After restoring alignment I could see again strong 00 flashes (about 250-300 counts on ALS-TRY). So I locked the arm with IR and after enabling the PDH servo for the green locking, I also locked the green on the Y arm in 00 mode. Then I moved the two mode matching lenses to maximize the power into the 00 mode, but I didn't reach more than 30-35 counts.
Green power injected into the Y arm 0.680mW
Green power reflected back 0.090mW
Green power transmitted on the PSL few uW
I would expect more power on the PSL table (maybe 10x more).