[Koji / Kiwamu]
We have realigned the interferometer except the incident beam.
The REFL beam is not coming out from the chamber and is likely hitting the holder of a mirror in the OMC chamber.
So we need to open the chamber again before trying to lock the recycled interferometers at some point.
--- What we did
---- things to be fixed
- Align the steering mirrors in the faraday rejected beam path (requires vent)
- SRM oplev (this is out of the QPD range)
- ITMX oplev (out of the range too)