Not logged in
Tue Aug 19 17:00:19 2008
Wiener TF calibration - update
This is an update for post 814
I added the calibration gains I got for the accelerometers (I realize I am just calibrating the accelerometers to themselves and this is not m/m exactly since we don't really know which accelerometer is doing exactly what we want it to do. However, since we are talking on relative small numbers, this shouldn't really change much).
I also added another missing gain for the seismometer. Rana has previously installed a 4300 ohm resistor in the seismometer, which changed the gain to 4300/(4300+5000) = 0.46 (this is from the manual). Moreover, there is a gain of 100 on the SR560. This comes up to an extra gain of 46, meaning multiplying the seismometer's counts by 1/46.