I have calibrated MC_F. The conversion factor is 137.49 MHz/count.
The calibration data taken is attached, along with a calibrated power spectrum.
On the data plot, the x axis is volts from the C1:IOO-MC_FAST_MON channel, with the calibration between FAST_MON and MC_F = -788.18 volts/count.
The linear term of the fit line = -0.085MHz/volt. Error bars are +/- 1 in the last digit of what the spectrum analyzer gave me for frequency (+/- 0.01MHz).
The net conversion factor is then (-788.18)*(-0.085)*(2) = 137.49 MHz/count. The factor of 2 is because the light passes through the AOM twice.
On the power spectrum,
REF0 and REF1 = MC unlocked, HEPAs on, MC Refl gain = 22
REF2 and REF3 = MC locked, HEPAs on, MC Refl gain = 22
REF4 and REF5 = MC locked, HEPAs on, MC Refl gain = 19
REF6 and REF7 = MC locked, HEPAs off, MC Refl gain = 19 |