[Koji, Jamie, Jenne]
Koji did this, while we actuated on PRM in pos, and watched the oplev. Empirically, he found the following values for the POS column of the output matrix:
UL = 1.020
UR = 0.990
LL = 1.000
LR = 0.970
SD = 0.000
(The nominal values are all +1, except for Side, which is 0).
Actuation of PRM was through C1:SUS-PRM_LSC_EXC, f=0.1Hz, A=100 counts.
Ed by KA:
This means UL and UR are increased by 2% and UR and LR are decreased by 3%. More precisely UR should be 1.02*0.97.
This is just a quick hack which works only for the DC.