I made the signal box as described in eLog 7210. It adds the geophone signal and an external signal.
It has six switches, for x, y, and z signals from both an external and local (geophone) source. The x signals add if both x switches are flipped down (and the same for the other directions). For example, if you want to feed in only an external signal in the x direction, flip down the external x direction switch (it's labeled on the box), leaving all others flipped up.
The x, y, and z outputs are wired to the pins from the preamplifier that go to the high voltage board. These I disconnected from the preamplifier by cutting at their base (there are spare connectors if this wants to be undone, or, a wire can just be soldered from the pin to its old spot on the board). The power (plus/minus) and ground are wired to the respective pins from the geophone preamplifier (naturally, the STACIS must be turned on for the box to work since the box shares its power source). Below, the front (switches and geophone/external inputs) and back (power, ground, outputs) of the box are shown:
The preamplifier can plug into its regular connectors- the x,y,and z signals will all be redirected to the signal box with these modifications. The box sits outside the STACIS, there is room to feed the wires out from underneath the STACIS platform.
NOTE: The geophone z switch is a little different than the others, just make sure it's flipped all the way down if you want that signal to be seen in the z output.