Not logged in
Thu Sep 25 02:35:21 2008
FAST is moving alot
It looks like the FAST signal has started moving a lot - this is partly what inspired us to tune the SLOW loop.
Some of the spiking events happen when people go on the table or the MC loses lock. But at other times it just
spikes for no apparent reason. You can also see from the first plot (9 day 10-minute trend) that there is no
great change in DTEC so we shouldn't be worried about clogging in the NPRO head.
The second plot is a 1 day minute-trend.
Uploaded Thu Sep 25 03:40:48 2008