For Modelling of the OLTF, I measured the response of the BS suspension. I used the OSEM sensor for measurement. The attatchment1 is the measured TF from C1:SUSBS_LSC_EXC to C1:SUSBS_SUSPOS_IN1 with exciting with random force. The measured data was fitted and the resonant frequency is 1.029(±0.005) Hz and quality factor is 12.25 (± 0.2). Additionally I did same measurement for ITMX and ITMY. The attachment 2 and 3 are the results for ITMX and ITMY. Each eigenfrequency and Q are 1.063 (±0.008) Hz and 7.33 (±0.13) (ITMX), 1.022 (±0.005) Hz and 9.41 (±0.09) (ITMY).
After that, I locked the MICH with AS55, and measured the PSD of error signal. I compensated the that PSD by the modelled OLTF with this suspension TF and the servo TF. The result is in attachment 4. Above 1 Hz it is quite close to the previous data by Keiko (elog#6385) But below 1 Hz there is a large dip. The error signal has also this dip. I looked for a integral filter between 0.2 Hz and 1 Hz, but I connot find a such filter. And when I locked MICH with using ASDC, there was same dip at same frequency. I don't think it's true free running noise, and I will try to fix it.
I completely forgot to mention that I fitted the modelled OLTF into the measured OLTF. I used the fitted OLTF for compensation.
