Not logged in
Tue Jul 29 13:54:08 2008, rana, Update, SUS, ETMY and PRM have EQ related problems
Tue Jul 29 14:38:02 2008, rob, Update, SUS, ETMY and PRM have EQ related problems
Tue Jul 29 19:53:19 2008, Koji, Update, SUS, PRM photos from the south window
Tue Jul 29 13:54:08 2008
Reply to this:
ETMY and PRM have EQ related problems
The attached trend shows that ETMY and PRM both had large steps in their sensors
around the time of the EQ and didn't return afterwards. The calibration of the
OSEM sensors is ~0.5 mm/V. The PRM sensors respond when we give it huge biases
but there is very little change in the ETMY. Almost certainly true that the
optics have shifted in their wire slings and that we will have to vent to
examine and repair at least ETMY.
Jenne is looking at the spectra of the other suspensions to see if there is
other more subtle issues.