Following the circuit design in elog 8748, I constructed a prototype for the servo portion of the ISS (not including the differential amp) to be used in the CTN experiment. The device was built on a breadboard and its transfer function was measured with the Swept Sine measurement group of an SR785. For various excitation amplitudes, the transfer function (TF) was not consistent.
Recall the ideal transfer function for this particular servo and consider the following comparisons.
This gain limitation is problematic for characterizing prototypes as my particular servo has very large gain at low frequencies.
At the risk of looking too deeply into the above data,
Unfortunately, the noise was too large for lower excitation amplitudes to be used to any effect. I'll try this again tomorrow, just as a sanity check, but otherwise I will proceed with learning Altium and drawing up schematics for this servo.