Not logged in
Wed Jul 30 01:08:50 2008
SUS Drift Screen
This is a snap of the SUS Drift screen with all of the optics biases set back to their nominal
values except for the MC which Rob aligned and I didn't feel like mis-aligning. The reference
on the screen is from 3/25 when Andrey felt that Rob had a good IFO alignment.
Anything more than a few thousand is significant and more than 10k means something is wrong:
I wailed on the PRM for awhile and was able to loosen it up a little. The LL & LR sensors now
show some life one the dataviewer. The UL & UR are still railed ~1.6 V so that means that the
optic is pitched back. With aggressive pitch wailing I can see the PRM's ULR/UR sensors go
rail to rail so that means that the magnets are still on - although they may be half busted.
If they're OK we should be able to just re-sling this guy.
Did the same on SRM. The OSEM values have shifted on these, but not disastorously. The SIDE,
however, is completely unresponsive. The little signal I see when driving is is probably just
capacitive pickup in the cables. Have to vent to fix this one.
ITMX Has good life in all but the LR & UR channels. They respond, but the signal is very weak.
Seems like these magnets have not fallen off but that they are not between the LED/PD anymore.
ITMY seems ok. Check the spectra to be sure.
BS seems ok as well. Swings freely and no kinks in the swinging sensor waveforms. Check the spectra.