 1,000 less than REFL 11 or REFL165. The analog whitening gain for REFL11 is +18dB, and for REFL55 is +21dB, so it's not that we have significantly less analog gain (that we think). We need to look into this tomorrow. As of now, we don't think there's much hope for transitioning PRMI to REFL55 without a health checkup.
The PMC is seated on 3 SS balls and it is free to move. I'm sure it will move in an earthquake. Not much, because the input and output K1 mirror frame will act as an earthquake stop.Atm2
Are there a touch of super glue on the balls? No, but there are V grooves at the bottom and on the top of each ball.Atm3
The 40m fenced area will start storing this large ~ 8000 lbs chamber on April 14. The asphalt will be cut, jack hammered the next 2-3 days in order to lay concrete.
Their schedule is from 8 to 5 starting tomorrow. We are asking them to work from 6 to 3pm
ETMX is about 12-15 ft away
ITMX, ETMY, BS and SRM are oscillating ?
I saw this kicking before
The BS oplev has been misbehaving and kicking the optic from time to time since noon. The kicks are not strong enough to trip the watchdogs (current watchdog max counts for the sensors is 135).
I took a look at the spectrum of the BS oplev error in pit and yaw with both loops enabled while the optic was stable. There is nothing alarmingly big except for some additional noise above 4Hz.
I have turned the BS oplev servo OFF for now.
The 40m fenced area will start storing this large chamber on April 14. The asphalt will be cut, jack hammered the next 2-3 days in order to lay concrete.
Jackhammering was happening around 7:30am
It looks like it did no harm. It is too early to say what may have moved. Rana's worrisome email was late.
The ground preparation is completed.
There was a period of short unexpected jackhammering this morning. I asked them to stop.
The good mood of GTRX was not changed.
It is here.
The IFO_overview of oplevs seems ok, The servos are working fine. The green arms are locked, but master and oplev_summary monitoring screens are not.
I'm proposing to Erick G. to postpone the oplev noise measurement.
Manasa and Steve,
Is this what you want? Dashed lines are dark.
BS and PRM oplevs are blocked for this measurement. I will restore to normal operation at 4pm today.
BS & PRM oplev is restored. Note: the F -150 lens was removed right after the first turning mirror from the laser. This helped Rana to get small spot on the qpd.
It also means that the oplev paths are somewhat different now.
ETMX sus damping restored.
RFpds box is moved from RF cabinet E4 to clean cabinet S15
Inventory updated at https://wiki-40m.ligo.caltech.edu/RF_Pd_Inventory
Large Area InGaAs PIN Photodiode -- C30642GH 6 pieces in stock
Large Area InGaAs PIN Photodiode with a 2,0 mm active diameter chip in TO-5 package with flat glass window
Large area InGaAs PIN photodiode with useful diameter of 2,0 mm in a T0-5 package with a flat glass window. The C30642GH provides high quantum efficiency from 800nm to 1700nm. It features high responsivity, high shunt resistance, low dark current, low capacitance for fast response time and uniformity within 2% across the detector active area.
Few 1/4 -20 socket cap head screw with washers were tested for optimum torque.
QJR 117E Snap On torque wrench was used. I found that 40 lb in was enough.
These numbers will varie with washers, material it's going into and so on!
The standard among high-strength fasteners, these screws are stronger than Grade 8 steel screws. They have a minimum tensile strength of 170,000 psi. and a minimum Rockwell hardness of C37. Length is measured from under the head.
Inch screws have a Class 3A thread fit. They meet ASTM A574.
Black Oxide—Screws have been heat-treated for hardness, which results in a dark surface color.
Rana is next to calibrate his feelings and declare the right number.
Than Koji....and so on
Once we a number, than I buy more torque wrenches to fit it.
Air cond filters checked by Chris. The 400 days plot show 3 bad peaks at 1-20, 2-5 & 2-19
Based on Jenne's chiara disk usage monitoring script, I made a script that checks the N2 pressure, which will send an email to myself, Jenne, Rana, Koji, and Steve, should the pressure fall below 60psi. I also updated the chiara disk checking script to work on the new Nodus setup. I tested the two, only emailing myself, and they appear to work as expected.
The scripts are committed to the svn. Nodus' crontab now includes these two scripts, as well as the crontab backup script. (It occurs to me that the crontab backup script could be a little smarter, only backing it up if a change is made, but the archive is only a few MB, so it's probably not so important...)
This watch script gives little time to replace N2 cylinder. When the regulated supply drops below 60 psi the cylinder pressure is 60 psi too.
It is more of a statement that V1 is closed and act accordingly. It's only practical if you are in the lab.
Rana pointed it out correctly that we need this message 24 hrs before it happens. This requires monitoring of total supplies , not the regulated one.
So we need pressure transducers on each nitrogen cylinder, before the regulator. The sum of the two N2 cylinder when they are full 4000 - 4500 psi
The first email should be send out at 1000 psi as sum of the two cylinders. This means that you have 1 day to replace nitrogen cylinder.
Most of the time the daily consumption is 750 +-50 psi
However sometimes this variation goes up ~750 +-150 psi
We have 2 transduser PX303-3KG5V http://www.omega.com/pressure/pdf/PX303.pdf They will be installed on the out put of the N2 cylinders to read the supply pressure.
I will order one DC power supply http://www.omega.com/pptst/PSU93_FPW15.html PSU-93
One full cylinder pressure is ~ 2400 PSI max so two of them will give us ~9Vdc
The email reminder should be send at 1000 PSI = 1.8 V
As it was requested by the Bos.
It would be nice to read from the epic screen C1:Vac-state_mon.......Current State: Vacuum Normal, valve configuration
Maglev is the main pump of our vacuum system below 500 mTorr
It's long term pressure has to be <500 mTorr
Each chamber has it's own annulos. These small volumes are indipendent from main volume. Their pressure ranges are <5 mTorr at vac. normal valve configuration.
CC1=cold cathode gauge (low emmision), Pressure range: 1e-4 to 1e-10 Torr,
In vac- normal configuration CC1= 2e-6 Torr
The N2 supply is regulated to 60-80 PSI out put at the auto cylinder changer.
Each cylinder pressure will be measured before the regulator and summed for warning message to be send
at 1000 PSI
Why is the Y arm drifting so much?
The " PSL FSS Slow Actuator Adjust " was brought back to range from 1.5 to 0.3 yesterday as ususual. Nothing else was touched.
I'm not sure if the timing scale is working correctly on theses summery plots. What is the definition of today?
The y-arm became much better as I noticed it at 5pm
CC1s are not reading any longer. It is an attempt to clean them over the weekend at 85C
These brand new gauges "10421002" sn 11823-vertical, sn 11837 horizontal replaced 11 years old 421 http://nsei.missouri.edu/manuals/hps-mks/421%20Cold%20Cathode%20Ionization%20Guage.pdf on 09-06-2012 http://nodus.ligo.caltech.edu:8080/40m/7441
We have two cold cathode gauges at the pump spool and one signal cable to controller. CC1 in horizontal position and CC1 in vertical position.
CC1 h started not reading so I moved cable over to CC1 v
Baking both CC1 at 85 C for 60 hrs did not help.
The temperature is increased to 125 C and it is being repeated.
COD_Sugar napolion is due to Steve: Item delivered, model CMG-SCU-0013, sn G9536
Reward being offered for the safe return of this thing:
Vacuum Operation Guide is up loaded into the 40m-wiki. This is an old master copy. Not exact in terms of real action, but it is still a good guide of logic.
Rana has promissed to watch the N2 supply and change cylinder when it is empty. I will be Hanford next week.
The cable tray holder cross beam is removed and cut short for good access to seismometer.
Tried swapping cables at the Guralp interface box side. It seems that all of our seismic signal problems have to do with the GUR2 cable being flaky (not surprising since it looks like it was patched with Orange Electrical tape!! rather than proper mechanical strain relief).
After swapping the cables today, the GUR2 DAQ channels all look fine: i.e. GUR1 (the one at the Y end) is fine, as is its cable and the GUR2 analog channels inside the interface box.
OTOH, the GUR1 DAQ channels (which have GUR2 (EX) connected into it) are too small by a factor of ~1000. Seems like that end of the cable will need to be remade. Luckily Jenne is still around this week and can point us to the pinout / instructions. Looks like there could be some shorting inside the backshell, so I've left it disconnected rather than risk damaging the seismometer. We should get a GUR1 style backshell to remake this cable. It might also be possible that the end at the seismometer is bad - Steve was supposed to swap the screws on the granite-aluminum plate on Thursday; I'll double check.
The Guralps were swapped.
What I did: turned DC power off at 1X1, hand carried them, centered each leveling bubbles, gently locked the jack screws and turned power back on.
ETMY at the east end now has CMG-T40-0008, sn T4157 as channel C1:PEM-SEIS_STS_1_Y_OUT_DQ.........
ETMX at south end now has CMG-T40-0053, sn T4Q17 as channel C1:PEM-SEIS_STS_2_Y_OUT_DQ.........
Conclusion: Guralps are fine. X cable is bad. It was bad 6 months ago when it was made.
We can still swap the 3ft short cables at the granite base if Rana has not done it.
Koji and Steve,
The result: bad Guralp x-arm cable.
I will swap the short cables tomorrow at the base.
Local earthquake 3.8 Mag tripped only PRM
Vac monitor is not communicating.
PSL HEPA turned on
Short 46" long cables at the base plates were swapped. Their solderings looked horrible.
This cable actually worked at 5-5-2015
Bad cable at ETMY station now. The new cable should be a little bit longer ~52"
I'm getting ready change the Newport Ultima U100-AC to SS-Polaris-K1 LOW DRIFT MIRROR MOUNTS
Note: there is only one lens in the PRM lunching path ( only realized later ) , so the spots are large ~ 3 mm at PRM qpd and ~4.5 mm at BS qpd
The spots are well centered.
Atm3, the spots were well centered yesterday ( the PRM is misaligned in pitch and retsore does not work today )
Lenses removed from oplev beam path at elog entry 11246