Message ID: 8041
Entry time: Fri Feb 8 19:29:44 2013
In reply to: 8040
Reply to this: 8059

Author:

yuta

Type:

Summary

Category:

General

Subject:

arbcav of half PRC with flipped PR2

We need expected finesse and g-factor to compare with mode-scan measurement. Can you give us the g-factor of the half-PRC and what losses did you assumed to calculate the finesse?

Also, flipped PR2 should have RoC of - R_HR * n_sub (minus measured RoC of HR surface multiplied by the substrate refractive index) because of the flipping.
According to Jenne dictionary, HR curvature measured from HR side is;

PRM: -122.1 m
PR2: -706 m
PR3: - 700 m
TM in front of BS: -581 m

Please use these values to calculate expected g-factor so that we don't get confused.

Quote:

Arbcav with half PRC (flat temporary mirror in front of BS), PR2 RoC = 600m, PR3 RoC = -600m: