[The 40m Family]
The access connector and all heavy doors are back on.
Jamie put the regular viton EQ stops back on PRM, since he had to adjust the distance between the EQ stops and the PRM anyway. Jamie also waved an IR card near the IPANG steering mirrors in ETMY, but it was not possible to take a good photo. Jamie certifies that the beam is centered on both of those 2 optics.
I have centered IPPOS and IPANG QPDs.
All oplevs need a little realignment, especially ETMY, which had it's lens removed (Rana has a Wall of Shame photo of this, which is why it was removed by him). Steve will look into this tomorrow, after he starts pumping.
I have turned off all PZT high voltage supplies for in-vac PZTs: The input PZTs, the output PZTs, and the OMC PZTs (which weren't on, but I confirmed they were off).
I have also prepared the 3 low-power items for high power: MC refl's path was changed back to regular BS, AS camera was moved to its nominal position, and IPPOS has its ND filters back. MC refl and the AS camera will need to be realigned once we're actually at high power tomorrow afternoon.
Long vent, but good work everyone.